Journal of Analytical Methods in Chemistry
Volume 2018 (2018), Article ID 8461380, 7 pages
https://doi.org/10.1155/2018/8461380

Research Article

A Simple Separation Method of the Protein and Polystyrene Bead-Labeled Protein for Enhancing the Performance of Fluorescent Sensor

Hye Jin Kim,¹ Dong-Hoon Kang,² Seung-Hoon Yang,³ Eunji Lee,⁴ Taewon Ha,⁵ Byung Chul Lee,⁶ Youngbaek Kim,⁵ Kyo Seon Hwang,¹ Hyun-Joon Shin,² and Jinsik Kim⁴

¹Department of Clinical Pharmacology, Kyung Hee University, Seoul 02447, Republic of Korea
²Center for Bionics, Korea Institute of Science and Technology, Seoul 02792, Republic of Korea
³Systems Biotechnology Research Center, Korea Institute of Science and Technology, Gangneung 25451, Republic of Korea
⁴Department of Medical Biotechnology, Dongguk University, Seoul 04620, Republic of Korea
⁵Center for Nano-Photonics Convergence Technology, Korea Institute of Industrial Technology (KITECH), Gwangju 61012, Republic of Korea
⁶Center for BioMicrosystems, Korea Institute of Science and Technology (KIST), Seoul 02792, Republic of Korea

Correspondence should be addressed to Jinsik Kim; lookup2@dongguk.edu

Received 23 March 2018; Revised 31 May 2018; Accepted 7 June 2018; Published 11 July 2018

Academic Editor: Subhankar Singha

Copyright © 2018 Hye Jin Kim et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

Abstract.

Dielectrophoresis- (DEP-) based separation method between a protein, amyloid beta 42, and polystyrene (PS) beads in different microholes was demonstrated for enhancement of performance for bead-based fluorescent sensor. An intensity of was relative to a diameter of a microhole, and the diameters of two microholes for separation between the protein and PS beads were simulated to 3 μm and 15 μm, respectively. The microholes were fabricated by microelectromechanical systems (MEMS). The separation between the protein and the PS beads was demonstrated by comparing the average intensity of fluorescence (AIF) by each molecule. Relative AIF was measured in various applying voltage and time conditions, and the conditions for allocating the PS beads into 15 μm hole were optimized at 80 mV and 15 min, respectively. In the optimized condition, the relative AIF was observed approximately 4.908 ± 0.299. Finally, in 3 μm and 15 μm hole, the AIFs were approximately 3.143 and −1.346 by 2 nm of protein and about −2.515 and 4.211 by 30 nm of the PS beads, respectively. The results showed that 2 nm of the protein and 30 nm of PS beads were separated by DEP force in each microhole effectively, and that our method is applicable as a new method to verify an efficiency of the labeling for bead-based fluorescent sensor .

1. Introduction

Labeling is one of the essential processes for analyzing and tracking the biomolecules and proceeds by conjugating the molecules with various materials such as isotope markers [1], photochromic compounds [2], and fluorescence polystyrene (PS) beads [3–5]. Especially, fluorescent PS beads have competitive price, high accessibility, and controllability so that various biomolecules such as protein [6], cell [3], and deoxyribonucleic acid (DNA) [7] have been conjugated with fluorescent PS beads followed by being quantified and qualified [8–10]. Qin et al. separated the protein which conjugated with surface-modified fluorescent PS [11], and Fakih et al. multidetected the viral DNA using gold nanoparticle-coated fluorescent PS beads [12]. However, fluorescent PS beads are not conjugated with biomolecules perfectly; in other words, their labeling efficiency is under 100%, and consequently, not only biomolecules conjugated with fluorescent PS beads but also nonconjugated biomolecules existed in the analyte. The nonconjugated biomolecules decrease the accuracy and sensitivity in analyzing and tracking of the biomolecules; hence, a method is required for separating the biomolecules conjugated with PS beads and nonconjugated biomolecules, namely, the residue molecules. So, residue biomolecules after labeling need to be separated from the biomolecules, which are conjugated with labels, ideally. Although centrifugation approaches [13, 14] and fluidic-based approaches [15] are suitable for separating the residue molecules, these approaches are complex and require an additional process.

