BioMed Research International
Volume 2016 (2016), Article ID 4567146, 8 pages
http://dx.doi.org/10.1155/2016/4567146

Research Article

A Comparison of Conventional Collagen Sponge and Collagen-Gelatin Sponge in Wound Healing

Chizuru Jinno,¹ Naoki Morimoto,² Ran Ito,¹ Michiharu Sakamoto,¹ Shuichi Ogino,¹ Tsuguyoshi Taira,³ and Shigehiko Suzuki¹

¹Department of Plastic and Reconstructive Surgery, Graduate School of Medicine, Kyoto University, 54 Kawahara-machi, Syougoin, Sakyou-ku, Kyoto City, Kyoto 606-8507, Japan
²Department of Plastic and Reconstructive Surgery, Kansai Medical University, 2-5-1 Shin-machi, Hirakata City, Osaka 573-1010, Japan
³GUNZE Ltd. QOL Research Center Laboratory, 1 Ishiburo, Inokura Shinmachi, Ayabe, Kyoto 623-8512, Japan

Received 5 January 2016; Revised 23 February 2016; Accepted 3 March 2016

Academic Editor: Francesco Piraino

Copyright © 2016 Chizuru Jinno et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

Abstract.

The objective of this study was to compare the effectiveness of the collagen-gelatin sponge (CGS) with that of the collagen sponge (CS) in dermis-like tissue regeneration. CGS, which achieves the sustained release of basic fibroblast growth factor (bFGF), is a promising material in wound healing. In the present study, we evaluated and compared CGSs and conventional CSs. We prepared 8 mm full-thickness skin defects on the backs of rats. Either CGSs or CSs were impregnated with normal saline solution (NSS) or 7 μg/cm² of bFGF solution and implanted into the defects. At 1 and 2 weeks after implantation, tissue specimens were obtained from the rats of each group (, total ). The wound area, neoepithelial length, dermis-like tissue area, and the number and area of capillaries were evaluated at 1 and 2 weeks after implantation. There were no significant differences in the CGS without bFGF and CS groups. Significant improvements were observed in the neoepithelial length, the dermis-like tissue area, and the number of newly formed capillaries in the group of rats that received CGSs impregnated with bFGF. The effects on epithelialization, granulation, and vascularization of wound healing demonstrated that, as a scaffold, CGSs are equal or superior to conventional CSs.

1. Introduction

We developed a bilayered acellular artificial dermis (Pelnac®, Gunze Co. Ltd., Ayabe, Japan) consisting of an upper silicone sheet and a lower collagen sponge (CS) [1]. After the CS is grafted onto a full-thickness skin defect, fibroblasts and new capillaries spread throughout the lower layer of the sponge. New collagen fibers are synthesized by the penetration of fibroblasts and the collagen sponge biodegrades and are gradually replaced with regenerated dermis-like tissue within a period of 2-3 weeks [1, 2]. Artificial dermis has been used for the treatment of full-thickness skin defects caused by burns, in the waiting period for tumor excision, and for the treatment of intractable ulcers for more than 10 years [3]. However, until the capillaries infiltrate the collagen sponge and the vascular network is formed, the artificial dermis exposes the patient to a high risk of infection [4]. It is therefore difficult to apply artificial dermis to chronic ulcers, such as decubitus, diabetic, and leg ulcers [4]. Basic FGF, which was identified in 1974, promotes the proliferation of fibroblasts and the formation of capillaries and accelerates tissue regeneration [5–9]. In Japan, human recombinant bFGF (Fibrast Spray®, Kaken Pharmaceutical Co., Ltd., Tokyo, Japan) has been available since 2001, and its clinical effectiveness has been verified [7]. In combination with bFGF, artificial dermis has been reported to accelerate dermis-like tissue formation [10]. However, bFGF is rapidly diffused and inactivated after application in vivo [11]. To overcome this disadvantage, we developed a CGS that contains a 10 wt% concentration of acidic gelatin which is capable of sustaining positively charged growth factors, such as bFGF, via the formation of ion complexes between bFGF and gelatin [4, 12, 13]. In our previous study, we reported that CGSs impregnated with bFGF (7 μg/cm²) accelerated neovascularization and wound healing with less contraction on the backs of mice, in the femoral trochanters of diabetic mice, and in the palatal mucosa of dogs [12, 14]. We also reported that CGSs impregnated with bFGF accelerated the healing process of chronic ulcers in the first-in-human clinical trial [13]. However, the equivalence of CS and CGS as a scaffold has not yet been confirmed. Therefore, in this study, the effects of the application of CSs and CGSs with and without bFGF were evaluated and compared using the same full-thickness skin defects in rats in terms of dermis-like tissue formation, epithelization, and capillary formation.

