Advances in High Energy Physics
Volume 2016 (2016), Article ID 8561743, 8 pages
http://dx.doi.org/10.1155/2016/8561743

Research Article

Analytic Approximation of Energy Resolution in Cascaded Gaseous Detectors

Dezső Varga

Institute for Particle and Nuclear Physics, Wigner Research Centre for Physics of the HAS, 29-33 Konkoly-Thege Miklós Street, Budapest 1121, Hungary

Received 3 February 2016; Accepted 28 March 2016

Academic Editor: Ming Liu

Copyright © 2016 Dezső Varga. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited. The publication of this article was funded by SCOAP³.

Abstract.

An approximate formula has been derived for gain fluctuations in cascaded gaseous detectors such as Gas Electron Multipliers (GEMs), based on the assumption that the charge collection, avalanche formation, and extraction steps are independent cascaded processes. In order to test the approximation experimentally, a setup involving a standard GEM layer has been constructed to measure the energy resolution for 5.9 keV gamma particles. The formula reasonably traces both the charge collection and the extraction process dependence of the energy resolution. Such analytic approximation for gain fluctuations can be applied to multi-GEM detectors where it aids the interpretation of measurements as well as simulations.

1. Introduction

The energy resolution of gaseous detectors, that is, the precision with which a specific energy deposit can be measured, is a fundamental parameter and indication of instrumental quality. From the broad literature of gaseous detectors, such as classical [1] or recent review documents [2, 3], one can understand the key contributing factors to the energy resolution. The fluctuation from the ionization process defines the “intrinsic” limit, quantified by the “Fano factor.” The main contribution however stems from the fluctuation of the amplification of single electrons: as avalanches are formed, the amplitude (detected signal) fluctuates.

The energy resolution formula includes the single avalanche fluctuation: the signal amplitude , usually expressed in units of input electrons, will have the following relative precision:

Here the key is , the single electron avalanche fluctuation, defined as the standard deviation relative to the mean value. depends not only on gas composition but on amplification geometry. The energy resolution scales with the primary electron number from the ionization, whereas instrumental effects further broaden the response.

Assuming that an avalanche process (which can be quite complicated) initiated by a single electron finally results in detected electrons with probability of , one can express . The mean gain is the mean value of the probability distribution. If the standard deviation is , the relative fluctuation is defined as . As an example, if the electron number distribution is purely exponential, then .

Experimental and theoretical estimates for have been a subject for many studies. For proportional chambers, Alkhazov predicts around 0.8 [4], depending on gain. A classical calculation was proposed by Legler [5]. Micropattern Gaseous Detectors (MPGDs) perform considerably better in energy resolution, due to the smaller values: this is true for GEMs (Gas Electron Multipliers) [6], whereas micromesh detectors are particularly outstanding [7] in this respect.

For GEM detectors involving multiple amplification layers, is referring to the whole process of the cascaded amplification. The electrons undergo various steps until the avalanche reaches the readout anode: they can be lost before entering the GEM holes or may end up on the other side of the GEM, following the field lines. The question addressed in this paper is as follows: can we approximate for a cascaded process? How can we include collection and extraction (in) efficiencies? This question not only is related to detector physics, but has relevance for practical instrument design, where simulations and measurements have a very broad range of parameters. Direct calculation can help in finding guidelines and assist search for optimized designs. We will find that the approach can be applied to any cascaded gaseous detectors, including hybrids of different MPGD types.

2. Collection and Extraction from a GEM Layer

The avalanche process in a GEM detector takes place in the high field of the GEM holes. The focusing field geometry guides the electrons to the holes, whereas the field lines which emerge from the hole extract a certain fraction of the electrons [8]. The situation is shown in the cartoon of Figure 1. In this sense the avalanche formation can be separated in four steps:

(1)Collection: the (single) electron from an ionization process drifts towards the GEM, being focused to a hole after leaving the approximately homogeneous drift region with field strength of . As some field lines end up on the GEM, the probability of entering the hole is . This means that fraction of the electrons is lost for the subsequent amplification.(2)Avalanche: those electrons which arrive in the high field region undergo a Townsend-avalanche. The total number of electrons coming out of the avalanche is , with relative fluctuation of .(3)A fraction of electrons is extracted: with a probability of , depending strongly on the transfer field below the GEM, the electron leaves the vicinity of the hole and drifts towards the subsequent gain stages. Actually most of the electrons ( fraction) end up on the bottom of the GEM.(4)The extracted electrons will be further amplified leading to signal formation in the detector. For this step, the relative fluctuation will be denoted by .

