Evidence-Based Complementary and Alternative Medicine
Volume 2018 (2018), Article ID 5708393, 9 pages
https://doi.org/10.1155/2018/5708393

Research Article

Safety Assessment of the Auto Manipulation Device for Acupuncture in Sprague-Dawley Rats: Preclinical Evaluation of the Prototype

Geng-Hao Liu,1,2,3,4 Meng-Yen Tsai,¹ Gwo-Jyh Chang,² Chao-Min Wu,⁵ Sheng-Kai Lin,⁶ Yu-Sheng Chen,1,2 and Tzung-Yan Lee7,8

¹Division of Acupuncture and Moxibustion, Department of Traditional Chinese Medicine, Chang Gung Memorial Hospital, Linkou, Taiwan
²Graduate Institute of Clinical Medical Sciences, College of Medicine, Chang Gung University, Taoyuan, Taiwan
³School of Traditional Chinese Medicine, Chang Gung University, Taoyuan, Taiwan
⁴Sleep Center, Chang Gung Memorial Hospital, Taoyuan, Taiwan
⁵Department of Electrical Engineering, National Central University, Jhongli, Taiwan
⁶Delta Electronics, Inc., Taoyuan, Taiwan
⁷Graduate Institute of Traditional Chinese Medicine, College of Medicine, Chang Gung University, Taoyuan, Taiwan
⁸Department of Traditional Chinese Medicine, Chang Gung Memorial Hospital, Keelung, Taiwan

Correspondence should be addressed to Yu-Sheng Chen; cuspat@yahoo.com.tw and Tzung-Yan Lee; joyamen@mail.cgu.edu.tw

Geng-Hao Liu and Meng-Yen Tsai contributed equally to this work.

Received 6 June 2018; Accepted 19 July 2018; Published 6 August 2018

Academic Editor: Sang-Hoon Shin

Copyright © 2018 Geng-Hao Liu et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

Abstract.

Background. The Auto Manipulation Device for Acupuncture (AMDA) is designed for providing stable, quantified effects and higher frequency when doing lifting and thrusting manipulation. The purpose of this study is to investigate the safety of manipulation by AMDA in different frequency and duration in healthy rats. Methods. The study was divided into two parts: single intervention and once a day for a week. 12 rats and 15 rats were randomly allocated to different groups: Control (needle insertion only), AMDA (2Hz/10Mins), AMDA (2Hz/20Mins), AMDA (20Hz/10Mins), and AMDA (20Hz/20Mins) for single and repeated interventions. Real-time physiological functions, laboratory data, and the bilateral muscle tissue of acupoint (ST 36) were obtained after the intervention. Results. We found neither real-time physiological functions nor laboratory data differences between control group and AMDA groups in both parts. In the muscle tissue samples, the slight damage had been observed in the AMDA group with a frequency of 2 Hz for 20 minutes after once intervention, and the repeated session groups noted more obvious tissue damage with fibrotic change. Although the period was shorter, higher frequency manipulation caused more damage that fibroblast nuclei became more slender and obvious. However, no significant adverse effect was noted such as crippled and molting in the whole process. Conclusion. Our study suggested that the safety issue of AMDA operation in rats is feasible because there was no difference between control group and AMDA groups among real-time physiological functions and laboratory data. However, manipulation with higher frequency should be more preserved.

1. Introduction

Acupuncture is one of the most frequently requested complementary therapies [1]. Various disorders can effectively be cured by inserting long, fine needles into specific “acupuncture points” (acupoints) on the skin of the patient’s body. Besides China, acupuncture has spread to over 160 countries and regions. The World Health Organization recommends the use of acupuncture treatment for 43 diseases [2]. However, acupuncture needle manipulation is one of the most fundamental yet widely variable components of acupuncture treatments [3].

Because of the variability forms of manipulation and individual difference of acupuncturists, there exist artificial errors that make scientific studies more difficult to quantize. For providing stable and quantified effects, many researchers provide devices to monitor the frequency and amplitude of manipulation [4]. The Auto Manipulation Device for Acupuncture (AMDA) is designed for providing stable, quantified effects, and higher frequency when doing lifting and thrusting manipulation. The preliminary results have demonstrated the developed AMDA and its plausibility in the clinical application of acupuncture in simulated tissues [5].

