Journal of Pharmaceutics
Volume 2017 (2017), Article ID 9495732, 8 pages
https://doi.org/10.1155/2017/9495732

Research Article

Novel HPLC Analysis of Hydrocortisone in Conventional and Controlled-Release Pharmaceutical Preparations

Ofosua Adi-Dako,1,2 Samuel Oppong Bekoe,³ Kwabena Ofori-Kwakye,¹ Enoch Appiah,³ and Paul Peprah³

¹Department of Pharmaceutics, Faculty of Pharmacy and Pharmaceutical Sciences, Kwame Nkrumah University of Science and Technology (KNUST), Kumasi, Ghana
²Department of Pharmaceutics and Microbiology, School of Pharmacy, University of Ghana, Legon, Accra, Ghana
³Department of Pharmaceutical Chemistry, Faculty of Pharmacy and Pharmaceutical Sciences, Kwame Nkrumah University of Science and Technology (KNUST), Kumasi, Ghana

Correspondence should be addressed to Kwabena Ofori-Kwakye; koforikwakye@yahoo.com

Received 7 March 2017; Accepted 26 April 2017; Published 4 June 2017

Academic Editor: Nerea Ferreirós Bouzas

Copyright © 2017 Ofosua Adi-Dako et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

Abstract.

An isocratic sensitive and precise reverse phase high-performance liquid chromatography (RP-HPLC) method was developed and validated for the determination and quantification of hydrocortisone in controlled-release and conventional (tablets and injections) pharmaceutical preparations. Chromatographic separation was achieved on an ODS (C18), 5 m, 4.6 × 150 mm, with an isocratic elution using a freshly prepared mobile phase of composition methanol : water : acetic acid (60 : 30 : 10, v/v/v) at a flow rate of 1.0 ml/min. The detection of the drug was successfully achieved at a wavelength of 254 nm. The retention time obtained for the drug was 2.26 min. The proposed method produced linear detectable responses in the concentration range of 0.02 to 0.4 mg/ml of hydrocortisone. High recoveries of 98–101% were attained at concentration levels of 80%, 100%, and 120%. The intraday and interday precision (RSD) were 0.19–0.55% and 0.33–0.71%, respectively. A comparison of hydrocortisone analyses data from the developed method and the official USP method showed no significant difference () at a 95% confidence interval. The method was successfully applied to the determination and quantification of hydrocortisone in six controlled-release and fifteen conventional release pharmaceutical preparations.

1. Introduction

Hydrocortisone is a naturally occurring corticosteroid hormone secreted by the adrenal cortex and released during times of stress. The synthetic drug is employed in the management of inflammatory and rheumatoid diseases, allergic conditions, and autoimmune disorders such as Addison’s disease (adrenal insufficiency disease). Hydrocortisone is commercially available in pharmaceutical formulations such as tablets, capsules, creams, ointments, and injections. Hydrocortisone may exist commercially as the unchanged hormone or as the acetate, cypionate, sodium phosphate, butyrate, valerate, and the sodium succinate forms [1].

A search of the literature revealed that few analytical methods such as ultraviolet spectrophotometry [2], electrokinetic capillary chromatography [3, 4], photochemically enhanced fluorescence [5], thin-layer chromatography- (TLC-) densitometry [1, 6], and high-performance thin-layer chromatography (HPTLC) [7] have been used for the estimation of hydrocortisone in pharmaceutical formulations. However, modern chromatographic techniques such as high-performance liquid chromatography (HPLC) [8–12] and ultra-performance liquid chromatography (UPLC) [13] are preferentially employed for the determination of hydrocortisone in conventional pharmaceutical products such as tablets, creams, and injections. The different HPLC methods employ different chromatographic conditions including the use of complex and sometimes expensive solvent systems. Thus, access to some of these solvent systems in resource-poor countries in the developing countries can be challenging. The development of a simple and economical HPLC method using readily available reagents for the identification and quantification of hydrocortisone in pharmaceutical preparations would be an advantage in resource-poor countries.

