BioMed Research International
Volume 2016 (2016), Article ID 7075464, 10 pages
http://dx.doi.org/10.1155/2016/7075464

Research Article

Motor Ingredients Derived from a Wearable Sensor-Based Virtual Reality System for Frozen Shoulder Rehabilitation

Si-Huei Lee,¹ Shih-Ching Yeh,² Rai-Chi Chan,¹ Shuya Chen,³ Geng Yang,² and Li-Rong Zheng²

¹Department of Physical Medicine and Rehabilitation, Taipei Veterans General Hospital and National Yang-Ming University, Taipei, Taiwan
²School of Information Science and Technology, Fudan University, Shanghai, China
³Department of Physical Therapy, China Medical University, Taichung, Taiwan

Received 7 March 2016; Revised 30 June 2016; Accepted 26 July 2016

Academic Editor: Emanuele Marzetti

Copyright © 2016 Si-Huei Lee et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

Abstract.

Objective. This study aims to extract motor ingredients through data mining from wearable sensors in a virtual reality goal-directed shoulder rehabilitation (GDSR) system and to examine their effects toward clinical assessment. Design. A single-group before/after comparison. Setting. Outpatient research hospital. Subjects. 16 patients with frozen shoulder. Interventions. The rehabilitation treatment involved GDSR exercises, hot pack, and interferential therapy. All patients first received hot pack and interferential therapy on the shoulder joints before engaging in the exercises. The GDSR exercise sessions were 40 minutes twice a week for 4 weeks. Main Measures. Clinical assessments included Constant and Murley score, range of motion of the shoulder, and muscle strength of upper arm as main measures. Motor indices from sensor data and task performance were measured as secondary measures. Results. The pre- and posttest results for task performance, motor indices, and the clinical assessments indicated significant improvement for the majority of the assessed items. Correlation analysis between the task performance and clinical assessments revealed significant correlations among a number of items. Stepwise regression analysis showed that task performance effectively predicted the results of several clinical assessment items. Conclusions. The motor ingredients derived from the wearable sensor and task performance are applicable and adequate to examine and predict clinical improvement after GDSR training.

1. Introduction

Frozen shoulder is considered a common intrinsic shoulder dysfunction that affects patients’ daily living activities (ADL). Aside from medication, physical therapy is also needed to effectively restore shoulder joint functionality [1]. A crucial criterion for successful physical therapy involves the appropriate use of exercise, such as range of motion (ROM) exercises, stretching [2], pulley therapy, wall-climbing exercises involving the frontal and sagittal movement planes, exercises using towels, and joint mobilization, to sever capsule adhesions and improve joint mobility and performance. In addition, sufficient amount of muscle strengthening exercises must be provided to prevent muscle atrophy that results from a lack of exercise over an extended period.

With advancements in technology, researchers have begun to adopt virtual reality (VR) techniques in rehabilitation [3–7]. In addition, wearable sensors, based on inertial measurement unit (IMU), can be used to measure the movement and provide immediate feedback to alert patients about the movement errors [8–12]. Previous studies aforementioned mostly employed VR techniques or wearable sensor to provide systems for motor training. However, motor ingredients using data mining technique from the sensor or the task performance during training were not considered as a way to evaluate the therapeutic effect; instead, clinical assessment tools (CATs) were adopted [13, 14].

In this study, we presented an innovative goal-directed shoulder rehabilitation (GDSR) system by combining VR techniques with wearable IMU (WIMU) to design rehabilitation exercises (specifically, shoulder ROM exercises, shoulder muscle strengthening, and core muscle strengthening) for patients with frozen shoulder. Based on the motor training by GDSR, we further proposed a variety of motor ingredients, including motor indices and task performance, extracted from the raw data of WIMU. The motor ingredients might provide clues revealing motor functions. The objective of this study was to investigate the effects of proposed motor indices and task performance on motor functions of shoulder. Therefore, the proposed motor indices and task performance were examined in three ways. (1) Were they significantly improved from pretest to posttest? (2) Were they significantly correlated with clinical assessment items? (3) Which one of them was more influential to the clinical assessment items?

