ISRN Radiology

Volume 2013 (2013), Article ID 318425, 7 pages

http://dx.doi.org/10.5402/2013/318425

Clinical Study

Foetal Radiation Dose and Risk from Diagnostic Radiology Procedures: A Multinational Study

Ernest K. Osei1,2 and Johnson Darko3,4

¹Department of Medical Physics, Grand River Regional Cancer Center, Kitchener, Canada, N2G 1G3

²Department of Physics and Astronomy, University of Waterloo, Waterloo, Canada, N2L 3G1

³Cancer Center of Southeastern Ontario, Kingston General Hospital, Kingston, Canada, K7L 5P9

⁴Department of Oncology, Queens University, Kingston, Canada, K7L 5P9

Received 4 June 2012; Accepted 8 July 2012

Academic Editors: M. G. Andreassi, I. Seimenis, and H.-X. Xu

Copyright © 2013 Ernest K. Osei and Johnson Darko. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

Abstract.

In diagnostic radiology examinations there is a benefit that the patient derives from the resulting diagnosis. Given that so many examinations are performed each year, it is inevitable that there will be occasions when an examination(s) may be inadvertently performed on pregnant patients or occasionally it may become clinically necessary to perform an examination(s) on a pregnant patient. In all these circumstances it is necessary to request an estimation of the foetal dose and risk. We initiated a study to investigate fetal doses from different countries. Exposure techniques on 367 foetuses from 414 examinations were collected and investigated. The FetDoseV4 program was used for all dose and risk estimations. The radiation doses received by the 367 foetuses ranges: <0.001–21.9 mGy depending on examination and technique. The associated probability of induced hereditary effect ranges: <1 in 200000000 () to 1 in 10000 () and the risk of childhood cancer ranges <1 in 12500000 () to 1 in 500 (). The data indicates that foetal doses from properly conducted diagnostic radiology examinations will not result in any deterministic effect and a negligible risk of causing radiation induced hereditary effect in the descendants of the unborn child.

1. Introduction

In diagnostic radiology examinations, there is a benefit that the patient derives from the resulting diagnosis, provided that they are fully justified. However, given that so many examinations are performed each year, it is inevitable that there will be occasions when an examination will be performed on a woman who subsequently discovers that she was pregnant at the time of her examination(s). It may also occasionally become clinically necessary to perform a radiological examination(s) on a woman who is known to be pregnant. In the later case, there must be rigorous justification of the examination and the procedure itself must be optimized to minimise the foetal dose [1–3]. In order to avoid the former, some special rules have been developed to apply to the exposure of potentially pregnant women (who are or who may be pregnant) in which radiological examinations of such women are restricted to a certain period following menstruation [4–7].

For the protection of the foetus from occupational exposure of the pregnant worker, the International Commission on Radiological Protection (ICRP) [1, 2] considers that if a female worker has declared (i.e., notified her employer) that she is pregnant, additional controls have to be considered to protect the embryo/foetus. It is the Commission’s policy that the methods of protection at the workplace for women who are pregnant should provide a level of protection for the embryo/foetus, that is, broadly similar to that provided for members of the public. Therefore, the working conditions of a pregnant worker, after declaration of pregnancy, should be such as to ensure that the additional dose to the embryo/foetus would not exceed about 1 mSv during the remainder of the pregnancy [1, 2]. Irrespective of how these protective measures are applied in practice, it is almost inevitable that inadvertent foetal radiation exposures do occur. In these circumstances, it is necessary to consult an expert in medical physics and request that the foetal dose and hence the projected risks be estimated. This may require the medical physicist to calculate the dose based on the knowledge of the technique factors used, or a simulation of the examination using a phantom loaded with dosimeters or review the data from published scientific literature [7–20].