Dielectrophoresis (DEP), resulting from inhomogeneous electric fields, has been utilized for the specific manipulation of the particles, cells, and viruses as well as biomolecules such as DNA and even single protein, because of its simplicity, efficiency, and usability [16–18]. The intensity of the DEP force is dominated by the size of the molecules and the strength of the electric fields, which occurred between the electrodes, so that various molecules are affected by a different intensity of the force according to the size of the molecules and the structure of the electrode. Lapizco-Encinas et al. concentrated and separated the live and dead bacteria with insulator-based DEP (iDEP) [19]. But two types of bacteria were separated according to different types of DEP force, negative DEP and positive DEP, and Chen et al. suggested a simplified dielectrophoretic-based microfluid device for particle separation [20]. But these approaches were limited for observing the various molecules simultaneously or consisted of the complex structure.

Here, we suggest a simple method to separate the nonconjugated protein, namely, the residue protein, and the protein conjugated with PS beads with the DEP force, which is applicable for verifying the efficiency of labeling between protein and PS beads. The protein and PS beads were separated into two microholes with different diameters and formed on a single electrode according to the intensity of DEP force induced. The intensity of the DEP force increased when the diameter of the microhole was smaller, and thus, stronger repulsive force and attractive force occurred in the small microhole than the large one. Consequently, the smaller molecules, residue protein, were allocated into the small microhole, whereas the bigger molecules, PS beads, were expelled from that small microhole followed by allocating into the large microhole; it means that the protein and PS beads were separated. To verify the separation between protein and PS beads, approximately 2 nm of the protein, amyloid beta 42, and 30 nm fluorescent PS beads were used. The diameter of the two microholes for separating the protein and PS beads was optimized to 3 μm and 15 μm, respectively, by calculating the intensity of in each microhole with COMSOL simulation. Also, an applied voltage to induce the DEP force was optimized to 80 mV, which induced a difference in the force approximately 9.059-fold between two microholes. The microholes were fabricated by microelectromechanical systems (MEMS) technique, and separation between the protein and polystyrene beads was demonstrated by comparing a relative averaged intensity of fluorescence (AIF) by each protein and each PS bead.

2. Materials and Methods

2.1. Theory

The molecule present in an inhomogeneous electric field, , is influenced by the DEP force, , which is expressed as follows [21]:

where , , and represent the radius of molecules, the effective permittivity of liquid, and the Clausius–Mossotti factor, respectively. The and gradient of the electric field are described as follows:

where and are the applying voltage and the distance between the electrodes, respectively. On the basis of (1) and (3), the intensity of the DEP force can be modified through (4) as follows:

The intensity of is calculated with a finite-element model (FEM) in the AC/DC module of COMSOL Multiphysics software 5.2 (COMSOL Inc., USA).

2.2. Materials

Thirty nanometres of the carboxylate-modified polystyrene (PS) bead labeled with fluorescence (Sigma-Aldrich Inc., Korea) and 2 nm of the TAMRA-labeled beta-amyloid (1–42) protein (AnaSpec Inc., USA) were used to verify the separation of molecules, whose excitation/emission wavelength () was ∼470/505 nm and ∼544/572 nm, respectively. The protein and PS beads were diluted with 1 mM PBS buffer (Corning Korea Co. Ltd., Korea) to create a 1 ng·mL⁻¹ solution.

2.3. Fabrication of Microholes

Microholes were fabricated by a standard MEMS process. First, an insulation layer, 300 nm of SiO₂, and an electrode layer, 30 nm of tantalum (Ta) and 150 nm of platinum (Pt), were sequentially deposited on the 4-inch silicon (Si) wafer by thermal oxidation and sputtering, respectively. Next, an AZ GXR 601 photoresist (AZ Electronic Materials, Luxembourg) was coated by a spin coater (30 s, 3000 rpm) and exposed (3.8 s, 12 mW·cm⁻²). Then, the hole patterns were etched by inductively coupled plasma etching (Oxford Instruments), and the photoresist was stripped by Microwave Plasma Asher (Plasma-Finish, Germany).

2.4. System Setup for Molecules Separation and Fluorescence Analysis

A DG4062 Series waveform generator (Rigol Technologies Inc., USA) (frequency range: up to 60 MHz; voltage range: up to 10 V), which applies a sinusoidal AC voltage for inducing the DEP force in the microhole on a single electrode, was used. The intensity of fluorescence was observed via an electron-multiplying charge-coupled device (ANDORiXonEM), an oil immersion 100x lens (Nikon Corp., Japan) (NA: 1.4), and an Eclipse Ti inverted microscope (Nikon Corp.) equipped with a halogen lamp and a 593 nm (bandwidth: 40 nm) filter and was analyzed by Image-Pro Plus 6.0 (Media Cybernetics Inc., USA). The average intensity of fluorescence (AIF) was calculated by values measured at five random positions in the microhole, and a value of a relative AIF was calculated by dividing the AIF values measured at each condition by the value in the reference condition.