Along with the growth of techniques of internet, cloud-computing, and big-data processing, Internet of Things (IoT) based telerehabilitation was proposed in which VR system was able to be an end-system serving home-based motor training. Beyond that, the data collected from end-system (e.g., WIMU) was further proposed to undergo big-data processing on cloud system in order to provide home-based assessment equivalent to existing clinical assessment tool (e.g., CATs), while clinical assessment was only available in hospitals instead of at home. More importantly, the assessment results might be a reference for the plan of daily training program. As a result, this was a pilot study to investigate the relevance between the data collected from WIMU and the clinical assessment items. The results of this study might contribute to understanding the feasibility of WIMU data intended to apply to home-based motor assessment.

2. Methods and Materials

2.1. Participants

In this experiment, we recruited 16 patients with frozen shoulders who met the inclusion criteria. The inclusion criteria were as follows: patients who (1) were over 20 years of age; (2) never received physical therapy; (3) exhibited normal cognition and could follow instructions when using the system; (4) were clinically confirmed to have frozen shoulders; and (5) had signed a consent form. The exclusion criteria were as follows: patients who (1) had history of a proximal humeral fracture or dislocation of glenohumeral joint; (2) received hyaluronic acid injection in the shoulder; (3) experienced cervical radiculopathy or had been diagnosed with shoulder degenerative joint diseases; (4) were at the final stage of a malignant disease; or (5) were pregnant. Overall, 6 males and 10 women satisfied the required conditions. The mean age was 58.6 years, and the average duration of injury was 22.7 months.

2.2. Interactive VR System

We designed an interactive motor training system involving shoulder joint stretching, shoulder muscle strengthening, and core muscle strengthening. The unity 3D game-engine software was adopted to formulate a goal-directed shoulder rehabilitation (GDSR), which assists patients with motor rehabilitation. Concurrently, the WIMU sensors were secured to the shoulder joints to measure and record shoulder ROM while patients were undergoing various exercises (see Figure 1). The overall system architecture is presented in Figure 2. In each rehabilitation task, patients were instructed to meet or exceed the target angle predetermined by clinicians and maintain that posture for at least 10 seconds to be considered as successfully completing the task. The GDSR featured a hierarchy of challenges, in which the difficulty level was determined by the target angles set for the rehabilitation tasks and by the number of sets required for each task.

Figure 1: The rehabilitation tasks.

Figure 2: The system architecture.

2.3. Experimental Procedures

The shoulder rehabilitation involved GDSR exercises, hot pack, and interferential therapy. All patients first received hot pack and interferential therapy on the shoulder joints before engaging in each exercise session. The exercise sessions were 40 minutes twice a week for 4 weeks. Patients were required to employ the GDSR system and complete a series of exercises using various physical objects. The exercise protocol practiced by the patients was as follows: shoulder ROM exercises, shoulder muscle strengthening, and core muscle strengthening. The practice time for each exercise was based on patients’ progress and was determined by the therapist. However, the total number of practice for three types of exercise was around 120 times. In addition, before and after the entire rehabilitation course, an independent therapist performed clinical assessments including the Constant-Murley score (CMS), ROM of active and passive shoulder movements, and upper arm muscle strength. Based on patients’ severity of injury and their rehabilitation progress, the therapists adjusted the rehabilitation goal in the GDSR system (i.e., the target angles and the total number of exercise repetitions). The experimental procedures are shown in Figure 3.

Figure 3: Experimental procedures.

2.4. Outcome Measures

In this study, we used CATs, task performances, and motor indices to examine the effect of shoulder VR rehabilitation system.

2.4.1. CATs

The CATs were the CMS assessment [15], active and passive shoulder ROM (flexion, abduction, internal rotation, and external rotation) measurement, and upper arm muscle strength test. The CMS assessment was conducted by a therapist not involved in the patient training. Active and passive shoulder ROM were measured using the goniometer in the unit of degree. Arm muscle strength was measured by muscle testing device (microFET3™) in the unit of kg (see Figure 4).