The purpose of this study is to investigate the new method of manual manipulation (MA) by AMDA and the safety of different frequency and duration after acupuncture intervention in healthy rats.

2. Material and Methods

The study was divided into two parts: single intervention and once a day for a week (the detailed protocol was shown in Figure 1). Manual lifting-thrusting acupuncture was given with Auto Manipulation Device for Acupuncture (AMDA, prototype).

Figure 1: Protocol of experiment 1 (once intervention) and experiment 2 (repeated sessions).

2.1. Animal Preparation and Recording Procedures

27 healthy male Sprague-Dawley rats weighing 280±50 g and aged 6-8 weeks were provided by the Bio LASCO animal centre. The animals were maintained in a controlled environment (22±2°C and 50±5% humidity) and under a 12 h/12h light/dark cycle with free access to food and water. This study was performed in accordance with the Guidance Suggestions for the Care and Use of Laboratory Animals of the Animals in Science Regulation Unit of UK. All experimental procedures were approved by the Chang Gung University Institutional Animal Care and Use Committee (IACUC Approval no. CGU15-088) and were conducted in a manner that minimized the number of animals used and the number of procedures per animal.

In experiment 1 (once intervention), 12 rats were randomly allocated to different groups: Control (needle insertion only, AMDA 0 Hz/20Mins, n=4), (2Hz/10Mins), (2Hz/20Mins), (20Hz/10Mins), and (20Hz/20Mins) (n=2 each AMDA group). Real-time physiological functions, including heart rate, systolic blood pressure, mean arterial pressure, diastolic blood pressure, are measured using the tail-cuff method, recorded before the manual acupuncture with AMDA intervention and every five minutes during the course. Then, blood samples, including hepatic, renal function (AST/ALT, BUN/Cr), electrolytes (Na/K), and hemogram (CBC/DC), were also collected after sacrifice. Third, the bilateral muscle of ST36 acupoint was harvested and the histological sections were stained with hematoxylin and eosin (H&E) and were observed under a light microscope (40X, 100X) after the intervention.

In experiment 2 (repeated sessions), all 15 rats received daily manual ST36 acupuncture with AMDA intervention for 7 days and were randomly allocated to different groups: Control (0Hz/20Mins, n=3), (2Hz/10Mins), (2Hz/20Mins), (20Hz/10Mins), and (20Hz/20Mins) (n=3 each AMDA group). Real-time physiological functions, body weight change, the muscle histological sections, and blood samples were also collected in the seventh days.

2.2. Experimental Procedures

The rats were kept in supine position under anesthesia with 4% isoflurane inhalation and maintain the depth of anesthesia as stage III, which was assessed by pedal reflex, preserve normal body temperature using warm thermal pads. The ST36 point in the rat is located at a point 5 mm lateral and inferior to the tibial tubercle. Based on the comparative anatomical localization in rats as compared with that in human, selected points (the location of acupoints is shown in Figure 2) [6, 7].

(a)

(b)

(c)

(d)

Figure 2: Manual acupuncture at ST36 with AMDA and tissue sampling from ST 36. (a) ST 36 was located 4 mm bellowed and 1-2 mm lateral to the midpoint of the knee. Manual acupuncture with AMDA (b) was performed after well anesthesia. (c) After rechecking the margin of muscle (yellow) and bone (blue), the muscle tissues of ST 36 (red) were obtained (d).

The region of acupoint was shaved and disinfected; and then a sterilized single-use stainless steel needle measuring 0.27 mm ×13 mm (0.27 mm in diameter and 13 mm in length; Ching-Ming Medical Co., Ltd., Taiwan) was placed on the left side ST36 by a single experienced acupuncturist. The insertion depth was about 6 mm. After the de qi sensation, the AMDA was connected to the handle of the acupuncture needle. Afterward, adjust the device to the specific frequency and start the lifting-thrusting manipulation for each group.

In experiment 2, acupuncture was repeated for 7 days. All rats were sacrificed by decapitation, and tissue samples were collected and analyzed [8].

The procedure was carried out in accordance with the IACUC Guidelines.

2.3. Statistical Analysis

Statistical analysis was conducted using IBM SPSS Statistics 21.0 Software (IBM Corp. Released 2012. IBM SPSS Statistics for Windows, Version 21.0. Armonk, NY: IBM Corp.). All data were analyzed using Kruskal-Wallis ANOVA test and presented as mean±SD of the mean (SEM). Significance was considered when p<0.05.

