ISRN Medicinal Chemistry

Volume 2013 (2013), Article ID 178269, 9 pages

http://dx.doi.org/10.1155/2013/178269

Research Article

New Stability Indicating Method for Quantification of Impurities in Amlodipine and Valsartan Tablets by Validated HPLC

Rama Joga Venkata Eranki,1,2 Gopichand Inti,¹ Venkatasubramanian Jayaraman,¹ Sudhakar Rao Vidiyala,¹ and J. SreeRamulu²

¹Quality and Analytical Development Department, InvaGen Pharmaceuticals, Inc., Hauppauge, NY 11788, USA

²Department of Chemistry, Sri Krishnadevaraya University, 515003 Anantapur, India

Received 31 March 2013; Accepted 8 May 2013

Academic Editors: Y. Onuki and U. G. Spizzirri

Copyright © 2013 Rama Joga Venkata Eranki et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

Abstract.

A stability indicating LC method was developed for simultaneous determination of amlodipine and valsartan in pharmaceutical dosage form. Efficient chromatographic separation was achieved on C₈ stationary phase with simple combination of mobile phase-A (70 : 20 : 10 v/v/v of water : acetonitrile : methanol with 2 mL of Octylamine adjusted the pH to 2.50 + 0.05 with orthophosphoric acid) and mobile phase-B (Acetonitrile) delivered in gradient mode. Quantification was carried out using ultraviolet detection at 240 nm at flow rate of 1.0 mL/min with Injection Volume of 100 μL and ambient column temperature. This method was capable to detect both the drug components of Amlodipine and Valsartan in presence of their degradation products (Amlodipine Imp-A and Valsartan Impurity-B) with the detection level of 0.05%. Amlodipine/Valsartan and their combination drug product were exposed to thermal studies, photolytic, hydrolytic and oxidative stress conditions, and samples analysed. Peak homogeneity data of Amlodipine and Valsartan is obtained using PDA detector, demonstrating the specificity. The method shows excellent linearity over range of 0.05–2.0% for Amlodipine; Amlodipine Impurity-A and 0.05–1.0% for Valsartan and Valsartan Impurity-B. The correlation coefficient for Amlodipine and Valsartan are 0.9999. The proposed method was found to be suitable and accurate for quantitative determination and the stability study of Amlodipine and Valsartan in pharmaceutical preparations.

1. Introduction

The amlodipine besylate component of amlodipine/valsartan tablets is chemically described as 3-ethyl-5-methyl (±)-2-[(2-aminoethoxy) methyl]-4-(2-chlorophenyl)-1, 4-dihydro-6-methyl-3, 5-pyridinedicarboxylate, monobenzenesulphonate [1]. Amlodipine is a dihydropyridine calcium antagonist (calcium ion antagonist or slow-channel blocker) that inhibits the movement of calcium ions into vascular smooth muscle cells and cardiac muscle cells.

Amlodipine besylate has empirical formula of C₂₀H₂₅ClN₂O₅·C₆H₆O₃S, and its molecular weight is 567.1, and its structural formula is shown in Scheme 1.

Scheme 1: Amlodipine chemical structure.

Valsartan is a nonpeptide, orally active and specific angiotensin II antagonist acting on the AT₁ receptor subtype. valsartan is a white to practically white fine powder, soluble in ethanol and methanol and slightly soluble in water. Valsartan’s chemical name is N-(1-oxopentyl)-N-[[2′-(1 H-tetrazol-5-yl) [1, 1′-biphenyl]-4-yl] methyl]-L-valine [1].

Its empirical formula is C₂₄H₂₉N₅O₃, molecular weight is 435.5, and its structural formula is shown in Scheme 2.

Scheme 2: Valsartan chemical structure.