Osei and Faulkner [11] estimated the foetal dose of 50 pregnant women in the north of England. Doses to the embryo/foetus varied between less than 0.01 μGy and 117 mGy, depending on the examination and the gestational ages ranged between 2 and 24 weeks. Angel et al. [18] used Monte Carlo simulations to investigate foetal radiation dose resulting from an abdominal and pelvic examination for a range of patient gestational age and maternal size. The normalized foetal dose estimates from the Monte Carlo simulations ranged from 7.3 to 14.3 mGy/100 mAs, with an average of 10.8 mGy/100 mAs. Doshi et al. [19] also investigated foetal radiation dose from CT pulmonary angiography in late pregnancy. Dose measurements were made on three helical CT scanners, with an anthropomorphic phantom representing the chest and abdomen in late gestation. The estimated foetal doses from the CT scans of the maternal chest were in the range of 60–230 mGy, however by modulating the mA, shielding with a lead coat and scanning 5 cm shorter lengths, foetal doses were reduced by 10%, 35% and 56%, respectively. The Health Protection Agency, UK [20] has recently published foetal doses for some common diagnostic medical exposures. Foetal doses were derived from doses to the uterus observed in recent UK surveys. The foetal doses ranged from about 0.001 mGy to 50 mGy depending on the examination.

An accurate approach to provide an estimate of foetal dose, either prospectively or retrospectively is to assess the dose for the individual patient using the technique parameters that were used for the patient’s examination and taking into account both foetal depth and size. Simply looking up the average uterus dose for a specific examination may lead to underestimation or over estimation of the foetal dose [8–11]. Requests for foetal absorbed dose and associated radiation risk estimates are usually sent to medical physics departments where qualified medical physicists with expertise in foetal dose calculations will estimate the dose and risks associated with the exposure. In this study, we contacted several of these departments in different countries where requests for foetal dose estimate have been sent. Specific values for the following parameters were requested: projection and view (e.g., AP, PA) for each examination, beam quality (e.g., kVp, filtration), source-to-imager receptor distance, machine outputs, and techniques parameters (kVp, tube current, exposure time, etc.) used for the examination(s). In all, we collected and studied exposure data of 367 foetuses and 414 examinations from five institutions in five countries. The purpose of this paper is to present the results of this study.

2. Materials and Methods

2.1. Foetal Dose Calculations

We contacted several departments in different countries where requests for foetal dose estimate are usually sent. Specific values for the exposure parameters for simple examinations (e.g., chest, etc.), complex (fluoroscopic) examinations (e.g., barium meal, etc.), and CT scans were requested using (see the tables (1A–1C) in Appendix 1 in supplementary material available online at http://dx.doi.org/10.5402/2013/318425). All foetal doses and risks were calculated using the most recent version of FetDose [8] (FetDose V4). FetDose is a computer program developed for the estimation of the radiation dose to the foetus. It calculates the absorbed dose and risk to the foetus from conventional radiography, fluoroscopy, computed tomography, and radiation therapy procedures performed on the pregnant patient. It also calculates the dose to the foetus from occupational exposure of the pregnant worker in the radiological department. The normalized uterus doses (NUD) used for foetal dose calculations are taken from NRPB SR262 [12] and NRPB-SR250 [17]. A more detailed information on the methods used for foetal absorbed dose calculation and hence risk estimations using the FetDose program can be found elsewhere [8], however a short summary is given here. The foetal absorbed dose from a series of radiographic examinations, , when the quantity supplied is the entrance surface dose () for each radiograph is calculated as [8]

where

is the number of radiographs, is the uterus dose at a mean foetal depth normalized to free-in-air entrance surface dose, and is the foetal size factor (i.e., uterus to foetus dose conversion factor). For complete examinations, including fluoroscopy (e.g., barium enema, barium meal, etc.), involving both spot films and screening procedures of different areas of the body, the total foetal dose is calculated by summing the contributions from both the spot films and screening procedures as

where

If the quantity provided is the dose-area product per examination, in Gy-cm², then the foetal absorbed dose from a series of radiographic examinations is calculated from

where is the uterus dose at a mean foetal depth normalized to the dose-area product, and is the dose-area product for each examination, .

The foetal dose from a series of computed tomography scans is calculated from

where (mGy/100 mAs) is the to ICRU muscle () used as an approximation to the dose to soft tissue within the body, NUDV is the sum of normalized doses for all 5 mm slabs lying within the scan volume. The National Radiological Protection Board (NRPB) report SR250 [17] provides normalised organ dose data of 23 series of Monte Carlo calculations modelling the conditions of exposure relevant to 27 common models of scanners. Each set of calculation involved 5 mm transverse section (slab) of a mathematical phantom. Estimates of normalised dose to 27 organs or regions of the phantom are calculated for 208 contiguous slabs of the phantom. The contribution of the normalised dose to the uterus is used in FetDose for the estimation of foetal doses.