Figure 4: Measurement of upper arm muscle strength.

2.4.2. Task Performance

Task performance was defined as the patients’ abilities to complete the VR tasks in the GDSR system. Task performance was determined based on ROM goal achievable by patients and the ability to maintain postures for 10 seconds and complete a set number of repetitions.

2.4.3. Motor Indices

Based on data extracted from the IMU sensors, the following motor indices were used to assess the exercise performance.

Angular Velocity. The angular velocity was calculated via the angular trajectory time history. An increased angular velocity suggested an improvement of joint movement efficiency. The calculation formula was expressed as follows:

where is angular velocity, is angle, and is time.

Number of Interruptions during Joint Rotation. When performing the exercises, patients were asked to continuously rotate their joints to meet the target angles and complete the tasks. However, during this process, movement might be interrupted because of limited shoulder joint mobility or endurance. The number of interruptions during joint rotation was calculated via the angular trajectory time history, and it could be used as an indicator for assessing joint mobility or endurance. A decreased number of interruptions during joint rotation signified an improved joint mobility or endurance.

Varying Rate of Muscle Strength. Based on the elastic coefficient and the length change of the elastic band stretched by the patients, clinicians first calculated the shoulder muscle strength at each time-point and subsequently determined the force time history of the shoulder muscle strength. By using the force time history, the clinicians calculated the strength change per unit time and used this as an indicator to show the rate varying in muscle strength in order to determine whether the muscle strength was improved. The varying rate in muscle strength equation is expressed as follows:

where is force and is time.

2.5. Analysis Methods

Statistical analysis was performed in three ways to investigate whether the task performances and the motor indices were applicable to evaluate the therapeutic effect of rehabilitation as clinical assessment items. First, we used Wilcoxon rank sum test to examine if the difference between pretest and posttest was significant in task performances, motor indices, and clinical assessment items. Second, we performed correlation analysis to determine whether correlations of task performance and motor indices with various clinical assessment items did exist. Third, we utilized stepwise regression analysis to examine the predictability of task performance for the clinical assessment results and to examine which variable dominated the prediction.

3. Results

3.1. CATs

The three CATs (CMS assessment, active and passive shoulder ROM measurement, and upper arm muscle strength measurement) before and after the GDSR treatment are presented in Table 1. The results all indicated significant improvements ().

Table 1: CAT results ().

CATs

Mean
SD
value
Significance (2-tailed)

CMS
Before
58.4
14.3
3.41

After
79.5
9.47

ROM (degree)

    Flexion
Before
146.5
17.6
3.52

After
161.9
13.1

    Abduction
Before
134.5
33.5
3.52

After
149.6
26.1

    External rotation
Before
66.4
22.4
3.52

After
78.2
16.9

    Internal rotation
Before
49.5
21.3
3.41

After
61.1
16.8

Muscle strength (kg)

    Flexion
Before
16.45
5.47
3.52

After
22.3
6.65

    Abduction
Before
14.38
4.52
3.21

After
19.23
6.80

    External rotation
Before
14.7
3.39
2.92

After
17.6
3.01

    Internal rotation
Before
16.20
7.60
3.35

After
21.36
6.87

Significance level = 0.05.

.

3.2. Task Performance

Task performance included the maximum achievable angle of shoulder joint and the number of exercises completed. Given the small sample size, we used the Wilcoxon signed ranks test to analyze task performance between the first and eighth (last session) rehabilitation sessions. The results in Table 2 indicated that the patients’ shoulder ROM and numbers of completed exercise improved significantly after eight rehabilitation sessions over 4 weeks (). Thus, the GDSR rehabilitation significantly improved shoulder joint mobility and endurance.

Table 2: Task performance results ().