The upsurge of autoimmune diseases such as adrenal insufficiency is a major health concern and the long-term use of conventional hydrocortisone tablets in the management of such a condition is problematic. The twice or thrice daily dosing of conventional hydrocortisone tablets to patients with adrenal insufficiency disease is incapable of mimicking the unique diurnal cortisol circadian pattern. Thus, most persons with adrenal insufficiency continue to suffer from poor therapeutic management resulting in poor quality of life and increased mortality. There is therefore the need for the development of creative and innovative treatment models for hydrocortisone replacement therapy in such dire situations. The use of controlled-release hydrocortisone oral dosage forms holds great promise with the capability to replicate the unique physiological pattern of hydrocortisone. In addition, such formulations are better able to manage and control the levels of morning androgen levels. The accurate monitoring of the quality of these promising new drug therapies is an important prerequisite to obtain quality healthcare [14]. The improved therapy will enhance patient compliance and ensure the delivery of controlled amounts of hydrocortisone at the absorption site compared to the immediate release pattern of conventional tablets [15].

The aim of this study was to develop and validate a simple, sensitive, and reproducible isocratic reverse phase HPLC method with ultraviolet/visible (UV) detection capable of determination and quantification of hydrocortisone in conventional and controlled-release pharmaceutical preparations. The application of the method in determining the content of hydrocortisone in controlled-release and commercially available conventional dosage forms was also studied.

2. Materials and Methods

2.1. Standards and Reagents

The reference material of hydrocortisone (Sigma Aldrich, USA), HPLC grade methanol (Fisher Scientific, UK), glacial acetic acid (Fluka, Germany), and water (double distilled) were used. All stock and working solutions were freshly prepared with distilled water for HPLC analyses.

2.2. Hydrocortisone Pharmaceutical Preparations

Fifteen commercially available hydrocortisone pharmaceutical preparations, comprising nine tablet and six injectable formulations, were sampled from retail pharmacies within the Kumasi and Accra Metropolitan areas of Ghana. Additionally, four controlled-release hydrocortisone tablet formulations, two controlled-release capsule formulations, and one conventional release hydrocortisone tablet formulation were obtained from the Department of Pharmaceutics and Microbiology, University of Ghana School of Pharmacy, Accra, Ghana.

2.3. Instrumentation and Optimized Chromatographic Conditions

A chromatograph comprising a P100 Spectra Series pump with a 785A Programmable Perkin Elmer UV/VIS absorbance detector was employed in the study. Isocratic mode of elution was employed. Chromatographic separation was successfully achieved on a JT Baker® ODS C18 column (150 × 4.60 mm ID, 5 μm particle size). Ultraviolet detection at a wavelength of 254 nm was able to ensure good resolution of peaks. The injection volume was 20 μL and all analyses were undertaken at ambient temperature. A mobile phase consisting of methanol/water/acetic acid (60 : 30 : 10; v/v/v) at a flow rate of 1.00 ml/min was employed. All chromatographic data were acquired using PowerChrom series 280 integrator software.

2.4. Preparation of Standard Solution

Fifty milligrams of hydrocortisone reference standard was accurately weighed and dissolved in 25 ml of methanol as the diluent. The solution was transferred into a 50 ml dry volumetric flask and was then sonicated for 5 min. The volume was made up to the mark with methanol and mixed thoroughly (1 mg/ml stock solution). A one in ten (1 in 10) dilution was made to obtain a working solution with concentration of 0.10 mg/ml of hydrocortisone. This solution with the required working concentration was then injected to determine the retention time of the analyte of interest using the developed method.

2.5. Analytical Method Validation

The new HPLC method was validated in terms of linearity, limits of detection and quantification, accuracy, precision (intraday and interday), specificity, robustness, and stability, in compliance with International Conference on Harmonization (ICH) guidelines [16].

