BioMed Research International
Volume 2015 (2015), Article ID 173872, 10 pages
http://dx.doi.org/10.1155/2015/173872

Clinical Study

Clinical Performance and Safety of 108 SpineJack Implantations: 1-Year Results of a Prospective Multicentre Single-Arm Registry Study

David Noriega,¹ Gianluca Maestretti,² Christian Renaud,³ Natale Francaviglia,⁴ Mourad Ould-Slimane,⁵ Steffen Queinnec,⁶ Helmut Ekkerlein,⁷ Frank Hassel,⁸ Rainer Gumpert,⁹ Pascal Sabatier,¹⁰ Hervé Huet,¹¹ Miguel Plasencia,¹² Nicolas Theumann,¹³ Alexander Kunsky,¹⁴ and Antonio Krüger¹⁵

¹Valladolid University Hospital, Royal Academy of Medicine and Surgery, 47008 Valladolid, Spain
²Department of Orthopedic Surgery, Spine Unit, HFR Fribourg-Cantonal Hospital, 1708 Fribourg, Switzerland
³Department of Orthopedics, Toulouse Lautrec Clinic, 81030 Albi, France
⁴Department of Neurosurgery, A.R.N.A.S. Civico Di Cristina Benfratelli, 90127 Palermo, Italy
⁵Department of Orthopedic Surgery, University Hospital of Rouen, 76031 Rouen, France
⁶Department of Orthopedic Surgery, Hospital Beaujon, 92110 Clichy, France
⁷Department of Trauma and Orthopedic Surgery, Südostbayern AG Clinic, Traunstein Clinic, 83278 Traunstein, Germany
⁸Department of Spine Surgery, Loretto Hospital, 79100 Freiburg, Germany
⁹LKH-Univ. Klinikum Graz, 8036 Graz, Austria
¹⁰Department of Neurosurgery, Clinic of Cèdres, 31700 Cornebarrieu, France
¹¹Department of Neuroradiology, Regional University Hospital, 14033 Caen, France
¹²Department of Traumatology and Orthopedic Surgery, University Hospital Principe De Asturias, 28805 Madrid, Spain
¹³Department of Radiology, Bois-Cerf Clinic, 1006 Lausanne, Switzerland
¹⁴Department of Neurosurgery Spine, Private Hospital Sévigné, 35567 Cesson-Sévigné, France
¹⁵Department of Trauma and Reconstructive Surgery, Philipps University of Marburg, University Hospital of Giessen and Marburg, 35043 Marburg, Germany

Received 26 October 2015; Accepted 2 December 2015

Academic Editor: Nicholas Dunne

Copyright © 2015 David Noriega et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

Abstract.

This prospective, consecutive, multicentre observational registry aimed to confirm the safety and clinical performance of the SpineJack system for the treatment of vertebral compression fractures (VCF) of traumatic origin. We enrolled 103 patients (median age: 61.6 years) with 108 VCF due to trauma, or traumatic VCF with associated osteoporosis. Primary outcome was back pain intensity (VAS). Secondary outcomes were Oswestry Disability Index (ODI), EuroQol-VAS, and analgesic consumption. 48 hours after surgery, a median relative decrease in pain intensity of 81.5% was observed associated with a significant reduction in analgesic intake. Improvements in disability (91.3% decrease in ODI score) and in quality of life (increase 21.1% of EQ-VAS score) were obtained 3 months after surgery. All results were maintained at 12 months. A reduction in the kyphotic angulation was observed postoperatively (; ), remained at 12 months (, ). No adverse events were implant-related and none required device removal. Three patients (2.9%) experienced procedure-related complications. The overall adjacent fracture rate up to 1 year after surgery was 2.9%. The SpineJack procedure is an effective, low-risk procedure for patients with traumatic VCF allowing a fast and sustained improvement in quality of life over 1 year after surgery.

1. Introduction

Vertebroplasty and kyphoplasty are minimally invasive accepted procedures in the therapy of painful osteoporotic compression fractures [1–5]. Both techniques are indicated after inefficient conservative pain therapy in “stable” vertebral compression fractures (VCF). Acute traumatic fractures have to be differentiated. Traumatic fractures are related to acute trauma and can be witnessed also in osteoporotic patients. The involvement of the vertebra’s posterior wall is a relative contraindication. Theoretical risks of posterior wall displacement are potential cement leakages into the spinal canal and the further dislocation of bone fragments into the spinal canal. Both incidents could lead to neurological deficits [3, 5].

Balloon kyphoplasty has been used for the treatment of fresh traumatic fractures [6, 7]. Several clinical and biomechanical studies have shown that there is a height loss after deploying the balloons [8, 9]. To improve anatomical restoration of the endplates of the vertebral body (VB) a new intravertebral reduction device was designed. The aim of this study was to confirm the safety and clinical performance of the SpineJack system in combination with Cohesion bone cement for the treatment of traumatic VCF. The study protocol calls for a final analysis of the results after a 2-year postoperative follow-up for all patients. We present 1-year results focusing on 103 patients assessed peroperatively and 48 h postoperatively, 92 patients at 3-month follow-up, and 78 patients at 12-month follow-up.