Figure 1: Cartoon for avalanche amplification processes in GEM. Blue dots represent electrons, which follow the field lines.

The effective gain of the GEM, that is, the ratio of the electrons extracted relative to the number of those which arrived from the drift region, is . Evidently the same notation may be used for GEMs which are in the intermediate steps of a cascade, and one can even use this formulation for a broad range of gaseous detectors.

The question arises: can we determine the gain fluctuation for the complete process, ()–(), knowing the two probabilities and , the intrinsic gain fluctuation , and the fluctuation from the subsequent stages ?

3. Fluctuation of a Cascaded Processes

For cascaded process, one can calculate the fluctuation directly, derived from the probability distributions of each step. This section gives a reasonable mathematical formulation which is useful to understand the specific result. Let us assume two cascaded steps. First, from a single electron, one creates electrons with probabilities of . From each of these electrons, the second step creates electrons with probabilities of . The cascading leads to electrons, as depicted in Figure 2. The mathematical question is the following: how can we calculate fluctuation of the combination, from the fluctuations of each of the steps and ?

Figure 2: Cartoon for probability distributions in cascaded processes.

Let us denote the probability distribution of the combined process as . The fluctuation (RMS/mean, as already introduced) from the second process is , where and denote the first and second moments; a similar notation will be used for the first and combined processes, and , resulting in their respective fluctuations of and .

In case that two electrons are emitted from the first process, the number of electrons after the second process is the convolution of the respective probability distribution, , with itself:

whereas if electrons are emitted from the first process, the -fold convolution of the distribution, denoted by , is to be taken. The latter contains multiple sums and is most easily expressed as an iterative formula

In the general case, the first process emits electrons with probability distribution of ; therefore can be directly calculated as follows:

Direct evaluation of the fluctuation of , that is, , gives the key equation for the results in the present paper:

where is the mean number of electrons from the first process. For completeness, one can also derive an expected result, stating that : that is, the mean number of electrons from the cascaded process is the product of the mean electrons from the individual steps.

Equation (5) states that the cascaded process inherits its fluctuation from the first step and increased with adding in quadrature the fluctuation of the second step, suppressed by the gain of the first step. The formula resembles, but is distinct from, the summing of independent variables. Equation (5), which is conveniently simple, can be used to evaluate the cascaded gain fluctuation.

4. Fluctuation from Specific Processes in GEMs

The collection, avalanche, and extraction steps can be seen formally as a cascaded statistical process; therefore the formulation introduced can be applied. One needs however to directly calculate the corresponding fluctuations.

4.1. Collection

Starting out from a single electron, the probability of capturing or losing the electron is and , respectively. This means that the probability distribution is

The mean is certainly , whereas the fluctuation .

4.2. Avalanche

The mean number of electrons created in the avalanche, as introduced earlier, is , whereas the fluctuation of the avalanche is denoted by . We take both values as parameters.

4.3. Extraction

The fluctuation from the extraction can be calculated similarly to the collection process. Assuming a single electron just leaving the avalanche region, it is extracted to the transfer region with probability ; that is,

Again, the mean of the probability distribution is , whereas its fluctuation .

4.4. Formula for GEMs

Having the individual processes, one can set up the cascade steps ()–() to calculate the total fluctuation. For collection (), we have

The next step of the cascade is avalanche (), which is to be combined with the first step according to (5). Substituting , , and , one arrives at

Combining steps , one gets

where is the mean electron number after the first two steps. This last formula is actually a single GEM contribution. And finally the complete process, all steps , including the contribution from all subsequent stages, is as follows:

Having introduced the effective gain , one can rewrite the formula in the following convenient way:

In this formula, the first term of the sum contains collection, whereas the second term, being suppressed by , contains extraction. The surprisingly simple formula gives an easy access to experimental comparison.

5. Measurements of Energy Resolution at 5.9 keV

5.1. Experimental Setup

In order to obtain a systematic and consistent measurement set, a simple experimental setup has been constructed, including a single GEM layer and an MWPC as a high gain stage. The setup is shown in Figure 3, indicating the detector geometry. The GEM is a standard CERN product, 10 cm by 10 cm in size, with 140-micron pitch, 70-micron hole diameter, and 60-micron thickness. For the MWPC, a version of a “Close Cathode Chamber” [9] has been applied.

Figure 3: Detector outline, with a single GEM and an MWPC as a high gain stage.