2.5.1. Linearity

Serial dilutions were prepared from the stock solution of hydrocortisone reference standard (1 mg/ml) by pipetting 8.0 ml, 4.0 ml, 2.0 ml, 1.6 ml, 0.8 ml, and 0.4 ml of the stock solution into separate 20 ml volumetric flasks. The solutions were made up to volume with the diluent and mixed thoroughly. Solutions prepared were injected and analyzed using the developed method. The linearity of detector response was established by plotting a graph of concentration versus peak area of hydrocortisone standard and determining the correlation coefficient.

2.5.2. Limits of Detection and Quantification

The limit of detection (LOD) and limit of quantification (LOQ) of the new HPLC method were determined from the linear regression curve obtained from the linearity of hydrocortisone. The slope and standard deviations of the responses of the linear curve were used. The formulae for the calculation of the LOD and LOQ are given below:

where δ is standard deviation of the response; is slope of the calibration curve.

2.5.3. Accuracy

Into three separate 10 ml volumetric flasks 1 ml of a 1 mg/ml standard solution of hydrocortisone was pipetted. 0.8 ml, 1.0 ml, and 1.2 ml of formulated hydrocortisone solution (0.1 mg/ml) were added to the three volumetric flasks, respectively, to obtain concentrations at 80%, 100%, and 120% levels. The solutions were analyzed using the developed method. Triplicate determinations were made for each solution and mean percentage recovery calculated.

2.5.4. Precision

Both intraday and interday precision of the HPLC method were assessed. For intraday precision, three solutions of different concentrations were prepared in separate 20 ml volumetric flasks from the stock solution of hydrocortisone reference standard (1 mg/ml) by pipetting 8.0 ml, 2.0 ml, and 0.4 ml of the stock solution. Solutions were made up to volume with methanol. The various solutions where analyzed thrice during a particular day to obtain chromatograms from which recoveries and relative standard deviations (RSD) were calculated. For interday precision, a one in ten (1 in 10) dilution of the stock standard solution was prepared on three different occasions and analyzed. The solutions were analyzed on three consecutive days using the developed HPLC method to obtain chromatograms and the hydrocortisone content and relative standard deviation (RSD) were determined.

2.5.5. Specificity

The developed HPLC method was investigated for specificity to ensure there was no or minimal interference of analyte of interest from the solvent system. The mobile phase (blank) was analyzed, followed by an injection of a sample solution of hydrocortisone. After a ten-minute wash period, the blank was analyzed again.

2.5.6. Robustness

The robustness of the method was examined by introducing small changes in the composition of the mobile phase. Variation by ±10% was done with respect to the methanol component of the mobile phase while keeping the composition of the others constant. Chromatograms obtained were analyzed using the RSD of the responses obtained.

2.5.7. Stability

The stability of solutions of hydrocortisone were studied over a period of 8 hours. Triplicate injections were made each hour and the chromatograms recorded.

2.6. Assay of Hydrocortisone Preparations

An amount of finely powdered hydrocortisone powder from each of the nine commercial conventional tablet formulations equivalent to 2.5 mg hydrocortisone was individually weighed accurately and transferred to a 25 ml volumetric flask containing 10 ml of methanol. The solution was sonicated for 5 min and made up to volume with methanol. The resulting solution was then filtered using Whatman filter paper number 41. The filtrate obtained was analyzed by making triplicate injections. The controlled-release hydrocortisone tablet and capsule formulations were analyzed in a similar way. In the analysis of the hydrocortisone injections, aliquots equivalent to 2.5 mg hydrocortisone were pipetted from the six (6) injectable preparations into 25 ml volumetric flasks and treated similarly to above. The peak areas of sample were determined and the amount of hydrocortisone was estimated from the linear regression calibration curves using the developed method. The official United States Pharmacopoeia method [11] was employed in analyzing the commercial samples. Statistical comparison between the official and developed methods was done using Student’s -test.

