Validated RP-HPLC Method for the Determination of Buspirone in Pharmaceutical Formulations

A simple, selective, linear, precise, and accurate RP-HPLC method was developed and validated for the rapid assay of the Buspirone in tablet dosage form. Isocratic elution at a flow rate of 1.0 mL/min was employed on a symmetry C18 (250 × 4.6 mm, 5 μm in particle size) at ambient temperature. The mobile phase consisted of water : acetonitrile : methanol 45 : 35 : 20(V/V). The UV detection wavelength was 210 nm, and 20 μL sample was injected. The retention time for the Buspirone was 7.057 min. The percentage RSD for precision and accuracy of the method was found to be less than 2%. The method was validated as per the ICH guidelines. The method can be successfully applied for the routine analysis of Buspirone in tablet dosage form.


Chemicals and Reagents.
HPLC grade acetonitrile and methanol were purchased from Merck Specialities Pvt. Ltd.

Instrumentation and Analytical
Conditions. The analysis of drug was carried out on a PEAK HPLC system equipped with a reverse phase C18 column (210 × 4.6 mm, 5 μm in particle size), a LC-P7000 isocratic pump, a 20 μL injection loop, and a LC-UV7000 absorbance detector and running on PEAK Chromatographic Software version 1.06. Isocratic elution with water : acetonitrile : methanol 45 : 35 : 20 (V/V) (P H -5.5) was used at a flow rate of 1.0 mL/min. The mobile phase was prepared freshly and degassed by sonicating for 5 min. before use.  2.3. Stock and Working Standard Solutions. 10 mg of Buspirone working standard was transferred into a 10 mL volumetric flask, and diluent was added and sonicated to dissolve it completely, and then the volume was made up to the mark using the same solvent. Further 1 mL of the above stock solution was pipette into a 10 mL volumetric flask and diluted up to the mark with diluent and was filtered through 0.45 μm nylon filter paper. From the above filtered sample 30 μg/mL solution was prepared. The calibration curve was plotted with the five concentrations of the 5.0-30.0 μg/mL working standard solutions. Calibration solutions were prepared daily and analyzed immediately after preparation.

Assay of Buspirone
Tablets. 20 mg of Buspirone tablets were weighed and the average weight was calculated. An Accurately weighed amount of 10 mg of Buspirone was transferred into a 10 mL volumetric flask. Diluent added to buspirone and sonicated to dissolve buspirone in diluent and add diluent up to the mark of volumetric flask. The constituents were mixed well and filtered through 0.45 um filter. Further 1 mL of the above stock solution was pipetted out into a 10 mL volumetric flask and diluted up to mark with diluent. From the above filtered sample 30 μg/mL solution was prepared. An aliquot of this solution was injected into HPLC system. Peak area of Buspirone was measured for the determination.

Optimization of the Chromatographic Conditions.
Proper selection of the stationary phase depends upon the nature of the sample, molecular weight, and solubility. The drug Buspirone in polar compounds preferably analyzed by reverse phase columns. Among C 8 and C 18 , C 18 column was selected. Nonpolar compound is very attractive with reverse phase columns. So the elution of the compound from the column was influenced by polar mobile phase. Mixture of water: acetonitrile: methanol was selected as mobile phase, and the effect of composition of mobile phase on the retention time of Buspirone was thoroughly investigated. The concentrations of the water: acetonitrile: methanol were optimized to give symmetric peak with short run time.

Method Validation
3.2.1. Linearity. Five-point graph was constructed covering a concentration range 5.0-25.0 μg/mL (three independent determinations were performed at each concentration.) Linear relationships between the peak area signal of Buspirone and the corresponding drug concentration were observed.   The standard deviation of the slope and intercept were low. The statistical analysis of calibration is shown in Table 1.

Precision.
The validated method was applied for the assay of commercial tablets containing Buspirone. Sample was analyzed for five times after extracting the drug as mentioned in assay sample preparation of the experimental section. The results presented good agreement with the labeled content. Low values of standard deviation denoted very good repeatability of the measurement. Thus, it was shown that the equipment used for the study was correct, and hence the developed analytical method is highly repetitive. For the intermediate precision, a study carried out by the same analyst working on the same day on three consecutive days indicated a RSD of 1.352. This indicates good method precision.

Stability.
The stability of Buspirone in standard and sample solutions containing different solutions was determined by storing the solutions at ambient temperature (20 ± 10 • C). The solutions were checked in triplicate after three successive days of storage, and the data were compared with freshly prepared samples. In each case, it is noticed that solutions were stable for 48 hrs, as during this time the results did not decrease below 98%. This denotes that Buspirone is stable and standard for at least 2 days at ambient temperature. Results are furnished in Table 2.

System Suitability.
System suitability tests were carried out on freshly prepared standard stock solutions of Buspirone and were calculated by determining the standard deviation of Buspirone standards by injecting standards in five replicates at 6 mins intervals and the values were recorded. The system suitability parameters, like capacity factor, asymmetry factor, tailing factor, and number of theoretical plates were calculated. It was observed that all the values are within the limits ( Table 1). The statistical evaluation of the proposed method revealed its good linearity, reproducibility, and its validation for different parameters and leads us to the conclusion that it could be used for the rapid and reliable determination of Buspirone in tablet formulation. The results are furnished in Table 2.

Conclusion
A RP-HPLC method has been developed and validated for the determination of Buspirone in tablet dosage form. This method is simple, rapid, accurate, precise, and specific. Its chromatographic runtime of 10 mins allows the analysis of a large number of samples in short period of time. Therefore, it is suitable for the routine analysis of Buspirone in pharmaceutical dosage forms ( Figure 2, Table 3).