The measurements have been performed in two gas mixtures, at 80 : 20 and at 90 : 10 proportions. Using an ⁵⁵Fe isotope as X-ray source for 5.9 keV gamma particles, examples of amplitude spectra obtained by the apparatus are shown in Figure 4. One can clearly observe the expected structures: the main 5.9 keV peak, the escape peak in Ar, and also the peak corresponding to conversions below the GEM (i.e., only amplified by the MWPC). Figures 4 and 5 show Gaussian fits as examples, demonstrating that clear evaluation of the energy resolution is achievable with the presented setup.

(a)

(b)

Figure 4: Example of a 5.9 keV peak in the detector, in argon-based gas mixture. The GEM effective gain is different in the two cases, 1.8 for (a) and 7 for (b). The main peak, amplified by the GEM, is marked as (M1), whereas the escape peak (E1) is halfway down. The conversion below the GEM results in the main peak (M2) at lower amplitude, with the corresponding escape peak (E2).

(a)

(b)

Figure 5: Examples of a 5.9 keV peak in the detector, in neon-based gas mixture. The main peaks with GEM amplification (M1) and only with MWPC amplification (M2) are apparent, at two different GEM effective gains.

The GEM effective gain can be directly evaluated by taking the ratio of the peak position from gamma conversion above and below the GEM. The effective gain as a function of GEM voltage in the various gases is shown in Figure 6. The measurements were taken at two different transfer fields, value of 0.4 kV/cm and 1 kV/cm.

Figure 6: Effective gain as a function of GEM voltages for the various measurements.

5.2. Resolution versus Drift: Access to Collection Process

The collection efficiency depends on the drift field [8]. The collection efficiency can be approximated by measuring the gain as a function of the transfer field, shown in Figure 7. One clearly observes a plateau up to about 1.6 kV/cm, which is a starting point of a gradual decrease: here can be estimated as the gain change relative to the plateau. The resolution, shown in Figure 7(b), features a corresponding structure: an increase (broadening) of the energy resolution.

(a)

(b)

Figure 7: The effective gain (a) and the energy resolution (b) dependence on the drift field.

5.3. Resolution versus Gain: Access to the Extraction Process

The energy resolution of the 5.9 keV peak has been evaluated for the two transfer field values, at fixed drift field of 0.8 kV/cm (i.e., safely on the collection plateau with ), as a function of GEM effective gain. The results, shown in Figure 8, show a decreasing trend for both Ar and Ne. It is indicative to observe that the resolution is nearly the same for the different transfer fields: indeed from (12), one expects that the effective gain, , is the most relevant parameter in the second term of the formula.

(a)

(b)

Figure 8: The measured energy resolution (RMS) as a function of effective gain. Note the improvement of the resolution with gain, both for the argon-based mixture (a) and for the neon-based mixture (b).

6. Predictive Power of the Analytic Formula

In this section, the quantitative comparison of the energy resolution will be given between that predicted by (12) and the measurements. Since this paper is concerned more about the formulation and trends, rather than the specific parameters, some assumptions are needed. First, for (1), the Fano factor will be set to 0.2 [3], whereas the primary electron number for and mixture is assumed to be 210 and 165, respectively [3]. , that is, the gain fluctuation in the MWPC stage, is assumed to be (pure exponential), which is reasonable for high gain MWPCs [4]. The extraction efficiency is well below 1, and various values between 0 and 0.2 will be assumed, values which are based on experimental observations [8] for the applied transfer fields. An important and actually the most interesting figure from the point of view of detector physics is , the intrinsic avalanche fluctuation of GEMs. In this paper, we will tentatively use 0.9 for and 0.8 for figures which are motivated by simulations [6] and will be subject of our own detailed later studies as well. The reason why is smaller than (or, equivalently, that GEM detectors have potentially better energy resolution than MWPCs) can be understood by the classical argument from Legler [5]: in GEMs, the field is strongly inhomogeneous at the early phase of the avalanche development, leading to suppression of small avalanches.

The energy resolution calculation requires the instrumental fluctuation ; this is now set to 2%, which is estimated from the electronics and digitization noise and is therefore a lower limit. We will observe, however, that the various parameters are strongly correlated as seen in the formula, and most of the qualitative predictions will not depend on the specific parameter settings.

