Advances in Orthopedics
Volume 2018 (2018), Article ID 2735634, 5 pages
https://doi.org/10.1155/2018/2735634

Research Article

Outcome after 52 Salto Ankle Prostheses Implanted by a Single Surgeon

Frank W. M. Faber,¹ Monique J. L. Mastboom,² Sabine T. van Vliet-Koppert,² Ilse C. E. Bouman,³ and Paulien M. van Kampen¹

¹Department of Orthopaedic Surgery, Bergman Clinics, Rijswijk, Netherlands
²Department of Orthopaedic Surgery, LUMC, Leiden, Netherlands
³Department of Orthopaedic Surgery, HAGA Hospital, The Hague, Netherlands

Correspondence should be addressed to Frank W. M. Faber; f.faber@bergmanclinics.nl

Received 5 February 2018; Revised 2 July 2018; Accepted 10 July 2018; Published 1 August 2018

Academic Editor: Panagiotis Korovessis

Copyright © 2018 Frank W. M. Faber et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

Abstract.

While ankle arthrodesis was traditionally the gold standard method of treatment for disabling end-stage ankle arthritis, total ankle replacement (TAR) has been an acceptable alternative. The satisfaction rate of patients with TAR however differs. The purpose of our study is to investigate whether implant survival and results with special emphasis on the satisfaction rate of patients treated with a TAR implanted by a single surgeon were comparable to the literature. This was a retrospective cohort study in a teaching hospital. Data was collected from 52 patients who received a total ankle replacement (TAR) between 05/2002 and 06/2014. The mean follow-up time was 4.2 years (95% CI 3.3 – 5.0). Results showed a high satisfaction rate of 94% and 94% survival of the TAR after 5 years. We conclude that TAR with the Salto prosthesis is, in our hands, a reliable solution for end-stage ankle arthritis, with results comparable to the literature.

1. Introduction

While ankle arthrodesis was traditionally the gold standard method of treatment for disabling end-stage ankle arthritis, total ankle replacement (TAR) has been an acceptable alternative [1–4]. Drawbacks of ankle fusion are nonunion, quite long postoperative immobilization, and in the long-term arthritis in the adjacent joints [5, 6]. Disadvantages of TAR are the technical difficulty of the procedure, causing a considerable learning curve [7–9] and uncertain long-term survival of the implant [3, 10, 11]. The satisfaction rate of patients with TAR also differs. Conflicting reports exist about the cause for this difference in satisfaction. Spirt et al. [12] and Henricson et al. [13] found a worse result in younger patients, focusing on implant survival, complication rates, and revision rates. Tenenbaum et al. [14] and Dematracopoulous et al. [15] reported equal results in different age groups, considering clinical outcomes, gait improvement, and patient reported outcome. Schenk et al. [16] reported 6.9% unsatisfied patients, mostly because of conversion of the prosthesis to an arthrodesis. Other factors influencing implant survival, and with that patient satisfaction, are preoperative alignment, body weight, and proper operative technique. Furthermore, management of the expectations of the patients and explicit information in the preoperative phase is of influence [17]. The purpose of our study is to investigate whether implant survival and results with special emphasis on the satisfaction rate of our patients treated with a TAR were comparable to the literature.

2. Patients and Methods

2.1. Study Design and Setting

This was a retrospective cohort study. Data was collected from patients who received a total ankle replacement (TAR) between 05/2002 and 06/2014 in the HAGA Hospital, the Hague, in the Netherlands. All surgeries were performed by the same Orthopedic Surgeon specialized in foot and ankle surgery (FF). The study was enlisted by the Medical Ethical Committee Southwest Holland (15-103) and was declared not to subdue to Medical Research Involving Human Subjects Act. The board of directors of the hospital approved the study.

2.2. Participants

Patients who received a TAR in the previous mentioned time window were included in the study. Patients with the following criteria were included: symptomatic ankle osteoarthritis (Kellgren-lawrence score of 3+ or 4 despite conservative treatment), age≥ 65 years (except rheumatoid patients), BMI ≤ 30, American Society of Anesthesiologists (ASA) stage 1, 2, or 3, adequate bone-quality, ≤10° varus or valgus deformity, and no (extreme) sport wish. Exclusion criteria for a TAR were neurological disorders, bad peripheral circulation, osteonecrosis of talus or tibia, and infection present. The operation was performed under spinal or general anesthesia. Half an hour prior to surgery and 24 hours postoperatively intravenous antibiotics were administered. Standard anterior approach was used and an uncemented SALTO® prosthesis (Tornier SA, Saint Ismier, France) was placed (a third-generation ankle prosthesis, three components, and mobile bearing). Postoperative treatment consisted of a lower leg cast: first 2 weeks non-weight-bearing followed by 2 weeks full-weight-bearing. After cast removal, an intensive physical therapy program was followed for at least 6 weeks. The questionnaires were distributed with a minimum of 1 year follow-up and the postoperative X-rays were made the first day after surgery and 4 weeks postoperatively. The follow-up X-rays were performed with AP and lateral weight-bearing X-rays on a 2-yearly routine base.

