International Journal of Biomaterials

Volume 2012 (2012), Article ID 615018, 7 pages

doi:10.1155/2012/615018

Research Article

Assessment of the Quality of Newly Formed Bone around Titanium Alloy Implants by Using X-Ray Photoelectron Spectroscopy

Hiroshi Nakada,1,2 Toshiro Sakae,2,3 Yasuhiro Tanimoto,2,4 Mari Teranishi,⁵ Takao Kato,2,5 Takehiro Watanabe,¹ Hiroyuki Saeki,¹ Yasuhiko Kawai,1,2 and Racquel Z. LeGeros⁶

¹Department of Removable Prosthodontics, Nihon University School of Dentistry at Matsudo, 2-870-1 Sakaecho-Nishi Matsudo City, Chiba 271-8587, Japan

²Research Institute of Oral Science, Nihon University School of Dentistry at Matsudo, 2-870-1 Sakaecho-Nishi Matsudo City, Chiba 271-8587, Japan

³Department of Histology, Nihon University School of Dentistry at Matsudo, 2-870-1 Sakaecho-Nishi Matsudo City, Chiba 271-8587, Japan

⁴Department of Dental Biomaterials, Nihon University School of Dentistry at Matsudo, 2-870-1 Sakaecho-Nishi Matsudo City, Chiba 271-8587, Japan

⁵Department of Oral Implantology, Nihon University School of Dentistry at Matsudo, 2-870-1 Sakaecho-Nishi Matsudo City, Chiba 271-8587, Japan

⁶Department of Biomaterials and Biomimetics, New York University College of Dentistry, 345 East 24th Street, New York, NY 10010, USA

Received 7 December 2011; Accepted 29 February 2012

Academic Editor: Yo Shibata

Copyright © 2012 Hiroshi Nakada et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

Abstract.

The aim of this study was to evaluate differences in bones quality between newly formed bone and cortical bone formed around titanium alloy implants by using X-ray photoelectron spectroscopy. As a result of narrow scan measurement at 4 weeks, the newly formed bone of C1s, P2p, O1s, and Ca2p were observed at a different peak range and strength compared with a cortical bone. At 8 weeks, the peak range and strength of newly formed bone were similar to those of cortical bone at C1s, P2p, and Ca2p, but not O1s. The results from this analysis indicate that the peaks and quantities of each element of newly formed bone were similar to those of cortical bone at 8 weeks, suggestive of a strong physicochemical resemblance.

1. Introduction

Dental implantation is a treatment method in which fixtures are implanted in the jawbone, followed by prosthetic implantation after a resting period of approximately 3–6 months, at which time new bone is formed around the fixtures [1]. This process sets the cortical bone as the primary anchorage unit. Bone modeling and remodeling processes are important (1) to induce long-term stability of the implants; (2) to develop osseointegration between implant materials and the bone; (3) to allow the maturation of new bone around the implants. It has been reported that the maximum occlusal force in adults with a natural dentition is 430 N [2], and similar loads are likely to be applied to implants as well as normal prostheses. Therefore, to achieve long-term retention and stability of implants under such conditions, the quality of newly formed bone around implants is important.

Many studies have reported that newly formed bone around implants is spongy bone [3]. However, although the morphology of newly formed bone is reportedly like spongy bone, it is difficult to discriminate whether the bone quality is mature like cortical bone, or immature like spongy, osteoid, or cartilaginous bone; therefore, evaluation of the bone quality is required.

The quality of bone forming around implants has been investigated by various groups. Nakano et al. [4] evaluated bone density and alignment of biological apatite (BAp), and Boskey and Pleshko [5] used Fourier-transform infrared (FTIR) imaging to assess bone and cartilage quality and composition. In addition, our group has reported on the use of polarized microscopy [6] and scanning electron microscopy (SEM) [7] to evaluate new bone and cortical bone quality, as well as microscopic Raman spectroscopy to analyze phosphate peaks of bone apatite [6], and microcomputed tomography (micro-CT) to assess trabecular microarchitecture and bone mineral density (BMD) [8–10]. However, basic research on new bone formation has been sparse, and molecular and elemental characterization of BAp, the basic building block of bone, has generally been overlooked. Moreover, responses to early dynamic loading in implants, with analysis of changes in new bone quality associated with mineralization, remain an important issue for further research.