3. Results and Discussion

Controlled-release hydrocortisone preparations remain formulations of choice in the therapeutic management of adrenal insufficiency disease. This disease condition is a potentially life-threatening autoimmune disorder that requires prompt diagnosis and management to avoid fatality. The use of controlled-release formulations results in more stable cortisol concentrations during the diurnal cortisol circadian pattern than conventional hydrocortisone products [17]. It is however essential that suitable and accurate analytical methods are made available to assess the quality of the products with regard to their content. The development and validation of a simple isocratic RP-HPLC method for the determination of hydrocortisone in both conventional and controlled-release pharmaceutical formulations were the focus of this study. Review of literature indicates the employment of normal phase stationary support material [18, 19] and reverse phase stationary support material [20] for the development of HPLC method and further analyses of hydrocortisone products. There is yet no reported reverse phase HPLC with UV detection method capable of analyzing hydrocortisone in both conventional and controlled-release pharmaceutical formulations. The inclusion of acetic acid as a modifier in the mobile phase system ensured efficient resolution of hydrocortisone with minimal matrix effect from both conventional and controlled-release formulations. Readily available and cost effective solvents employed in the mobile phase ensured an efficient resolution and separation of hydrocortisone on a reverse phase column. Table 1 presents the optimized chromatographic conditions employed in the determination of hydrocortisone.

Table 1: Optimized chromatographic conditions for the developed HPLC method.

Parameters
Optimized conditions

Mobile phase
Methanol/water/acetic acid (60 : 30 : 10, v/v/v)

HPLC column
JT Baker ODS C18, 5 m, 4.6 × 150 mm

Flow rate
1 ml/min

Wavelength detection
254 nm

Diluent
Methanol

Injection volume
20 

3.1. Analytical Method Validation

This method provides an option for the analyses of hydrocortisone in various salt forms compared to the individual methods available in some official compendia such as the United States Pharmacopoeia [11]. The analytical method being reported gave good linear detectable responses in the range of 0.02 to 0.4 mg/ml with coefficient of correlation () of 0.9989 with well-resolved hydrocortisone and excipient peaks (Figures 1–4). The limit of detection and limit of quantification of 1.0662 × 10−2 mg/ml and 3.23076 × 10−2 mg/ml, respectively, obtained are an indication of the sensitivity of the method. The method showed good accuracy with good recoveries in the range of 98 to 101% recorded at three concentration levels of 80%, 100%, and 120% (Table 2). The method was reproducible with good intraday and interday precision of less than 1% RSD (Tables 3 and 4). The new HPLC method showed high specificity and the robustness was less than 1% RSD (Tables 5 and 6). The method also demonstrated good stability over a period of 8 h (<6% RSD) (Table 7).

Table 2: Accuracy of the HPLC method showing the mean percentage recovery of hydrocortisone at three concentration levels.

Concentration levels
Amount taken (mg/ml)
Amount added (mg/ml)
Amount recovered
Percentage recovery (%)

80%
0.10
0.08
0.07820
97.75

0.10
0.08
0.07840
98.01

0.10
0.08
0.07836
97.92

100%
0.10
0.10
0.10073
100.73

0.10
0.10
0.10120
101.20

0.10
0.10
0.10052
100.52

120%
0.10
0.12
0.11998
99.98

0.10
0.12
0.12062
100.05

0.10
0.12
0.11934
99.45

Table 3: Intraday precision of proposed HPLC method showing the mean percentage recovery and RSD of hydrocortisone at different concentrations.

Number of runs
Concentration levels (mg/ml)
Amount (mg/ml)
Percentage recovery (%)
% RSD

1
0.02
0.01990
99.52
0.55

2
0.01997
99.83

3
0.01975
98.76

1
0.10
0.10052
100.52
0.54

2
0.09945
99.45

3
0.10000
100.03

1
0.40
0.39952
99.88
0.19

2
0.39828
99.57

3
0.39964
99.91

of three replicate determinations.

Table 4: Interday precision of proposed HPLC method showing the mean percentage recovery and RSD of hydrocortisone on different days.