2.3. Variables

Patients’ satisfaction was measured by asking one question: “would you choose the TAR again in a similar situation?”

Revision of the TAR was defined as any secondary operation in which a prosthetic component was exchanged and/or conversion to an arthrodesis had to be performed, for any reason.

Functional outcome was measured by the Foot Function Index (FFI) [18]. We used the validated Dutch Version using verbal rating scales provided by Kuyvenhoven et al. [19]. The second outcome measure was the American Orthopaedic Foot and Ankle Society (AOFAS) questionnaire [20].

2.4. Measurement

Radiological outcome for the position of the prosthesis components is measured in degrees on first weight-bearing postoperative X-ray compared to last postoperative X-ray (Figure 1). Measurements, calculated as angular deviation according to the perfect position, are performed as described by Valderrabano et al. [21].

Figure 1: Radiological angles. Perfect tibial position is defined as γ angle 90 degrees and in the sagittal plane position of the tibial component with an inclination till 7 degrees (α = +7; according to the manufacturer of the Salto prosthesis). Perfect talar position is placement of this component parallel to the sole of the foot (β angle 0 degrees). An angle-difference over time of 5 degrees or more is considered significant and is classified as migration. Malposition is defined as an angulation of a component of more than 10 degrees from the perfect position as defined by Schimmel [7].

Radiographic analysis were performed by one independent senior orthopaedic surgeon (SvVK).

2.5. Statistical Methods

Continuous baseline characteristics were presented as mean with 95% confidence interval between square brackets. Nominal data was presented in counts and percentages. Statistical analyses were conducted in SPSS version 17 (IBM Co., Armonk, NY, USA). For all statistical analyses, a two-tailed p-value less than 0.05 was considered to be statistically significant.

Patient satisfaction was presented in percentages. A Kaplan-Meier curve was presented for survival analysis with revision for any reason as endpoint. All functional outcomes were presented as mean with 95% confidence interval (CI) and the radiological outcomes as median and IQR.

3. Results

3.1. Participants

A total 52 patients were included in the study. The mean follow-up time was 4.2 years (95% CI 3.3 – 5.0, range 0.1 – 13.7).

All baseline characteristics are presented in Table 1.

Table 1: Baseline characteristics.

Age, mean
Years, range
70

Gender
Male
26

Female
26

Side
Left
19

Right
33

ASA
1
21

2
26

3
2

Missing
3

Smoking
No
44

Yes
7

Missing
1

Indication
Post-traumatic
29

Primary Osteoarthritis
18

Rheumatoid Arthritis
4

Rheumatoid Arthritis and Post-traumatic
1

3.2. Outcome

In total 48 of the 51 patients would choose the TAR again in a similar situation (94% satisfaction rate). One answer was missing, because we were unable to contact this patient. Two patients were dissatisfied with the TAR; one patient required a conversion to an arthrodesis for septic loosening and the other dissatisfied patient had persistent pain, but had good function and perfect radiological images. This TAR was converted to an ankle arthrodesis for suspicion of low grade infection, which was not confirmed. Pain persisted even after an uneventful solid ankle fusion with allograft.

Three of all patients had an ankle arthrodesis on the contralateral side. These patients all preferred their TAR over their ankle arthrodesis.

In Figure 2 the Kaplan-Meier curve is presented. The survival of the TAR after 5 years is 94% with 15 patients at risk.

Figure 2: Survival of the TAR in years. At 2,5 years 29 were at risk and at 5 years 15 were at risk. The overall survival time is 11 with a 95% confidence interval of 10-13.

The functional and radiological results are presented in Table 2. The mean total AOFAS score was 85 out of a maximum of 100 points. The mean total FFI was 36 out of a maximum of 115 (with 23 as lowest and best score).

