International Journal of Surgical Oncology

Volume 2012 (2012), Article ID 868623, 4 pages

doi:10.1155/2012/868623

Clinical Study

Analysis of the Impact of Intraoperative Margin Assessment with Adjunctive Use of MarginProbe versus Standard of Care on Tissue Volume Removed

Ronald J. Rivera,¹ Dennis R. Holmes,² and Lorraine Tafra³

¹Los Angeles Center for Women's Health and David Geffen School of Medicine, University of California, Los Angeles, CA 90095, USA

²Los Angeles Center for Women's Health and Keck School of Medicine, University of Southern California, Los Angeles, CA 90015, USA

³Anne Arundel Medical Center, Annapolis, MD 21401, USA

Received 11 August 2012; Accepted 22 October 2012

Academic Editor: Joseph P. Crowe

Copyright © 2012 Ronald J. Rivera et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

Abstract.

Breast conserving surgery has been accepted as the optimal local therapy for women with early breast cancer, emphasizing the necessity to balance oncologic goals with patient satisfaction and cosmetic outcomes. In the move to enhance a surgeon's ability to achieve histologically clear margins intraoperatively at the initial surgery, the MarginProbe (Dune Medical Devices, Caesarea, Israel) has emerged as an effective tool to accomplish that task. Based on previously reported success using the device, we assessed cosmesis and tissue resection volumes among participants in a randomized-controlled trial comparing the standard of care lumpectomy performed with and without the MarginProbe. The use of the MarginProbe device resulted in a 57% reduction in reexcision rates compared to the control group with a small increase in tissue volume removed at the primary lumpectomy. When total tissue volumes removed were analyzed, the device and control groups were still very similar after normalization to bra cup size. We concluded that the MarginProbe is an effective device to assist surgeons in determining margin status intraoperatively while allowing for better patient cosmetic outcomes due to the smaller volumes of tissue resected and the reduction in patient referrals for second surgeries due to positive margins.

1. Introduction

Since the acceptance of breast conserving surgery with radiotherapy as a standard of care for early stage breast cancer [1], the National Comprehensive Cancer Network guidelines for breast cancer treatment recommend the assessment of surgical margins to aide in local control of disease. Any margins deemed positive should be considered for additional surgery according to these guidelines since clear surgical margins have been shown to minimize the risk of local recurrence [2].

The optimal time to identify positive margins is at the time of the initial operation since it affords the opportunity to reexcise positive or questionable margins without subjecting the patient to a second operation. The effort to achieve clear surgical margins intraoperatively is aided by surgeon’s judgment, specimen palpation, gross sectioning, imaging [3], wire localization, frozen section, and touch prep analysis [4], among other techniques. In spite of these efforts, up to 40% of women in the USA continue to undergo multiple operations due to initial failure to achieve clear margins.

To further reduce the need for reexcisions, the MarginProbe (Dune Medical Devices, Caesarea, Israel) was developed to provide real-time, intraoperative assessment of the presence of disease at the surgical margins. The MarginProbe is a handheld device that utilizes radiofrequency spectroscopy to detect electromagnetic changes in malignant tissue within 1 mm of the margin surface. A 21-center randomized, controlled trial (Pivotal Trial) was conducted to determine if the adjunctive use of the MarginProbe would enhance standard of care practices employed by surgeons to reduce the need for reexcision procedures. While the complete results of this study are still awaiting publication, the use of the MarginProbe has been validated in other studies as an effective way to assess margins intraoperatively because of its high sensitivity in identifying malignant tissue and high specificity in distinguishing between normal and malignant tissues [5, 6].

A central question in the Pivotal Trial was whether or not the use of the MarginProbe device would result in the resection of excessively wide margins, thereby producing an adverse effect on cosmesis. Herein, we present the analysis of the cosmetic impact of intraoperative margin assessment using the MarginProbe on the participants in the Pivotal Trial.