Exercise

Mean
SD
value
Significance (2-tailed)

Maximum achievable angles (degree)
Shoulder flexion
1st
136.50
15.59
−3.52

8th
160.12
10.65

Shoulder abduction in scapular plane
1st
128.93
16.72
−3.52

8th
157.00
11.34

Shoulder abduction
1st
118.37
30.02
−3.52

8th
146.68
26.23

Shoulder external rotation
1st
45.56
14.60
−3.41

8th
61.93
10.07

Shoulder internal rotation (towel)
1st
61.93
19.78
−3.52

8th
85.56
15.60

Shoulder external rotation (crutch)
1st
49.31
15.87
−3.52

8th
67.00
11.15

Shoulder external rotation (chair)
1st
86.50
12.82
−3.52

8th
110.18
9.07

Muscle strength of the abdomen and the front of the thigh
1st
57.06
14.63
−3.30

8th
79.06
5.66

Muscle strength of the abdomen and the back of the thigh
1st
16.93
5.07
−3.52

8th
27.06
5.24

Core muscle strengthening
1st
18.68
3.21
−3.47

8th
26.62
3.93

Cumulative number of repetitions of exercises
Shoulder flexion
1st
10.50
2.58
−2.96

8th
13.12
1.89

Shoulder abduction in scapular plane
1st
10.50
2.58
−2.96

8th
13.12
1.89

Shoulder abduction
1st
10.50
2.58
−2.97

8th
13.06
1.80

Shoulder external rotation
1st
9.31
3.35
−2.46

8th
12.06
2.40

Shoulder internal rotation (towel)
1st
8.75
3.15
−2.94

8th
12.12
1.45

Shoulder external rotation (crutch)
1st
9.00
3.30
−2.15

8th
11.18
2.37

Shoulder external rotation (chair)
1st
9.00
3.30
−2.00

8th
10.93
2.37

Muscle strength of the abdomen and the front of the thigh
1st
8.87
3.13
−2.52

8th
11.06
2.40

Muscle strength of the abdomen and the back of the thigh
1st
9.50
2.47
−2.39

8th
11.62
1.66

Core muscle strengthening
1st
9.25
3.13
−2.17

8th
11.06
2.40

Significance level = 0.05.

.

.

3.3. Motor Indices

The motor indices included the angular velocity, the number of interruptions during joint rotation, and the varying rate of muscle strength. We used the Wilcoxon signed ranks test to analyze the motor index between the first and eighth rehabilitation sessions. The results show that, after completing the GDSR treatment, the patients’ angular velocity for all exercises increased, the majority of which increased significantly () (Table 3). This finding suggested a significant improvement in joint mobility. The difference in the number of interruptions during joint rotation pre- and posttests was not significant; however, it indicated a downward trend. The varying rate of muscle strength was significantly increased after completing the GDSR treatment (). This finding indicated that rehabilitation significantly increased muscle strength.

Table 3: Motor indices results ().

Exercise

Mean
SD
value
Significance (2-tailed)

Angular velocities (degree/sec)
Shoulder flexion
1st
66.71
35.16
−2.20

8th
81.71
30.57

Shoulder abduction in scapular plane
1st
70.76
35.77
−2.10

8th
87.84
38.29

Shoulder abduction
1st
48.50
29.59
−2.17

8th
63.07
26.18

Shoulder external rotation
1st
62.00
63.77
−2.03

8th
83.57
76.55

Shoulder internal rotation (towel)
1st
28.21
14.18
−1.98

8th
42.64
19.27

Shoulder external rotation (crutch)
1st
37.37
38.19
−2.38

8th
67.37
44.44

Shoulder external rotation (chair)
1st
45.78
26.55
−1.76
0.079

8th
63.35
23.07

Muscle strength of the abdomen and the front of the thigh
1st
62.93
20.76
−2.79

8th
79.86
28.98

Muscle strength of the abdomen and the back of the thigh
1st
56.84
17.65
−2.48

8th
88.92
30.26

Core muscle strengthening
1st
34.81
29.14
−0.10
0.918

8th
35.68
26.47

Number of interruptions as angle increased
Shoulder flexion
1st
1.04
1.54
−1.33
0.182