2.2. Consequence of Foetal Irradiation

The FetDose program presents the consequence of foetal irradiation in four different ways: (1) risks (the probability that a consequence has to be expected), (2) safety (the probability that a consequence is not expected), (3) equivalent number of chest X-rays, and (4) equivalent period of exposure to natural background radiation. In this paper, the consequence is expressed only as the probability that a consequence is expected (risk). Foetal radiation risk () is calculated as

where RC is the risk coefficient for the consequence of interest [8]. Radiation effects considered include childhood cancer induction (risk coefficient used is 8.0 × 10⁻⁵ per mGy [4–6]), hereditary effects (risk coefficient used is 0.5 × 10⁻⁵ per mGy [4–6]), decline in IQ (risk coefficient used is 25 × 10⁻³ IQ points per mGy [21]) and severe mental retardation (risk coefficient used is 43 × 10⁻⁵ per mGy [22]). These risk coefficients have been mainly derived from high doses and doserates, and the extrapolation to low doses is far from verified. Nevertheless, most experts accept that the linear no-threshold (LNT) model best fits available data and should remain the foundation of radiation protection.

3. Results and Discussion

The categorization of the total number of foetuses exposed in utero into various gestational age (weeks) intervals is shown in Figure 1. Figure 2 shows the categorization of the total number of foetus exposed in utero into various absorbed dose intervals. Table 1 shows a summary of the maternal and foetal parameters. Estimated mean and range of entrance surface doses to female adult patients from diagnostic radiology examinations is shown in Table 2. Tables 3 and 4 show a summary of the mean and range of foetal absorbed dose per examination from some simple and special types of diagnostic radiology examinations, respectively. The foetal radiation risks of childhood cancer and hereditary effects per examination are shown in Table 5. Table 6 also shows the risks of childhood cancer and hereditary effects for the 367 foetuses investigated in this work.

Table 1: Summary of maternal and foetal parameters.

Parameter
Range
Mean

Maternal height (m)
1.4–1.8
1.6

Maternal weight (kg)
40.9–119.8
68.7

Maternal AP thickness (cm)
16.5–31.0
23.7

Maternal age (years)
20–39
30.9

GA (weeks)
2–38
19.2

Foetal depth (cm)*
4.9–9.3
7.1

*If the foetal depth is not provided, a mean value of 9 cm is used [9].

Table 2: Estimated mean and range of entrance surface doses to female adult patients from diagnostic radiology examinations.

Projection
Examination
No. of cases
Estimated entrance surface dose (mGy)

Range
Mean

AP
Abdomen
56
0.2–10.5
2.62

AP
Chest
38
0.034–0.41
0.12

LAT
Chest
36
0.07–2.51
0.72

PA
Chest
52
0.01–1.62
0.26

PA
Colon
3
4.69–6.24
5.23

Hip joint
5
0.28–3.13
1.35

AP
Lumbar spine
39
0.4–16.1
4.62

PA
Lumbar spine
4
0.45–0.80
0.56

LAT
Lumbar spine
41
0.29–38.30
12.06

Lumbosacral joint
30
1.03–28.53
12.24

AP
Pelvis
3
0.59–2.76
1.40

AP
Stomach
16
0.52–6.34
2.50

LAO
Stomach
2
1.59–3.35
2.47

LAT
Stomach
2
2.10–2.60
2.35

LPO
Stomach
6
3.31–3.35
3.33

PA
Stomach
12
0.26–6.34
2.40

AP
Thoracic spine
3
1.25–2.94
2.10

LAT
Thoracic spine
1
3.20
3.20

Table 3: The mean and range of foetal absorbed dose per examination from some simple diagnostic radiology examinations.

Projection
Examination
Foetal absorbed dose per examination (mGy)

Range
Mean

AP
Abdomen
0.05–3.74
0.80

AP
Chest
<0.001–0.001
0.001

LAT
Chest
<0.001–0.01
0.002

PA
Chest
<0.001–0.01
0.002

PA
Colon
1.66–2.20
1.85

Hip joint
0.01–0.05
0.03

AP
Lumbar spine
0.11–4.42
1.14

PA
Lumbar spine
0.02–0.80
0.23

LAT
Lumbar spine
0.03–3.89
0.92

Lumbosacral joint
0.02–0.80
0.25

AP
Pelvis
0.18–1.07
0.49

AP
Stomach
0.01–1.33
0.18

LAO
Stomach
<0.01–0.01
0.01

LAT
Stomach
<0.01–0.01
0.01

LPO
Stomach
<0.01–0.01
0.01

PA
Stomach
<0.01–0.01
0.02

AP
Thoracic spine
<0.001–0.85
0.28

LAT
Thoracic spine
<0.001
<0.001

Table 4: The mean and range of foetal absorbed dose per examination from some special types of diagnostic radiology examinations.