2. Materials and Methods

Six hundred and sixty-four (664) women with nonpalpable invasive cancer and/or DCIS undergoing lumpectomy were enrolled in the Pivotal Trial at 21 institutions. Following “standard of care (SOC)” lumpectomy, 596 women were randomized (1 : 1) intraoperatively to MarginProbe device use or control (i.e., SOC only with no device use). The definition of SOC varied by institution, but typically involved wire localization, specimen palpation, specimen radiography, and reexcision of questionable margins. Frozen section, touch prep analysis, and gross sectioning were not permitted to avoid confounding subsequent margin analysis. In women randomized to the device arm, the MarginProbe was used to assess each margin of the resected surgical specimen. The 7 mm sensor footplate at the probe tip was applied to a minimum of 5 sites and a maximum of 8 sites on each margin surface, depending on the area of each margin surface. A vacuum mechanism ensured full contact of the 7 mm sensor with the margin surface (Figure 1). An auditory and visual binary signal (positive/negative) was produced when the MarginProbe detected the presence or absence of malignant tissues within 1 mm of the margin surface at any of the 5–8 examined sites (Figure 2). By protocol requirements, any margin producing a positive reading required the reexcision of the entire affected margin. The thickness of each margin was left to the discretion of the operating surgeon. Skin margins and muscle margins did not require reexcision. Women randomized to the SOC arm underwent no additional margin resection following randomization. Surgeons were discouraged from taking additional shave margins as a safeguard against randomization to the SOC arm. Excision of shave margins following randomization to the SOC arm was considered a protocol violation and resulted in censuring of the data.

(a)

(b)

Figure 1: Photograph (a) of MarginProbe device showing vacuum mechanism that ensures full contact of the 7 mm sensor with the margin surface. Schematic (b) showing 7 mm diameter sensor and radiofrequency field that penetrates margin surface to detect cancer-associated electromagnetic changes.

Figure 2: Visual binary signal display showing either positive (Pos.) or negative (Neg.) margin at multiple sites measured on each specimen margin surface [deep (D), anterior (A), inferior (I), superior (S), lateral (L), and medial (M)].

All primary and reexcision specimens in both arms were submitted for standard histopathological examination by pathologists who were blinded to the study arm. Device readings were compared per specimen for histological assessment of the initially excised lumpectomy specimens. True positive device readings occurred when invasive breast cancer or DCIS was detected histologically less than 1 mm from original specimen margin. True negative device readings occurred when histopathology of the primary specimen revealed no malignant cells within 1 mm of the corresponding margin surface.

The ability to correctly and intraoperatively identify all of the involved margins on the main specimen and reexcise them was defined as a correct Complete Surgical Resection (CSR). Correctness or incorrectness of CSR was defined based on permanent histology data. CSR was defined as correct only when all main specimen margins detected as positive by histology were reexcised intraoperatively.

3. Results

The breakdown of tissue volume removed is shown in Table 1. When analyzing the impact of the MarginProbe on reexcision rates, the use of the device resulted in a 57% reduction in reexcision compared to the control group (device: 42/298 (14.1%), Control: 98/298 (29.9%), 57% reduction, ). As a result of true positive and false positive device readings, there was a small increase in tissue volume removed at primary lumpectomy (15.6 cc and less than 2 shavings per patient). Among patients requiring reexcision of positive margins at a second operation, less tissue was ultimately removed in the device arm (device: 28.4 cc, control: 49.5 cc, a 43.4% reduction). When analyzing the total tissue volumes removed (all operations combined), resected tissue volume was only slightly greater (8.5 cc) in the device arm (2.6% greater when normalized to bra cup size).

Table 1: Total tissue volume removed.

