Case Reports in Gastrointestinal Medicine

Volume 2013 (2013), Article ID 652597, 3 pages

http://dx.doi.org/10.1155/2013/652597

Case Report

Delayed Migration of a WallFlex Enteral Stent Resulting in Jejunal Perforation

Phillip S. Ge,¹ Rabindra R. Watson,² David C. Chen,³ and V. Raman Muthusamy²

¹Department of Medicine, UCLA Medical Center, Los Angeles, CA 90095, USA

²Division of Digestive Diseases, UCLA Medical Center, Los Angeles, CA 90095, USA

³Department of Surgery, UCLA Medical Center, Los Angeles, CA 90095, USA

Received 7 March 2013; Accepted 4 April 2013

Academic Editors: E. Altintas, F. H. Mourad, and M. Neri

Copyright © 2013 Phillip S. Ge et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

Abstract.

Enteral stents are increasingly utilized to palliate malignant gastrointestinal obstruction; however, they can be associated with significant complications. We describe an unusual case of a 67-year-old male with gastric adenocarcinoma who underwent placement of a WallFlex metallic enteral stent to relieve a malignant gastric outlet obstruction. Four months later, while actively undergoing chemotherapy, he developed acute abdominal pain and was found to have delayed stent migration and jejunal perforation. He required emergent surgical resection of the perforated segment of jejunum. Delayed migration of the WallFlex enteral stent with subsequent visceral perforation has yet to be reported in the literature. Chemotherapy after stent placement has been associated with an increase in maintenance of stent patency; however, shrinkage of the local tumor by chemoradiation may increase the risk of stent migration. Care should be taken in placing enteral stents in patients undergoing continued treatment of their malignancy, as delayed migration of even uncovered stents may occur.

1. Introduction

Endoscopic enteral stent placement is commonly performed for palliative treatment of malignant gastric outlet obstruction [1]. This minimally invasive procedure, first described in 1992, involves the trans-endoscopic placement of a self-expandable uncovered metallic stent across a malignant obstruction [2, 3]. A number of studies have since demonstrated the safety and feasibility of this technique [4–9]. However, enteral stents can occasionally be associated with significant complications including perforation, bleeding, stent migration (immediate or delayed), stent malposition, and stent occlusion from tumor growth or food impaction [1]. The combination of delayed stent migration and perforation is extremely rare [9, 10]. Here we describe a case of delayed stent migration and jejunal perforation occurring four months after the insertion of a WallFlex metallic enteral stent to relieve malignant gastric outlet obstruction.

2. Case Report

A 67-year-old male with stage IIIB gastric adenocarcinoma treated with Billroth-II gastrojejunostomy, adjuvant chemotherapy with oxaliplatin and 5-fluorouracil, and external beam radiation therapy presented one year following completion of adjuvant therapy with abdominal pain and inability to tolerate oral intake. Imaging studies including computed tomography (CT) of the abdomen and pelvis revealed a malignant gastric outlet obstruction at the gastrojejunal anastomosis (Figure 1(a)). Following nasogastric decompression, he underwent endoscopic placement of a 90 mm × 22 mm WallFlex Enteral Stent (Boston Scientific, Natick, MA) across the stricture (Figures 1(b)–1(d)). There were no immediate complications, his abdominal pain resolved, and he resumed oral intake. He was initiated on docetaxel therapy and discharged home.

(a)

(b)

(c)

(d)

Figure 1: (a) Coronal CT image reveals a malignant gastric outlet obstruction at the gastrojejunal anastomosis. (b) The patient underwent endoscopic placement of a 90 mm × 22 mm WallFlex Enteral Stent across the malignant stricture. (c) Fluoroscopy immediately following stent placement confirms the enteral stent in place. (d) Coronary CT image obtained two months following stent placement confirms the enteral stent in place.

Four months later, after four cycles of docetaxel, he presented with severe acute abdominal pain with peritoneal signs. Laboratories revealed white blood cell count of 17,720/μL (reference: 3.28–9.29 × 10³/μL) with 93.0% neutrophils. CT imaging (Figure 2(a)) showed migration of the stent 20 cm distally into the mid-jejunum, resulting in visceral perforation, stent erosion into the anterior abdominal wall, leakage of enteric contents, and peritonitis. He emergently underwent operative resection of the perforated segment of jejunum (Figure 2(b)) with primary re-anastomosis. His postoperative course was unremarkable and he was discharged home tolerating a regular diet. He has subsequently done well on continued chemotherapy.