Number of runs
Percentage (%)

Day 1
Day 2
Day 3

1
100.52
98.52
100.16

2
99.45
99.67
99.51

3
100.03
98.36
99.72

Mean
100.00
98.85
99.80

% RSD
0.54
0.71
0.33

of three replicate determinations.

Table 5: Results of specificity studies of the proposed HPLC method.

Parameter
Response

Mobile phase
No peak (−)

Hydrocortisone
Single peak obtained

Mobile phase
No peak (−)

(−): not detected.

Table 6: Robustness of proposed HPLC method showing the mean percentage content of hydrocortisone when conditions were varied.

Sample
Original condition
Varied condition

% content
% content

1
100.52
101.32

2
100.75
100.65

3
101.34
99.99

Mean
100.87
100.65

% RSD
0.42
0.67

Table 7: Stability of HPLC method showing the mean recoveries of hydrocortisone in solution over an eight-hour period.

Time (h)
Mean concentration
% RSD

0
100.72
0.56

2
100.29
0.63

4
98.40
1.73

6
95.82
3.54

8
93.11
5.79

Figure 1: Calibration curve of hydrocortisone showing the linear detectable range (0.02–0.4 mg/ml).

Figure 2: A chromatogram of hydrocortisone standard.

Figure 3: A chromatogram of an excipient in the hydrocortisone tablet.

Figure 4: A chromatogram of the hydrocortisone tablet.

3.2. Application of Developed Method to the Analysis of Commercial Samples

Table 8 presents the characteristics, labelled strength, and assay results of conventional hydrocortisone preparations which were analyzed using both the developed HPLC method and the official USP method. Statistical analysis (-test) of data from the two methods (official/USP and developed) showed no significant difference () at a 95% confidence interval (Table 9). The analytical method developed provides a relatively cheaper alternative technique for routine postmarket analyses of hydrocortisone formulations in resource-poor countries such as Ghana. The validated method was also successfully applied to the analysis of six controlled-release hydrocortisone tablet and capsule formulations (Table 10). The application of the method in the analyses of tablets (conventional and controlled-release) and parenteral formulations indicated the versatility of the analytical procedure. The analyses of the hydrocortisone formulations yielded percentage hydrocortisone content of 95–104%, 96–103%, and 87–103% for the four controlled-release tablet formulations, two controlled-release capsule formulations, and fifteen conventional hydrocortisone formulations, respectively. All the controlled-release products were of good quality in terms of their content (percentage of content ≥ 90%). The percentage of content of hydrocortisone powder was 100.44%. Twelve products (all 6 injections, 6 tablets) out of the 15 commercial hydrocortisone products analyzed were determined to contain the right amounts of hydrocortisone as per United States Pharmacopoeia (USP) method and the developed HPLC method [11]. This study did not cover other pharmacopoeia specifications such as disintegration and dissolution tests.

Table 8: Characteristics and labelled strength of commercial hydrocortisone preparations analyzed using the developed HPLC method (DM) and the USP method.

Code
Dosage
Strength
Country of manufacture
Batch number
Date of manufacture
Expiry date
(%) (DM)
(%)
(USP)