Average per patient
Control
Device
Difference

Initial surgery

    Main specimen
61.3 cc
59.7 cc
−1.6 cc

    True positive shavings
2.7 cc
6.7 cc
4.0 cc

    False positive shavings
7.7 cc
21.1 cc
13.4 cc

    Total volume
71.9 cc
87.5 cc
15.6 cc

Reexcision surgeries

    Tissue volume
49.5 cc
28.4 cc
−21.1 cc

    Normalized tissue volume

    (normalized to breast volume)
5.6%
4.0%
−1.6%

All surgeries

    Total tissue volume
84.8 cc
93.3 cc
8.5 cc

    Normalized total tissue volume

    (normalized to breast volume)
12.5%
15.1%
2.6%

4. Discussion

It is well established that cosmesis after breast conserving surgery is affected by multiple variables. Among the most important are the need for reexcision as well as surgery the amount of tissue removed at the primary and secondary surgeries [7–9]. Accurate intraoperative assessment of surgical margins allows the tumor to be removed in one surgical procedure, thereby sparing patients the burden of a second breast operation. However, standard of care approaches for intraoperative margin assessment (e.g., palpation, gross-sectioning of the specimen, specimen imaging, wire localization, and frozen section or touch prep analysis) continues to be commonly associated with margin reexcision rates of 20–40%. Reexcision has been associated with the risk of postoperative infection, delays in the onset of adjuvant therapy, lower patient satisfaction, lower rates of cosmetic acceptability, increased medical costs, and stress for patients who sometimes needlessly elect mastectomy rather than risk another positive margin [7–11]. Based on the results of the Pivotal Trial, breast reexcisions can be significantly reduced with the use of the MarginProbe device which should, in turn, significantly improve the safety and feasibility of breast conserving surgery.

In spite of its benefits in reducing breast reexcisions, concerns have been expressed that the adjunctive use of the MarginProbe might compromise breast cosmesis due to the excessive resection of breast tissue, particularly when false positive readings are encountered. However, comparison of the device and control arms in the Pivotal Trial showed minimal impact on breast cosmesis when analyzed by the volume of resected breast tissue. In fact, only 2.6% greater volume was resected in the device arm when normalized to breast size. This corresponded to 2 additional margin shavings per patient, which is less than the 4–6 margins that may be indiscriminately reexcised by some surgeons who routinely harvest margin shavings. Furthermore, among patients who ultimately required reexcision at a second operation, there was essentially no difference in the two study arms (−1.6% difference, normalized) in the total volume of resected breast tissue. Collectively, these findings resolved concerns that the use of the MarginProbe may adversely affect cosmesis.

5. Conclusion

The primary goal of breast conserving surgery is resection of breast malignancy with clear margins and acceptable cosmesis. While this goal is not always achievable at the initial operation, every reasonable effort should be made avoid multiple surgeries, undesirable cosmetic outcomes, increased treatment burden, and increased medical costs associated with these factors. The MarginProbe represents a practical advancement in the field of surgical specimen margin evaluation. When combined with the standard of care techniques, the MarginProbe may significantly lower the rates of reexcision for breast cancer patients, achieve comparable tissue volume removal at the first surgery, and reduce the amount of tissue removed among patient requiring a second operation. The end result is significant quality improvement in the management of conservatively treated breast cancer patients.

Conflict of Interests

Drs. D. R. Holmes and L. Tafra were funded researchers as a part of the Pivotal Trial, the multicenter, randomized controlled trial evaluating the MarginProbe device. Neither has any ongoing financial interest in the MarginProbe device.

