Journal of Parasitology Research

Volume 2014 (2014), Article ID 213745, 6 pages

http://dx.doi.org/10.1155/2014/213745

Research Article

Evaluation of an Immunoassay-Based Algorithm for Screening and Identification of Giardia and Cryptosporidium Antigens in Human Faecal Specimens from Saudi Arabia

Yousry Hawash1,2

¹Department of Medical Parasitology, National Liver Institute, Menoufia University, Shebin-El Koum 32513, Menoufia, Egypt

²Department of Medical Laboratory Science, College of Applied Medical Science, Taif University, Taif 2425, Saudi Arabia

Received 25 October 2013; Revised 4 December 2013; Accepted 15 December 2013; Published 29 January 2014

Academic Editor: Alvin A. Gajadhar

Copyright © 2014 Yousry Hawash. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

Abstract.

An immunoassay-based algorithm, involving three commercial kits, was introduced and evaluated for screening and identification of Giardia/Cryptosporidium antigens in human stool specimens. Initially, Giardia/Cryptosporidium Chek kit (TechLab), an enzyme-linked immunosorbent assay (ELISA), was adopted for screening. The ELISA-positive reactions were subsequently characterised by RIDA Quick Giardia and RIDA Quick Cryptosporidium immunochromatographic kits (R-Biopharm). A gold standard test comprising PCR and microscopy was used for preparing control samples. Performance of individual kits was tested against these samples which included 50 Giardia-positive, 40 Cryptosporidium-positive, and 70 Cryptosporidium/Giardia-negative. For Cryptosporidium, specificities of the ELISA and RIDA Quick Cryptosporidium kits were 95.71% and 100%, respectively. Both kits demonstrated sensitivity of 95%. For Giardia, the ELISA and RIDA Quick Giardia kits showed sensitivities of 100% and 97.5%, respectively. Specificities obtained by the ELISA and RIDA Quick Giardia were 95.7% and 100%, respectively. Based on the results of two reference PCRs, on 250 random samples, the algorithm exhibited sensitivity, specificity, positive predictive value, and negative predictive value of 97.06%, 100.00%, 100.00%, and 98.91%, respectively. In conclusion, this immunoassay-based algorithm can be used as routine test in diagnostic laboratories for screening and identification of a large number of samples.

1. Introduction

Cryptosporidium and Giardia are two common intestinal protozoa causing gastroenteritis in humans [1, 2]. Many outbreaks of diarrhoea have been frequently attributed to these two protozoa [3]. Infections are often common in children and patients with weakened immunity [4, 5]. Definitive diagnosis of these protozoa requires the identification of their morphological stages, trophozoite or cyst stages of Giardia, and oocyst stage of Cryptosporidium in faecal specimens. In developing countries, this is routinely done through classical microscopic-based techniques, namely, the iodine-saline mount for Giardia and the modified Ziehl-Neelsen (mZN) staining for Cryptosporidium [6, 7]. However, these methods lack sensitivity due to the intermittent shedding or the presence of organisms in low numbers [8]. Accordingly, antigen detection immunoassays have been developed and adopted in many hospitals and public health laboratories for the diagnosis of these two protozoa [9]. The direct fluorescent-antibody tests (DFA) detecting intact organisms and the enzyme immunoassays (EIAs) detecting protozoan soluble antigens in stool are the two widely used antigen detection immunoassays [9, 10]. The sensitivity and specificity of DFA have been reported to be 96 to 100% and 99 to 100%, respectively [9–13]. In the other hand, the sensitivity and the specificity of EIAs have been reported to be 94 to 97% and 99 to 100%, respectively [9, 11, 13]. However, a large number of faecal specimens can be screened at one time with EIAs with less technical skills, less costs, less labour, and less laboratory turnaround times [14, 15]. In addition, rapid immunochromatographic-based kits have been developed and became widely used for detection of Cryptosporidium and Giardia antigens in stool samples. These lateral flow immunoassays can be accomplished within 10 minutes with a sensitivity more than 97% and specificity of 100% [16].

In Saudi Arabia, few studies have used these immunoassay-based tests for detection of Giardia and Cryptosporidium in human faecal specimens [17, 18]. In contrary, the traditional microscopic-based techniques, have been widely adopted, in many studies [19–22]. In this study, it was aimed to introduce and subsequently evaluate an immunoassay-based algorithm for screening and identification of Giardia and Cryptosporidium in human stool samples from Taif, Saudi Arabia. This algorithm is based on three commercial coproantigen kits, namely, TechLab, Inc., Giardia/Cryptosporidium Chek ELISA kit for Giardia and/or Cryptosporidium screening in a large number of samples [23], RIDA Quick Giardia, and RIDA Quick Cryptosporidium cassettes (R-Biopharm) to identify the ELISA-positive samples.

2. Material and Methods

2.1. Samples Collection and Storage

One hundred sixty diarrhoeal and nondiarrhoeal fresh stool samples were used for preparation of Giardia/Cryptosporidium positive and negative control samples. Samples were collected from those submitted to the Microbiology Laboratory at King Faisal Public Hospital in Taif, Saudi Arabia, for routine parasitological examination. Additionally, 250 faecal specimens were randomly collected between January and August 2013 for further evaluation and validation of the combined immunoassay-based algorithm. Faecal samples, without any preservatives, were properly labelled and transmitted to the Medical Laboratory at College of Applied Medical sciences, Taif University, within 2-3 hours of collection. Upon arrival to the laboratory, samples were stored at 4°C for microscopic and immunoassays testing. An aliquot of each specimen was prepared, marked, and stored at –20°C for PCR examination.

