Hindawi Publishing Corporation
International Journal of Biomaterials
Volume 2009 (2009), Article ID 765813
doi:10.1155/2009/765813

Research Article

Adhesion of Respiratory-Infection-Associated Microorganisms on Degradable Thermoplastic Composites

Teemu Tirri,¹ Eva Söderling,¹ Minna Malin,² Matti Peltola,³ Jukka V. Seppälä,² and Timo O. Närhi1,4

¹Department of Prosthetic Dentistry and Biomaterials Science, Institute of Dentistry, University of Turku, 20520 Turku, Finland
²Laboratory of Polymer Technology, Department of Chemical Technology, Helsinki University of Technology, 02150 Espoo, Finland
³Department of Otorhinolaryngology - Head and Neck Surgery, Turku University Central Hospital, Turku, 20520 Turku, Finland
⁴Clinic of Oral Diseases, Turku University Central Hospital, Turku, 20520 Turku, Finland

Correspondence should be addressed to Timo O. Närhi, timo.narhi@utu.fi

Received 5 October 2008; Accepted 3 March 2009

Recommended by Toshiki Miyazaki

The purpose of this study was to evaluate bacterial adhesion and early colonization on a composite consisting of bioactive glass (BAG) particles and copolymer of -caprolactone/D,L-lactide. Materials were incubated with suspensions of both type strains and clinical isolates of Streptococcus pneumoniae, Haemophilus influenzae, and Pseudomonas aeruginosa for 30 minutes (adhesion) and 4 hours (colonization). Clear differences exist in the microorganisms' ability to adhere on the experimental materials. However, the presence of BAG particles does not inhibit bacterial adhesion, but early colonization of the materials with P. aeruginosa was inhibited by the addition of 90–315 m BAG particles.

Copyright © 2009 Teemu Tirri et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

1. Introduction

Tissue integrating biomaterials can be utilized in various applications in head and neck and craniomaxillofacial surgeries such as the sinus augmentations and obliterations, repair of fractured orbital floor, and occluding nasal septum perforations. Bioactive glass (BAG) plates and granules are currently available for clinical use for these applications [1–5]. Good biocompatibility and osteopromotive properties of bioactive glasses are well known, which make them interesting materials for reconstructive surgery. However, glass itself is a brittle material and it cannot be easily shaped during the surgery. Composites of BAG and biodegradable polymers may offer better handling properties without losing glasses bioactive function [6, 7].

A composite of copolymer of poly(-caprolactone and D,L-lactide) and BAG (S53P4) was recently developed. The composite degrades via hydrolysis and it forms hydroxyapatite layer on its surface in simulated body conditions [8, 9]. The composite is thermoplastic and it can be shaped during the surgery. Materials' osteoconductive capacity has been proven in experimental animal study [7]. Thus, the composite of poly(-caprolactone and D,L-lactide) and BAG is a potential new material for surgical corrections of tissue defects in craniofacial area.

Biofilm forms usually on implant surfaces within few hours when they are exposed to an environment where they are continuously bathed in bacteria-containing fluids. Bacterial colonization can either be beneficial or destructive for the implanted device [10]. Materials that enhance biofilm formation may cause chronic infections in the implantation site. No data is available about early colonization of polymer composites containing BAG. In principle, in the upper respiratory tract area, there is a risk to bacterial colonization of implants. The present study aimed to evaluate the adhesion and colonization reflecting early biofilm formation of clinically important head and neck area pathogens Streptococcus pneumoniae, Haemophilus influenzae, and Pseudomonas aeruginosa [11, 12] on a composite of copolymer of poly(-caprolactone and D,L-lactide) and BAG.

2. Materials and Methods

2.1. Composite Substrates

Bioactive thermoplastic composites were made by incorporating 60 wt% of bioactive glass granules (BAG) (S53P4, Vivoxid Ltd., Turku, Finland) into the copolymer of poly(-caprolactone and D,L-lactide) through blending [8]. The composition of the BAG by wt% was Si 53%, NO 23%, CaO 20%, and 4%, while the particle size was 90–315 m (C90) and 45 m (C45). Copolymer without BAG was used as a control (C). Round test specimens (surface area 78.5 m, thickness 2 mm) for bacterial cultivation were punched out of the cast composite plates. Specimens were packed and sterilized with -radiation (dose 25 Gy, Gammaster, Waageningen, The Netherlands). A representative sample of the specimens was characterized prior to the microbial cultivation by measuring their surface roughness and wetting properties. Wetting performance was determined by contact angle measurement.