2. Material and Methods

2.1. Animals and Operations

The animals were maintained at the Bioscience Department of KAC Co., Ltd. (Rittou, Japan). The number of animals used in this study was kept to a minimum, and all possible efforts were made to reduce their suffering in compliance with the protocols established by the Animal Research Committee Determination of KAC.

2.2. Preparation of CS and CGS

We used conventional CSs (Pelnac, Gunze Co. Ltd., Ayabe, Japan) and CGSs produced according to the procedure shown in our previous study [1, 4]. Briefly, a collagen and gelatin solution with a gelatin (isoelectric point (IEP): 5.0) concentration of 10 wt% of the total solute was prepared and freeze-dried. A silicone solution (Shin-Etsu Chemical, Tokyo, Japan) was used to make silicone sheets of 200 μm in thickness containing a polyester mesh, and the top of the CGS was covered with the silicone sheet.

2.3. The Impregnation of CS and CGS with NSS and bFGF

We prepared CSs and CGSs of 8 mm in diameter on a dish (NUNC DISH 100X20 VENTS NUNCLON D SI (Thermo Fisher Scientific Inc., Yokohama, Japan)). We previously reported that CGSs impregnated with bFGF at doses of 7 μg/cm² and 14 μg/cm² accelerated the healing processes in mice and diabetic mice, respectively [12]. We therefore adopted the 7 μg/cm² dosage in the present study. At the time of impregnation, 500 μg of human recombinant bFGF, in the form of a dry powder (Fibrast Spray; Kaken Pharmaceutical, Tokyo, Japan), was dissolved in 7.14 mL of normal saline solution (NSS, Otsuka Pharmaceutical Co., Ltd., Tokyo, Japan), and 70 μg/mL (3.5 μg/50 μL) of bFGF solution was prepared. Fifty microliters of NSS was applied to CSs or CGSs in the NSS group; the same amount of bFGF solution was applied to impregnate CSs or CGSs with 7 μg/cm² of bFGF. They were incubated for 30 minutes at room temperature. Four experimental groups were generated as follows: (1) CSs impregnated with NSS (CS-NSS group); (2) CGSs impregnated with NSS (CGS-NSS group); (3) CSs impregnated with bFGF (7 μg/cm²) (CS-bFGF group); and (4) CGSs impregnated with bFGF (7 μg/cm²) (CGS-bFGF group). Eighteen samples were taken from each experimental group.