2. Patients and Methods

2.1. Patient Population

Between October 2011 and December 2012, 103 patients from 14 centers throughout Europe were enrolled in this prospective, consecutive, multicenter observational registry. All of them met the indication as listed on the products’ labelling, namely, “Patients age > 18 years presenting a mobile spinal fracture that may result from trauma (Magerl group A1, A2, or A3.1) and/or osteoporosis (IFU), with a minimum internal pedicle diameter > 5.8 mm to allow placement of the device.” In order to ensure overall homogeneity and relevance of the results, we decided to include patients with acute fresh traumatic VCF, a traumatic fracture being defined as a fracture of the VB resulting from a high or low-energy impact occurring during a traumatic event. In this registry, data of interest (at baseline, peroperatively and 48-h postoperatively, and 3 and 12 months) were collected without requiring the physician to perform additional investigations. As surgeons must follow their own standard care practice follow-up, the patients analyzed might have no complete datasets at 3 or 12 month.

2.2. Ethics

Local Ethics Committee submissions were performed as per local regulations requirements. All patients gave their informed consent in accordance with ISO 14155 and the Declaration of Helsinki.

2.3. Operative Technique

The 5 mm diameter SpineJack was implanted using a percutaneous or minimally invasive posterior surgical approach using surgical tools supplied with the device [10]. With the patient being in prone position, the device was inserted into the fractured VB. Then, the implant was expanded using a specially designed tool which locks into the device and pulls the two ends of the implant towards each other. Longitudinal compression of the device causes the implant to open in the inferior-superior direction only due to the machined grooves (Figure 1).

Figure 1: SpineJack expandable intravertebral body implant, closed and fully expanded.

A simple mechanism locks the implant into the desired position as controlled by the physician. Once the desired expansion obtained, the device was left in place inside the restored vertebra and polymethylmethacrylate (PMMA) bone cement was injected into and around the implant. Regular fluoroscopic controls throughout the operative procedure ensured correct implantation (Figure 2). Postoperative rehabilitation was per standard of care at the treating institution.

Figure 2: Functional principle of the SpineJack procedure.

2.4. Clinical Assessment

The main outcome was pain evolution over time as assessed using a 10-cm visual analogue scale (VAS) where 0 is no pain and 10 is the worst imaginable pain. Secondary outcomes included analgesic intake, functional capacity (assessed using the self-administered Oswestry Disability Index, ODI) [11] and quality of life (assessed using the EQ-VAS from the self-administered European Quality of life-5 dimensions, EQ-5D, score) [12] evaluated at baseline, 3, and 12 months (and at discharge for analgesic intake). Complications were recorded throughout the follow-up period (adverse events device-related or not, surgery-related complications, technical incidents, device removal, subsequent compression fractures, and cement leakages).

2.5. Radiographic Assessment

X-rays were performed following routine clinical practice at each study site (usually 48 hours postoperatively, 3, and 12 months) and performed as per described in the protocol imaging. All available X-ray images were collected to assess the evolution of vertebral kyphotic angle.

Quantitative radiographic analysis was done on X-ray by an independent, qualified core lab using the validated FXA software developed by ACES Ing. -GmbH, Filderstadt, Germany [13]. This interim analysis focused on 48-hour, 3-month, and 12-month postoperative results.

2.6. Statistical Analysis

Statistical analyses were performed at the 0.05 global significance level using two-sided tests (Statistical Analysis System software, version 9.2, SAS Institute Inc., Cary, NC, USA). Within-group tests were used to test evolution between visits. Depending on the normality of the distribution, Wilcoxon’s test or Student’s test for pairwise comparisons were used.

Assuming a mean evolution of pain from baseline to 12 months of 5 (SD 2.5) with a precision of 0.6 and a relative precision of 12%, 66 evaluable patients were required.

Considering a lost-to-follow-up rate of 20%, 80 patients were to be included.