Examination
No. of cases
Foetal absorbed dose per examination (mGy)

Range
Mean

Barium enema
3
1.14–16.27
7.23

Barium meal
7
0.08–0.19
0.11

Cholangiography
7
1.57–6.65
4.04

Barium follow through
2
0.17–0.64
0.41

IVU
3
0.05–1.33
0.71

KUB
7
0.01–0.59
0.21

Urinary bladder (AP)
1
1.49
1.49

Nephrostomy
2
<0.01–0.37
0.18

CT-abdomen
12
1.04–3.65
1.99

CT-chest
3
0.02
0.02

CT-lumbar spine
10
13.61–21.85
19.05

CT-pelvis
8
1.32–17.06
10.64

Table 5: Foetal absorbed dose and risks of childhood cancer and hereditary effects for some common diagnostic radiology examinations.

Projection and examination
Foetal dose range (mGy)
Risk of childhood cancer
Risk of hereditary effects

AP
Abdomen
0.05–3.74
1 in 250000–1 in 3300
1 in 3300000–1 in 50000

AP
Chest
<0.001–0.001
<1 in 12500000–1 in 1250000
<1 in 200000000–1 20000000

LAT
Chest
<0.001–0.01
<1 in 12500000 to 1 in 1250000
<1 in 200000000–1 20000000

PA
Chest
<0.001–0.01
<1 in 12500000 to 1 in 1250000
<1 in 200000000–1 20000000

PA
Colon
1.66-2.20
1 in 10000–1 in 5000
1 in 125000–1 in 100000

Hip joint
0.01–0.05
1 in 1250000–250000
1 in 20000000–1 in 3300000

AP
Lumbar spine
0.11–4.42
1 in 110000–1 in 2500
1 in 1700000–1 in 50000

PA
Lumbar spine
0.02–0.80
1 in 500000–1 in 17000
1 in 10000000–1 in 250000

LAT
Lumbar spine
0.03–3.89
1 in 500000–1 in 3300
1 in 5000000–1 in 50000

Lumbosacral joint
0.02–0.80
1 in 500000–1 in 17000
1 in 10000000–1 in 250000

AP
Pelvis
0.18–1.07
1 in 100000–1 in 11000
1 in 1100000–1 in 200000

AP
Stomach
0.01–1.33
1 in 1250000–1 in 10000
1 in 20000000–1 in 140000

LAO
Stomach
<0.01–0.01
<1 in 1250000–1 in 1250000
<1 in 20000000–1 in 20000000

LAT
Stomach
<0.01–0.01
<1 in 1250000–1 in 1250000
<1 in 20000000–1 in 20000000

LPO
Stomach
<0.01–0.01
<1 in 1250000–1 in 1250000
<1 in 20000000–1 in 20000000

PA
Stomach
<0.01–0.01
<1 in 1250000–1 in 1250000
<1 in 20000000–1 in 20000000

AP
Thoracic Spine
<0.001–0.85
1 in 12500000–1 in 14000
<1 in 200000000–1 in 250000