2.2. Microorganisms

The following microorganisms were used in the experiments: S. pneumoniae ATCC 49619, S. pneumoniae clinical isolate 1186; H. influenzae ATCC 49247, H. influenzae clinical isolate 1258; P. aeruginosa ATCC 27853, P. aeruginosa clinical isolate 1218. The clinical isolates were kind gifts from Dr. Erkki Eerola, Department of Microbiology, University of Turku, Finland.

2.3. Cultivation of the Microorganisms

For the adhesion experiments the microorganisms were cultured in Brain Heart Infusion medium (BHI; Difco Laboratories, Mich, USA). H. influenzae and P. aeruginosa were grown overnight at , producing log-phase cells with the inoculum used. For H. influenzae the medium was supplemented with -nicotinamide adenine dinucleotide (10 mg/L) and hemin (20 mg/L) according to Kuo et al. [13]. The fast-growing S. pneumoniae was grown in BHI supplemented with heat-inactivated fetal calf serum (10% v/v; Promocell, Heidelberg, Germany). It was first grown overnight whereafter an inoculum was transferred to fresh medium in the morning and the cells were cultured for 3–4 hours resulting in log-phase cells. The cells were harvested by centrifugation, washed once with saline (0.9% NaCl w/v) and suspended in saline to an optical density of appr. 0.25 at the wavelength 660 nm. The S. pneumoniae cell suspension was prepared as described before but the optical density of the suspension was adjusted to appr. 0.5. The cell densities were chosen based on preliminary adhesion experiments showing in scanning electron microscopy (SEM) even distribution of the cells on the material surfaces with abundant space for surface colonization.

2.4. Adhesion Tests

The cell suspensions were prepared for the adhesion experiments as described above. The adhesion experiments were essentially performed as described earlier [14]. Each tested disc was incubated with 1.5 mL of the cell suspension at room temperature for 30 minutes using gentle rolling (The Coulter Mixer, Luton, UK) in a 14 mL capped plastic test tube with an inner diameter of 16 mm (Falcon, BD Biosciences, Bedford, Mass, USA). The mixer was tilted in a 15-degree angle to ensure that the materials were covered by the cell suspension at all times. After the rolling the material was rinsed gently 3 times in 50 mL saline. The cells adhering to the two flat surfaces of each disc were carefully scraped from the surfaces in 900 L of transport medium (Tryptic Soy Broth, Difco Laboratories), each side of the disc in a separate vial. Three applicators dipped into fresh transport medium before the procedure (Quick-Stick, Dentsolv AB, Sweden) were used to scrape the cells from one side on the disc, the brush ends of the applicators were cut into the transport media. In preliminary experiments increasing the number of scrapings did not increase the cell yield. Vortexing removed efficiently the cells from the ends of the applicators. The experiments were performed with 3–4 replicates and repeated at least once.

For the enumeration of cells on the disc surfaces as colony-forming units (CFU), the microbe samples collected from the surfaces were thoroughly Vortexed and grown on agar plates after serial dilutions of the samples. H. influenzae was grown on chocolate agar, S. pneumoniae and P. aeruginosa on blood agar overnight at in air supplemented with 7% C.

In some experiments, the materials were subjected to fixation followed by SEM as described in what follows.

2.5. Early Surface Colonization

For testing early surface colonization the materials were, after the 30 minutes rolling in the bacterial suspensions described earlier, dipped in 50 mL saline to remove cells not attached to the surfaces. The materials were then transferred to fresh growth media described in Section 2.3. The culturing was performed using rolling in 1.5 mL of growth medium in the tube used in the adhesion experiments at for 4 hours. The numbers of cells on the material surfaces were determined as described in Section 2.4. The pHs of the growth media were determined at the end of the incubation to ensure that the pH of the medium was not affected by the material, the cells growing on the material surface or shed to the medium.

The early colonization experiments were performed with 3–4 replicates. The experiments were repeated at least once.

2.6. SEM

For the SEM examinations the samples were fixed for 5 minutes (2% glutaraldehyde and 2% formaldehyde in phosphate-buffered saline, pH 7.4), rinsed once in distilled water and dried in an ascending ethanol series: 50% EtOH for 5 minutes, 70% EtOH for 10 minutes, two times 96% EtOH for 10 minutes, and absolute EtOH for 5 minutes. Finally the specimens were sputter coated with gold and examined with SEM (Model JSM 5500, JEOL Ltd., Tokyo, Japan).

2.7. Statistical Analysis

Statistical analysis was done with StatView and Graphics Program (SAS institute, Cary, NC, USA). Differences in the mean numbers of the microbes (CFUs) harvested form the experimental materials were tested with an ANOVA after logarithmic transformation. Post Hoc comparisons were made with Fisher's PLSD test after the test for equal means was found to be significant at level.