3. Results

3.1. Experiment 1 (Once Intervention)

In experiment 1 (once intervention), real-time physiological functions before the intervention were collected and showed no significant difference in each group (all P>0.05). After the intervention, we found neither real-time physiological functions nor blood samples differences between control group and AMDA groups (Table 1).

Table 1: Baseline vital signs, hemogram, and biochemistry laboratory data in single intervention.

Baseline vital signs
Control group (n=4)
AMDA group (n=8)
P

    HR (bpm)
377.8±15.9
355.9±16.8
.056

    SBP (mmHg)
89.0±4.5
85.6±6.5
.379

    MBP (mmHg)
70.3±4.6
63.1±7.3
.109

    DBP (mmHg)
61.0±5.0
52.0±9.5
.110

Vital signs
Control group (n=4)
AMDA group (n=8)
P

    HR (bpm)
431.3±43.4
423.4±40.1
.734

    SBP (mmHg)
103.5±20.6
101.4±18.5
.932

    MBP (mmHg)
80.5±9.7
80.1±12.9
.865

    DBP (mmHg)
69±4.8
69.8±10.3
1.000

Hemogram

    WBC (1000/uL)
5.25±3.78
5.79±3.32
.610

    RBC (million/uL)
5.013±1.513
5.28±1.496
.610

    Hb (g/dL)
10.63±3.43
11.19±3.25
.552

    Hct (%)
33.65±11.01
35.83±10.34
.396

    MCV (fL)
66.7±2.28
67.81±2.68
.497

    MCH (pg/Cell)
21.08±0.67
21.18±0.64
.495

    MCHC (gHb/dL)
31.6±0.22
31.23±0.73
.200

    Plt (1000/uL)
436.5±494.4
470.3±420.6
.552

    Seg (%)
21.98±12.26
15.13±6.04
.308

    Lym (%)
74.3±14.29
82.56±6.1
.396

    Mono (%)
3.45±2.34
1.75±0.99
.234

    Eosin (%)
0.1±0.2
0.38±0.43
.096

    Baso (%)
0.18±0.21
0.19±0.15
.930

Biochemistry laboratory data

    BUN (mg/dL)
11.95±3.72
11.24±2.33
.609

    Cr (mg/dL)
0.27±0.062
0.244±0.041
.670

    Na (mEq/L)
146±0.8
146.1±1.8
1.000

    K (mEq/L)
5.93±0.57
6.59±0.29
.061

    AST (U/L)
112.3±5.3
111.3±7.8
.932

    ALT (U/L)
44.5±4.2
49.6±7.5
.147

HR, heart rate; SBP, systolic blood pressure; MBP, mean blood pressure; DBP, diastolic blood pressure. WBC, white blood cell; RBC, red blood cell; Hb, hemoglobin; Hct, hematocrit; MCV, mean corpuscular volume; MCH, mean corpuscular hemoglobin; MCHC, mean corpuscular hemoglobin concentration; Plt, platelet; Seg, segment; Lym, lymphocyte; Mono, monocyte; Eosin, eosinophil; Baso, basophil. BUN, blood urea nitrogen; Cr, Creatinine; Na, sodium; K, potassium; AST, aspartate aminotransferase; ALT, alanine aminotransferase.

Mean ± standard deviation was presented for vital signs (HR, SBP, MBP, and DBP), hemogram (complete blood count and differential count), and biochemistry laboratory data (BUN/Cr, Na/K, and AST/ALT). No significant difference between the 2 groups was found at baseline and once intervention study.

In the muscle tissue samples, the slight damage had been observed in the AMDA group with a frequency of 2 Hz for 20 minutes (Figure 3(a)). As long as the period prolonged, the damage increased in an order: control (AMDA 0 Hz/20Mins) < (2Hz/20Mins) < (20Hz/10Mins) < (20Hz/20Mins) (Figure 3(a)).