3. Results and Discussion

Protein, nonconjugated with PS beads after labeling, does not emit the fluorescence but binds to the receptor specifically so that it impedes the specific binding between the ideally conjugated protein with PS beads and receptor, followed by decreasing the accuracy and reliability in the molecules’ analyzing and tracking process (Figure 1(a)). Thus, the nonconjugated protein, namely, the residue protein, should be separated from the protein conjugated with PS beads. When alternating current (AC) voltage is applied to the electrode with microholes, the protein is allocated into each microhole according to the intensity of the DEP force that occurred in each microhole. The intensity of the DEP force is related to the diameter of the molecules and the distance between the electrodes, namely, the size of microholes, as described in (4), and consequently allocates different molecules into each microhole, respectively: residue protein, smaller than the PS beads, is allocated into the small microhole, whereas the PS beads are placed in the large microhole—two molecules separate into small and large microholes, respectively (Figure 1(b)).

(a)

(b)

Figure 1: Schematic illustration of a simple separation method of the protein and protein conjugated with polystyrene (PS) beads. (a) Intensity of fluorescence by a specific binding of the protein conjugated with PS beads decreased due to a specific binding of the nonconjugated protein, expressed as residue protein. (b) Residue protein and protein conjugated with PS beads were separated by the dielectrophoresis (DEP) force in different microholes, respectively.

In order to separate the residue protein and conjugated protein with PS beads in each microhole, 4.5 kDa of amyloid beta, whose diameter was calculated to be approximately 2 nm, was used as a residue protein, and the conjugated protein with PS beads was simplified to just PS beads. The length and width of the electrode were fixed to 27 μm and 21 μm, respectively, and the diameter of the small microhole, d, and pitch between two microholes, p, were fixed to 3 μm. The intensity of the applied AC voltage and size of microholes were optimized via the COMSOL simulation (Figure 2(a)). Firstly, maximum intensity of in the small microhole was simulated according to the applied AC voltage (Figure 2(b)). Black line and scatter showed the maximum intensity of that occurred in the small microhole, and red line and scatter indicated the size of protein, which was allocated into the small microhole, depending on the intensity of the applied AC voltage. The intensity of increased parabolically and size of the protein decreased accordingly. The results signified that approximately 30 mV voltage, which resulted in with intensity approximately 2.310 × 10¹³ V²·m⁻³, was required to place 2 nm of the protein in the 3 μm hole. Also, the maximum intensity of that occurred in the other microhole was simulated according to the size of the other microhole at the condition that applied 30 mV AC voltage (Figure 2(c)). The diameter of the other microhole was expressed as a ratio to the diameter for the 3 μm hole, and the intensity of decreased according to the increase in the ratio of the diameter. In the 6 μm microhole (the ratio was 2), intensity of was about 0.861 × 10¹³ V²·m⁻³ and it is too strong to allocate 30 nm of the PS beads into the hole. was approximately 0.528 × 10¹³ V²·m⁻³, 0.347 × 10¹³ V²·m⁻³, 0.255 × 10¹³ V²·m⁻³, and 0.194 × 10¹³ V²·m⁻³ in each value of the ratio, and an optimized size of the microhole for placing 30 nm of the PS beads into the hole was verified to be 15 μm. Thus, the two microholes for separating the residue protein and protein conjugated with PS beads were optimized to 3 μm and 15 μm, respectively, whose difference in the intensity was approximately 9.059-fold.

(a)

(b)

(c)

Figure 2: Simulation for separating the molecules by DEP force in the two microholes. (a) Distribution of at the top view of the electrode. The diameter of the small microhole and pitch between two microholes were represented as “d” and “p,” respectively. (b) According to the applied voltage, maximum intensity of occurring in the 3 μm hole increased, whereas molecular weight of the protein, allocated into the microhole, decreased. (c) Maximum intensity of in the other microhole and the diameter of PS beads, allocated into the hole, decreased according to the increase of the diameter.

Two microholes in the electrode were produced via a standard MEMS process on a 4-inch silicon (Si) wafer. The fabrication process consisted of 3 steps (Figure 3(a)), and details of the process are described in Material and Methods. The diameter of the fabricated small and large microholes was approximately 3 μm and 15 μm, respectively, and pitch between two microholes was about 3 μm (Figure 3(b)).

(a)

(b)

Figure 3: Fabrication of the single electrode consisting of two different-sized microholes by MEMS technology. (a) Schematic illustration of the fabrication process of the microholes. (b) Microscopic image of the two different microholes.