HP1
PFI
100 mg
Belgium
14HB24
04/2014
03/2017
98.41
98.60

HP2
PFI
100 mg
China
131255
12/2013
12/2016
102.60
102.34

HP3
PFI
100 mg
Belgium
14HB52
12/2014
11/2017
100.64
100.42

HP4
PFI
100 mg
India
5AB01007
10/2015
09/2018
101.20
101.27

HP5
PFI
100 mg
Belgium
15HB22
09/2015
08/2018
102.80
102.83

HP6
PF1
100 mg
India
5AE04010
10/2015
09/2018
100.63
100.35

HT1
Tablet
10 mg
China
150801
08/2015
08/2018
86.80
86.45

HT2
Tablet
10 mg
France
5EH3C
05/2015
05/2018
98.90
98.94

HT3
Tablet
5 mg
France
5ET4B
04/2014
04/2017
101.55
101.72

HT4
Tablet
10 mg
India
KC004
02/2014
02/2017
101.27
101.65

HT5
Tablet
5 mg
India
KD157
04/2015
04/2018
87.69
87.36

HT6
Tablet
5 mg
India
KD150
04/2015
04/2018
89.05
89.03

HT7
Tablet
10 mg
United Kingdom
13J11/C
10/2013
10/2016
94.38
94.27

HT8
Tablet
10 mg
United Kingdom
13J14/B
10/2013
10/2016
100.28
100.11

HT9
Tablet
10 mg
United Kingdom
13J12/A
08/2013
08/2016
98.22
98.09

= powder for injection (im/iv). criterion = 90–110%.

Table 9: Statistical comparison between the developed method and the USP method.

USP method (column B) versus developed method (column A)

-test

value
0.9742

value summary
Ns

Significantly different? ()
No

One- or two-tailed value?
Two-tailed

Mean ± SEM of column A
97.63 ± 1.415 = 15

Mean ± SEM of column B
97.56 ± 1.440 = 15

Difference between means
−0.06600 ± 2.019

95% confidence interval
−4.202 to 4.070

Table 10: Characteristics and labelled strength of pectin-based modified release hydrocortisone preparations analyzed using the developed HPLC method.

Code
Dosage form
Strength
Assay

HM1
Tablet
100 mg
103.36

HM2
Capsule
100 mg
95.99

HM3
Capsule
100 mg
103.29

HM4
Tablet
100 mg
102.41

HM5
Tablet
100 mg
104.38

HM6
Tablet
100 mg
95.34

criterion = 90–110 %.

4. Conclusion

It can be concluded that the developed and validated simple, isocratic, sensitive RP-HPLC method is useful for the determination and quantification of hydrocortisone in conventional and controlled-release pharmaceutical formulations. The method was successfully applied to commercial samples which indicate the ability of the analytical method to distinguish between good and poor quality hydrocortisone products.

Conflicts of Interest

The authors declare that there are no conflicts of interest regarding the publication of this paper.

Acknowledgments

The authors gratefully acknowledge the University of Ghana Office of Research, Innovation and Development (ORID) for providing a Faculty Development Grant to Ofosua Adi-Dako. They also appreciate the technical assistance provided by the technical staff of the Department of Pharmaceutical Chemistry, Faculty of Pharmacy and Pharmaceutical Sciences, KNUST, Kumasi, Ghana.

References

A. Pyka, M. Babuska-Roczniak, and P. Bochenska, “Determination of hydrocortisone in pharmaceutical drug by TLC with densitometric detection in UV,” Journal of Liquid Chromatography and Related Technologies, vol. 34, no. 9, pp. 753–769, 2011.

J. M. L. Gallego and J. P. Arroyo, “Spectrophotometric determination of hydrocortisone, nystatin and oxytetracycline in synthetic and pharmaceutical preparations based on various univariate and multivariate methods,” Analytica Chimica Acta, vol. 460, pp. 85–97, 2002.

J. M. L. Gallego and J. P. Arroyo, “Micellar electrokinetic capillary chromatography as an alternative method for determination of hydrocortisone and its most important associated compounds in local pharmaceutical preparations,” Chromatographia, vol. 56, no. 6-7, pp. 455–462, 2002.

J. M. L. Gallego and J. P. Arroyo, “Determination of hydrocortisone and associated compounds in pharmaceutical preparations by micellar electrokinetic chromatography,” Journal of Liquid Chromatography, vol. 26, no. 7, pp. 1011–1025, 2003.

C. E. Cardoso, R. O. R. Martins, C. A. S. Telles, and R. Q. Aucelio, “Sequential determination of hydrocortisone and epinephrine in pharmaceutical preparations via photochemically enhanced fluorescence,” Microchimica Acta, vol. 146, no. 1, pp. 79–84, 2004.