In this study, differences in the bone quality between newly formed bone and cortical bone formed around titanium (Ti) alloy implants were investigated using X-ray photoelectron spectroscopy (XPS).

2. Materials and Methods

2.1. Experimental Animals

Six 18-week-old New Zealand White Rabbits (Sankyo Labo Service Co., Tokyo, Japan) were used in experiments. The rabbits were housed in individual metal cages at a room temperature of C and humidity of %, with ad libitum access to food and water. The experimental protocol was approved by an animal experimentation ethics committee (approval number ECA 07-0016). All experiments were conducted according to the Guidelines for the Treatment of Animals, Nihon University, Chiba, Japan.

2.2. Materials

Implants (diameter, 3.0 mm; length, 7.0 mm) were fabricated using Ti-15%Zr-4%Nb-4%Ta (Ti-15-4-4) alloy [11–15] (Table 1 displays its chemical composition). Surface treatment of implants consisted of sulfuric acid-etching and grit-blasting with apatitic abrasive (particle size 250 μm, HiMed Co., NY, USA) (Figure 1).

Table 1: Chemical composition (mass %) of the Ti-15-4-4 alloy used in the present study.

Titanium alloy
(mass %)

Zr
Nb
Ta
Pd
Fe
O
N
H
C
Ti

Ti-15-4-4
15.24
3.90
3.92
0.22
0.022
0.162
0.048
0.011
0.002
Bal.**

**Bal.: balance.

(a)

(b)

Figure 1: Scanning electron microscope photographs of implant surface at (a) low magnification and (b) high magnification.

2.3. Implantation

Rabbits underwent general anesthesia with 2.0 mg/kg of intravenous Ketalar (Daiichi Sankyo, Tokyo, Japan). Implant cavities were surgically created in the tibia 10 mm distal to the knee joint, one each bilaterally, by using a 1.0 mm and 3.0 mm diameter round bar, while irrigating the area with sterile saline. Implants were inserted into the right tibia, with each rabbit receiving one implant (8 implants were used in total). After surgery, the areas were disinfected with tincture of iodine for 3 days. Rabbits were sacrificed by anesthesia overdose at 4 or 8 weeks after-implantation, and the tibias were resected.

2.4. XPS Analysis

For XPS analysis, newly formed bone in close proximity to implants and cortical bone (control) which not in close proximity to implants were analyzed using thin-cut nondecalcified histological specimens. Since both bone resorption by osteoclasts and bone formation by osteoblasts progress concurrently, differences in measurement of bone tissue results are apt to occur; therefore, it is difficult to specify the measurement area. In the present study, more than 3 areas of newly formed bone close to implants and those of cortical bone were measured, and sites within which appropriate average values were obtained were determined as the measurement areas (Figure 2). An XPS analysis was performed under the following conditions: X-ray source: monochromatic AlKα (1,486.6 eV), detection region: 20 μmθ, and detection depth: approximately 4-5 nm (take-off angle: 45°).

Figure 2: Measurement areas for XPS analysis (upper: thin section specimens; lower: enlarged images; +: measurement areas). Newly bone: newly formed bone.

Qualitative analysis was performed using wide scan measurement, and the chemical bonding conditions of detected elements were analyzed using narrow scan measurement. Hydroxyapatite powders were used as the standard specimens, followed by performance of measurements with a correction for relative sensitivity factors (RSFs).

3. Results

Qualitative analysis of the peak strengths on newly formed bone around implants and cortical bone at 4 and 8 weeks is shown in Figure 3. Although peaks of Ca, O, P, C, Mg, N, and Na were detected, a marked unevenness in these peaks was observed at 4 weeks, in contrast to 8 weeks. Based on the results of wide scan measurement, narrow scan measurement of elements related to Ca₁₀(PO₄)₆OH₂ was performed. The resulting overlap of C1s, O1s, Ca2p, and P2p of newly formed and cortical bone at 4 weeks is shown in Figures 3 and 4, respectively; the results at 8 weeks are shown in Figure 5.