3. Results

Ra value of the samples with small glass particles was lower than that of samples with big glass particles ( m versus  m). Contact angle of the samples with small glass particles was also smaller than that with big glass particels ( versus ).

Clear differences were observed in bacterial adhesion and early stages of surface colonization of all experimental materials when using the type strains. H. influenzae (ATCC 49247, clinical isolate 1258) adhered significantly better on the copolymer C when compared to P. aeruginosa (ATCC 27853, clinical isolate 1218; mean difference 0.944, ) and S. pneumoniae (ATCC 49619, clinical isolate 1186; mean difference 0.430, ). Significant difference was also noticed between S. pneumoniae and P. aeruginosa (mean difference 0.514, ).

Adhesion of P. aeruginosa was significantly higher to C90 as compared to the C (mean difference 1.000; ). The adhesion was also higher to C90 as compared to C45 (mean difference 0.756, ). The adhesion of H. influenzae and S. pneumoniae did not differ between the tested materials (Figure 1).

(a)

(b)

(c)

Figure 1: Attachment of (lg CFU/mL) P. aeruginosa (ATCC 27853), H. influenzae (ATCC 49247), and S. pneumoniae (ATCC 49619) on the experimental materials C, C45, and C90.

Similar results were obtained with the type strains and the clinical isolates of the microorganisms in the adhesion experiments. Thus the surface colonization experiments were performed with the type strains only.

P. aeruginosa showed rapid surface colonization during the culturing and was superior to H. influenzae in this respect (Figures 1 and 2). S. pneumoniae was not able to colonize the material surfaces in the experimental conditions used and the number of viable cells harvested from the materials after the 4-hour culturing actually decreased ().

(a)

(b)

Figure 2: Early colonization (lg CFU/mL) by P. aeruginosa (ATCC 27853) and H. influenzae (ATCC 49247) on the experimental materials C, C45, and C90.

For P. aeruginosa the differences observed in the cell colonization evened up after the 4-hour culturing indicating that C90 inhibited surface colonization (Figure 2): no differences among the materials were observed although significantly more viable cells had adhered on C90 during the 30 minutes adhesion. H. influenzae showed surface colonization of all materials (Figure 2). No colonization inhibition was observed, actually, slightly higher numbers of cells were harvested from the C45 than the C discs after the 4-hour culturing.

Presence of BAG in the material had no effect on the final pH of the culture media. No differences among the experimental materials were noticed in this respect. No visible turbidity indicating growth in the medium was observed after the 4-hour culturing.

3.1. SEM

The SEM examinations verified the results obtained by culturing the cells harvested from the material surfaces. Roughness of the sample surfaces varied according to the size of BAG particles incorporated into the copolymer matrix. The smoothest surface was noticed on the copolymer surface C, while the material C90 had the roughest surface texture. Microbes seemed to attach evenly on the materials' surfaces, but no microbes were detected directly on the surfaces of the exposed BAG granules as is shown in Figure 3 for P. aeruginosa after 4 hours of culturing. The rough surface of C45 and C90 appeared to favor adhesion of P. aeruginosa but this was not seen for H. influenzae and S. pneumoniae. The results of the SEM analyses appeared to be in agreement with the quantitative CFU assessments of the cells from the material surfaces.

(a)

(b)

Figure 3: SEM image of the P. aeruginosa (ATCC 27853) attachment on the composite (a) and on the C90 disc (b) after 30-minute immersion (white arrows). Bioactive glass granules are visible on the surface of C90 material (black arrows). Images are inverted into negatives to show the attached microbes.

4. Discussion

Primary adhesion between bacteria and implant devices is mediated by nonspecific interactions [10]. Before the adhesion, implant surface is conditioned by the adsorption of water and proteins, which alters its original surface properties. Thus, in clinical environment bacterial adhesion and subsequent biofilm development is strongly determined by materials ability to adsorb fluids from their surroundings.

In the present study all the experimental materials consisted of poly(-caprolactone/D,L-lactide) or its composites with BAG. Due to the polycaprolactone blocks the copolymer itself is hydrophobic and has relatively poor wettability [8]. The presence of BAG increases water adsorption, but the adsorption occurs mainly through the interface of the BAG and copolymer matrix. As formed, all BAG granules are immediately embedded in the copolymer matrix and even near the surface granules are covered by a thin polymer film. The samples used in this study were prepared by casting, which resulted in different topographies on the specimen surfaces. Casting was chosen in order to achieve circumstances which resemble clinical situation after material application. Due to the thermoplastic nature of the materials, a thin polymer skin covers the embedded galss granules even after the material is shaped into its desired form. Grinding and polishig would have resulted in more uniform surface texture, but thermoplastic nature and softness of the material prevented using this method.