LAT
Thoracic Spine
<0.001
<1 in 12500000
<1 in 200000000

Barium enema
1.14–16.27
1 in 11000–1 in 1000
1 in 170000–1 in 12500

Barium meal
0.08–0.19
1 in 170000–1 in 50000
1 in 2500000–1 in 1000000

Cholangiography
1.57–6.65
1 in 10000–1 in 2000
1 in 125000–1 in 33000

Barium follow through
0.17–0.64
1 in 100000–1 in 20000
1 in 1100000–1 in 330000

IVU
0.05–1.33
1 in 250000–1 in 10000
1 in 3300000–1 in 140000

KUB
0.01–0.59
1 in 1250000– 1 in 20000
1 in 20000000–1 in 330000

AP
Urinary bladder
1.49
1 in 10000
1 in 140000

Nephrostomy
<0.01–0.37
1 in 1250000–33000
<1 in 20000000–1 in 500000

CT-Abdomen
1.04–3.65
1 in 12500–1 in 3300
1 in 200000–1 in 50000

CT-Chest
0.02
1 in 500000
1 in 10000000

CT-Lumbar spine
13.61–21.85
1 in 1000–1 in 500
1 in 14000–1 in 10000

CT-Pelvis
1.32–17.06
1 in 10000–1 in 1000
1 in 140000–1 in 11000

Table 6: Foetal dose-averaged and risk of childhood cancers and hereditary effects for the 367 foetuses exposed during diagnostic radiology examinations of their mothers.

Foetal dose-averaged
No. of foetuses
Risk of childhood cancer
Risk of hereditary effects

<0.1
166
<1 in 125000
<1 in 2000000

0.1–<1
119
1 in 125000–<1 in 12500
1 in 2000000–<1 in 200000

1–<10
66
1 in 12500–<1250
1 in 200000–<1 in 20000

10–<50
16
1 in 1250–<1 in 250
1 in 20000–<4000

Figure 1: The categorization of the total number of foetuses (367) exposed in utero into various gestational age (weeks) intervals.

Figure 2: The categorization of the total number of foetus (367) exposed in utero into various absorbed dose intervals.

3.1. Foetal Radiation Dose

In this study, exposure parameters from 367 pregnant patients from five institutions in five counties were investigated to estimate the radiation dose and risk to the foetuses. The radiation doses received by the 367 foetuses ranges from less than 0.001 mGy from chest X-ray examination of the pregnant mother to 21.9 mGy from CT of lumbar spine examination (Tables 3 and 4). Some of the women had multiple examinations (2 to 4 different examinations performed on the same patient) and others had repeat examinations (2 to 3 repeated examinations on the same patient), thus contributing to increased foetal doses. According to Osei and Faulkner [11], foetal doses are influenced by the maternal size and foetal depth and size. As a consequence, simply looking up the average uterus dose for a specific examination may lead to underestimation or over estimation of the foetal dose. Foetal dose should therefore be estimated based on the actual technique parameters used for the examination and taking into account the foetal depth and size. However, if the AP thickness of the patient and the foetal depth are not known, an average value of 25 cm and 9 cm, respectively, are good approximations to use [9]. Data in Tables 3 and 4 shows that, for the same examination, the accrual foetal dose can vary significantly from hospital to hospital and from patient to patient depending on several factors including the imaging equipment, technique parameters, patient size, and foetal depth and size. About 78% (285) of the foetus investigated received doses less than 1 mGy, and about 4% (16) of them received radiation doses greater than 10 mGy.

3.2. Risks of Death, Malformation, and Mental Retardation

The principal deterministic effects of exposure of the developing embryo or foetus to ionising radiation are death, malformation, growth retardation, and mental impairment. The ICRP [1] has reviewed the risks of harmful tissue reactions and malformations after prenatal irradiation and concluded that no deterministic effects of practical significance would be expected to occur in humans below a dose of at least 100 mGy. The threshold dose [1, 6] for the induction of these deterministic effects following in utero exposure of the 367 foetuses all lie very well above the estimated foetal doses, with the maximum foetal dose being 21.9 mGy. There is therefore no risk of death, malformation, growth retardation, or mental impairment from properly conducted diagnostic X-ray examination(s) in this group. In comparison, for the natural course of ordinary pregnancy it is estimated that the general population’s total risk of spontaneous abortion, major malformation, mental retardation, and childhood malignancy is about 28.6 percent [23]. According to Brent [24], every healthy woman without personal or family history of reproductive or developmental problems begins her pregnancy with a 1 in 33 (3%) risk for birth defects and 1 in 7 (15%) risk for miscarriage.