Amlodipine/valsartan Tablets Brand name called as Exforge. Exforge contains 2 of the no. 1 prescribed high blood pressure medicines of their classes: amlodipine, a calcium channel blocker (CCB), and valsartan, an angiotensin receptor blocker (ARB). In clinical studies, EXFORGE was proven to be more effective in lowering high blood pressure than either of its components alone [2]. Amlodipine and valsartan are official in USP [3] but their combination drug product is not official in any pharmacopoeia. Exforge was proven to significantly lower high blood pressure in adults, regardless of age or gender. The blood pressure-lowering effects of Exforge have been studied in several clinical trials that included more than 5,000 patients with high blood pressure.

Exforge is available in a variety of dosage combinations that can be adjusted to best fit your needs. The dosage combinations are given as 5/160 mg, 5/320 mg, 10/160 mg, and 10/320 mg.

Amlodipine and valsartan tablets are developed in reference to (brand) Exforge tablets. Similar invagen has 4 strengths, namely, 5/160 mg, 10/160 mg; 5/320 mg, and 10/320 mg with the approximate tablet weights being 475 mg, 603 mg, 835 mg, and 950 mg, respectively.

The inactive ingredients and colorants for all strengths are standard excipients used in generic products.

Since this combination product has not been published in USP, a new method has been developed and validated as per ICH guideline and monitoring of these impurities with good separation of peaks and quantification of impurities in amlodipine/valsartan tablets as shown in Scheme 3.

Scheme 3: Amlodipine impurity-A, valsartan impurity-B.

Stability testing forming an important part of the process of drug product testing is to provide evidence on how quality of a drug substance or drug product varies with time under the influence of a variety of environmental factors such as temperature, humidity, and light and enables recommendation of storage conditions, retest periods, and shelf life to be established. The two main aspects of drug products that play an important role in shelf life determinations are assay of active drug and degradants generated during the stability study. Stability-indicating methods have been reported for assays of various drugs in drug products containing only one active drug substance. Only few stability-indicating methods are reported for the impurity assay of combining drug products containing two or more active drug substances. The objective of this work was to develop an analytical LC procedure, which would serve as stability-indicating impurity assay method for combining drug products of amlodipine and valsartan.

The literature survey reveals that several methods were reported for the individual estimation of amlodipine and valsartan. Various methods use HPLC [4–6], RP-HPLC [7, 8], HPTLC [9, 10], LC-MS [11], and LCMS/MS [12] and simultaneous UV spectrophotometric.

Methods [13, 14] are reported for the estimation of amlodipine alone or in combination with other antihypertensive agents. Methods such as HPLC [15–17], LC-MS [18–20], protein precipitation [21], capillary electrophoresis [22], and simultaneous UV spectrophotometer methods [23, 24] are reported for estimation of valsartan alone or in combination with other agents. None of the reported analytical procedures describe a stability indicating method for simultaneous determination of amlodipine and valsartan in combined pharmaceutical dosage form in the presence of their degradates.

If the reported individual methods are applied for the analysis of the tablets containing amlodipine and valsartan it would require UPLC to have shorter runtime and it is not possible for all firms to afford the same, and the method would not be rapid, less expensive, or economical, whereas the simultaneous determination of the ingredients of the tablets would be rapid, stability indicative, and also economical and can be afforded by all.

In the present study attempts were made to develop a rapid, economical, precise, and accurate method for the simultaneous estimation of the ingredients of this combination in the presence of their degradants.

2. Experimental

2.1. Chemicals and Reagents

Samples of amlodipine impurity-A and the valsartan impurity-B were synthesized and characterized at Hetero Drugs and Hetero Labs Limited, India. HPLC grade acetonitrile and methanol were procured from Honeywell: Burdick & Jackson, muskegon, MI, and analytical grade octylamine was procured from Sigma Aldrich Company. High purity water was generated in-house from Siemens water purification system.

2.1.1. Chromatographic Conditions

The chromatographic system used was Shimadzu LC 2010 HPLC system comprised of degasser, quaternary pump, autoinjector, column compartment, and UV detector, and the system was controlled through EZ chrome software. Zorbax SB C8 column (i.d. 4. 6 · 150 mm, 3.5 μm, Advance Chromatography, USA), maintained at 25°C using a column oven, was eluted with mobile phase at the flow rate of 1.0 mL min⁻¹ with gradient program (see Table 1).