Acknowledgments

The MarginProbe Study Group was acknowledged for cooperation and assistance while preparing the paper. The investigators and IRBs names are as follows. Tanir Allweis, Hadassah Medical Organization, Israel (research institution: Hadasit Medical Research Services and Development Ltd. of Hadassah Ein-Kerem Hospital); Moshe Carmon, Shaare Zedek Medical Center, Israel; Tami Karni, Assaf Harofeh Medical Center, Israel; Alison Estabrook, Milton and Carroll Petrie Division, IRB Beth Israel Medical Center, USA (research institution: St. Luke’s Roosevelt Hospital Center); Freya Schnabel, IRB NYU School of Medicine, USA (research institution: NYU School of Medicine); Rache M. Simmons, IRB Weill Cornell Medical College, USA (research institution: Cornell University on behalf of its Joan & Sanford I. Weill Medical College); Sheldon Feldman, IRB Columbia University Medical Center (CU IRB), USA (research institution: New York and Presbyterian Hospital); Mark Gittelman, Western IRB, USA (research institution: Westfield Hospital); Neil B. Friedman, Mercy Medical Center IRB, USA; Kristen Fernandez, MedStar Research Institute, Georgetown University Oncology IRB, USA (research institution: MedStar Research Institute); Shawna C. Willey, MedStar Research Institute, Georgetown University Oncology IRB, USA (research institution: Georgetown University); Lorraine Tafra, Anne Arundel Health System Research Institute, USA (research institution: Anne Arundel Health System); Karen Lane, UC Irvine IRB Office of Research Administration, USA; Jay Harness, St. Joseph Hospital IRB, USA (research institution: St. Joseph Hospital of Orange); Alice Police, Western IRB, USA (research institution: Newport Beach Surgery Center); Dennis Holmes, University of South California Health Science Campus IRB, USA (research institution: Health Research Association); Susan Boolbol, Milton and Carroll Petrie Division, IRB Beth Israel Medical Center, USA (research institution: Beth Israel Medical Center); Scott Karlan, Cedars Sinai IRB, Office of Research Compliance and Quality Improvement, USA; Stephani Akbari, Schulman Association IRB, USA (research institution: Virginia Hospital Center); Thomas Frazier, Main Line Hospitals IRB, USA; Lisa Guerra, Western IRB, USA.

References

U. Veronesi, N. Cascinelli, L. Mariani et al., “Twenty-year follow-up of a randomized study comparing breast-conserving surgery with radical mastectomy for early breast cancer,” New England Journal of Medicine, vol. 347, no. 16, pp. 1227–1232, 2002.

“The NCCN breast cancer clinical practice guidelines in oncology,” 2012, http://www.nccn.org

A. Chagpar, T. Yen, A. Sahin et al., “Intraoperative margin assessment reduces reexcision rates in patients with ductal carcinoma in situ treated with breast-conserving surgery,” The American Journal of Surgery, vol. 186, no. 4, pp. 371–377, 2003.

T. P. Olson, J. Harter, A. Muñoz, D. M. Mahvi, and T. M. Breslin, “Frozen section analysis for intraoperative margin assessment during breast-conserving surgery results in low rates of re-excision and local recurrence,” Annals of Surgical Oncology, vol. 14, no. 10, pp. 2953–2960, 2007.

T. Karni, I. Pappo, J. Sandbank et al., “A device for real-time, intraoperative margin assessment in breast-conservation surgery,” The American Journal of Surgery, vol. 194, no. 4, pp. 467–473, 2007.

I. Pappo, R. Spector, A. Schindel et al., “Diagnostic performance of a novel device for real-time margin assessment in lumpectomy specimens,” Journal of Surgical Research, vol. 160, no. 2, pp. 277–281, 2010.

R. A. Cochrane, P. Valasiadou, A. R. M. Wilson, S. K. Al-Ghazal, and R. D. Macmillan, “Cosmesis and satisfaction after breast-conserving surgery correlates with the percentage of breast volume excised,” British Journal of Surgery, vol. 90, no. 12, pp. 1505–1509, 2003.

S. W. W. Chan, P. S. Y. Chueng, and S. H. Lam, “Cosmetic outcome and percentage of breast volume excision in oncoplastic breast conserving surgery,” World Journal of Surgery, vol. 34, no. 7, pp. 1447–1452, 2010.

J. A. Dewar, S. Benhamou, E. Benhamou et al., “Cosmetic results following lumpectomy axillary dissection and radiotherapy for small breast cancers,” Radiotherapy and Oncology, vol. 12, no. 4, pp. 273–280, 1988.

M. Morrow, “Margins in breast-conserving therapy: have we lost sight of the big picture?” Expert Review of Anticancer Therapy, vol. 8, no. 8, pp. 1193–1196, 2008.

A. Munshi, S. Kakkar, R. Bhutani, R. Jalali, A. Budrukkar, and K. A. Dinshaw, “Factors influencing cosmetic outcome in breast conservation,” Clinical Oncology, vol. 21, no. 4, pp. 285–293, 2009.