8th
0.39
0.55

Shoulder abduction in scapular plane
1st
1.44
1.71
−0.08
0.937

8th
1.17
1.98

Shoulder abduction
1st
2.88
3.26
−2.66

8th
0.32
0.52

Shoulder external rotation
1st
0.31
0.74
0.00
1.000

8th
0.09
0.22

Shoulder internal rotation (towel)
1st
2.35
2.04
−2.69

8th
0.48
0.80

Shoulder external rotation (crutch)
1st
3.06
8.21
−1.60
0.109

8th
0.22
0.60

Shoulder external rotation (chair)
1st
2.15
2.91
−1.96

8th
0.70
1.14

Muscle strength of the abdomen and the front of the thigh
1st
0.28
0.70
−0.42
0.674

8th
0.29
0.72

Muscle strength of the abdomen and the back of the thigh
1st
0.21
0.40
−1.36
0.176

8th
0.03
0.06

Core muscle strengthening
1st
0.68
1.25
−0.53
0.594

8th
0.80
1.06

∆F/∆t (kg/sec)
Shoulder flexion
1st
5.57
2.63
−2.98

8th
7.70
2.65

Shoulder abduction in scapular plane
1st
6.04
2.82
−2.83

8th
8.29
3.35

Shoulder abduction
1st
4.58
2.50
−2.73

8th
6.69
2.46

Shoulder external rotation
1st
8.93
9.00
−2.37

8th
15.20
11.95

Significance level = 0.05.

.

.

3.4. Correlation Analysis

We investigated the correlation between the clinical assessments and task performances using the aforementioned results. The correlations between the clinical assessments and task performances were examined by the Pearson product-moment correlation coefficients for the differences between the pre- and posttest results. Table 4 shows the correlation coefficients between the maximum achievable angle of the task performances and the clinical assessments. Significant correlations were observed (1) between the task performance of shoulder flexion and the clinical assessment of external rotation (ROM) and (2) between the task performance of shoulder external rotation (crutch) and the clinical assessment of external rotation (muscle strength) (). Table 5 presents the correlation between the number of exercise repetitions and the clinical assessments. Shoulder flexion, shoulder abduction in scapular plane, and shoulder abduction were all significantly correlated with the clinical assessment of shoulder flexion (ROM) and shoulder external rotation (ROM) (). Shoulder internal rotation (towel) was significantly correlated with the clinical assessment of shoulder abduction (muscle strength).

Table 4: Correlations between task performances (angles) and clinical assessments.

⁢CAT
⁢Task performance

Shoulder flexion
Shoulder abduction in scapular plane
Shoulder abduction
Shoulder internal rotation (towel)
Shoulder external rotation (crutch)

External rotation (ROM)
0.504

0.178

0.102

−0.151

0.465

External rotation (muscle strength)
0.398

0.250

0.366

−0.103

0.497

Significance level = 0.05.

.

Correlation level:

: low.

: medium.

: high.

Table 5: Correlation between task performances (number of repetitions) and clinical assessments.

⁢CAT
⁢Task performance

Shoulder flexion
Shoulder abduction in scapular plane
Shoulder abduction
Shoulder internal rotation (towel)
Shoulder external rotation (crutch)

Flexion
(ROM)
0.580

0.564

0.580

0.291

−0.050

External rotation
(ROM)
0.514

0.514

0.524

0.233

0.165

Abduction
(muscle strength)
−0.158

−0.158

−0.145

−0.507

−0.427

Significance level = 0.05.

.

Correlation level:

: low.

: medium.

: high.

3.5. The Predictability of Task Performances for Clinical Assessment Results

We used stepwise regression analysis to further investigate the predictability of task performances for the clinical assessment results. Clinical assessments involved three testing items: CMS, ROM, and muscle strength. ROM and muscle strength were further divided into shoulder flexion, shoulder abduction, and internal and external shoulder rotation. Task performance involved maximum angles for various exercises and the number of exercises completed. The results for the stepwise regression analysis are presented in Tables 6 and 7.