3. Results

3.1. Baseline Characteristics

The 103 patients analyzed (49.5% female) had a median age of 61.6 years, with a median BMI of 25.8 kg/m². Eight patients (7.8%) presented with previous traumatic VCF; 5 of them had been already treated surgically at a level different from the one treated in this study. For 77 patients (74.8%), a previous treatment had been administered: bed rest (65.1%), bracing (9.7%), and walking aid (4.9%). A total of 108 VCF were treated (5 patients, 4.9%, had 2 fractures treated). Most fractures were due to high energy trauma (, i.e., 79.6% concerning 81 patients, i.e., 78.6% of population) and the remaining were traumatic fractures with associated osteoporosis (, i.e., 20.4% concerning 22 patients, i.e., 21.4% of population). As osteoporosis evaluation by DEXA was not performed in routine practice, it was not requested by the protocol. Osteoporotic patients were already known by the investigator, but patients with severe osteoporosis were excluded from the study. Median time from trauma to surgery was 6 days for pure traumatic fractures and 12 days in osteoporotic patients. The Magerl classification [14] showed the following distribution: 44.2% of type A3 fractures (27.9% A3.1, 4.8% A.3.2, and 11.5% A.3.3), 41.4% of type A1 fractures (1.0% A.1.3, 40.4% A1.2), 8.7% of type A2 fractures (1.0% A.2.1, 2.9% A.2.2, and 4.8% A.2.3), and 5.9% of type B fractures (2.9% B.1.3, 1.0% B.2.3, 1.0% B.3.1, and 1.0% B.3.3). Three-quarters of fractures (76.8%) were located at T12-L2 and the remaining between L3 and L5 (16.8%) or between T9 and T11 (6.4%) (Figure 3).

Figure 3: Operated vertebral levels.

Mean duration of hospital stay was days. Prolonged hospitalization (27 days) was observed in one obese, osteoporotic patient treated out of IFU indications at the time of this registry as he presented with a severe A3.3 fracture. He underwent decompression and posterior instrumentation after dislocation of posterior wall at Day 4 (procedure-related); cut-out of screws in L3 occurred at 3 months, but the patient refused revision surgery. Mean follow-up period of the 103 patients was months. Table 1 contains demographics and main baseline characteristics on the study population.

Table 1: Baseline patient characteristics.

Traumatic VCF with osteoporosis
Traumatic VCF
Total

Age, mean (SD) (years)
69.2 (12.1)
59.5 (15.5)
61.6 (15.3)

Female, (%)
20 (90.9%)
31 (38.3%)
51 (49.5%)

BMI, mean (SD) (kg/m²)
25.9 (4.6)
25.8 (3.8)
25.8 (3.9)

Number of augmented vertebrae
22
86
108

Time from fracture to procedure, mean (SD) (days)
13.9 (8.9)
8.1 (7.1)
9.2 (7.8)

Pain, VAS score, mean (SD) (cm)
6.8 (1.9)
6.6 (2.8)
6.6 (2.6)

ODI score, mean (SD)
70.5 (16.0)
78.3 (19.9)
76.6 (19.4)

EQ-VAS
47.7 (21.4)
51.1 (26.6)
50.4 (25.5)

Twenty-three (23) patients withdrew from the study before the 12-month visit: 2 patients died due to renal failure and acute respiratory syndrome, respectively; 11 patients refused medical follow-up because of complete relief of their symptoms; 1 patient was withdrawn because of severe aggravation of a preexisting osteoporosis at inclusion, with four consecutive spontaneous fractures after surgery on Day 19, Day 49 and two fractures on Day 86; 9 patients were lost to follow-up. Flowchart of visits according to type and age of fracture is presented in Table 2.

Table 2: Flowchart of visits according to type and age of fracture.

Inclusion

= 103

Traumatic and osteoporosis associated
Traumatic only

= 22
= 81

<15 days
[15; 30] days
[30; 45] days
<15 days
[15; 30] days
[30; 45] days

= 14
= 7
= 1
= 70
= 9
= 2

Procedure

= 103

Traumatic and osteoporosis associated
Traumatic only

= 22
= 81

<15 days
[15; 30] days
[30; 45] days
<15 days
[15; 30] days
[30; 45] days

= 14
= 7
= 1
= 70
= 9
= 2

48 hours after surgery

= 103

Traumatic and osteoporosis associated
Traumatic only

= 22
= 81

<15 days
[15; 30] days
[30; 45] days
<15 days
[15; 30] days
[30; 45] days

= 14
= 7
= 1
= 70
= 9
= 2

3 months after surgery

= 92

Traumatic and Osteoporosis associated
Traumatic Only

= 21
= 71

<15 days
[15; 30] days
[30; 45] days
<15 days
[15; 30] days
[30; 45] days

= 13
= 7
= 1
= 63
= 6
= 2

12 months after surgery

= 78

Traumatic and Osteoporosis associated
Traumatic Only

= 16
= 62

<15 days
[15; 30] days
[30; 45] days
<15 days
[15; 30] days
[30; 45] days

= 11
= 4
= 1
= 53
= 7
= 2

3.2. Surgical Data

All surgeries were performed by surgeons, according to their standard procedures. Most fractures (94.2%) were treated under general anesthesia, 3.9% under local anesthesia, 1% by both local and general anesthesia, and the remaining 1% by spinal anesthesia. Mean duration of general anesthesia was  min; all local anesthesia lasted 60 min. Of the 108 treated vertebrae, 106 (98.1%) were treated by a percutaneous approach, while 2 (1.9%) were treated by open surgery. Mean operating time was  min (range: 17–105). Mean quantity of cement injected was 6.7 mL 2.2 mL (range: 3.0–10.8).