(a)

(b)

Figure 3: Once intervention (a) and repeated sessions (b) of AMDA effect on muscle tissue at acupoint ST36 in different needle frequency and retention time. The bar of proportional scale is 100 um. AMDA with higher frequency (20Hz) make clearly visible tissue tear (black arrowhead). Regardless of the needle frequency in repeated sessions, fibrotic change (yellow arrowhead) was noted. As the needle retention time was extended to 20 minutes, the fibroblast nuclei (yellow arrow) are also more slender and obvious. (a) Once intervention. ls, low frequency (2Hz) and short duration (10mins); ll, low frequency (2Hz) and long duration (20mins); hs, high frequency (2Hz) and short duration (10mins); hl, high frequency (2Hz) and long duration (20mins). (b) Repeated sessions. LS, low frequency (2Hz) and short duration (10mins); LL, low frequency (2Hz) and long duration (20mins); HS, high frequency (2Hz) and short duration (10mins); HL, high frequency (2Hz) and long duration (20mins).

3.2. Experiment 2 (Repeated Sessions)

After 7 days of repeated acupuncture intervention, the data presented in Table 2 showed that it still had no significant change in the real-time physiological function, including heart rate, systolic blood pressure, mean arterial pressure, and diastolic blood pressure in the rats. Also, it did not affect hepatic or renal function (AST/ALT, BUN/Cr), electrolyte (Na/K), or hemogram (CBC/DC).

Table 2: Vital signs, hemogram, and biochemistry laboratory data in repeated sessions study.

Control group (n=3)
AMDA group (n=12)
P

Rat body weight gain (%)
12.85±2.25
12.69±2.65
.942

Vital signs

    HR (bpm)
411.3±17.4
382.7±29.7
.149

    SBP (mmHg)
84.7±11.2
87.1±9.8
.771

    MBP (mmHg)
68.3±13
67.3±10.6
1.000

    DBP (mmHg)
59.3±14.2
57.6±11.8
.828

Hemogram

    WBC (1000/uL)
7.4±2.96
8.43±4.09
.613

    RBC (million/uL)
6.357±1.384
6.801±0.698
.773

    Hb (g/dL)
13.47±2.8
14.21±1.3
.772

    Hct (%)
41.83±8.82
44.48±3.89
.773

    MCV (fL)
65.87±0.81
65.58±3.11
.563

    MCH (pg/Cell)
21.2±0.26
20.93±0.89
.347

    MCHC (gHb/dL)
32.23±0.21
31.94±0.36
.128

    Plt (1000/uL)
516.7±373.2
523.7±322.2
.885

    Seg (%)
16.33±2.25
16±7.41
.248

    Lym (%)
81.5±2.43
81.45±7.45
.278

    Mono (%)
1.4±0.1
1.53±0.93
.884

    Eosin (%)
0.5±0.2
0.76±0.6
.563

    Baso (%)
0.27±0.06
0.26±0.12
.939

Biochemistry laboratory data

    BUN (mg/dL)
15.2±3.87
14.3±3.43
.773

    Cr (mg/dL)
0.213±0.015
0.23±0.045
.716

    Na (mEq/L)
145.3±1.5
146.8±2.2
.210

    K (mEq/L)
6.57±0.12
6.93±0.61
.346

    AST (U/L)
121.7±16.3
125.8±23.7
.885

    ALT (U/L)
50.7±2.3
54.1±8.1
.469

Mean ± standard deviation was presented for the percentage of rat body weight gain, vital signs (HR, SBP, MBP, and DBP), hemogram (complete blood count and differential count), and biochemistry laboratory data (BUN/Cr, Na/K, and AST/ALT). No significant difference between the 2 groups was found at chronic AMDA study.

In addition, there was no significant body weight change between those groups (P=0.220). No significant adverse effect was noted such as crippled and molting in the whole process.

In the muscle tissue samples, the damage had been observed in all AMDA groups. In comparison to once intervention group with a frequency of 2Hz for 10 minutes, the repeated session groups noted more obvious tissue damage with fibrotic change (Figure 3(b)). Our study found that although the period was shorter, higher frequency manipulation causes more damage that fibroblast nuclei became more slender and obvious (Figure 3(b)). In conclusion, the damage increased in an order: control (AMDA 0 Hz/20Mins) < (2Hz/10Mins) < (2Hz/20Mins) < (20Hz/10Mins) < (20Hz/20Mins).