2.4. The Implantation of CS and CGS into Full-Thickness Skin Defects in Rats

We prepared seven-week-old male rats (, Slc: Wistar, SLC Japan Co., Ltd., Fukuoka, Japan). All of the rats had their backs and abdomens shaved and depilated under anesthesia by the intraperitoneal injection of pentobarbital (30 mg/kg) (Somnopenthyl®, Kyoritsu Seiyaku Corporation, Tokyo, Japan) and the inhalation of isoflurane (Escain® Pfizer Japan Inc., Tokyo, Japan). Three full-thickness skin defects measuring 8 mm in diameter were created in the back (longitudinally) of each rat, and 9 samples from three animals were included in each group. Therefore, each group had 9 samples that were adequate for the statistical analysis. Three full-thickness skin defects were created at an interval of 16 mm using an 8 mm diameter skin punch biopsy tool (Kai Industries, Gifu, Japan), a scalpel, and scissors. The panniculus carnosus was preserved. CSs or CGSs ( in each group) impregnated with either NSS or bFGF were implanted into three skin defects created on the backs of each rat (6 rats in each group). The CSs and CGSs were sutured to the marginal skin using 5-0 nylon sutures (Medical U&A Inc., Osaka, Japan), covered with gauze, and fixed in place with adhesive tape (SILKYTEX®, Alcare Co. Ltd., Tokyo, Japan).

2.5. The Assessments of the Wound Area and the Neoepithelial Length

At 1 and 2 weeks after implantation, 3 rats per group were sacrificed via the inhalation of carbon dioxide. After the removal of the silicone sheets, the wounds () were photographed, and the wound area was measured using the Image J software program (version 1.47, National Institute of Health, USA). The wound area was expressed as the percentage of the original wound area. Skin specimens, including the implanted CSs and CGSs, were harvested using scalpels and scissors and were sectioned axially at the center of each specimen. The specimens were then fixed with 20% formalin fluid (Mildfolm®, Wako Pure Chemical Industries, Osaka, Japan), paraffin-embedded, and sliced into 4 μm thick sections. The sections were then stained with hematoxylin and eosin (H&E). Using a fluorescence microscope (Biorevo BZ-9000; Keyence, Co., Osaka, Japan), the length of the neoepithelium, from the innermost hair root of the marginal skin to the end of the neoepithelium, was measured on each side of each cross-sectional area in each specimen at a magnification of ×100. The sum of the lengths of the epithelium was evaluated on both sides.

2.6. Evaluation of the Area of Dermis-Like Tissue

Azan staining sections were used to assess the newly formed dermis-like tissue. Azan stain is a complex staining method for general histology. Nuclei are stained bright red with azocarmine G and connective tissues, such as collagen fibers, are stained blue with aniline blue or orange G. The area of newly formed dermis-like tissue was measured by using the image stitching function of a fluorescence microscope (Biorevo BZ-9000). The lateral border of the regenerated area was determined by the transitional boundary layer of the continuous parallel structure of the dermis and the fragmental structure of newly formed dermis-like tissue. The lower border of the regenerated area was above the panniculus carnosus, and the upper epithelium was not included in the measurement. The area of newly formed dermis-like tissue was statistically analyzed.

2.7. Evaluation of Newly Formed Capillaries in the Dermis-Like Tissue

Immunohistological staining with von Willebrand factor antigen was performed to detect newly formed capillaries in the dermis-like tissue. After deparaffinization and rehydration, the sections were incubated in phosphate-buffered saline (PBS Life Technologies Co. Japan, Tokyo, Japan) with 0.1% trypsin (Vector Laboratories Inc., Burlingame, GA, USA) for 15 min at 37°C for antigen retrieval. Anti-von Willebrand factor rabbit polyclonal antibody (1 : 500 dilutions, Dako Japan Co., Tokyo, Japan) and EnVision⁺ Rabbit/HRP (Dako Japan Co., Tokyo, Japan) were used as the secondary antibody. These sections were exposed to DAB (3,3′-diaminobenzidine-4HCL; Dako Japan Co., Tokyo, Japan) for 2 min at room temperature. Counterstaining was performed with hematoxylin.

In each section, the newly formed capillaries were counted and the area of newly formed capillaries was measured manually in the regenerated area at a magnification of ×200, using the measurement module of a fluorescence microscope (Biorevo BZ-9000).

2.8. Statistical Analysis

All of the data were expressed as the mean + standard error (SE) and were analyzed using Tukey’s significant difference test (Tukey Kramer test). values of <0.05 were considered to indicate statistical significance.