(a)

(b)

Figure 2: (a) Axial CT image showing migration of the stent distally into the mid-jejunum, with resultant visceral perforation. (b) Surgical specimen showing dislodged enteral stent with perforation of the mid-jejunum.

3. Discussion

Gastric outlet obstruction is a feared late complication of unresectable periampullary, distal gastric, pancreatic, and duodenal malignancy. Symptoms of gastric outlet obstruction include intractable nausea and vomiting, cachexia, and a progressive decline in clinical status and in quality of life [9]. Endoscopic enteral stent placement is becoming increasingly utilized for the minimally invasive palliative treatment of malignant gastrointestinal obstructions, with success rates favorable to surgical bypass [11]. A prospective randomized trial comparing laparoscopic gastrojejunostomy versus duodenal stenting for malignant gastric outlet obstruction showed that duodenal stenting provided a superior means of palliation compared to laparoscopic gastrojejunostomy, with decreased in-hospital complications, decreased pain, and increased quality of life, without significant difference in cumulative survival [12]. The reported technical success rate of endoscopic stenting is approximately 94–97%, with clinical success rates of 87–94%; patients are typically able to resume oral intake within 4 days [8, 13, 14].

Despite widespread use, severe complications of perforation and bleeding have only been reported in 0.7% and 0.5% of patients, respectively [8]. Nonsevere complications (stent obstruction, stent migration, pain, etc.) have been reported in 26.7% of patients, including stent migration in 5.1% [8]. Most of the existing published data reflect experience with the enteral Wallstent (Boston Scientific, Natick, MA) [7–9, 14, 15]. The limited flexibility of the Wallstent’s metal wire mesh can contribute to stent migration, and the sharp ends can cause mucosal ulceration with subsequent risk of bleeding and perforation [10]. The newer WallFlex stent has a nickel titanium alloy mesh to improve flexibility while maintaining lumen integrity, looped ends to reduce the risk of mucosal injury, and a proximal flared end to minimize the risk of stent migration [13].

Prospective studies involving the WallFlex stent have only reported rare incidences of stent migration or perforation within the first month following placement [13, 16]. Delayed migration of the WallFlex stent with visceral perforation has yet to be reported in the literature. Chemotherapy following stent placement has been associated with an increase in maintenance of stent patency [7, 17]; however, shrinkage of the local tumor by chemoradiation may increase the risk of stent migration [18]. This may be due to a reduction in luminal compression resulting in decreased contact of the intestinal wall with the stent, or to regression of tissue overgrowth of the mesh of the stent, leading to dislodgment of the device. Subsequent mechanical shear of the stent against the intestinal mucosa may cause ulceration and perforation.

As this case illustrates, care should be taken in placing enteral stents in patients undergoing continued treatment of their malignancy, as delayed migration of even uncovered stents may occur. When evaluating a patient with a history of an enteral stent presenting with acute abdominal pain, it is important to maintain a high degree of suspicion for stent migration resulting in either visceral perforation and/or intestinal obstruction. Immediate imaging should be undertaken to rule out stent-related complications, even if several months have elapsed since stent placement.

Conflict of Interests

The authors disclose no conflict of interests.

References

T. H. Baron, “Expandable metal stents for the treatment of cancerous obstruction of the gastrointestinal tract,” New England Journal of Medicine, vol. 344, no. 22, pp. 1681–1687, 2001.

R. A. Kozarek, T. J. Ball, and D. J. Patterson, “Metallic self-expanding stent application in the upper gastrointestinal tract: caveats and concerns,” Gastrointestinal Endoscopy, vol. 38, no. 1, pp. 1–6, 1992.

S. Truong, V. Bohndorf, H. Geller, V. Schumpelick, and R. W. Gunther, “Self-Expanding metal stents for palliation of malignant gastric outlet obstruction,” Endoscopy, vol. 24, no. 5, pp. 433–435, 1992.