Table 2

N

Range

FFI
Activity
45
Mean
18 [16 -20]
[9-43]

Pain
45
Mean
16 [14 – 18]
[9-39]

Restriction
45
Mean
8 [7 -9]
[5-16]

Total
45
Mean
42 [37 – 47]
[25-94]

AOFAS
Alignment
45
Mean
9 [8 – 10]
[0-10]

Function
45
Mean
41 [38 – 45]
[5-50]

Pain
45
Mean
34 [32 – 37]
[20-40]

Total
45
Mean
85 [80 – 90]
[26-100]

Radiological outcome
First alpha
49
Median [IQR]
6
[ -3 - 12]

Last alpha
47
Median [IQR]
5
[-3-14]

First beta
49
Median [IQR]
-4
[-15-9]

Last beta
47
Median [IQR]
-4
[-15-8]

First gamma
49
Median [IQR]
0
[-4-5]

Last gamma
47
Median [IQR]
0
[-5-7]

Not for all the 52 patients clinical and radiological data could be collected. The patients that were converted to an arthrodesis were excluded. In addition, we were unable to reach one patient (like mentioned above) and one patient was not willing to come to the hospital and only answered the satisfaction question on the phone. This explains the different numbers in the first column.

3.3. Complications

Five talar component malpositions were noted at postoperative weight-bearing X-rays. Four malpositions of the tibial part were discovered, of which one was treated with a reoperation within two weeks: repositioning of the tibial component leading to a satisfactory result afterwards. Preoperatively, 2 medial malleolus fractures and 1 lateral malleolus fracture occurred. All fractures were fixed during the ankle replacement procedure.

Three patients suffered from prolonged wound healing: more than 4 weeks postoperatively wound leakage was present: these all resolved with prolonged immobilization and antibiotics.

During the follow-up period in 4 cases, conversion to an arthrodesis was necessary. Two patients needed conversion because of aseptic loosening caused by multiple cysts; one patient suffered from severe wound problems and an acute deep infection. One conversion was performed because of persistent pain and suspicion of low grade infection, which was not confirmed during reoperation.

4. Discussion

TAR is known as a technically demanding procedure and a considerable learning curve is described by several authors [7–9]. Yet, equally functional results are described by Reuver et al. [22] in low volume centres: the AOFAS score of these authors had an average of 75 (SD ± 15). Our study, single surgeon and low volume, shows an even higher average AOFAS score of 85 (SD ± 17). This difference could be explained because the senior author of this series (FF) already had considerable experience in TAR with another type ankle prosthesis (STAR) before he started implanting the Salto prosthesis. The functional outcome of this study is comparable to other series that also described the FFI and/or the AOFAS score as an outcome parameter. The FFI found by Kerkhoff et al. [23] and Schimmel et al. [7] were 33 and 32, respectively. The AOFAS score varied from 73 to 85 [16, 24–27]. Bonnin, who is actually one of the inventers of the Salto prosthesis, described an AOFAS score of 79 points, with a follow-up of 7–11 years [28]. So, in our hands, the functional results are certainly not inferior.

In our study 48 of the 51 patients (94%) would have the TAR performed again in a similar situation, of whom some are even revised and converted to an arthrodesis. Yet, 2 of the 2 dissatisfied patients had a revision. So implant survival and satisfaction are definitely related. Implant survival can be improved by proper technique, for instance, component placement in proper alignment [3, 29]. Our survival rate is 94% after 5 years with 15 patients at risk. This is better than the results described by Henricsson et al. [13], that is, 78% survival at 5 years. One study found a 94% implant survival at 5 years, even 87% survival at 10 years, which dropped quite steep to 64% after 15 years [30]. One of our exclusion criteria for a TAR is age: we excluded patients younger than 65 years, in whom we prefer an arthrodesis, except for rheumatoid patients. These patients often have other joint impairments (hindfoot, midfoot), which probably benefit when the ankle joint is kept mobile instead of fused. Gait analysis showed a more normal gait in TAR patients than in patients who had an arthrodesis [31]. Although the age selection criterion seems justifiable in terms of long-term implant survival, reports in the literature are conflicting. Kofoed and Lundberg-Jensen [32] and Skyttä et al. [33] described no influence of age; Tenenbaum et al. [14] reported equal functional improvement in patients over 70 years patients and aged 50-60 years. In contrast with this, Henricson et al. [13] and Raikin et al. [34] all reported worse results in younger patients. Because of our high patient satisfaction and the good survival rate compared with the literature, we consider performing TARs in younger patients. We realize we have to inform this younger group about the risk of a conversion to an arthrodesis later in life.

The strong points of this study were the independent investigators who performed the patient investigations at follow-up, the high follow-up percentage, and the use of one and the same prosthesis, with an unchanged design by the same surgeon. A weak point is its retrospective design, so no preoperative clinical scores were available.

5. Conclusion

We conclude that TAR with the Salto prosthesis is, in our hands, a reliable solution for end-stage ankle arthritis, with results comparable to the literature.

Data Availability

The data used to support the findings of this study are available from the corresponding author upon request.