4. Discussion

The main purpose addressed by this study was whether AMDA was available for providing a stable and safe new method acupuncture manipulation with different frequency and duration in rats. As most patients receive more than one acupuncture therapy in the treatment course, this study was conducted in two phases for investigating short-term and repeated effects. We hypothesized that MA manipulation by AMDA may affect the rats in three ways: real-time physiological functions, laboratory finding, and tissue damage. It follows that neither real-time physiological functions nor laboratory blood test differences between control group and AMDA groups in both time courses. Moreover, there was no significant adverse effect noted during the course. The results presented here reveal the safety issue of AMDA operation in rats is feasible.

However, according to the histologic review, a  single  AMDA  operation with 20 Hz will have a slight damage to muscle tissue. After daily manipulation with AMDA, regardless of the needle frequency, muscle tissue over acupoint (ST36) became fibrosis. Moreover, AMDA with higher frequency (20Hz) make clearly visible tissue tear. As the needle retention time was extended to 20 minutes, the fibroblast nuclei are also more slender and obvious.

For fibroblasts and endothelial cells, focal adhesions form mechanical links between extracellular collagen matrix and intracellular cytoskeleton. It was evident that the needle movement was an effective mechanical stimulus leading to tissue displacement [9]. Tissue tension is likely sensed by fibroblast by their adhesion to collagen fibers [10, 11]. Langevin observed that mechanical coupling between the needle and connective tissue with a winding of tissue around the needle during needle manipulation transmits a mechanical signal to connective tissue cells via mechanotransduction [9, 12].

We knew that the therapeutic effectiveness of acupuncture could be influenced by multiple factors. Basic science experiments, mostly in animals and healthy human subjects, show that acupuncture needling has demonstrable physiological effects that are dependent on needling parameters, including needle insertion depth, type, amplitude, and frequency of needle stimulation [13]. Between therapeutic effectiveness of acupuncture and connective tissue damage, the control of parameters of manipulation was quite important. In this study, we can deduce that the most important factor cause tissue damage is the frequency of manipulation, and then it is the time of needle retention and repeated times of manipulation.

Another study that evaluates the variable frequencies of manual acupuncture at ST36 in rats with atropine-induced inhibition of gastric motility also found that twirling manipulations with frequencies of 1, 2, and 3 Hz had better therapeutic effects than a frequency of 4 Hz on the recovery of the gastric motility amplitude [14]. Therefore, the frequency of manual acupuncture influences not only the therapeutic effects but the safety of acupuncture intervention.

On the other hand, electroacupuncture at ST36 increases the concentration and reorganization of collagen in the rat model of tendon healing [15]. The subtle differences in acupuncture needle manipulation techniques can affect cellular responses in mouse subcutaneous connective tissue [16]. Further studies will be needed to determine whether those manipulations are related to therapeutic responses.

Now, modern imaging and cell biology techniques have been employed to study the nature of acupuncture and many researchers build many models to explain acupuncture such as mechanistic function, neural response, or electrical response. For example, earlier studies have shown that rotation of an inserted acupuncture needle stretches nearby connective tissue by pulling collagen fibers from the periphery toward the needle [11, 17]. On the other hand, current study found that the acoustic shear wave, being a mechanical energy, is capable of mechanotransduction, stimulating cytosolic Ca2+ rise in both excitable and nonexcitable cells, producing Ca2+ oscillations and memory, and giving rise to in vivo calcium fluorescence and endorphin release into blood plasma in mice [18]. However, there were some current studies which suggest that the initial action of acupuncture appears to be mechanical and not neural or electrical. The mechanism of acupuncture still needs further evidence to be proved.

As for the manipulation effect for fibroblasts, previous studies have shown that both acupuncture needle rotation and simple tissue stretching cause fibroblasts to increase their cross-sectional area, as their cell bodies expand and spread out [11, 19]. The fibroblast responsiveness along a plane of connective tissue could be the source of purines that led to adenosine-mediated acupuncture analgesia some distance away from the needle [20]. The response of fibroblasts to acupuncture still needs further study to distinguish the benefit and the adverse effect of a high frequency intervention.

However, the size proportion of the needle to the body of a rat was different from that of a human. Thus, the muscle tissue damage may be overrated in the animal model. Even though our study only focuses on rats, the damage of repeated needle intervention still was an important issue. Skin changes such as localized lipoatrophy and hypertrophic scar have been reported in some review articles, especially during a relatively long treatment period [21]. In some case reports, epithelioid granuloma, pseudolymphoma, and scars at needling sites were also mentioned [22].