In this study the hydrophobic nature of copolymer did not prevent the adhesion of microorganisms as all tested microbes attached on the experimental materials. Copolymer samples without BAG showed relatively smooth surface texture. However, the presence of BAG particles induced irregularities on the sample surfaces, which increased the surface area of all the samples containing BAG.

The purpose of this work was to study attachment of H. influenzae, S. pneumoniae, and P. aeruginosa to the composites as well as net accumulation of the bacteria to the materials during the 4-hour exposure to the growth medium. The growth media did not show turbidity after the experiments, thus cells shed from the material surfaces should not have interfered with the experiments. In this study, we used scraping of the cells and not labelling of them as earlier [14] since we wanted to collect both sides of the discoid material specimens for separate assessment. This enabled also the assessment of CFUs and SEM detection from the same specimen. The repeatability of the scraping method was good as judged by the small standard deviations of the replicates. Thus the differences in bacterial adhesion and early colonization are related with the differences in the composition and surface topography of the tested materials.

In the previous studies BAG has shown to possess direct antimicrobial properties against S. mutans, Porphyromonas gingivalis, Actinobacillus actinomycetemcomitans, and Actinomyces naeslundii [4, 15, 16] and even against Candida albicans [17]. In these studies a clear antimicrobial effect has been noticed when BAG has been used in the particulate form. In this form, BAG's active surface area is large and it increases significantly the pH of the aqueous environment on its surrounding. It is likely that the main antimicrobial effects of BAG are related to its reactivity and capability to increase the pH.

In our study all the materials were in solid discoid form. Although BAG particles are reactive and capable of changing pH in the vicinity of the composite surface [18] it did not prevent bacterial attachment in this study. The attachment of P. aeruginosa was increased by the presence of large BAG granules, most probably due to larger surface area, while the attachment of H. influenzae and S. pneumoniae was influenced by the chemical nature of the material. In a previous study H. influenzae and S. pneumoniae attached poorly on BAG plates [4, 19]. Embedding BAG in the copolymer apparently hinders the surface reactions of the BAG particles since only the granules close to the composite surface are exposed to dissolution in the short-term immersion. Despite that, in our study the presence of BAG appeared to inhibit colonization, reflecting early biofilm formation, as the expected values based on bacterial attachment did not increase. As expected, P. aeruginosa could colonize on the materials, and also H. influenzae was able to grow on the materials and stay attached to them, while S. pneumoniae did not colonize at all. The differences in the inhibiton of colonization among the studied microorganisms is propably related with the surface reactivity of the experimental materials. Prevention of P. aeruginosa colonization was especially clear on substrates containing large BAG particles.

Within the limitations of this study it can be concluded that composites of poly(-caprolactone and D,L-lactide) and BAG do not inhibit the adhesion of S. pneumoniae, H. influenzae or P. aeruginosa to the copolymer. However, the presence of BAG seems to inhibit their colonization. This is especially clear for P. aeruginosa, which is known to colonize easily on surfaces. One of the major drawbacks in the use of biomaterials is the occurrence of biomaterials centred infections [20] and, thus, any method to prevent them is beneficial. The findings in the present study suggest that in terms of bacterial colonization the composite of poly(-caprolactone and D,L-lactide) and BAG is a promising material for medical devices in head and neck and craniomaxillofacial applications.

Acknowledgments

The skilful technical assistance of the biomedical laboratory technologist, Oona Hällfors, is gratefully acknowledged. This study was conducted under the umbrella of the Center of Excellence of Bio- and Nanopolymers Research Group, which is jointly funded by the Academy of Finland (CoE program 77317), National Technology Agency, TEKES (no. 40696/01) and the participating institutes.

References

T. Furusawa and K. Mizunuma, “Osteoconductive properties and efficacy of resorbable bioactive glass as a bone-grafting material,” Implant Dentistry, vol. 6, no. 2, pp. 93–101, 1997.

M. Peltola, J. Suonpää, K. Aitasalo, M. Varpula, A. Yli-Urpo, and R.-P. Happonen, “Obliteration of the frontal sinus cavity with bioactive glass,” Head and Neck, vol. 20, no. 4, pp. 315–319, 1998.

K. Aitasalo, I. Kinnunen, J. Palmgren, and M. Varpula, “Repair of orbital floor fractures with bioactive glass implants,” Journal of Oral and Maxillofacial Surgery, vol. 59, no. 12, pp. 1390–1395, 2001.