Disclosure

Level of clinical evidence is retrospective cohort study, Level 3.

Conflicts of Interest

The authors declare that there are no conflicts of interest regarding publication of this paper.

References

C. L. Saltzman, R. G. Kadoko, and J. S. Suh, “Treatment of isolated ankle osteoarthritis with arthrodesis or the total ankle replacement: a comparison of early outcomes,” Clinics in Orthopedic Surgery, vol. 2, no. 1, pp. 1–7, 2010.

J. Morash, D. M. Walton, and M. Glazebrook, “Ankle Arthrodesis Versus Total Ankle Arthroplasty,” Foot and Ankle Clinics, vol. 22, no. 2, pp. 251–266, 2017.

J. A. Mann, R. A. Mann, and E. Horton, “STAR™ ankle: Long-term results,” Foot & Ankle International, vol. 32, no. 5, pp. 473–484, 2011.

R. Zaidi, S. Cro, K. Gurusamy et al., “The outcome of total ankle replacement: A systematic review and meta-analysis,” The Bone & Joint Journal, vol. 95, no. 11, pp. 1500–1507, 2013.

L. M. Coester, C. L. Saltzman, J. Leupold, and W. Pontarelli, “Long-term results following ankle arthrodesis for post-traumatic arthritis,” The Journal of Bone & Joint Surgery, vol. 83, no. 2, pp. 219–228, 2001.

R. A. Mann and K. M. Rongstad, “Arthrodesis of the ankle: A critical analysis,” Foot & Ankle International, vol. 19, no. 1, pp. 3–9, 1998.

J. J. P. Schimmel, L. H. B. Walschot, and J. W. K. Louwerens, “Comparison of the short-term results of the first and last 50 scandinavian total ankle replacements: Assessment of the learning curve in a consecutive series,” Foot & Ankle International, vol. 35, no. 4, pp. 326–333, 2014.

A. Haskell and R. A. Mann, “Perioperative complication rate of total ankle replacement is reduced by surgeon experience,” Foot & Ankle International, vol. 25, no. 5, pp. 283–289, 2004.

F. G. Usuelli, C. Maccario, A. Pantalone, N. Serra, and E. W. Tan, “Identifying the learning curve for total ankle replacement using a mobile bearing prosthesis,” Journal of Foot and Ankle Surgery, vol. 23, no. 2, pp. 76–83, 2017.

A. Henricson, J.-Å. Nilsson, and Å. Carlsson, “10-year survival of total ankle arthroplasties: A report on 780 cases from the Swedish Ankle Register,” Acta Orthopaedica, vol. 82, no. 6, pp. 655–659, 2011.

G. Labek, H. Klaus, R. Schlichtherle, A. Williams, and M. Agreiter, “Revision rates after total ankle arthroplasty in sample-based clinical studies and national registries,” Foot & Ankle International, vol. 32, no. 8, pp. 740–745, 2011.

A. A. Spirt, M. Assal, and S. T. Hansen Jr., “Complications and failure after total ankle arthroplasty,” The Journal of Bone & Joint Surgery, vol. 86, no. 6, pp. 1172–1178, 2004.

A. Henricson, A. Skoog, and Å. Carlsson, “The Swedish Ankle Arthroplasty Register: An analysis of 531 arthroplasties between 1993 and 2005,” Acta Orthopaedica, vol. 78, no. 5, pp. 569–574, 2007.

S. Tenenbaum, J. Bariteau, S. Coleman, and J. Brodsky, “Functional and clinical outcomes of total ankle arthroplasty in elderly compared to younger patients,” Journal of Foot and Ankle Surgery, vol. 23, no. 2, pp. 102–107, 2017.

C. A. Demetracopoulos, S. B. Adams, R. M. Queen, J. K. Deorio, J. A. Nunley, and M. E. Easley, “Effect of Age on Outcomes in Total Ankle Arthroplasty,” Foot & Ankle International, vol. 36, no. 8, pp. 871–880, 2015.

K. Schenk, S. Lieske, M. John et al., “Prospective study of a cementless, mobile-bearing, third generation total ankle prosthesis,” Foot & Ankle International, vol. 32, no. 8, pp. 755–763, 2011.

E. Pinsker, T. Inrig, T. R. Daniels, K. Warmington, and D. E. Beaton, “Symptom Resolution and Patient-Perceived Recovery Following Ankle Arthroplasty and Arthrodesis,” Foot & Ankle International, vol. 37, no. 12, pp. 1269–1276, 2016.

E. Budiman-Mak, K. J. Conrad, and K. E. Roach, “The foot function index: A measure of foot pain and disability,” Journal of Clinical Epidemiology, vol. 44, no. 6, pp. 561–570, 1991.