Table 6: Task performance ability to predict flexion (ROM) according to stepwise regression analysis.

Predictor variable

Adjusted  
Estimated
coefficient
value
Significance

Shoulder abduction in scapular plane (total number of repetitions)
0.337
0.289
1.559
0.580
2.666
0.01

(a) Predictor variable in the model: shoulder abduction in scapular plane (total number of repetitions).

(b) Dependent variable: flexion (ROM).

.

Table 7: Task performance ability to predict external rotation (muscle strength) according to stepwise regression analysis.

Predictor variable

Adjusted
Estimated
coefficient
value
Significance

Shoulder external rotation-crutch (angle)
0.247
0.194
0.131
0.497
2.146
0.05

(a) Predictor variable in the model: (a constant) shoulder external rotation-crutch (angle).

(b) Dependent variable: external rotation (muscle strength).

.

Table 6 reveals that the total number of completed shoulder abduction in the scapular plane had an explained variance of 0.289 in the prediction model of the shoulder flexion ROM. This result showed a statistically significant explanatory power of 28.9% ( = 7.107, ) for the shoulder flexion ROM.

Table 7 shows that, among the various variables for predicting the effectiveness of shoulder external rotation (muscle strength), the maximum achievable angle of shoulder external rotation (crutch) had an explained variance of 0.194. This result indicated a statistically significant explanatory power of 19.4% ( = 4.604, ) for the shoulder external rotation (muscle strength). The estimated results confirmed that the maximum achievable angle could effectively predict the results of shoulder external rotation (muscle strength).

4. Discussion

We used the pre- and posttest values obtained from the CATs, task performances, and motor indices to evaluate the therapeutic effects of the GDSR exercises. Most testing items in these three outcome measurements showed significant improvements, as shown in Tables 1–3. Therefore, the motor ingredients directly derived from the GDSR system including task performance and motor indices possessed similar ability of examining the intervention effects as the CATs.

Each shoulder task performance (including the number of exercises and the maximum achievable angles) showed significant improvement, as presented in Table 2. This finding indicated that shoulder joint mobility improved significantly, which also signified an increased ability to stretch the resistance bands when performing shoulder strengthening exercises. Therefore, there is a potential that the patients’ shoulder muscle strength gradually increased with the progress of rehabilitation. Significant improvements were observed in the core muscle task performance (i.e., the number of exercises and the maximum achievable angles). Moreover, the overall muscle strength training intensified as the number of core muscle exercises increased, which indirectly enhanced the therapeutic effect on shoulder muscle strength.

Table 3 shows that, for the external rotation (chair) and core muscle (bridge) exercises, both the angular velocities and the number of interruptions during joint rotation failed to achieve a significant level but exhibited improvements. This result may be attributable to the small sample; in addition, because the core muscles were healthy, the extent of improvement was limited.

We further investigated the correlation between the maximum achievable angle and the various clinical assessment items, which are presented in Table 4. Among all correlations between the maximum achievable angles and the shoulder-motion angles identified by the CAT, only the maximum angle achieved with shoulder flexion exhibited a significant correlation with shoulder external rotations (ROM); the remaining variables did not exhibit significant correlations. This finding was primarily because when patients engaged in various exercises, they used physical objects, such as resistance bands, towels, and crutches. However, when measuring the ROM, patients did not have any physical objects; thus, the measured shoulder ROM differed, producing no significant correlation. By contrast, only the maximum achievable angle of external rotation (crutch) and the external rotation (muscle strength) exhibited a significant correlation. This result indicated that the increased improvement in external rotation mobility increased the strength of the associated muscles. However, no significant correlations were observed for the other exercises, which was due to the different physical objects used and varying usage habits.

We then investigated the correlations of the number of performed exercises with the various assessment items of the CAT (Table 5). The number of repetitions of shoulder abduction in the scapular plane and shoulder abduction showed significant correlations with the shoulder-motion measurements (i.e., flexion (ROM) and external rotation (ROM)). Thus, we infer that the patient performance in shoulder flexion and external rotation improved by repeating these three exercises.