M. Dołowy, K. Kulpińska-Kucia, and A. Pyka, “Validation of a thin-layer chromatography for the determination of hydrocortisone acetate and lidocaine in a pharmaceutical preparation,” The Scientific World Journal, vol. 2014, Article ID 107879, 10 pages, 2014.

G. Genete, A. Hymete, and A. A. Bekhit, “Development and validation of HPTLC assay method for simultaneous quantification of hydrocortisone and clotrimazole in cream and applying for stability indicating test,” Journal of the Chilean Chemical Society, vol. 57, no. 3, pp. 1199–1203, 2012.

J. M. L. Gallego and J. P. Arroyo, “Comparison of HPLC and multivariate regression methods for hydrocortisone and lidocaine analysis of pharmaceutical preparations,” Analytical and Bioanalytical Chemistry, vol. 374, no. 2, pp. 282–288, 2002.

R. Hajkova, P. Solich, J. Dvorak, and J. Sicha, “Simultaneous determination of methylparaben, propylparaben, hydrocortisone acetate and its degradation products in a topical cream by RP-HPLC,” Journal of Pharmaceutical and Biomedical Analysis, vol. 32, no. 4-5, pp. 921–927, 2003.

B. Jancic-Stojanović, A. Malenović, S. Marković, D. Ivanović, and M. Medenica, “Optimization and validation of an RP-HPLC method for analysis of hydrocortisone acetate and lidocaine in suppositories,” Journal of AOAC International, vol. 93, no. 1, pp. 102–107, 2010.

The United States Pharmacopoeia, Rockville, Md, USA, 2011.

M. Pendela, G. Kahsay, I. Baekelandt, A. Van Schepdael, and E. Adams, “Simultaneous determination of lidocaine hydrochloride, hydrocortisone and nystatin in a pharmaceutical preparation by RP-LC,” Journal of Pharmaceutical and Biomedical Analysis, vol. 56, no. 3, pp. 641–644, 2011.

L. Nováková, L. Matysová, and P. Solich, “Advantages of application of UPLC in pharmaceutical analysis,” Talanta, vol. 68, no. 3, pp. 908–918, 2006.

S. Chan and M. Debono, “Replication of cortisol circadian rhythm: new advances in hydrocortisone replacement therapy,” Therapeutic Advances in Endocrinology and Metabolism, vol. 1, no. 3, pp. 129–138, 2010.

G. Johannsson, A. G. Nilsson, R. Bergthorsdottir, and etal., “Improved cortisol exposure-time profile and outcome in patients with adrenal insufficiency: a prospective randomized trial of a novel hydrocortisone dual-release formulation,” Journal of Clinical Endocrinology and Metabolism, vol. 97, no. 2, pp. 473–481, 2012.

in Proceedings of the International Conference on Harmonization specifications (ICH) Q2A, Validation of Analytical Procedures: Definitions and Terminology, Geneva, Switzerland, 1995, http://www.ich.org/fileadmin/Public_Web_Site/ICH_Products/Guidelines/Quality/Q2_R1/Step4/Q2_R1__Guideline.pdf.

E. Charmandari, N. C. Nicolaides, and G. P. Chrousos, “Adrenal insufficiency,” The Lancet, vol. 383, no. 9935, pp. 2152–2167, 2014.

M. J. Walters and W. E. Dunbar, “High‐performance liquid chromatographic analysis of hydrocortisone drug substance, tablets, and enema,” Journal of Pharmaceutical Sciences, vol. 71, no. 4, pp. 446–451, 1982.

M. Alvinerie and P. L. Toutain, “Simultaneous determination of corticosterone, hydrocortisone, and dexamethasone in dog plasma using high performance liquid chromatography,” Journal of Pharmaceutical Sciences, vol. 71, no. 7, pp. 816–818, 1982.

R. D. Toothaker, W. A. Craig, and P. G. Welling, “Effect of dose size on the pharmacokinetics of oral hydrocortisone suspension,” Journal of Pharmaceutical Sciences, vol. 71, no. 10, pp. 1182–1185, 1982.