3. Results

3.1. Wound Area

One animal died suddenly on the fifth day due to unknown reasons. The gross appearance of the wounds at 1 and 2 weeks after implantation is shown in Figure 1. At one week after surgery, wound closure was not completed in any of the four groups. At two weeks after surgery, the wound area was reduced and almost completely epithelized, and there were no signs of pus, redness, or inflammation in any of the four groups. The time course of the wound area is shown in Figure 2. At two weeks after implantation, the wound area in each group was significantly smaller than at 1 week (). There were no significant differences in wound areas of the experimental groups at 1 and 2 weeks after surgery (Figure 2).

Figure 1: The gross appearance of the full-thickness skin defects before CS and CGS implantation and wounds at 1 and 2 weeks after implantation. The gross appearance of wounds in the CS-NSS (a), CGS-NSS (b), CS-bFGF (c), and CGS-bFGF (d) groups at 1 week after implantation. The gross appearance of wounds in the CS-NSS (), CGS-NSS (), CS-bFGF (), and CGS-bFGF () groups at 2 weeks after implantation. Scale bar: 1000 μm.

Figure 2: The time course of the remaining wound area. The wounds were treated with CS-NSS, CGS-NSS, CS-bFGF, or CGS-bFGF at 1 and 2 weeks. At 1 and 2 weeks, the remaining wound area did not differ among the groups to a statistically significant extent. At 2 weeks after implantation, the areas of the wounds in all of the groups were significantly smaller than at 1 week ().

3.2. Histological Assessment of the Neoepithelial Length

Light microphotographs of the histological sections (H&E staining) at 1 and 2 weeks after implantation are shown in Figure 3. The time course of the neoepithelial length is shown in Figure 4. At two weeks, the neoepithelial length of the CGS-bFGF group was significantly longer than that of the CS-NSS and CGS-NSS groups at 1 week ( and , resp.). At two weeks, the neoepithelial length of the CGS-NSS group was significantly longer than that of the CS-NSS group at 1 week (). At two weeks after implantation, the neoepithelial length of the CGS-bFGF group was significantly longer than at 1 week () (Figure 4).

Figure 3: Light micrographs of full-thickness skin defects at 1 and 2 weeks after implantation. The wounds of the CS-NSS (a), CGS-NSS (b), CS-bFGF (c), and CGS-bFGF (d) groups at 1 week. The wounds of the CS-NSS (), CGS-NSS (), CS-bFGF (), and CGS-bFGF () groups at 2 weeks. The black arrow with a solid line indicates a hair root which is the edge of the regenerated dermis-like tissue. The black arrow with a broken line indicates the end of the neoepithelium. The neoepithelium is shown in the upper section as a black line. Scale bar: 500 μm.

Figure 4: The time course of the neoepithelial length. The wounds of the CS-NSS, CGS-NSS, CS-bFGF, and CGS-bFGF groups at 1 and 2 weeks. At 2 weeks, the neoepithelial length of the CGS-bFGF group was significantly longer than that of the CS-NSS and CGS-NSS groups at 1 week ( and , resp.). At 2 weeks, the neoepithelial length of the CGS-NSS group was significantly longer than that of the CS-NSS group at 1 week (). At 2 weeks after implantation, the neoepithelial length of the CGS-bFGF group was significantly longer than at 1 week ().

3.3. Evaluation of the Area of Dermis-Like Tissue

The regenerated area of dermis-like tissue at two weeks after implantation is shown in Figure 5. The area of dermis-like tissue in the CGS-bFGF group was significantly larger than that in the CS-NSS and CS-bFGF groups () (Figure 6).

(a)

(b)

(c)

(d)

Figure 5: Light micrographs (Azan staining) of the wounds at 2 weeks after implantation. The wounds of the CS-NSS (a), CGS-NSS (b), CS-bFGF (c), and CGS-bFGF (d) groups. The area surrounded with black line shows newly formed dermis-like tissue. Scale bar: 500 μm.