3.3. Clinical Outcome

As early as 48 hours after surgery, a significant improvement in back pain was obtained with a mean VAS score decreasing from  cm at baseline to  cm (mean change: − cm; ). This change corresponded to a median relative decrease in pain intensity of 81.5%. This improvement was maintained over the 12-month follow-up period. Mean changes in VAS scores versus baseline at 48 hours after surgery, 3 months, and 12 months are shown in Table 3.

Table 3: Absolute changes (cm) and relative (%) changes in VAS score.

Traumatic VCF with osteoporosis
Traumatic VCF
Total

48 h versus baseline

Mean (SD) absolute changes
−5.5 (1.9)
−5.1 (2.9)
−5.2 (2.7)

Median absolute changes
−5.7
−6.1
−6.0

Within-group test
<.001 (Student)
<.001 (Wilcoxon)
<.001 (Wilcoxon)

Median relative changes
−81.0
−81.8
−81.5

3 months versus baseline

Mean (SD) absolute changes
−5.5 (2.7)
−5.3 (3.0)
−5.3 (2.9)

Median absolute changes
−6.2
−6.0
−6.0

Within-group test
<.001 (Student)
<.001 (Wilcoxon)
<.001 (Wilcoxon)

Median relative changes
−90.0
−88.0
−88.0

12 months versus baseline

Mean (SD) absolute changes
−5.7 (2.3)
−5.5 (3.0)
−5.5 (2.9)

Median absolute changes
−5.2
−6.3
−5.8

Within-group test
<.001 (Student)
<.001 (Wilcoxon)
<.001 (Wilcoxon)

Median relative changes
−90.3
−92.2
−91.5

As can be seen from Table 3, similar results were observed with both pure traumatic VCF and traumatic VCF in patients with osteoporosis. Evolution of median VAS score over time is depicted in Figure 4.

Figure 4: Evolution of pain: median VAS score over time.

The decrease in pain allowed a significant reduction in the intake of analgesics within 48 hours after surgery. Indeed, before surgery, nearly three out of four patients needed strong (20.4%) or moderate (51.5%) analgesics. Two days after surgery, these percentages decreased to 1% and 5.8%, respectively. Evolution in analgesic requirement is illustrated in Figure 5.

Figure 5: Evolution in analgesic requirements.

A marked improvement in disability was observed at 3 months with a mean ODI score decreasing from at baseline to (mean change: −; ). This change corresponded to a median relative decrease of 91.3%. This improvement was maintained at 12 months. Mean changes in ODI scores at 3 months and 12 months versus baseline are shown in Table 4.

Table 4: Absolute and relative (%) changes in ODI score.

Traumatic VCF with osteoporosis
Traumatic VCF
Total

3 months versus baseline

Mean (SD) absolute changes
−54.3 (26.6)
−64.3 (24.1)
−62.0 (24.9)

Median absolute changes
−62.0
−73.3
−71.3

Within-group test
<.001 (Student)
<.001 (Wilcoxon)
<.001 (Wilcoxon)

Median relative changes
−91.0
−92.7
−91.3

12 months versus baseline

Mean (SD) absolute changes
−60.7 (18.8)
−67.1 (24.9)
−65.7 (23.8)

Median absolute changes
−64.1
−74.0
−73.3

Within-group test
<.001 (Student)
<.001 (Wilcoxon)
<.001 (Wilcoxon)

Median relative changes
−92.9
−95.0
−94.9

As indicated in Table 4, similar results were observed with both types of fractures. Evolution of median ODI score over time is depicted in Figure 6.

Figure 6: Evolution of functional disability: median ODI score over time.

A clear improvement in quality of life was observed at 3 months with a mean EQ-VAS score increasing from at baseline to (mean change: +; ). This change corresponded to a median relative increase in quality of life of 21.1%. A slightly further improvement was observed at 12 months. Mean changes in EQ-VAS scores at 3 months and 12 months versus baseline are shown in Table 5. As indicated in this table, similar results were observed with both types of fractures.

Table 5: Absolute and relative (%) changes in EQ-VAS score.

Traumatic VCF with osteoporosis
Traumatic VCF
Total

3 months versus baseline

Mean (SD) absolute changes
23.1 (25.3)
16.6 (31.6)
18.1 (30.2)

Median absolute changes
16.5
12.5
13.0

Within-group test
<.001 (Student)
<.001 (Student)
<.001 (Wilcoxon)

Median relative changes
27.0
17.4
21.1

12 months versus baseline

Mean (SD) absolute changes
29.3 (22.3)
23.4 (28.4)
24.6 (27.2)

Median absolute changes
26.5
16.0
19.0

Within-group test
<.001 (Student)
<.001 (Wilcoxon)
<.001 (Wilcoxon)

Median relative changes
44.2
33.3
38.3

3.4. Radiological Outcome

A significant and immediate reduction in the kyphotic angulation was observed 48 hours after the procedure (from ° to °, i.e., °; ). Despite a lower reduction observed at 3 and 12 months, the improvement of kyphosis remained statistically significant compared to baseline (−° at 3 months, ; −° at 12 months, ). Similar results were observed for both types of fractures.