P. Stoor, E. Söderling, and R. Grénman, “Bioactive glass S53P4 in repair of septal perforations and its interactions with the respiratory infection-associated microorganisms Heamophilus influenzae and Streptococcus pneumoniae,” Journal of Biomedical Materials Research Part B, vol. 58, no. 1, pp. 113–120, 2001.

T. Turunen, J. Peltola, A. Yli-Urpo, and R.-P. Happonen, “Bioactive glass granules as a bone adjunctive material in maxillary sinus floor augmentation,” Clinical Oral Implants Research, vol. 15, no. 2, pp. 135–141, 2004.

T. Tirri, T. Jaakkola, T. Närhi, J. Rich, J. Seppälä, and A. Yli-Urpo, “Pre-treated bioactive composite in rat soft tissue,” Key Engineering Materials, vol. 192–195, pp. 653–656, 2001.

T. O. Närhi, J. A. Jansen, T. Jaakkola, et al., “Bone response to degradable thermoplastic composite in rabbits,” Biomaterials, vol. 24, no. 10, pp. 1697–1704, 2003.

J. Rich, T. Jaakkola, T. Tirri, T. Närhi, A. Yli-Urpo, and J. Seppälä, “In vitro evaluation of poly(-caprolactone-co-DL-lactide)/bioactive glass composites,” Biomaterials, vol. 23, no. 10, pp. 2143–2150, 2002.

T. Jaakkola, J. Rich, T. Tirri, et al., “In vitro Ca-P precipitation on biodegradable thermoplastic composite of poly(-caprolactone-co-DL-lactide) and bioactive glass (S53P4),” Biomaterials, vol. 25, no. 4, pp. 575–581, 2004.

W. M. Dunne Jr., “Bacterial adhesion: seen any good biofilms lately?” Clinical Microbiology Reviews, vol. 15, no. 2, pp. 155–166, 2002.

R. J. Schlosser, S. D. London, J. M. Gwaltney Jr., and C. W. Gross, “Microbiology of chronic frontal sinusitis,” Laryngoscope, vol. 111, no. 8, pp. 1330–1332, 2001.

I. Brook, “Bacteriology of acute and chronic frontal sinusitis,” Archives of Otolaryngology—Head & Neck Surgery, vol. 128, no. 5, pp. 583–585, 2002.

J. S.-C. Kuo, V. W. Doelling, J. F. Graveline, and D. W. McCoy, “Evidence for covalent attachment of phospholipid to the capsular polysaccharide of Heamophilus influenzae type b,” Journal of Bacteriology, vol. 163, no. 2, pp. 769–773, 1985.

J. Tanner, A. Carlén, E. Söderling, and P. K. Vallittu, “Adsorption of parotid saliva proteins and adhesion of Streptococcus mutans ATCC 21752 to dental fiber-reinforced composites,” Journal of Biomedical Materials Research Part B, vol. 66, no. 1, pp. 391–398, 2003.

P. Stoor, V. Kirstilä, E. Söderling, I. Kangasniemi, K. Herbst, and A. Yli-Urpo, “Interactions between bioactive glass and periodontal pathogens,” Microbial Ecology in Health and Disease, vol. 9, no. 3, pp. 109–114, 1996.

P. Stoor, E. Söderling, and J. I. Salonen, “Antibacterial effects of a bioactive glass paste on oral microorganisms,” Acta Odontologica Scandinavica, vol. 56, no. 3, pp. 161–165, 1998.

H. Yli-Urpo, T. Närhi, and E. Söderling, “Antimicrobial effects of glass ionomer cements containing bioactive glass (S53P4) on oral micro-organisms in vitro,” Acta Odontologica Scandinavica, vol. 61, no. 4, pp. 241–246, 2003.

T. O. Närhi, J. E. de Ruijter, T. Jaakkola, et al., “Proliferation of osteoblast like cells on a new thermoplastic bioactive resin,” Journal of Dental Research, vol. 79, p. 3329, 2000.

P. Stoor, E. Söderling, Ö. H. Andersson, and A. Yli-Urpo, “Interaction between the frontal sinusitis-associated pathogen Heamophilus influenzae and the bioactive glass S53P4,” Key Engineering Materials, vol. 8, pp. 253–258, 1995.

B. Gottenbos, H. J. Busscher, H. C. van der Mei, and P. Nieuwenhuis, “Pathogenesis and prevention of biomaterial centered infections,” Journal of Materials Science: Materials in Medicine, vol. 13, no. 8, pp. 717–722, 2002.