(a)

(b)

(c)

(d)

Figure 3: Wide scan peaks of each element at 4 and 8 weeks. Newly bone: newly formed bone.

(a)

(b)

(c)

(d)

Figure 4: Narrow scan peaks of each element at 4 weeks. Newly bone: newly formed bone.

(a)

(b)

(c)

(d)

Figure 5: Narrow scan peaks of each element at 8 weeks. Newly bone: newly formed bone.

The results of narrow scan measurement at 4 weeks (Figure 4) indicate that, although the peak strengths of newly formed and cortical bone were almost equal at O1s, the half-width became smaller in newly formed bone. At P2p and C1s, although a shifting of the peak in newly formed bone was observed, the half-width became smaller in newly formed bone; the peak strength of newly formed bone at C1s were lower than that of cortical bone. At Ca2p (Ca2p 3/2), although the peak strengths of newly formed bone was lower than that of cortical bone, the half-width became smaller in newly formed bone, and a shifting of the peak was observed. The chemical bonding condition observed for each element in newly formed bone differed from that of cortical bone.

As a result of narrow scan measurement at 8 weeks (Figure 5), the half-width and strength of newly formed bone were almost equal to those of cortical bone at C1s, P2p, and Ca2p. At O1s, the half-width became small in newly formed bone, and a shifting of the peak was observed. The chemical bonding condition observed for each element of newly formed bone was similar to that of cortical bone.

Table 2 shows the results of the quantitative analysis of each element and the Ca/P ratio. The results for newly formed bone were Ca: weight percent and P: weight percent at 4 weeks; Ca: weight percent and P: weight percent at 8 weeks. These values gradually became similar to those of cortical bone at 4 weeks (Ca: weight percent and P: weight percent) and at 8 weeks (Ca: weight percent and P: weight percent).

Table 2: Results of the quantitative analysis of newly formed bone around implants at 4 and 8 weeks, and cortical bone. Newly bone: newly formed bone.

(Weight %)

Ca
P
Ca/P

4 weeks newly bone

4 weeks cortical bone

8 weeks newly bone

8 weeks cortical bone

4. Discussion

The use of XPS analysis in the present experiment has been employed in industry since the 1960s and is now being clinically applied to the qualitative analysis of several nanometer-sized areas on material surfaces. Elemental analysis showed complex spectra based on chemical bond arrangement and elemental compositional changes. Changes in spectra reveal alterations in chemical bonds based on changes in interatomic transition and valence-band state density.

In the present study, newly formed bone at 4 weeks showed differences in the Ca/P ratio (Table 2), and in peak position, peak height, and half-width on the narrow scan measurement of each element (Figures 4 and 5). Many mineral elements were present in newly formed bone at 4 weeks, suggesting that the arrangement of chemical bonding and element composition in newly formed bone differed from those in cortical bone. Furthermore, the peak position, peak height, and half-width on narrow scan measurement at 8 weeks in newly formed and cortical bone showed similar traces, and the quantitative analysis of new and cortical bone at 8 weeks (Table 2) showed similar results. These results suggest that newly formed bone at 8 weeks showed a similar bone quality to that of cortical bone, due to progressive bone maturation and bone metabolism of minerals.