3.5. Complications

Postoperatively, 15 patients (14.6%) experienced a total of 21 adverse events which are detailed in Table 6. None of these adverse events were considered implant-related and none required device removal.

Table 6: Adverse events.

Patient number
Type of event
Time after surgery (days)
Relationship to device
Relationship to procedure
Outcome at 12 M

11R01
Death due to acute kidney failure aggravation with vascular obliteration of leg
52
No
No
Death

11R04
Sigma diverticulitis (preexisting condition at inclusion) aggravation: hospitalization for diverticulitis resection
8
No
No
Resolved

11R07
Asymptomatic adjacent vertebral fracture T11: No treatment needed (treated vertebra T12)
99
No
Yes
Ongoing (fracture remains stable)

13R06
Prolongation of hospitalization for dislocation of posterior wall secondary to surgery which leads to sensorial deficit → surgery with wrong indication
4
No
Yes
Resolved

Breakage of screws in L3: recommendation of revision surgery but patient declined surgery
82
No
No
Ongoing
Patient lost to follow-up

31R04
Posterior articular conflict due to a discopathy L4/L5 grade IV associated to a degenerative spondylolisthesis grade II Meyerding
147
No
No
Resolved at Day 455

31R05
Algoneurodystrophic syndrome due to calcaneum fracture resolved by decompressive surgery
100
No
No
Resolved

31R06
Prostatic cancer
257
No
No
Ongoing

31R15
Bleeding at the point of the skin incision just after the surgery
2
No
No
Resolved

31R16
Spontaneous adjacent fracture T12 (treated vertebra L1) in a context of major osteoporosis aggravation: hospitalization for vertebroplasty
19
No
No
Resolved

Spontaneous adjacent fracture L2 (treated vertebra L1) in a context of major osteoporosis aggravation treatment: analgesics and Zoledronic acid monohydrate
49
No
No
Patient discontinued

Spontaneous new fracture L4 (treated vertebra L1) in a context of major osteoporosis aggravation treatment: analgesics and Zoledronic acid monohydrate
86
No
No
Patient discontinued

Spontaneous new fracture L5 (treated vertebra L1) in a context of major osteoporosis aggravation treatment: analgesics and Zoledronic acid monohydrate
86
No
No
Patient discontinued

35R08
Death due to acute respiratory syndrome
204
No
No
Death

35R11
Hospitalization in psychiatric department
Not available
No
No
Resolved

36R01
Subsequent fracture at T12 level (previous treated level L2): hospitalization for vertebroplasty
18
No
No
Resolved

Subsequent fracture at T11 level (previous treated level L2): hospitalization for vertebroplasty
126
No
No
Resolved

Adjacent fracture at L3 level (previous treated level L2): hospitalization for vertebroplasty
126
No
No
Resolved

36R02
Lumbar pain
418
No
No
Resolved

38R07
Shoulder fracture due to a fall discovered lately and treated by physiotherapy
276
No
No
Ongoing

41R02
Collapse of treated vertebral body associated with canal compromise and hematoma leading to neurological symptoms
16
No
Yes
Improved

Among these adverse events, 13 serious adverse events concerning 10 patients (9.7%) were reported. Two patients died because of renal failure with lower limb vascular obliteration on Day 52 in one patient and acute respiratory syndrome at 6.8 months in the other patient.

Three patients (2.9%) experienced procedure-related complications. Eight subsequent compression fractures concerning 3 osteoporotic patients were reported. The overall adjacent fracture rate up to 1 year after surgery was 2.9% (4 fractures concerning 3 out of 103 patients). The adjacent fracture rate in the osteoporotic group was 13.6% (4 fractures concerning 3 out of 22 patients).

Cement leakage was shown in 43 out of 108 (39.8%) treated vertebrae with no clinical consequences. Leaks were detected by peroperative fluoroscopy (28.6% of cases), postoperative X-rays (28.6% of cases) or CT scans (42.8% of cases).

4. Discussion

Treatment of traumatic VCF without neurological symptoms and intact posterior ligament complex is still controversial. Vaccaro et al. proposed a classification system that should be helpful for decision making [15]. One major problem is that spine surgeons are faced with different classification systems for fractures [14, 16–18] and a great inhomogeneity in patient populations and treatment strategies. The Magerl classification is widely used in Europe but has shown its limits [19]. Moreover, treatment philosophies diversify impressively. In some countries, traumatic incomplete cranial burst fractures are considered to be unstable [20]. Others promote conservative treatment if posterior ligament complex and neurologic status are unaffected [15, 18]. The evidence level concerning treatment strategies is still low [21]. Yi et al. stated that there is no statistically significant difference on functional outcome two years or more after therapy between operative and nonoperative treatment [22]. Conservative treatment consisted inter alia of an average 4–6-week bed rest and an additional 6–12-week TLSO bracing. There is a consensus in the aims of operative treatment. Operative treatment should prevent neurologic symptoms, minimize spinal deformity and complications, allow fracture healing, and insure the best possible function.