Figure 6: The comparison of the area of newly formed dermis-like tissue at 2 weeks. The newly formed dermis-like tissue in the wounds of the CGS-bFGF group was significantly larger than that of the CS-NSS and CS-bFGF groups at 2 weeks ().

3.4. Evaluation of Newly Formed Capillaries in the Dermis-Like Tissue

Light micrographs (von Willebrand factor antibody staining), which were captured at 2 weeks after surgery, are shown in Figure 7. The images show the center area of dermis-like tissue specimens of 250 μm in width and height. The number of newly formed capillaries in the CGS-bFGF group was significantly larger than that in the CS-NSS and CGS-NSS groups ( and , resp.) (Figure 8).

(a)

(b)

(c)

(d)

Figure 7: Immunohistochemical staining (von Willebrand factor antibody) of newly formed capillaries at 2 weeks after implantation. The wounds of the CS-NSS (a), CGS-NSS (b), CS-bFGF (c), and CGS-bFGF (d) groups. The open arrowheads indicate newly formed capillaries. Scale bar: 50 μm.

Figure 8: The comparison of the number and area of newly formed capillaries at 2 weeks after implantation. The number of newly formed capillaries in the wounds treated of the CGS-bFGF group was significantly higher than that of the CS-NSS and CGS-NSS groups ( and , resp.).

4. Discussion

In the present study we showed the efficacy of CGSs impregnated with bFGF in the acceleration of the wound healing process. We applied conventional CSs and CGSs (a novel scaffold that allows for the sustained release of growth factors) with and without bFGF to skin defects in rats and compared the wound healing process. A significantly higher number of capillaries were observed in the CGS-bFGF group; however, the area of newly formed capillaries was not significantly higher. This is probably because the number of specimens in the CS-NSS group was small due to the unexpected death of an animal. In other respects, CGSs impregnated with bFGF (7 μg/cm²) accelerated angiogenesis and dermis-like tissue formation. This finding is in line with the findings of our previous studies using mice, diabetic mice, rabbits, and beagle dogs [4, 12, 14]. We emphasize the formation of dermis-like tissue in the healing process, because it accelerates epithelization and prevents wound contracture [14]. Two mechanisms are involved in the wound healing process: wound contraction and epithelization. Murine wounds close mostly through wound contraction. In humans, wounds close mostly through epithelization and wound contraction causes scar contracture, decreasing the range of joint motion and/or causing hypertrophic scarring. The CGS-bFGF group was the only group in which the neoepithelial length was observed to grow from the first week to the second week. This growth was the result of dermis-like tissue formation (Figure 4). Conventional CSs have been used for the preparation of dermis-like tissue, which prevent scar contracture. In this point, the CGS-bFGF group showed the highest level of dermis-like tissue formation, which indicates that CGSs impregnated with bFGF can prevent wound contracture. In our previous studies, we developed CGSs impregnated with bFGF mainly to accelerate the healing of chronic ulcers (such as diabetic foot ulcers and venous leg ulcers); however, the present results indicate that, in addition to conventional CSs, CGSs impregnated with bFGF can be used to promote dermis-like tissue formation in the treatment skin defects.

CGS contains 10 wt% of gelatin, which degrades more easily than collagen. Another concern with regard to CGSs is the possibility that its earlier degradation (in comparison to CSs) after application in vivo is not a desirable characteristic for a scaffold. However, we did not observe a significant difference in the area of dermis-like tissue formation in the CS-NSS and CGS-NSS groups and the CGS-bFGF group showed the largest area of dermis-like tissue. These results indicate that CGSs are equal to CSs in their efficacy as scaffolds and that CGSs were not easily degraded in the accelerated angiogenesis that occurred as a result of their impregnation with bFGF.