Table 1: Elution: gradient.

Time (min)
Composition of mobile phase
Elution

% A
% B

0.01
100
0
Isocratic

20
100
0
Isocratic

60
40
60
Linear gradient

62
100
0
Reequilibration

75
100
0
Reequilibration

Mobile Phase A: Preparation involved transfer of 700 mL of DI Water, 200 mL of Acetonitrile and 100 mL of Methanol into suitable container, with addition of 2.0 mL of Octylamine and mixed well. Adjusted the pH to using Ortho Phosphoric acid. Filtered through 0.45 μm nylon membrane filter and degassed. Mobile Phase B: Acetonitrile.

Measurements were made with injection volume 100 μL and ultraviolet (UV) detection at 240 nm.

For standard and sample solution preparation were done using the diluents consisting of Methanol and DI water in the ratio of 50 : 50 v/v.

For analysis of forced degradation samples, the photodiode array detector (Model no. 2998) and Empower Software were used in scan mode with a scan range of 200–400 nm. The peak homogeneity was expressed in terms of peak purity and was obtained directly from the spectral analysis report using the previously-mentioned software.

2.1.2. Standard Stock Solutions

Standard solutions were prepared by dissolving the drugs in the diluent and diluting them to the desired concentration.

2.1.3. Amlodipine

17.36 mg amlodipine standard (99.8%) was accurately weighed, transferred into a 250 mL volumetric flask, and dissolved with diluent.

2.1.4. Valsartan

320.0 mg valsartan standard (99.3%) was accurately weighed, transferred into a 100 mL volumetric flask, and dissolved with diluent.

2.1.5. Low-Level Standard Preparation

The concentration of Low-level standard preparation contains 0.0003472 mg/mL of amlodipine besylate and 0.016 mg/mL of valsartan.

2.1.6. Detectability Level Standard Preparation

The concentration of detectability level standard preparation contains 0.00003472 mg/mL of amlodipine besylate and 0.0016 mg/mL of valsartan.

2.1.7. Impurity Stock Solution

Transfer  accurately Weighed 1.80 mg of Amlodipine Impurity-A Standard in to 100 mL of volumetric flask, and dissolved with diluent.

2.1.8. Identification Solution

Transfer  accurately Weighed 6.94 mg of Amlodipine Standard in to 100 mL of volumetric flask, and added 5.0 mL of the above impurity stock solution dissolved with diluent. The concentration of identification solution contains 0.0009 mg/mL of amlodipine besylate impurity-A and 0.0694 mg/mL of amlodipine besylate.

2.1.9. Preparation of Sample

Twenty tablets were weighed for average weight and finely powdered. A quantity of powder equivalent to 320.0 mg of valsartan was transferred into a 100 mL volumetric flask. To this flask, 75 mL of diluent was added, and the solution was sonicated for about 15 min with intermittent shaking and with mechanical shaking for about 15 min. The solution was cooled to ambient temperature. Then the volume was made up with diluent and centrifuged at 10,000 rpm for about 15 min. Then the solution was used for injection.

2.1.10. Analytical Method Validation

The developed chromatographic method was validated for selectivity, linearity, range, precision, accuracy, sensitivity, robustness, and system suitability.

2.1.11. Selectivity/Specificity

Selectivity of the developed method was assessed by performing forced degradation studies. According to ICH stress testing of the drug substance can help the intrinsic stability of the molecule and validate the stability indicating power of the analytical procedure used. Photo stability testing should be an integral part of stress testing. The standard conditions for photo stability testing are described in ICH Q1B. The specificity of the developed LC method for amlodipine and valsartan was determined in the presence of its related compounds amlodipine impurity-A, valsartan impurity-B. The stress conditions employed for degradation study include photolysis (ICH Q1B), thermal studies (80°C), acid hydrolysis (1N HCl), base hydrolysis (1N NaOH) and oxidation (30% H₂O₂). For thermal study period was 7 days at 80°C, whereas for acid, base, and oxidation it was 2 hours. The results of data are presented in Tables 2(a) and 2(b) and related chromatograms and spectra are represented in Figures 1 through 10.