Operative treatment ranges from combined anteroposterior approaches to minimal invasive procedures like cement augmentation [20]. Balloon kyphoplasty was used in combination with dorsal instrumentation to restore the sagittal balance [6, 23, 24]. Treatment of osteoporotic burst fractures by standalone kyphoplasty has become a standard procedure [25]. Several authors have used balloon kyphoplasty in young patients with traumatic unstable fractures [7, 26, 27]. The 10-year results of Maestretti et al. seem very promising [7]. Nevertheless, there are some studies showing that the endplate fracture reduction gained by inflation of bone tamps could not be maintained after deflation [8, 9].

The study implant is designed to work as an intravertebral reduction device directing its forces in the craniocaudal direction. The technical possibilities and improved height restoration have been shown in several biomechanical studies [28–30]. All clinical results from this study, especially the ones reported 48 h after surgery with significant reduction in pain and analgesic intake (from 71.9% to 6.8% of patients requiring strong or moderate analgesics) cannot be reached by conservative treatment [4]. A marked improvement in disability (91.3% decrease in ODI score) and in quality of life (increase in 21.1% of EQ-VAS score) was obtained 3 months after surgery and maintained at 12 months. The biggest argument in favor of cement augmenting procedures in unstable fractures is the immediate pain reduction. Change in pain intensity is nearly thrice the change considered as clinically meaningful. Indeed, Ostelo et al. stated that a 30% change from baseline may be considered a clinically significant improvement [31]. Concerning radiological outcome, a significant and immediate reduction of the kyphotic angulation was observed 48 hours after surgery. Despite a lower reduction observed at 3 and 12 months, global improvement in kyphosis remained statistically significant compared to baseline.

In patients with VCF, clinical and radiological results of this minimal invasive technique have to be balanced against its complications. In the study performed by the Task Force “Wirbelsäule” (spinal column) German Society of Trauma Surgery, the complication rates (recessing and nonrecessing complications) of the different procedures varied between 14.1 and 29.7% [32–34]. Compared to these values, the procedure herein described is a safe technique. We did not observe any neurological complications. Only three patients (2.9%) experienced procedure-related complications. In patients with potential unstable fractures and poor bone quality sufficient cement has to be injected to ensure stability [29]. The adjacent fracture rate up to 1 year after surgery was 2.9% (4 fractures concerning 3 out of 103 patients). In osteoporotic patients, the risk of developing a new fracture is around 19% [35, 36]. Compared to these values, the rate of adjacent fractures seems to be low. It has to be pointed out that the average age of treated patients (59.5 years) is lower than the one reported in most studies about osteoporotic fractures. Cement leakage was detected in 39.8% of treated vertebrae with no clinical consequences. These numbers are comparable to other studies on unstable fractures treatment [25]. Symptomatic cement leakages were not seen in any patient.

5. Conclusion

The present study supports the growing interest in minimally invasive techniques in the management of spinal injuries with no neurological deficit. These long-term results confirm the stability of the correction over time. The device described allows an effective, low-risk procedure for patients with vertebral fractures of traumatic origin with a significant reduction in pain and analgesic consumption achieved immediately after surgery and maintained over time. Additionally, this procedure allows a fast and sustainable improvement in quality of life. Main complications included asymptomatic cement extravasation and adjacent fractures, which were caused by the cementation technique or resulted from underlying osteoporosis but were not due to the procedure itself. One-year results from this registry are very promising and will be confirmed with the 2-year outcomes and have to be further proven by comparative randomized study results.

Conflict of Interests

Relevant financial activities outside the submitted work include consultancy, expert testimony, payment for lecture, and payment for the development of educational presentations.

Acknowledgments

The authors thank all the coinvestigators participating at the investigational centers: Francisco Ardura, M.D., Spain; Julien Baud, M.D., France; Emmanuel Foulongne, M.D., France; Guillaume Perot, M.D., France; Thomas Poglitsch, M.D., Austria; Paul Puchweine, M.D., Austria; Riccardo Ciarpaglini, M.D., Switzerland; and Vito Fiorenza, M.D., Italy. Statistical analyses have been performed by Statitec (Labège, France). This work is sponsored by Vexim SA, Hills Plaza, 8 Rue Vidailhan, 1er Etage, 31130 Balma, France.