Recently, skin substitutes containing living cells have been reported to be effective in the treatment of chronic skin ulcers [15–18]. We have reported that CSs cultured with autologous fibroblasts and CGSs impregnated with bFGF were equally effective in the treatment of diabetic ulcers [19]. We have not evaluated the combination of CGSs with living cells; however, it is thus expected that CGSs could be used in combination with cells as well as CSs. Various types of cells, including stem cells, have been used in the treatment of skin ulcers [20–22]. CGSs can achieve the sustained delivery of positively charged growth factors and can also be used as a cell carrier. In the future, other growth factors and cells will be developed. It is possible that these growth factors and cells will be combined with our CGS and used for treatment; thus, our CGS has the potential to be used in the coming era of regenerative medicine.

5. Conclusions

Wound healing after the treatment of rodent skin defects with conventional CSs and CGSs with or without bFGF was compared. The results showed that, as a scaffold, CGSs can equal the efficacy of conventional CSs without bFGF. Furthermore, we found that CGSs impregnated with bFGF (7 μg/cm²) accelerated the wound healing process which indicates that they may supersede conventional CSs in the clinical setting.

Competing Interests

The authors declare that they have no competing interests.

Authors’ Contributions

Naoki Morimoto and Chizuru Jinno conceived and designed the experiments. Naoki Morimoto, Chizuru Jinno, Michiharu Sakamoto, and Shuichi Ogino performed the experiments. Chizuru Jinno and Naoki Morimoto analyzed the data. Tsuguyoshi Taira did the preparation of materials. Chizuru Jinno, Naoki Morimoto, Ran Ito, and Shigehiko Suzuki wrote the paper.

Acknowledgments

This work was supported by Grant-in-Aid for Scientific Research B (Grant no. 24390399).

References

S. Suzuki, K. Matsuda, N. Isshiki, Y. Tamada, and Y. Ikada, “Experimental study of a newly developed bilayer artificial skin,” Biomaterials, vol. 11, no. 5, pp. 356–360, 1990.

A. C. C. Wosgrau, T. Da Silva Jeremias, D. F. Leonardi, M. J. Pereima, G. Di Giunta, and A. G. Trentin, “Comparative experimental study of wound healing in mice: pelnac versus integra,” PLoS ONE, vol. 10, no. 3, Article ID e0120322, 2015.

I. V. Yannas, D. S. Tzeranis, B. A. Harley, and P. T. C. So, “Biologically active collagen-based scaffolds: advances in processing and characterization,” Philosophical Transactions of the Royal Society A: Mathematical, Physical and Engineering Sciences, vol. 368, no. 1917, pp. 2123–2139, 2010.

S. Takemoto, N. Morimoto, Y. Kimura et al., “Preparation of collagen/gelatin sponge scaffold for sustained release of bFGF,” Tissue Engineering—art A., vol. 14, no. 10, pp. 1629–1638, 2008.

D. Gospodarowicz, “Localization of a fibroblast growth factor and its effect alone and with hydrocortisone on 3T3 cell growth,” Nature, vol. 249, no. 453, pp. 123–127, 1974.

G. S. McGee, J. M. Davidson, A. Buckley et al., “Recombinant basic fibroblast growth factor accelerates wound healing,” Journal of Surgical Research, vol. 45, no. 1, pp. 145–153, 1988.

H. Uchi, A. Igarashi, K. Urabe et al., “Clinical efficacy of basic fibroblast growth factor (bFGF) for diabetic ulcer,” European Journal of Dermatology, vol. 19, no. 5, pp. 461–468, 2009.

G. F. Pierce, J. E. Tarpley, D. Yanagihara, T. A. Mustoe, G. M. Fox, and A. Thomason, “Platelet-derived growth factor (BB homodimer), transforming growth factor-beta 1, and basic fibroblast growth factor in dermal wound healing. Neovessel and matrix formation and cessation of repair,” The American Journal of Pathology, vol. 140, no. 6, pp. 1375–1388, 1992.