Guidelines of World Health Organization on basic training and safety in acupuncture have proposed the safety in acupuncture, including prevention of infection, contraindications, management of accidents, and untoward reactions [23]. We suggested avoiding high frequency and repeated manipulation, based on our result of muscle pathology and clinic observation of patients with acupoint-fibrosis. At the same time, try to avoid the same acupoint in the treatment course for decreasing the stimulation of same muscle tissue and the incidence of muscle tissue fibrosis.

One advantage of the study is that we try to provide a new and safe method for acupuncture study to solve the variability forms of manipulation and individual difference of acupuncturists, which is another way to reduce artificial errors and quantified the effects. The limit of our study includes the related small amount sample and may need further study for evaluation the fibrosis tissue and the differences between manipulations. On the other hand, even though the experienced acupuncturist performed the needle manipulation to the de qi sensation first, the followed AMDA could not sense and maintain it in the whole process. Another question that may be asked is whether the finding from rats is applicable to humans.

5. Conclusions

Our study suggested that the safety issue of AMDA operation in rats is feasible because there was no difference between control group and AMDA groups among real-time physiological functions and laboratory sample test in both intervention courses. However, lifting-thrusting manipulation with higher frequency should be more preserved, especially in patients that need more often acupuncture intervention, such as chronic arthritis, sciatica, and cerebrovascular disease. Further studies will be needed to investigate the potential of AMDA.

Data Availability

The data used to support the findings of this study are available from the corresponding author upon request.

Conflicts of Interest

The authors declared no conflicts of interest.

Authors’ Contributions

Details of the each author with his/her contribution in this paper are as follows: Geng-Hao Liu contributed to acupuncture, sample collection, statistics, and writing of manuscript; Meng-Yen Tsai contributed to writing of manuscript; Gwo-Jyh Chang contributed to tail-cuff equipment lending; Chao-Min Wu contributed to AMDA prototype supervisor; Sheng-Kai Lin contributed to AMDA prototype maker; Tzung-Yan Lee is supervisor; Yu-Sheng Chen contributed to origin idea of AMDA and is supervisor. Geng-Hao Liu and Meng-Yen Tsai contributed equally to this study.

Acknowledgments

The authors thank Chin-Chang Chen, Ph.D., for animal care and euthanasia. This study was supported by grants from the Chang Gung Memorial Hospital (Grant no. CMRPG3E1041, YS-Chen, and CMRPG5E0141, GH-Liu) in Taiwan

Supplementary Materials

The low and high frequency (2/20 Hz) manipulation of the AMDA prototype have been recorded as a video file and as a supplementary document. (Supplementary Materials)

References

M. Silvert, “Acupuncture wins BMA approval,” BMJ, vol. 321, no. 7252, pp. 11–11, 2000.

Z.-Q. Zhao, “Neural mechanism underlying acupuncture analgesia,” Progress in Neurobiology, vol. 85, no. 4, pp. 355–375, 2008.

R. T. Davis, D. L. Churchill, G. J. Badger, J. Dunn, and H. M. Langevin, “A new method for quantifying the needling component of acupuncture treatments,” Acupuncture in Medicine, vol. 30, no. 2, pp. 113–119, 2012.

Xiaomei Li, Yanqi Li, Jingzi Chen et al., “The Influence of Skin Microcirculation Blood Perfusion at Zusanli Acupoint by Stimulating with Lift-Thrust Reinforcing and Reducing Acupuncture Manipulation Methods on Healthy Adults,” Evidence-Based Complementary and Alternative Medicine, vol. 2013, Article ID 452697, 7 pages, 2013.

. Chao-Min Wu, . Sheng-Kai Lin, . Yu-Sheng Chen, and . Geng-Hao Liu, “Development of Automatic Manipulation Device for Acupuncture (AMDA),” in Proceedings of the 2014 36th Annual International Conference of the IEEE Engineering in Medicine and Biology Society (EMBC), pp. 518–521, Chicago, IL, August 2014.

Y.-Q. Li, B. Zhu, P.-J. Rong, H. Ben, and Y.-H. Li, “Effective regularity in modulation on gastric motility induced by different acupoint stimulation,” World Journal of Gastroenterology, vol. 12, no. 47, pp. 7642–7648, 2006.

R. Zhang, L. Lao, K. Ren, and B. M. Berman, “Mechanisms of acupuncture-electroacupuncture on persistent pain,” Anesthesiology, vol. 120, no. 2, pp. 482–503, 2014.