Each element showed a complex spectrum with changes in the arrangement of chemical bonding regarding the peak and element composition. The spectrum is related to changes in the chemical bonding of elements caused by interatomic transition and changes in the density or state of the valence band. Under the atomic arrangement of BAp crystals in immature new bone, various minor and trace elements, including , Na⁺, and Mg²⁺, were substituted for Ca²⁺, , and . Such substitutions have been shown to affect the properties of BAp crystals [16]. For example, substitution of for in the apatite lattice can cause changes in chemical bonding that leads to strain and reduced crystallite size in BAp [16, 17]. Such changes in chemical bonds can result in changes in binding energy in the magnitude of several electron volts (eV). In chemical bonds, trace element peaks shift to higher energy as differences in the electronegativity of bond-forming elements and the electron valence of elements increase. Peak shifts are easily influenced by the surrounding molecular environment, and peak shift and half-width variations signify the presence of compounds with different molecular weights and atomic arrangement. The peak value in a chemical bond is clear with many reports [18–21]. The main combined states of each element and a corresponding peak are shown in Table 3.

Table 3: Main chemical bonding of each element and peak position (eV).

Peak
Binding energy (eV)
Peak position

Metal—O
530-531

O1s
Metal—POx, −OH
531-532

C=O, C–O–C
532-533

C–C, C–H
284.8

C1s
C–O, C–N
286

C=O
287-288

O=C–O
289

Ca
346

Ca2p 3/2
CaO
346-347

Ca₁₀ (PO₄)₆ (OH)₂
346-347

Metal—P
129

P2p
Metal—POx
133.5

P₂O₅
135

Ca₁₀ (PO₄)₆ (OH)₂
133–133.5

C–N, C=N, N–H
398.5–401

Nls
Metal–N
396–398

NO
401–403

Si
99.4

Si2p
SiOx, SiOC
101–103

SiO₂
103–103.6

In the present study, differences were observed in XPS spectra between new bone at 4 weeks postimplantation and cortical bone, indicating immature bone quality due to BAp imperfection. In the chemical bonds of each spectrum, changes in the order of 0.1 eV affected BAp crystallinity [22].

The XPS spectra of new bone at 8 weeks postimplantation resembled those of cortical bone, and quantitative analysis showed higher Ca and P in newly formed bone compared to 4 weeks. This indicated that the composition of newly formed bone was closer to that of cortical bone at 8 weeks. Osseointegration around implants occurs as mineralization progresses, involving the accumulation of mineral components or BAp nanocrystals [23]. However, the present study was unable to analyze in detail the chemical bonding present in BAp nanocrystals, a subject for future analysis.

Acknowledgments

This study was supported in part by Grants-in-Aid for Scientific Young Scientists (B) (22791942 and 23792297) and a Grant for Supporting Project for Strategic Research (S0801032) 2008–2012 from The Ministry of Education, Culture, Sports, Science and Technology, Japan.

References

P. I. Branemark, “Osseointegration and its experimental background,” The Journal of Prosthetic Dentistry, vol. 50, no. 3, pp. 399–410, 1983.

A. Van Der Bilt, A. Tekamp, H. Van Der Glas, and J. Abbink, “Bite force and electromyograpy during maximum unilateral and bilateral clenching,” European Journal of Oral Sciences, vol. 116, no. 3, pp. 217–222, 2008.

T. Albrektsson, “Direct bone anchorage of dental implants,” The Journal of Prosthetic Dentistry, vol. 50, no. 2, pp. 255–261, 1983.

T. Nakano, K. Kaibara, Y. Tabata et al., “Unique alignment and texture of biological apatite crystallites in typical calcified tissues analyzed by microbeam X-ray diffractometer system,” Bone, vol. 31, no. 4, pp. 479–487, 2002.

A. Boskey and N. Pleshko Camacho, “FT-IR imaging of native and tissue-engineered bone and cartilage,” Biomaterials, vol. 28, no. 15, pp. 2465–2478, 2007.

Y. Numata, H. Nakada, T. Sakae et al., “Assessments of the new bone quality around implants using microscopic FTIR and polarizing microscope,” Journal of Hard Tissue Biology, vol. 17, no. 3, pp. 131–140, 2008.

H. Nakada, T. Sakae, R. Z. Legeros et al., “Early tissue response to modified implant surfaces using back scattered imaging,” Implant Dentistry, vol. 16, no. 3, pp. 281–289, 2007.