J. K. J. Gaidos and P. V. Draganov, “Treatment of malignant gastric outlet obstruction with endoscopically placed self-expandable metal stents,” World Journal of Gastroenterology, vol. 15, no. 35, pp. 4365–4371, 2009.

M. O. Baerlocher, M. R. Asch, A. Vellahottam, G. Puri, K. Andrews, and A. Myers, “Safety and efficacy of gastrointestinal stents in cancer patients at a community hospital,” Canadian Journal of Surgery, vol. 51, no. 2, pp. 130–134, 2008.

I. Maetani, H. Isayama, and Y. Mizumoto, “Palliation in patients with malignant gastric outlet obstruction with a newly designed enteral stent: a multicenter study,” Gastrointestinal Endoscopy, vol. 66, no. 2, pp. 355–360, 2007.

J. J. Telford, D. L. Carr-Locke, T. H. Baron et al., “Palliation of patients with malignant gastric outlet obstruction with the enteral Wallstent: outcomes from a multicenter study,” Gastrointestinal Endoscopy, vol. 60, no. 6, pp. 916–920, 2004.

A. J. Dormann, S. Meisner, N. Verin, and A. Wenk Lang, “Self-expanding metal stents for gastroduodenal malignancies: systematic review of their clinical effectiveness,” Endoscopy, vol. 36, no. 6, pp. 543–550, 2004.

B. Bessoud, T. de Baere, A. Denys et al., “Malignant gastroduodenal obstruction: palliation with self-expanding metallic stents,” Journal of Vascular and Interventional Radiology, vol. 16, no. 2 I, pp. 247–253, 2005.

V. K. Thumbe, A. D. Houghton, and M. S. H. Smith, “Duodenal perforation by a wallstent,” Endoscopy, vol. 32, no. 6, pp. 495–497, 2000.

M. del Piano, M. Ballarè, F. Montino et al., “Endoscopy or surgery for malignant GI outlet obstruction?” Gastrointestinal Endoscopy, vol. 61, no. 3, pp. 421–426, 2005.

S. Mehta, A. Hindmarsh, E. Cheong et al., “Prospective randomized trial of laparoscopic gastrojejunostomy versus duodenal stenting for malignant gastric outflow obstruction,” Surgical Endoscopy and Other Interventional Techniques, vol. 20, no. 2, pp. 239–242, 2006.

J. E. van Hooft, M. J. Uitdehaag, M. J. Bruno, et al., “Efficacy and safety of the new WallFlex enteral stent in palliative treatment of malignant gastric outlet obstruction (DUOFLEX study): a prospective multicenter study,” Gastrointestinal Endoscopy, vol. 69, pp. 1059–1066, 2009.

A. P. Holt, M. Patel, and M. M. Ahmed, “Palliation of patients with malignant gastroduodenal obstruction with self-expanding metallic stents: the treatment of choice?” Gastrointestinal Endoscopy, vol. 60, no. 6, pp. 1010–1017, 2004.

P. Mosler, K. D. Mergener, J. J. Brandabur, D. B. Schembre, and R. A. Kozarek, “Palliation of gastric outlet obstruction and proximal small bowel obstruction with self-expandable metal stents: a single center series,” Journal of Clinical Gastroenterology, vol. 39, no. 2, pp. 124–128, 2005.

J. van Hooft, M. Mutignani, A. Repici, H. Messmann, H. Neuhaus, and P. Fockens, “First data on the palliative treatment of patients with malignant gastric outlet obstruction using th Wall-Flex enteral stent: a retrospective multicenter study,” Endoscopy, vol. 39, no. 5, pp. 434–439, 2007.

J. H. Kim, H. Y. Song, J. H. Shin et al., “Metallic stent placement in the palliative treatment of malignant gastroduodenal obstructions: prospective evaluation of results and factors influencing outcome in 213 patients,” Gastrointestinal Endoscopy, vol. 66, no. 2, pp. 256–264, 2007.

K. Mergener and R. A. Kozarek, “Stenting of the gastrointestinal tract,” Digestive Diseases, vol. 20, no. 2, pp. 173–181, 2002.