Table 2: Results of analysis of forced degradation study samples using the proposed method, indicating percentage of degradation and peak purity of amlodipine and valsartan.

(a)  

Degradation in
Degradation sample
Amlodipine imp-A

(%)
Valsartan imp-B

(%)
Maximum individual unknown impurity (%)
Total impurities

(%)

As such sample
Placebo + amlodipine and valsartan
0.12
0.07
BRL
0.24

1 N HCl (5 mL)
Placebo + amlodipine and valsartan
0.11
0.07
0.06
0.34

1 N NaOH (5 mL)
Placebo + amlodipine and valsartan
0.11
0.07
0.06
0.34

30% H₂O₂ (5 mL)
Placebo + amlodipine and valsartan
5.16
ND
0.31
6.35

UV light
Placebo + amlodipine and valsartan
0.10
0.07
0.06
0.28

Thermal condition at 80°C

Placebo + amlodipine and valsartan
0.11
0.07
0.06
0.34

ND: not detected; BRL: below reporting level (BRL = 0.05%).

(b)  

Degradation samples
Peak

area
Peak

area
Retention

time (min)
Purity

angle
Purity threshold
Match

angle
Match threshold

Placebo + drug degradation in 5 mL of 1 N HCl
Amlodipine
2598903
12.30
0.236
0.432
0.138
1.159

Valsartan
2943627
40.92
0.050
0.291
0.038
1.025

Placebo + drug degradation in 5 mL of 1 N NaOH
Amlodipine
2599981
12.36
0.233
0.448
0.128
1.164

Valsartan
2938404
40.91
0.047
0.287
0.042
1.025

Placebo + drug degradation in 5 mL of 30% H₂O₂
Amlodipine
2483045
12.38
0.223
0.431
0.138
1.168

Valsartan
2891144
40.92
0.063
0.319
0.055
1.038

Placebo + drug degradation in UV light
Amlodipine
2586948
12.05
0.234
0.438
0.220
1.162

Valsartan
2933545
40.76
0.059
0.303
0.109
1.031

Placebo + drug degradation in thermal conditionat 80°C
Amlodipine
2602008
12.46
0.247
0.448
0.193
1.161

Valsartan
2924035
40.88
0.049
0.288
0.051
1.025

(a)

(b)

Figure 1: Chromatogram of Placebo and sample of 5/320 mg on acid degradation.

(a)

(b)

Figure 2: Chromatogram of Placebo and sample of 5/320 mg on base degradation.

(a)

(b)

Figure 3: Chromatogram of placebo and sample of 5/320 mg on oxidation.

(a)

(b)

Figure 4: Chromatogram of placebo and sample of 5/320 mg on UV treatment.

(a)

(b)

Figure 5: Chromatogram of placebo and sample of 5/320 mg on thermal treatment.

Figure 6

Figure 7

Figure 8

Figure 9

Figure 10

3. Results and Discussion

3.1. Optimization of the Chromatographic Conditions

To develop the stability-indicating method different stationary phases like C18, CN, different mobile phases containing buffers like phosphate, ammonium acetate, and trifluoroacetic acid with different pH (3–5) and organic modifier (acetonitrile) were used.

Our objective of the chromatographic method development was to achieve a peak tailing factor <2, signal to noise ratio should be above 10, theoretical plates should be above 1000 for amlodipine and valsartan, % RSD for 5 consecutive injection should be less than NMT 10.0%, and very good separation between amlodipine impurity-A, valsartan impurity-B along with drug peak amlodipine and valsartan.

The chromatographic separation was achieved using a Zorbax SB C8 column (i.d. 4.6·150 mm, 3.5 μm). Changing the composition of mobile phase optimized the chromatographic method. Segregation of all of the 2 peaks (amlodipine and valsartan) was observed on any C18 or CN column, but it was difficult to separate both drug degradants on these columns (amlodipine impurity-A and valsartan impurity-B). The Zorbax SB C8 column shows better performance as compared to other columns.