References

P. A. Hulme, J. Krebs, S. J. Ferguson, and U. Berlemann, “Vertebroplasty and kyphoplasty: a systematic review of 69 clinical studies,” Spine, vol. 31, no. 17, pp. 1983–2001, 2006.

J. C. Eck, D. Nachtigall, S. C. Humphreys, and S. D. Hodges, “Comparison of vertebroplasty and balloon kyphoplasty for treatment of vertebral compression fractures: a meta-analysis of the literature,” Spine Journal, vol. 8, no. 3, pp. 488–497, 2008.

M. J. Lee, M. Dumonski, P. Cahill, T. Stanley, D. Park, and K. Singh, “Percutaneous treatment of vertebral compression fractures: a meta-analysis of complications,” Spine, vol. 34, no. 11, pp. 1228–1232, 2009.

D. Wardlaw, J. Van Meirhaeghe, J. Ranstam, L. Bastian, and S. Boonen, “Balloon kyphoplasty in patients with osteoporotic vertebral compression fractures,” Expert Review of Medical Devices, vol. 9, no. 4, pp. 423–436, 2012.

I. D. Papanastassiou, A. Filis, M. A. Gerochristou, and F. D. Vrionis, “Controversial issues in kyphoplasty and vertebroplasty in osteoporotic vertebral fractures,” BioMed Research International, vol. 2014, Article ID 934206, 12 pages, 2014.

L. Zhang, J. Zou, M. Gan, J. Shi, J. Li, and H. Yang, “Treatment of thoracolumbar burst fractures: short-segment pedicle instrumentation versus kyphoplasty,” Acta Orthopaedica Belgica, vol. 79, no. 6, pp. 718–725, 2013.

G. Maestretti, P. Sutter, E. Monnard et al., “A prospective study of percutaneous balloon kyphoplasty with calcium phosphate cement in traumatic vertebral fractures: 10-year results,” European Spine Journal, vol. 23, no. 6, pp. 1354–1360, 2014.

J.-J. Verlaan, E. B. van de Kraats, F. C. Oner, T. Van Walsum, W. J. Niessen, and W. J. A. Dhert, “The reduction of endplate fractures during balloon vertebroplasty: a detailed radiological analysis of the treatment of burst fractures using pedicle screws, balloon vertebroplasty, and calcium phosphate cement,” Spine, vol. 30, no. 16, pp. 1840–1845, 2005.

G. Voggenreiter, “Balloon kyphoplasty is effective in deformity correction of osteoporotic vertebral compression fractures,” Spine, vol. 30, no. 24, pp. 2806–2812, 2005.

S. S. Baeesa, A. Krueger, F. A. Aragón, and D. C. Noriega, “The efficacy of a percutaneous expandable titanium device in anatomical reduction of vertebral compression fractures of the thoracolumbar spine,” Saudi Medical Journal, vol. 36, no. 1, pp. 52–60, 2015.

D. Vogler, R. Paillex, M. Norberg, P. de Goumoëns, and J. Cabri, “Cross-cultural validation of the Oswestry disability index in French,” Annales de Readaptation et de Medecine Physique, vol. 51, no. 5, pp. 379–385, 2008.

R. Brooks, R. Rabin, and F. Charro, The Measurement and Valuation of Health Status Using EQ-5D: A European Perspective. Evidence from the EuroQol BIOMED. Research Programme, Kluwer Academic Publishers, Dordrecht, The Netherlands, 2003.

M. Schulze, F. Trautwein, T. Vordemvenne, M. Raschke, and F. Heuer, “A method to perform spinal motion analysis from functional X-ray images,” Journal of Biomechanics, vol. 44, no. 9, pp. 1740–1746, 2011.

F. Magerl, M. Aebi, S. D. Gertzbein, J. Harms, and S. Nazarian, “A comprehensive classification of thoracic and lumbar injuries,” European Spine Journal, vol. 3, no. 4, pp. 184–201, 1994.

A. R. Vaccaro, R. A. Lehman Jr., R. J. Hurlbert et al., “A new classification of thoracolumbar injuries: the importance of injury morphology, the integrity of the posterior ligamentous complex, and neurologic status,” Spine, vol. 30, no. 20, pp. 2325–2333, 2005.

F. Denis, “The three column spine and its significance in the classification of acute thoracolumbar spinal injuries,” Spine, vol. 8, no. 8, pp. 817–831, 1983.

T. McCormack, E. Karaikovic, and R. W. Gaines, “The load sharing classification of spine fractures,” Spine, vol. 19, no. 15, pp. 1741–1744, 1994.

A. R. Vaccaro, C. Oner, C. K. Kepler et al., “AOSpine thoracolumbar spine injury classification system: fracture description, neurological status, and key modifiers,” Spine, vol. 38, no. 23, pp. 2028–2037, 2013.

F. C. Oner, L. M. Ramos, R. K. Simmermacher et al., “Classification of thoracic and lumbar spine fractures: problems of reproducibility. A study of 53 patients using CT and MRI,” European Spine Journal, vol. 11, no. 3, pp. 235–245, 2002.