In summary, four pairs of testing items between CAT and task performance or motor indices were identified to be significantly correlated. This finding implied that the task performance and motor indices were able to detect the therapeutic effects measured by the CATs. Nonetheless, further analysis will be needed to construct more exact relationship between these four pairs of correlated indicators.

Stepwise regression analysis was conducted to investigate the explanatory power of task performance (Tables 6 and 7). The results showed that the total number of movement repetitions of shoulder abduction in the scapular plane effectively predicted the shoulder flexion (ROM). Thus, increasing the number of this motor rehabilitation substantially improved the therapeutic effect of flexion (ROM). In addition, the maximum achievable angle of external rotation (crutch) effectively predicted the external rotation (muscle strength). Thus, increasing the achievable angles during rehabilitation improved the therapeutic effect of external rotation (muscle strength).

Previous studies [13, 14] have integrated VR techniques with motor rehabilitation treatment. These studies used only the CATs to assess the therapeutic effects, neglecting the considerable data collected during the exercises. In this study, we not only used the CATs to assess the therapeutic effects of the exercises but also adopted data derived from the innovative VR rehabilitation system developed in this study to calculate and determine the task performance and motor indices and to evaluate the therapeutic effects. Previously, Pastor et al. [16] based improvements in therapeutic effects on improvements in game performance; however, their assessment results obtained according to the CATs indicated no improvement, impeding the ability of Pastor to assess therapeutic effects. By contrast, the values of task performances and motor indices derived from the GDSR exercise system showed an improvement, demonstrating the consistency with those obtained using the objective CATs. Moreover, by using the correlation analysis, we successfully determined the best task performance items which exhibited a significant correlation with the items of the CATs. Finally, we used stepwise regression analysis and successfully identified the best task performance items which could predict CAT results in order to assess the therapeutic effect.

The study showed that the data collected from the training sessions can be converted into the ingredients related to motor functions and could be used for motor assessment effectively. The clinical evaluation measured by the motor ingredients could be conducted immediately at the end of each training session instead of being performed before and after the training sessions. The motor ingredients examined in this study provided useful and important information for planning better and individualized training programs.

The limitation of our study was the small sample size. A large sample size could reinforce our study results. In addition, the wireless sensors used in our system encountered limitations regarding transmission speed and data retrieval accuracy. Moreover, the effects of size and weight of the wireless sensors on patient joint mobility require clarification.

5. Conclusion

In this study, we successfully combined WIMU sensors and interactive VR techniques to develop a novel GDSR exercise system, which could be applied to patients with frozen shoulders as rehabilitation training for shoulder joint mobility and shoulder endurance. We conducted clinical trials on 16 patients and used the motor trajectory of the system to develop the motor indices. By using the motor indices, task performances, and CATs, we statistically confirmed the effectiveness of the GDSR system for frozen shoulder rehabilitation. Moreover, through correlation analysis, we effectively verified the correlation between task performance and CAT items. Finally, we used stepwise regression analysis to identify the best task performance item which could be used to predict the CAT results. This study provided preliminary evidence showing that data collected from WIMU might apply to motor assessment as clinical assessment did. However, further study is needed to figure out how to convert WIMU data to clinical assessment items quantitatively.

Ethical Approval

This study project was approved by the Institutional Review Board of Taipei Veterans General Hospital with VGHIRB: 2012-07-004A.

Competing Interests

The authors declare that they have no competing interests.

Acknowledgments

The authors thank the funding from the Taipei Veterans General Hospital with project ID: DV101-06.

References

H. F. Moseley, “Disorders of the shoulder,” Clinical Symposia, vol. 11, no. 3, pp. 75–102, 1959.

H. Jampol, “Exercise treatment for the frozen shoulder,” The Physical Therapy Review, vol. 30, no. 6, pp. 221–229, 1950.

D. Meldrum, A. Glennon, S. Herdman, D. Murray, and R. McConn-Walsh, “Virtual reality rehabilitation of balance: assessment of the usability of the Nintendo Wii Fit Plus,” Disability and Rehabilitation: Assistive Technology, vol. 7, no. 3, pp. 205–210, 2012.