From the development studies, it was determined that using mobile phase-A as water : acetonitrile : methanol in the ratio of 70 : 20 : 10 (v/v) with 2.0 mL of octylamine and adjusted the pH to using Ortho Phosphoric acid and Mobile phase-B as Acetonitrile with Gradient flow rate of 1.0 mL/min and ambient temperature, yielded good results. The analytes of this combination had adequate retentions, peak shape, less tailing, and more resolution between drug and it’s degradants, and the chromatographic analysis time was about 70 min. In optimized conditions amlodipine, valsartan, and their degradants were well separated. Typical retention times of amlodipine and valsartan were about 12.03 and 44.92 min.

Even the retention time looks little higher but the separations were achieved to greater extent and the methods prove stability indicating. Instead of regular HPLC if we use UFLC the retention time can be reduced to 20 minutes with the same suitability parameters, but when compared to cost efficiency all cannot afford UFLC.

During the initial forced degradation experiments, it was observed that oxidation was a fast reaction for amlodipine and valsartan tablets, and almost complete degradation occurred when 5 mL of 30% H₂O₂  solution was used. Both drugs showed extensive degradation in alkali hydrolytic and oxidative condition and indicate homogeneous peaks and thus establishing the specificity of the impurity assay method.

3.2. Calibration and Linearity

Calibration curve obtained by the least square regression analysis between average peak area and concentration showed linear relationship with a regression coefficient of 0.999 over the calibration ranges tested.

The results of linearity and range obtained for the two potential impurities were tabulated. Linear calibration plot for this chromatographic method was obtained over the calibration ranges tested, that is, 0.05% to 1.0% for amlodipine impurity-A and 0.05% to 1.0% for valsartan impurity-B. The correlation coefficient obtained was greater than 0.999 for the two impurities and the major compounds amlodipine and valsartan (Figures 11 and 12). The method exhibited good linearity with correlation coefficient values greater than 0.999.

Figure 11

Figure 12

3.3. Precision (Repeatability)

The precision of the method was studied by determining the concentrations of each 0.85 and 0.89. The results of the precision study indicate that the method is reliable (RSD% < 10), in Tables 3(a) and 3(b).

Table 3: (a) % RSD of six (6) replicate injections of each impurity should be less than 10.0, theoretical plates should be NLT 1000 and tailing factor should be NMT 2.0 for system precision. (b) % RSD of six (6) sample preparations for each impurity should be less than 10.0 for method precision.

(a)  

 System precision

Amlodipine
Valsartan

% RSD

Theoretical plates (N)

Tailing factor (T)
0.85

12460.23

1.034
% RSD

Theoretical plates (N)

Tailing factor (T)
0.89

617503.71

1.045

(b)  

Method precision 5/320 mg

Amlodipine impurity-A

% RSD
1.06
Valsartan impurity-B

% RSD
0.83

3.4. Accuracy (Recovery Test)

The percentage recovery was established for all the analytes throughout the range concentration as explained under linearity studies, and obtained results are tabulated later in Table 4.

Table 4

Concentration

µg/mL
% Spiking level
Average peak area
% Recovery
Mean recovery %

Amlodipine imp-A

0.0310

0.3110

3.1100
0.05

0.5

2.0
2806

28745

121197
94.0

96.3

101.5
101.5

Valsartan imp-B

1.6475

16.475

164.75
0.05

0.5

1.0
339061

3398554

6795594
105.2

105.5

105.4
105.4

4. Robustness

Robustness study was conducted by making small but deliberate changes in the optimized method parameters.

To determine robustness of the method, experimental conditions were purposely altered and chromatographic resolution between amlodipine and valsartan was evaluated.

The flow rate of the mobile phase was 1.0 mL min⁻¹. To study the effect of flow rate on the retention time of amlodipine and valsartan, it was changed to 0.9 mL/min and 1.1 mL/min. The effects of pH variation were studied at pH 2.40 and pH 2.60 instead of pH 2.50, while other mobile phase components were held constant.