K. Hirai, Y. Tabata, S. Hasegawa, and Y. Sakai, “Enhanced intestinal anastomotic healing with gelatin hydrogel incorporating basic fibroblast growth factor,” Journal of Tissue Engineering and Regenerative Medicine, 2013.

K. Ito, S. Ito, M. Sekine, and M. Abe, “Reconstruction of the soft tissue of a deep diabetic foot wound with artificial dermis and recombinant basic fibroblast growth factor,” Plastic and Reconstructive Surgery, vol. 115, no. 2, pp. 567–572, 2005.

M. Watanabe, H. Li, A. G. Kim et al., “Complete tissue coverage achieved by scaffold-based tissue engineering in the fetal sheep model of Myelomeningocele,” Biomaterials, vol. 76, pp. 133–143, 2016.

N. Kanda, N. Morimoto, A. A. Ayvazyan et al., “Evaluation of a novel collagen-gelatin scaffold for achieving the sustained release of basic fibroblast growth factor in a diabetic mouse model,” Journal of Tissue Engineering and Regenerative Medicine, vol. 8, no. 1, pp. 29–40, 2014.

N. Morimoto, K. Yoshimura, M. Niimi et al., “Novel collagen/gelatin scaffold with sustained release of basic fibroblast growth factor: clinical trial for chronic skin ulcers,” Tissue Engineering Part A, vol. 19, no. 17-18, pp. 1931–1940, 2013.

A. Ayvazyan, N. Morimoto, N. Kanda et al., “Collagen-gelatin scaffold impregnated with bFGF accelerates palatal wound healing of palatal mucosa in dogs,” Journal of Surgical Research, vol. 171, no. 2, pp. e247–e257, 2011.

J. Rice, U. Desai, L. Ristovska et al., “Economic outcomes among medicare patients receiving skin substitutes for treatment of Diabetic foot Ulcers,” Value in Health, vol. 18, no. 3, p. A43, 2015.

T. D. S. Jeremias, R. G. Machado, S. B. C. Visoni, M. J. Pereima, D. F. Leonardi, and A. G. Trentin, “Dermal substitutes support the growth of human skin-derived mesenchymal stromal cells: potential tool for skin regeneration,” PLoS ONE, vol. 9, no. 2, Article ID e89542, 2014.

C. M. Zelen, T. E. Serena, L. Gould et al., “Treatment of chronic diabetic lower extremity ulcers with advanced therapies: a prospective, randomised, controlled, multi-centre comparative study examining clinical efficacy and cost,” International Wound Journal, vol. 13, no. 2, pp. 272–282, 2016.

N. Papanas, I. Eleftheriadou, N. Tentolouris, and E. Maltezos, “Advances in the topical treatment of diabetic foot Ulcers,” Current Diabetes Reviews, vol. 8, no. 3, pp. 209–218, 2012.

N. Morimoto, T. Ito, S. Takemoto et al., “An exploratory clinical study on the safety and efficacy of an autologous fibroblast-seeded artificial skin cultured with animal product-free medium in patients with diabetic foot ulcers,” International Wound Journal, vol. 11, no. 2, pp. 183–189, 2014.

A. Condé-Green, A. A. Marano, E. S. Lee et al., “Fat grafting and adipose-derived regenerative cells in burn wound healing and scarring,” Plastic and Reconstructive Surgery, vol. 137, no. 1, pp. 302–312, 2016.

R. Kosaraju, R. C. Rennert, Z. N. Maan et al., “Adipose-derived stem cell-seeded hydrogels increase endogenous progenitor cell recruitment and neovascularization in wounds,” Tissue Engineering Part A, vol. 22, no. 3-4, pp. 295–305, 2016.

J. Rodriguez, F. Boucher, C. Lequeux et al., “Intradermal injection of human adipose-derived stem cells accelerates skin wound healing in nude mice,” Stem Cell Research & Therapy, vol. 6, no. 1, article 241, 2015.