6.1. Energy Resolution Decrease with Collection Loss

The most prominent feature of the resolution broadening pattern is the correlation with loss of collection. From Figure 7 one can estimate the collection fraction by dividing the actual gain with the plateau gain. Using this value, the measured energy resolution as a function of is shown in Figure 9. Comparing this to the prediction from (12), shown as the estimated function, a reasonable agreement is apparent. The intuitive guess would be proportionality to (if half of the charge is collected, the fluctuation increases by ), whereas the measurements are considerably higher. One can note that the data is actually higher than the prediction (12): this may be due to a correlated loss of the charge cluster, that is, if the assumption that the avalanche processes are independent (leading to (5)) is not exactly correct. In conclusion, the predicted decrease of the resolution is well corresponding to the actual measurements.

Figure 9: The measured energy resolution at 5.9 keV, in , as a function of the estimated collection efficiency. The curves show the prediction by (12), as well as a proportionality to .

6.2. Energy Resolution Decrease with Gain

The formula, in case of (full collection), predicts dependence on , with other parameters being not so relevant individually (such as , or increase of with GEM voltage). According to (12), one should get a linear dependence of the square of the width on . The corresponding measurement, along with the prediction from (12) and (1), is shown in Figure 10.

(a)

(b)

Figure 10: The measured energy resolution (RMS) as a function of .

For both Ne and Ar based gases, the prediction gives a correct slope against and indicates that the gain indeed follows the rule from (12). Figure 10 shows the prediction for various values of , which means that the linear behaviour (and the steepness of the slope) is largely independent of the specific value.

7. Conclusions

Interpreting the avalanche formation in GEMs as a statistical process, one can calculate the gain fluctuation from the individual step fluctuations. This includes charge collection, avalanche development, extraction, and subsequent amplification. The final fluctuation, directly related to energy resolution, can be expressed as a formula represented by (12). Two steps, the collection and the extraction, have been studied experimentally. Estimating the collection efficiency from the gain loss at varying drift fields, the energy resolution can be reasonably predicted for 5.9 keV gamma rays. Similarly, in case of varying extraction field, a specific dependence of the resolution on is predicted, which is supported by the measurements. The approximate formula for energy resolution can be used as a basis for estimating figure of merits for gaseous detectors, especially when energy resolution worsening will balance improvement in other parameters such as ion blocking or overall gain. The formulation also helps to understand the avalanche formation, as a statistical process in cascaded gaseous detectors, and thus simplifies the interpretation of experimental or simulated data.

Competing Interests

The author declares that there are no competing interests.

Acknowledgments

This work has been supported by the Momentum Grant of the Hungarian Academy of Sciences (LP2013-60). The author wishes to acknowledge the support from the REGARD group members at the Wigner Research Centre in Budapest.

References

W. Blum and L. Rolandi, Particle Detection with Drift Chambers, Springer, Berlin, Germany, 1993.

F. Sauli, Gaseous Radiation Detectors : Fundamentals and Applications, vol. 36 of Cambridge Monographs on Particle Physics, Nuclear Physics and Cosmology, 2014.

T. Francke and V. Peskov, Innovative Applications and Developments of Micro-Pattern Gaseous Detectors, Advances in Chemical and Materials Engineering (ACME), Engineering Science Reference, Hershey, Pa, USA, 2014.

G. D. Alkhazov, “Statistics of electron avalanches and ultimate resolution of proportional counters,” Nuclear Instruments and Methods C, vol. 89, pp. 155–165, 1970.

W. Legler, “The influence of the relaxation of the electron energy distribution on the statistics of electron avalanches,” British Journal of Applied Physics, vol. 18, no. 9, p. 1275, 1967.

H. Schindler, Microscopic simulation of particle detectors [Ph.D. dissertation], T. U. Wien, Vienna, Austria, CERN-THESIS-2012-208, 2012.

T. Zerguerras, B. Genolini, V. Lepeltier, J. Peyré, J. Pouthas, and P. Rosier, “Single-electron response and energy resolution of a Micromegas detector,” Nuclear Instruments and Methods in Physics Research, Section A: Accelerators, Spectrometers, Detectors and Associated Equipment, vol. 608, no. 3, pp. 397–402, 2009.

M. Killenberg, S. Lotze, J. Mnich et al., “Modelling and measurement of charge transfer in multiple GEM structures,” Nuclear Instruments and Methods in Physics Research, Section A: Accelerators, Spectrometers, Detectors and Associated Equipment, vol. 498, no. 1–3, pp. 369–383, 2003.

D. Varga, G. Kiss, G. Hamar, and G. Bencédi, “Close cathode chamber: low material budget MWPC,” Nuclear Instruments and Methods in Physics Research A, vol. 698, pp. 11–18, 2013.