3.3. Risk of Hereditary Disease

The risk of hereditary effects resulting from exposure of the embryo or foetus is considered to be directly proportional to the radiation dose and fairly independent of the stage of pregnancy after the first three to four weeks of gestation [20]. The probability of induced hereditary effect on the basis of the doses received by the foetuses (Tables 3 and 4) in this study shows that the risk to the individual foetuses ranged from less than 1 in 200000000 (5 × 10⁻⁹) from chest X-ray examination of the mother to 1 in 10000 (1 × 10⁻⁴) from lumbar spine CT examination (Table 5). The majority (78%) of the foetuses carry risk less than 1 in 200000 (5 × 10⁻⁶) (Table 6). The highest foetal dose of 21.9 mGy in this study will carry a risk of hereditary effect of about 1 in 10000 (1 × 10⁻⁴). The risks values are very low compared to the natural incidence. The natural frequency of genetic disease manifested at birth in human populations has been estimated to be in the range 1 in 100 (1 × 10⁻²) to about 1 in 17 (6 × 10⁻²) if minor congenital abnormalities are included [6, 11, 20]. Thus, the increased genetic risk of 1 in 10000 (1 × 10⁻⁴) for an individual foetus associated with the highest dose in this study is very small compared with the natural risk of genetic disease.

3.4. Risk of Childhood Cancer

The risk of childhood cancer resulting from exposure of the embryo or foetus is considered to be directly proportional to the radiation dose and fairly independent of the stage of pregnancy after the first three to four weeks of gestation [20]. On the basis of the doses received by the foetuses, the risk of childhood cancer ranges from less than 1 in 12500000 (8 × 10⁻⁸) to about 1 in 500 (2 × 10⁻³). The risk of 1 in 500 (2 × 10⁻³) associated with the maximum absorbed dose of 21.9 mGy is very comparable to the natural baseline risk of childhood cancer (1 in 500 (2 × 10⁻³)) in the UK [20]. Foetal doses up to about 50 mGy (Table 6) could result in an approximate doubling of the natural baseline risk of childhood cancer. Therefore, those examinations resulting in foetal doses of some tens of mGy during pregnancy should be avoided unless the health of the mother (and indirectly that of the unborn child) would be compromised by delaying the examination until after the birth of the baby [20].

4. Conclusion

Exposure of the embryo or foetus to high dose ionising radiation can potentially lead to some adverse health effects. However, the radiation dose to the embryo or foetus that is likely to result from any diagnostic procedures should present no risk of causing any deterministic effects such as foetal death, malformation, growth retardation, or mental retardation. Data from this study indicate that foetal doses from properly conducted diagnostic radiology examination will not result in any deterministic effects. Furthermore, the foetal doses presented in this study shows a negligible risk of causing radiation induced hereditary effect in the descendants of the unborn child compared to the natural frequency of genetic disease manifested at birth in human populations. Despite that, all radiological examinations should be clinically justified and the foetal dose kept to a minimum consistent with the diagnostic requirements. It should be noted that doses from some diagnostic radiology examinations may present a very low risk of childhood cancer compared to the natural baseline risk but are certainly not sufficient to justify termination of pregnancy taking into account the health of the mother. It is therefore recommended that [20], any radiological examinations that can result in relatively high foetal doses of some tens of mGy should be avoided on pregnant women, if that can be achieved without serious detrimental effects to their health.

Acknowledgments

The authors would like to acknowledge with much gratitude the following people for collecting and supplying the diagnostic radiology examination parameters, and the foetal and maternal parameters for this study: Elin Rotstigen and Marius Ackenhausen in Norway, Dave Rawlings in England, Paulo R. Costa in Brazil, and Mark Smekhov in Israel.

References

International Commission on Radiological Protection (ICRP), “The 2007 Recommendations of the ICRP,” ICRP Publication 103, Elsevier, 2007, Annals of the ICRP.

International Commission on Radiological Protection (ICRP), “1990 Recommendations of the ICRP,” ICRP Publication 60, Pergamon Press, Oxford, UK, 1991, Annals of the ICRP 21, no. 1–3.

International Commission on Radiological Protection, “Radiological protection and safety in medicine,” ICRP Publication 73, Pergamon Press, Oxford, UK, 1990, Annals of the ICRP 26.

National Radiological Protection Board, “Statement by the National Radiological Protection Board: diagnostic medical exposures: advice on exposure to ionizing radiation during pregnancy,” Documents of the NRPB, vol. 4, no. 4, pp. 1–3, 1993.

National Radiological Protection Board, “Diagnostic medical exposures: exposure to ionizing radiation of pregnant women: biological Basis of the Board's Statement,” Documents of the NRPB, vol. 4, no. 4, pp. 7–14, 1993.