M. M. Kuyvenhoven, K. J. Gorter, P. Zuithoff, E. Budiman-Mak, K. J. Conrad, and M. W. M. Post, “The Foot Function Index with verbal rating scales (FFI-5pt): A clinimetric evaluation and comparison with the original FFI,” The Journal of Rheumatology, vol. 29, no. 5, pp. 1023–1028, 2002.

H. B. Kitaoka, I. J. Alexander, R. S. Adelaar, J. A. Nunley, M. S. Myerson, and M. Sanders, “Clinical rating systems for the ankle-hindfoot, midfoot, hallux, and lesser toes,” Foot & Ankle International, vol. 15, no. 7, pp. 349–353, 1994.

V. Valderrabano, B. Hintermann, and W. Dick, “Scandinavian total ankle replacement: A 3.7-Year average followup of 65 patients,” Clinical Orthopaedics and Related Research, no. 424, pp. 47–56, 2004.

J. M. Reuver, N. Dayerizadeh, B. Burger, L. Elmans, M. Hoelen, and N. Tulp, “Total ankle replacement outcome in low volume centers: Short-term followup,” Foot & Ankle International, vol. 31, no. 12, pp. 1064–1068, 2010.

Y. R. A. Kerkhoff, N. M. Kosse, W. P. Metsaars, and J. W. K. Louwerens, “Long-term Functional and Radiographic Outcome of a Mobile Bearing Ankle Prosthesis,” Foot & Ankle International, vol. 37, no. 12, pp. 1292–1302, 2016.

A. Frigg, B. Nigg, L. Hinz, V. Valderrabano, and I. Russell, “Clinical relevance of hindfoot alignment view in total ankle replacement,” Foot & Ankle International, vol. 31, no. 10, pp. 871–879, 2010.

E. P. McConnell and R. M. Queen, “Correlation of Physical Performance and Patient-Reported Outcomes Following Total Ankle Arthroplasty,” Foot & Ankle International, vol. 38, no. 2, pp. 115–123, 2017.

J.-L. Besse, N. Brito, and C. Lienhart, “Clinical evaluation and radiographic assessment of bone lysis of the AES total ankle replacement,” Foot & Ankle International, vol. 30, no. 10, pp. 964–975, 2009.

T. Kraal, H. J. L. Van Der Heide, B. J. Van Poppel, M. Fiocco, R. G. H. H. Nelissen, and H. C. Doets, “Long-term follow-up of mobile-bearing total ankle replacement in patients with inflammatory joint disease,” The Bone & Joint Journal, vol. 95, no. 12, pp. 1656–1661, 2013.

M. Bonnin, F. Gaudot, J.-R. Laurent, S. Ellis, J.-A. Colombier, and T. Judet, “The salto total ankle arthroplasty: Survivorship and analysis of failures at 7 to 11 years,” Clinical Orthopaedics and Related Research, vol. 469, no. 1, pp. 225–236, 2011.

N. Espinosa, M. Walti, P. Favre, and J. G. Snedeker, “Misalignment of total ankle components can induce high joint contact pressures,” The Journal of Bone & Joint Surgery, vol. 92, no. 5, pp. 1179–1187, 2010.

H. Koivu, I. Kohonen, K. Mattila, E. Loyttyniemi, and H. Tiusanen, “Long-term Results of Scandinavian Total Ankle Replacement,” Foot & Ankle International, vol. 38, no. 7, pp. 723–731, 2017.

S. Singer, S. Klejman, E. Pinsker, J. Houck, and T. Daniels, “Ankle arthroplasty and ankle arthrodesis: gait analysis compared with normal controls,” The Journal of Bone & Joint Surgery, vol. 95, no. 24, pp. 2195–2202, 2013.

H. Kofoed and A. Lundberg-Jensen, “Ankle arthroplasty in patients younger and older than 50 years: A prospective series with long-term follow-up,” Foot & Ankle International, vol. 20, no. 8, pp. 501–506, 1999.

E. T. Skyttä, H. Koivu, A. Eskelinen, M. Ikävalko, P. Paavolainen, and V. Remes, “Total ankle replacement: A population-based study of 515 cases from the Finnish Arthroplasty Register,” Acta Orthopaedica, vol. 81, no. 1, pp. 114–118, 2010.

S. M. Raikin, K. Sandrowski, J. M. Kane, D. Beck, and B. S. Winters, “Midterm Outcome of the Agility Total Ankle Arthroplasty,” Foot & Ankle International, vol. 38, no. 6, pp. 662–670, 2017.