Various conditions of DEP, intensity of the DEP force and applied time, were optimized by measuring the relative AIF resulting from placing the PS beads in the 15 μm hole. In order to optimize the intensity of the DEP force, the applied frequency required for the DEP force to occur was fixed at 50 MHz. Firstly, relative AIF was measured in various applying voltage conditions ranging from 0 V (ref.) to 500 mV, and consequently, it was maximized at 80 mV (Figure 4(a)). The values were approximately 1, 1.265, 1.655, 1.396, and 1.1604 in ref., 50 mV, 80 mV, 100 mV, and 500 mV, respectively. The results indicated that the PS beads were most effectively placed in the microhole by the DEP force induced by the applied voltage 80 mV; consequently, the intensity of the applied voltage was settled to 80 mV. Also, in order to optimize the applied time condition of DEP force, the relative AIF was measured according to time every 3 minutes up to 21 minutes (Figure 4(b)). The AIF increased gradually depending on the time up to 15 min and was saturated afterward: each value of the AIF was approximately 0.907, 1.048, 1.184, 1.450, 1.526, 1.535, and 1.536 according to the applied time. Hence, the applied voltage and time were optimized to 80 mV and 15 minutes, respectively, and the relative AIF was observed to be approximately 4.908 ± 0.299 in the optimized condition (Figure 4(c)). The result demonstrated that the PS beads were allocated into the 15 μm hole effectively.

(a)

(b)

(c)

Figure 4: Optimization of the DEP condition by measuring the average intensity of fluorescence (AIF) of the PS beads in the 15 μm hole. Relative AIF was verified according to (a) the applied voltage and (b) the applied time of AC voltage. (c) Relative AIF by the PS beads in the 15 μm hole was compared in each reference and optimized DEP condition.

Finally, based on these results, a separation of the protein and PS beads in 3 μm and 15 μm holes, respectively, was demonstrated (Figure (5)). It was also confirmed by comparing the AIF by each molecule in the two microholes at the previous optimized condition. The relative AIF by 2 nm of the protein in the 3 μm hole was a positive value, but the value by 30 nm of the PS beads was negative, and the values were approximately 3.143 and −1.346, respectively, whereas in the 15 μm hole, the relative AIFs by the protein and the PS beads showed an opposite sign compared with the previous values, and the values were approximately −2.515 and 4.211, respectively. The negative value of the AIF indicated that the molecules were moving far away owing to the strong DEP force in the microhole, and the positive value of the AIF signified that the molecules were attracted and trapped into the microhole by the DEP force. Thus, the results signified that the DEP force allocated 2 nm of the protein and 30 nm of the PS beads into 3 μm and 15 μm holes, respectively. The results demonstrated that 2 nm of the protein and 30 nm of the PS beads were separated by DEP force in each microhole, effectively.

Figure 5: Relative AIF by 2 nm of the protein and 30 nm of the PS beads in 3 μm and 15 μm holes, respectively.

4. Conclusions

In this paper, a simple method for separation between 2 nm of the protein and PS beads into different microholes, respectively, by the DEP force was demonstrated. In order to separate two molecules, the diameter of the two microholes was simulated and the intensity of the DEP force induced in the microholes was calculated via simulation. The optimized diameter of the two microholes was 3 μm and 15 μm, and a difference in the DEP force between two microholes was approximately 9.059-fold. The condition of the DEP force to separate two molecules was optimized experimentally: intensity of the AC voltage was 80 mV and the applied time was 15 minutes. The molecules which were separated by the DEP force in each microhole were verified by measuring the relative AIF by each molecule. In 3 μm and 15 μm holes, the AIFs were approximately 3.143 and −1.346 by 2 nm of the protein and about −2.515 and 4.211 by 30 nm of the PS beads, respectively. Consequently, the results demonstrated that 2 nm of the protein and 30 nm of the PS beads were separated by DEP force in each microhole, effectively. Our method has high expandability in separation of various-sized molecules, and furthermore, it is applicable for verification of the labeling efficiency.

Data Availability

The data used to support the findings of this study are available from the corresponding author upon request.

Conflicts of Interest

The authors declare that there are no conflicts of interest regarding the publication of this paper.

Acknowledgments

This work was mainly supported by the Dongguk University Research Fund of 2017. Taewon Ha and Youngbaek Kim are grateful for financial support from the Korea Institute of Industrial Technology (Project no. EO170047).

References

X. Chen, L. M. Smith, and E. M. Bradbury, “Site-specific mass tagging with stable isotopes in proteins for accurate and efficient protein identification,” Analytical Chemistry, vol. 72, no. 6, pp. 1134–1143, 2000.