National Radiological Protection Board, “Estimates of late radiation risks to the UK population. Chapter 6: Irradiation in utero,” Documents of the NRPB, vol. 4, pp. 105–125, 1993.

K. L. Wagner, R. G. Lester, and L. R. Saldana, Exposure of the Pregnant Patient to Diagnostic Radiations: A Guide to Medical Management, Medical Physics Publishing, Madison, Wis, USA, 1997.

E. K. Osei, J. B. Darko, K. Faulkner, and C. J. Kotre, “Software for the estimation of foetal radiation dose to patients and staff in diagnostic radiology,” Journal of Radiological Protection, vol. 23, no. 2, pp. 183–194, 2003.

E. K. Osei and K. Faulkner, “Fetal position and size data far dose estimation,” British Journal of Radiology, vol. 72, pp. 363–370, 1999.

E. K. Osei and C. J. Kotre, “Equivalent dose to the fetus from occupational exposure of pregnant staff in diagnostic radiology,” British Journal of Radiology, vol. 74, no. 883, pp. 629–637, 2001.

E. K. Osei and K. Faulkner, “Fetal doses from radiological examinations,” British Journal of Radiology, vol. 72, pp. 773–780, 1999.

D. Hart, D. C. Jones, and B. F. Wall, “Normalized organ doses for medical x-ray examinations calculated using Monte Carlo techniques,” NRPB SR262, NRPB, Madison, Wis, USA, 1991.

M. Rosenstein, Handbook of Selected Organ Doses for Projections Common in Diagnostic Radiology, Department of Health, Education and Welfare, Rockville, Md, USA, 1976.

IPSM, National Protocol for Patient Dose Measurements in Diagnostic Radiology, Dosimetry Working Party of the Institute of Physical Sciences in Medicine (IPSM), New York, NY, USA, 1992.

C. J. Martin, B. Farquhar, E. Stockdale, and S. MacDonald, “A study of the relationship between patient dose and size in paediatric radiology,” British Journal of Radiology, vol. 67, no. 801, pp. 864–871, 1994.

C. L. Chapple, D. A. Broadhead, and K. Faulkner, “A phantom based method for deriving typical patient doses from measurements of dose-area product on populations of patients,” British Journal of Radiology, vol. 68, no. 814, pp. 1083–1086, 1995.

D. G. Jones and P. C. Shrimpton, “Normalized organ doses for X-ray CT calculated using monte Carlo techniques,” NRPB Software Report 250, NRPB, 1993.

E. Angel, C. V. Wellnitz, M. M. Goodsitt et al., “Radiation dose to the fetus for pregnant patients undergoing multidetector CT imaging: Monte carlo simulations estimating fetal dose for a range of gestational age and patient size1,” Radiology, vol. 249, no. 1, pp. 220–227, 2008.

S. K. Doshi, I. S. Negus, and J. M. Oduko, “Fetal radiation dose from CT pulmonary angiography in late pregnancy: a phantom study,” British Journal of Radiology, vol. 81, no. 968, pp. 653–658, 2008.

Health Protection Agency (HPA), “Protection of pregnant patients during diagnostic medical exposures to ionising radiation: advice from the health protection agency,” Documents of the Health Protection Agency: Radiation, Chemical and Environmental Hazards RCE 9, The Royal College of Radiologists and the College of Radiographers, 2009.

M. Otake, W. J. Schull, Y. Fujikoshi, and H. Yoshimaru, “Effect on school performance of prenatal exposure to ionising radiation in Hiroshima,” Tech. Rep. RERF TR2-88, Radiation Effects Research Foundation, Hiroshima, Japan, 1988.

M. Otake, W. J. Schull, and H. Yoshimaru, “A review of forty-five years study of Hiroshima and Nagasaki atomic bomb survivors. Brain damage among the prenatally exposed,” Journal of Radiation Research, vol. 32, supplement, pp. 249–264, 1991.

R. L. Brent, “Utilization of developmental basic science principles in the evaluation of reproductive risks from pre- and postconception environmental radiation exposures,” Teratology, vol. 59, no. 4, pp. 182–204, 1999.

R. Brent, Pregnancy and Radiation Exposure, Health Physics Society, 2012, http://hps.org/hpspublications/articles/pregnancyandradiationexposureinfosheet.html.