J. P. Cuthbert, K. Staniszewski, K. Hays, D. Gerber, A. Natale, and D. O'Dell, “Virtual reality-based therapy for the treatment of balance deficits in patients receiving inpatient rehabilitation for traumatic brain injury,” Brain Injury, vol. 28, no. 2, pp. 181–188, 2014.

S.-S. Kim, W.-K. Min, J.-H. Kim, and B.-H. Lee, “The effects of VR-based Wii fit yoga on physical function in middle-aged female LBP patients,” Journal of Physical Therapy Science, vol. 26, no. 4, pp. 549–552, 2014.

A. C. Recio, D. Becker, M. Morgan, N. R. Saunders, L. P. Schramm, and J. W. McDonald, “Use of a virtual reality physical ride-on sailing simulator as a rehabilitation tool for recreational sports and community reintegration: a pilot study,” American Journal of Physical Medicine and Rehabilitation, vol. 92, no. 12, pp. 1104–1109, 2013.

M.-C. Huang, S.-H. Lee, S.-C. Yeh et al., “Intelligent frozen shoulder rehabilitation,” IEEE Intelligent Systems, vol. 29, no. 3, pp. 22–28, 2014.

S. Patel, H. Park, P. Bonato, L. Chan, and M. Rodgers, “A review of wearable sensors and systems with application in rehabilitation,” Journal of NeuroEngineering and Rehabilitation, vol. 9, article 21, 2012.

A. Salarian, F. B. Horak, C. Zampieri, P. Carlson-Kuhta, J. G. Nutt, and K. Aminian, “iTUG, a sensitive and reliable measure of mobility,” IEEE Transactions on Neural Systems and Rehabilitation Engineering, vol. 18, no. 3, pp. 303–310, 2010.

A. Weiss, T. Herman, M. Plotnik et al., “Can an accelerometer enhance the utility of the Timed Up & Go Test when evaluating patients with Parkinson's disease?” Medical Engineering and Physics, vol. 32, no. 2, pp. 119–125, 2010.

S. K. Prajapati, W. H. Gage, D. Brooks, S. E. Black, and W. E. McIlroy, “A novel approach to ambulatory monitoring: investigation into the quantity and control of everyday walking in patients with subacute stroke,” Neurorehabilitation and Neural Repair, vol. 25, no. 1, pp. 6–14, 2011.

A. Salazar, A. Silva, C. Silva et al., “Low-cost wearable data acquisition for stroke rehabilitation: a proof-of-concept study on accelerometry for functional task assessment,” Topics in Stroke Rehabilitation, vol. 21, no. 1, pp. 12–22, 2014.

J. M. Hijmans, L. A. Hale, J. A. Satherley, N. J. McMillan, and M. J. King, “Bilateral upper-limb rehabilitation after stroke using a movement-based game controller,” Journal of Rehabilitation Research and Development, vol. 48, no. 8, pp. 1005–1013, 2011.

A. Henderson, N. Korner-Bitensky, and M. Levin, “Virtual reality in stroke rehabilitation: a systematic review of its effectiveness for upper limb motor recovery,” Topics in Stroke Rehabilitation, vol. 14, no. 2, pp. 52–61, 2007.

C. R. Constant and A. H. G. Murley, “A clinical method of functional assessment of the shoulder,” Clinical Orthopaedics and Related Research, vol. 214, pp. 160–164, 1987.

I. Pastor, H. A. Hayes, and S. J. Bamberg, “A feasibility study of an upper limb rehabilitation system using Kinect and computer games,” in Proceedings of the Annual International Conference of the IEEE Engineering in Medicine and Biology Society, pp. 1286–1289, San Diego, Calif, USA, August-September 2012.