M. Lummer, F. Humpert, M. Wiedenlübbert, M. Sauer, M. Schüttpelz, and D. Staiger, “A new set of reversibly photoswitchable fluorescent proteins for use in transgenic plants,” Molecular Plant, vol. 6, no. 5, pp. 1518–1530, 2013.

H. Sahoo, “Fluorescent labeling techniques in biomolecules: a flashback,” RSC Advances, vol. 2, no. 18, pp. 7017–7029, 2012.

D. Jung, K. Min, J. Jung, W. Jang, and Y. Kwon, “Chemical biology-based approaches on fluorescent labeling of proteins in live cells,” Molecular BioSystems, vol. 9, no. 5, pp. 862–872, 2013.

M. M. Bonar and J. C. Tilton, “High sensitivity detection and sorting of infectious human immunodeficiency virus (HIV-1) particles by flow virometry,” Virology, vol. 505, pp. 80–90, 2017.

C. Obermaier, A. Griebel, and R. Westermeier, “Principles of protein labeling techniques,” Proteomic Profiling: Methods and Protocols, vol. 1295, pp. 153–165, 2015.

T. T. Weil, R. M. Parton, and I. Davis, “Making the message clear: visualizing mRNA localization,” Trends in Cell Biology, vol. 20, no. 7, pp. 380–390, 2010.

J. Lu, G. Getz, E. A. Miska et al., “MicroRNA expression profiles classify human cancers,” Nature, vol. 435, pp. 834–838, 2005.

B. S. Edwards, T. Oprea, E. R. Prossnitz, and L. A. Sklar, “Flow cytometry for high-throughput, high-content screening,” Current Opinion in Chemical Biology, vol. 8, no. 4, pp. 392–398, 2004.

X. H. Gao and S. M. Nie, “Quantum dot-encoded mesoporous beads with high brightness and uniformity: rapid readout using flow cytometry,” Analytical Chemistry, vol. 76, no. 8, pp. 2406–2410, 2004.

L. Qin, X. W. He, W. Zhang, W. Y. Li, and Y. K. Zhang, “Surface-modified polystyrene beads as photografting imprinted polymer matrix for chromatographic separation of proteins,” Journal of Chromatography A, vol. 1216, no. 5, pp. 807–814, 2009.

H. H. Fakih, M. M. Itani, and P. Karam, “Gold nanoparticles-coated polystyrene beads for the multiplex detection of viral DNA,” Sensors and Actuators B: Chemical, vol. 250, pp. 446–452, 2017.

E. Fernández-Vizarra, M. J. López-Pérez, and J. A. Enriquez, “Isolation of biogenetically competent mitochondria from mammalian tissues and cultured cells,” Methods, vol. 26, no. 4, pp. 292–297, 2002.

U. Michelsen and J. von Hagen, “Isolation of subcellular organelles and structures,” in Methods in Enzymology, pp. 305–328, Academic Press, Cambridge, MA, USA, 2009.

A. A. S. Bhagat, H. Bow, H. W. Hou, S. J. Tan, J. Han, and C. T. Lim, “Microfluidics for cell separation,” Medical and Biological Engineering and Computing, vol. 48, pp. 999–1014, 2010.

S. Paracha and C. Hestekin, “Field amplified sample stacking of amyloid beta (1-42) oligomers using capillary electrophoresis,” Biomicrofluidics, vol. 10, no. 3, Article ID 033105, 2016.

T. D. Mai, F. Qukacine, and M. Taverna, “Multiple capillary isotachophoresis with repetitive hydrodynamic injections for performance improvement of the electromigration preconcentration,” Journal of Chromatography A, vol. 1453, pp. 116–123, 2016.

L Zheng, J. P. Brody, and P. J. Burke, “Electronic manipulation of DNA, proteins, and nanoparticles for potential circuit assembly,” Biosensors and Bioelectronics, vol. 20, no. 3, pp. 606–619, 2004.

B. H. Lapizco-Encinas, B. A. Simmons, E. B. Cummings, and Y. Fintschenko, “Dielectrophoretic concentration and separation of live and dead bacteria in an array of insulators,” Analytical Chemistry, vol. 76, no. 16, pp. 1571–1579, 2004.

X. Chen, Y. Ren, W. Liu et al., “A simplified microfluidic device for particle separation with two consecutive steps: induced charge electro-osmotic prefocusing and dielectrophoretic separation,” Analytical Chemistry, vol. 89, no. 17, pp. 9583–9592, 2017.

A. Ramos, H. Morgan, N. G. Green, and A. Castellanos, “Ac electrokinetics: a review of forces in microelectrode structures,” Journal of Physics D: Applied Physic, vol. 31, no. 18, p. 2338, 1998.