S.-Y. Wu, W.-H. Chen, C.-L. Hsieh, and Y.-W. Lin, “Abundant expression and functional participation of TRPV1 at Zusanli acupoint (ST36) in mice: mechanosensitive TRPV1 as an ‘acupuncture-responding channel’,” BMC Complementary and Alternative Medicine, vol. 14, article 96, 2014.

E. S. Yang, P.-W. Li, B. Nilius, and G. Li, “Ancient Chinese medicine and mechanistic evidence of acupuncture physiology,” Pflügers Archiv - European Journal of Physiology, vol. 462, no. 5, pp. 645–653, 2011.

J. J. Tomasek, G. Gabbiani, B. Hinz, C. Chaponnier, and R. A. Brown, “Myofibroblasts and mechano: regulation of connective tissue remodelling,” Nature Reviews Molecular Cell Biology, vol. 3, no. 5, pp. 349–363, 2002.

N. Goldman, D. Chandler-Militello, H. M. Langevin, M. Nedergaard, and T. Takano, “Purine receptor mediated actin cytoskeleton remodeling of human fibroblasts,” Cell Calcium, vol. 53, no. 4, pp. 297–301, 2013.

H. M. Langevin, E. E. Konofagou, G. J. Badger et al., “Tissue displacements during acupuncture using ultrasound elastography techniques,” Ultrasound in Medicine & Biology, vol. 30, no. 9, pp. 1173–1183, 2004.

H. M. Langevin, P. M. Wayne, H. MacPherson et al., “Paradoxes in Acupuncture Research: Strategies for Moving Forward,” Evidence-Based Complementary and Alternative Medicine, vol. 2011, Article ID 180805, 11 pages, 2011.

L.-L. Gao, Y. Guo, T. Sha et al., “Differential effects of variable frequencies of manual acupuncture at ST36 in rats with atropine-induced inhibition of gastric motility,” Acupuncture in Medicine, vol. 34, no. 1, pp. 33–39, 2016.

M. Dos Santos De Almeida, K. M. De Freitas, L. P. Oliveira et al., “Acupuncture increases the diameter and reorganisation of collagen fibrils during rat tendon healing,” Acupuncture in Medicine, vol. 33, no. 1, pp. 51–57, 2015.

H. M. Langevin, N. A. Bouffard, D. L. Churchill, and G. J. Badger, “Connective tissue fibroblast response to acupuncture: dose-dependent effect of bidirectional needle rotation,” The Journal of Alternative and Complementary Medicine, vol. 13, no. 3, pp. 355–360, 2007.

H. M. Langevin, D. L. Churchill, and M. J. Cipolla, “Mechanical signaling through connective tissue: a mechanism for the therapeutic effect of acupuncture,” The FASEB Journal, vol. 15, no. 12, pp. 2275–2282, 2001.

G. Li, J.-M. Liang, P.-W. Li et al., “Physiology and cell biology of acupuncture observed in calcium signaling activated by acoustic shear wave,” Pflügers Archiv - European Journal of Physiology, vol. 462, no. 4, pp. 587–597, 2011.

H. M. Langevin, N. A. Bouffard, G. J. Badger, J. C. Iatridis, and A. K. Howe, “Dynamic fibroblast cytoskeletal response to subcutaneous tissue stretch ex vivo and in vivo,” American Journal of Physiology-Cell Physiology, vol. 288, no. 3, pp. C747–C756, 2005.

H. M. Langevin, T. Fujita, N. A. Bouffard et al., “Fibroblast cytoskeletal remodeling induced by tissue stretch involves ATP signaling,” Journal of Cellular Physiology, vol. 228, no. 9, pp. 1922–1926, 2013.

S. Park, W. Kim, J. Mun et al., “Adverse events associated with acupuncture: a clinicopathologic review,” International Journal of Dermatology, vol. 55, no. 7, pp. 757–763, 2016.

Shifen Xu, Lizhen Wang, Emily Cooper et al., “Adverse Events of Acupuncture: A Systematic Review of Case Reports,” Evidence-Based Complementary and Alternative Medicine, vol. 2013, Article ID 581203, 15 pages, 2013.

World Health Organization, “Guidelines on Basic Training and Safety in Acupuncture,” 1999.