At all conditions the relative retention time found for amlodipine impurity-A and valsartan impurity-B was found 0.46 and 0.93, respectively (see Figure 13).

Figure 13

4.1. Determination of Limit of Quantification

Prepared Amlodipine and Valsartan LOQ solution as per the method containing the concentration of about 0.0346 μg/mL of Amlodipine and 1.5900 μg/mL of Valsartan. Made five (5) replicate injections and calculated S/N ratio of 0.05% to establish LOQ (see Table 5).

Table 5: S/N ratio of amlodipine, valsartan and composite impurities at LOQ (0.05%) level.

Component name
Concentration µg/mL
% RSD
Signal to noise

ratio (S/N)

Amlodipine impurity-A
0.0253
5.43
75.672

Amlodipine
0.0346
0.70
100.156

Valsartan impurity-B
1.6145
0.71
9168.987

Valsartan
1.5900
0.66
9977.428

5. Conclusion

The gradient RP-LC method developed for the analysis of binary mixtures of amlodipine and valsartan in their pharmaceutical preparations is precise, accurate, and with a reasonable run time. The method was fully validated showing satisfactory data for all the method validation parameters tested. The developed method is stability indicating, separates degradants, and can be conveniently used by the quality control department to determine impurity assay of pharmaceutical preparations and also stability sample analysis.

Acknowledgments

The authors wish to thank the management of Invagen Pharmaceuticals, Inc., for supporting this work. Cooperation from colleagues of Analytical Research and Development of Invagen Pharmaceuticals, Inc., is appreciated.

References

S. Budavari, The Merck Index, Merck and Co. Press, Whitehouse Station, NJ, USA, 14th edition, 2006.

http://www.exforge.com/

The United States Pharmacopoeia Convention, Inc., Rockville, Md, USA, 2007.

A. Zarghi, S. M. Foroutan, A. Shafaati, and A. Khoddam, “Validated HPLC method for determination of amlodipine in human plasma and its application to pharmacokinetic studies,” Farmaco, vol. 60, no. 9, pp. 789–792, 2005.

G. Bahrami and S. H. Mirzaeei, “Simple and rapid HPLC method for determination of AML in human serum with fluorescence detection and its use in pharmacokinetic studies,” Journal of Pharmaceutical and Biomedical Analysis, vol. 36, pp. 163–168, 2004.

M. Ahmed, Y. Manohara, and M. Ravi, “RP-HPLC method development and validation for simultaneous estimation of atorvastatin calcium and amlodipine besylate,” International Journal of ChemTech Research, vol. 4, no. 1, pp. 337–345, 2012.

A. P. Agrekar and S. G. Powar, “Reverse phase high performance liquid chromatographic determination of ramipril and AML in tablets,” Journal of Pharmaceutical and Biomedical Analysis, vol. 21, pp. 1137–1142, 2000.

B. G. Chaudhari, N. M. Patel, and P. B. Shah, “Stability indicating RP-HPLC method for simultaneous determination of atorvastatin and amlodipine from their combination drug products,” Chemical and Pharmaceutical Bulletin, vol. 55, no. 2, pp. 241–246, 2007.

K. K. Pandya, M. Satia, T. P. Gandhi, I. A. Modi, R. I. Modi, and B. K. Chakravarthy, “Detection and determination of total amlodipine by high-performance thin-layer chromatography: a useful technique for pharmacokinetic studies,” Journal of Chromatography B, vol. 667, no. 2, pp. 315–320, 1995.

S. N. Meyyanathan and B. Suresh, “HPTLC method for the simultaneous determination of amlodipine and benazepril in their formulations,” Journal of Chromatographic Science, vol. 43, no. 2, pp. 73–75, 2005.

Y. Feng, L. Zhang, Z. Shen, F. Pan, and Z. Zhang, “Analysis of amlodipine in human plasma by liquid chromatography-mass spectrometry,” Journal of Chromatographic Science, vol. 40, no. 1, pp. 49–53, 2002.