H. Nakada, T. Suwa, Y. Numata et al., “Analysis of bone mineral density and bone area of newly formed bone around implants using micro-CT after pull-out tests,” International Journal of Oral-Medical Sciences, vol. 5, no. 2, pp. 97–106, 2007.

H. Nakada, Y. Numata, T. Sakae et al., “Comparison of bone mineral density and area of newly formed bone around Ti-15%Zr-4%Nb-4%Ta alloy and Ti-6%A1-4%V alloy implants,” Journal of Hard Tissue Biology, vol. 17, no. 3, pp. 99–108, 2008.

H. Nakada, S. Suzuki, T. Sakae et al., “Quantitative and qualitative analyses of low-mineral-diet ovariectomised rat femora using microscopic computed tomography,” Journal of Hard Tissue Biology, vol. 20, pp. 107–114, 2011.

Y. Okazaki, S. Rao, Y. Ito, and T. Tateishi, “Corrosion resistance, mechanical properties: corrosion fatigue strength and cytocompatibility of new Ti alloys without A1 and V,” Biomaterials, vol. 19, no. 13, pp. 1197–1215, 1998.

Y. Okazaki, E. Nishimura, H. Nakada, and K. Kobayashi, “Surface analysis of Ti-15Zr-4Nb-4Ta alloy after implantation in rat tibia,” Biomaterials, vol. 22, no. 6, pp. 599–607, 2001.

Y. Okazaki, “Effect of friction on anodic polarization properties of metallic biomaterials,” Biomaterials, vol. 23, no. 9, pp. 2071–2077, 2002.

Y. Okazaki, E. Gotoh, T. Manabe, and K. Kobayashi, “Comparison of metal concentrations in rat tibia tissues with various metallic implants,” Biomaterials, vol. 25, no. 28, pp. 5913–5920, 2004.

Y. Okazaki and E. Gotoh, “Comparison of metal release from various metallic biomaterials in vitro,” Biomaterials, vol. 26, no. 1, pp. 11–21, 2005.

R. Z. LeGeros, “Calcium phosphates in enamel,” in Calcium Phosphates in Oral Biology and Medicine, M. M. Howard, Ed., pp. 108–110, Karger, Basel, Switzerland, 1991.

R. Z. LeGeros, “Apatites in biological systems,” Progress in Crystal Growth And Characterization, vol. 4, no. 1-2, pp. 1–45, 1981.

C. D. Wagner, W. M. Riggs, L. E. Davis, J. F. Moulder, and G. E. Mullenberg, “A reference book of standard data for use in X-ray Photoelectron spectroscopy,” in Handbook of X-ray Photoelectron Spectroscopy, G. E. Muilenberg, Ed., Perkin-Elmer, Eden Prairie, Minn, USA, 1979.

A. Ide-Ektessabi, T. Yamaguchi, and Y. Tanaka, “RBS and XPS analyses of the composite calcium phosphate coatings for biomedical applications,” Nuclear Instruments and Methods in Physics Research B, vol. 241, no. 1–4, pp. 685–688, 2005.

M. Morra, C. Cassinelli, L. Meda, M. Fini, G. Giavaresi, and R. Giardino, “Surface analysis and effects on interfacial bone microhardness of collagen-coated titanium implants: a rabbit model,” International Journal of Oral and Maxillofacial Implants, vol. 20, no. 1, pp. 23–30, 2005.

T. F. Stoica, C. Morosanu, A. Slav et al., “Hydroxyapatite films obtained by sol-gel and sputtering,” Thin Solid Films, vol. 516, no. 22, pp. 8112–8116, 2008.

H. Nakada, Y. Numata, T. Sakae et al., “Changes in bone quality associated with the mineralization of new bone formed around implants—using XPS, polarized microscopy, and FTIR imaging,” Journal of Hard Tissue Biology, vol. 19, no. 2, pp. 101–110, 2010.

Y. Shibata and T. Miyazaki, “Anode glow discharge plasma treatment enhances calcium phosphate adsorption onto titanium plates,” Journal of Dental Research, vol. 81, no. 12, pp. 841–844, 2002.