A. P. Verheyden, A. Hölzl, H. Ekkerlein et al., “Recommendations for the treatment of thoracolumbar and lumbar spine injuries,” Unfallchirurg, vol. 114, no. 1, pp. 9–16, 2011.

T. Oh, J. K. Scheer, S. Fakurnejad, N. S. Dahdaleh, and Z. A. Smith, “Minimally invasive spinal surgery for the treatment of traumatic thoracolumbar burst fractures,” Journal of Clinical Neuroscience, vol. 22, no. 1, pp. 42–47, 2015.

L. Yi, B. Jingping, J. Gele, X. Baoleri, and W. Taixiang, “Operative versus non-operative treatment for thoracolumbar burst fractures without neurological deficit,” Cochrane Database of Systematic Reviews, no. 4, Article ID CD005079, 2006.

S. Fuentes, B. Blondel, P. Metellus, J. Gaudart, T. Adetchessi, and H. Dufour, “Percutaneous kyphoplasty and pedicle screw fixation for the management of thoraco-lumbar burst fractures,” European Spine Journal, vol. 19, no. 8, pp. 1281–1287, 2010.

J.-J. Verlaan, W. J. A. Dhert, and F. C. Oner, “Intervertebral disc viability after burst fractures of the thoracic and lumbar spine treated with pedicle screw fixation and direct end-plate restoration,” Spine Journal, vol. 13, no. 3, pp. 217–221, 2013.

A. Krüger, R. Zettl, E. Ziring, D. Mann, M. Schnabel, and S. Ruchholtz, “Kyphoplasty for the treatment of incomplete osteoporotic burst fractures,” European Spine Journal, vol. 19, no. 6, pp. 893–900, 2010.

R. de Falco, E. Scarano, D. Di Celmo, U. Grasso, and L. Guarnieri, “Balloon kyphoplasty in traumatic fractures of the thoracolumbar junction. Preliminary experience in 12 cases,” Journal of Neurosurgical Sciences, vol. 49, no. 4, pp. 147–153, 2005.

G. Maestretti, C. Cremer, P. Otten, and R. P. Jakob, “Prospective study of standalone balloon kyphoplasty with calcium phosphate cement augmentation in traumatic fractures,” European Spine Journal, vol. 16, no. 5, pp. 601–610, 2007.

F. Bolzinger, J. F. Oglaza, A. Krüger, and P. Swider, “Measurement of implant deployment and related forces in kyphoplasty by percutaneous approach,” Clinical Biomechanics, vol. 29, no. 4, pp. 463–467, 2014.

A. Krüger, G. Baroud, D. Noriega et al., “Height restoration and maintenance after treating unstable osteoporotic vertebral compression fractures by cement augmentation is dependent on the cement volume used,” Clinical Biomechanics, vol. 28, no. 7, pp. 725–730, 2013.

A. Krüger, L. Oberkircher, J. Figiel et al., “Height restoration of osteoporotic vertebral compression fractures using different intravertebral reduction devices: a cadaveric study,” Spine Journal, vol. 15, no. 5, pp. 1092–1098, 2015.

R. W. J. G. Ostelo, R. A. Deyo, P. Stratford et al., “Interpreting change scores for pain and functional status in low back pain: towards international consensus regarding minimal important change,” Spine, vol. 33, no. 1, pp. 90–94, 2008.

C. Knop, L. Bastian, U. Lange, M. Oeser, M. Zdichavsky, and M. Blauth, “Complications in surgical treatment of thoracolumbar injuries,” European Spine Journal, vol. 11, no. 3, pp. 214–226, 2002.

C. Knop, M. Blauth, V. Buhren et al., “Surgical treatment of injuries of the thoracolumbar transition—3: follow-up examination. Results of a prospective multi-center study by the “Spinal” Study Group of the German Society of Trauma Surgery,” Unfallchirurg, vol. 104, no. 7, pp. 583–600, 2001.

M. Reinhold, C. Knop, R. Beisse et al., “Operative treatment of 733 patients with acute thoracolumbar spinal injuries: comprehensive results from the second, prospective, internet-based multicenter study of the Spine Study Group of the German Association of Trauma Surgery,” European Spine Journal, vol. 19, no. 10, pp. 1657–1676, 2010.

R. Lindsay, S. L. Silverman, C. Cooper et al., “Risk of new vertebral fracture in the year following a fracture,” The Journal of the American Medical Association, vol. 285, no. 3, pp. 320–323, 2001.

C. Bliemel, L. Oberkircher, B. Buecking, N. Timmesfeld, S. Ruchholtz, and A. Krueger, “Higher incidence of new vertebral fractures following percutaneous vertebroplasty and kyphoplasty—fact or fiction?” Acta Orthopaedica Belgica, vol. 78, no. 2, pp. 220–229, 2012.