J. Bhatt, S. Singh, G. Subbaiah, B. Shah, S. Kambli, and S. Ameta, “A rapid and sensitive liquid chromatography-tandem mass spectrometry (LS-MS/MS) method for the estimation of amlodipine in human plasma,” Biomedical Chromatography, vol. 21, no. 2, pp. 169–175, 2007.

M. D. Malesuik, S. G. Cardoso, L. Baierski, and F. A. Lanzanova, “Determination of amlodipine in pharmaceutical dosage forms by liquid chromatograpy and ultraviolet spectrophotometry,” Journal of AOAC International, vol. 89, no. 2, pp. 359–364, 2006.

R. Sahu and V. B. Patel, “Simultaneous spectrophotometric determination of amlodipine besylate and atorvastatin calcium in binary mixture,” Indian Journal of Pharmaceutical Sciences, vol. 69, no. 1, pp. 110–111, 2007.

B. Koçyigit-Kaymakcoglu, S. Ünsalan, and S. Rollas, “Determination and validation of ketoprofen, pantoprazole and valsartan together in human plasma by high performance liquid chromatography,” Pharmazie, vol. 61, no. 7, pp. 586–589, 2006.

N. Daneshtalab, R. Z. Lewanczuk, and F. Jamali, “High-performance liquid chromatographic analysis of angiotensin II receptor antagonist valsartan using a liquid extraction method,” Journal of Chromatography B, vol. 766, no. 2, pp. 345–349, 2002.

L. González, J. A. López, R. M. Alonso, and R. M. Jiménez, “Fast screening method for the determination of angiotensin II receptor antagonists in human plasma by high-performance liquid chromatography with fluorimetric detection,” Journal of Chromatography A, vol. 949, no. 1-2, pp. 49–60, 2002.

N. Koseki, H. Kawashita, H. Hara et al., “Development and validation of a method for quantitative determination of valsartan in human plasma by liquid chromatography-tandem mass spectrometry,” Journal of Pharmaceutical and Biomedical Analysis, vol. 43, no. 5, pp. 1769–1774, 2007.

H. Li, Y. Wang, Y. Jiang et al., “A liquid chromatography/tandem mass spectrometry method for the simultaneous quantification of valsartan and hydrochlorothiazide in human plasma,” Journal of Chromatography B, vol. 852, no. 1-2, pp. 436–442, 2007.

P. S. Selvan, K. V. Gowda, U. Mandal, W. D. S. Solomon, and T. K. Pal, “Simultaneous determination of fixed dose combination of nebivolol and valsartan in human plasma by liquid chromatographic-tandem mass spectrometry and its application to pharmacokinetic study,” Journal of Chromatography B, vol. 858, no. 1-2, pp. 143–150, 2007.

J. Macek, J. Klíma, and P. Ptáček, “Rapid determination of valsartan in human plasma by protein precipitation and high-performance liquid chromatography,” Journal of Chromatography B, vol. 832, no. 1, pp. 169–172, 2006.

S. Hillaert and W. Van Den Bossche, “Simultaneous determination of hydrochlorothiazide and several angiotensin-II-receptor antagonists by capillary electrophoresis,” Journal of Pharmaceutical and Biomedical Analysis, vol. 31, no. 2, pp. 329–339, 2003.

E. Satana, S. Altinay, N. G. Goger, S. A. Ozkan, and Z. J. Senturk, “Simultaneous determination of VAL and Hydrochlorothiazide in tablets by first-derivative ultraviolet spectrophotometry and LC,” Journal of Pharmaceutical and Biomedical Analysis, vol. 25, pp. 1009–1013, 2001.

S. Tatar and S. Salk, “Comparison of UV- and second derivative-spectrophotometric and LC methods for the determination of valsartan in pharmaceutical formulation,” Journal of Pharmaceutical and Biomedical Analysis, vol. 30, no. 2, pp. 371–375, 2002.