2.2. Control Clinical Samples Preparation

The aforementioned 160 samples were subjected to a combined gold standard test comprising microscopic as well as PCR testing for Giardia and Cryptosporidium. According to this gold standard test’s results, positive and negative samples were classified into control groups.

2.3. Protozoan Coproscopic Diagnosis

Unpreserved faecal specimens without prior concentration procedure were subjected to microscopic examination within 2-3 hours after collection. Wet mount preparations, one with saline and the other with iodine, were prepared and each coverslip area was scanned for Giardia trophozoite or cyst stages [7]. A third, moderately thick, wet mount smear was prepared from each specimen and subjected to modified Ziehl-Neelsen staining procedure for Cryptosporidium oocysts identification [6]. The staining procedure was carried out on batch of samples collected at the same day.

2.4. Protozoan Copro-DNA Diagnosis

Aliquots, stored at −20°C, were subjected to DNA extraction within one to two weeks after arrival to the laboratory using QIAmp Stool Mini Kit (Qiagen) following the manufacturer’s protocol. DNA extracts were subsequently subjected to amplification by two published PCR assays. PCR assays were conducted following the protocol of Xiao et al. for Cryptosporidium and the protocol of Hopkins et al. for Giardia [24, 25]. Primers were synthesized by the VHBio (Gateshead, UK). On arrival, primers were dissolved in dH₂O for stock preparation (100 pmol/μL) and stored at −20°C until use. Reactions were carried out in Techne TC-4000 thermal cycler. GoTaq Hot Start Polymerase (Promega) and other PCR reagents were used in amplification reactions with final concentrations closely similar to the published protocols.

2.5. Protozoan Coproantigens Diagnosis

Unpreserved specimens, stored at 4°C, were subjected to examination by the three coproantigen detection kits, under evaluation. Samples were initially screened for Giardia and/or Cryptosporidium coproantigens using TechLab, Inc., Giardia/Cryptosporidium Chek ELISA kit. Subsequently, the ELISA-positive specimens were tested for the presence of Giardia coproantigen by the RIDA Quick Giardia and for Cryptosporidium oocysts surface antigen by RIDA Quick Cryptosporidium kits. All immunoassays were performed according to the corresponding manufacturers’ directions. The rapid tests’ results were interpreted visually by the naked eye, while the ELISA test’s results were analysed on a multiwell scanning spectrophotometer (ELISA reader) with ≥0.150 being the cutoff for the positive sample at an optical density (OD) of 450 nm.

2.6. Data Storage and Statistical Analysis

Results obtained from examination of the clinical stool samples were stored and analysed through Microsoft Excel TM 2007. The diagnostic sensitivity, specificity, positive predictive value, and negative predictive value of various diagnostic assays were determined by standard formulae [26].

3. Results

3.1. Selection Criteria for Positive and Negative Control Faecal Samples

Based on the combined gold standard test-results on the preceding 160 stool samples, three positive control groups were assigned. Samples were diagnosed as positives by microscopy (see Figure 1) and the two PCR assays were selected as a positive control group (group-1).

Figure 1: Representative bright field microscopic pictures for the Cryptosporidium oocysts stained with modified Ziehl-Neelsen dye with ×200 magnification.

This group included 25 Cryptosporidium- and 30 Giardia-positive stool samples. Samples diagnosed as negative by microscopy but positive by the PCR tests (see Figure 2) were selected as a positive control group (group-2). 15 Cryptosporidium- and 20 Giardia-positive stool samples were included in this group. Finally, stool samples () diagnosed as Cryptosporidium- and Giardia-negative by both microscopy and PCR tests were selected as a negative control group (group-3).

Figure 2: Representative ethidium bromide-stained 1% agarose gel pictures showing amplification products (≈825 bp) of Cryptosporidium spp., reference nested PCR. Lane 1 to 10, Cryptosporidium-positive DNA samples; Lane 11 to 15, Cryptosporidium-negative DNA samples; M, λ-HindIII DNA molecular marker.

3.2. The Diagnostic Performance of TechLab ELISA Kit

The screening kit successfully detected the target antigens of all Cryptosporidium- and Giardia-positive control faecal samples apart from two (see Table 1). These two samples belonged to group-2 and both samples were Cryptosporidium-positive by the reference nested PCR. Equally important, the kit showed positive reaction in three samples from group-3, the negative control group (). Interestingly, these three false-positive samples gave no reactions when individually tested by the two subsequent discriminatory rapid kits. Based on these results, the TechLab ELISA test exhibited different diagnostic performance rates for Cryptosporidium and Giardia as shown in Table 2.

Table 1: The ELISA kit’s results compared to the assembled gold standard test's results.

ELISA kit results
The combined gold standard test resultsa

Positive
Negative
Total

Positive
88
3c
91

Negative
2b
67
69

Total
90
70
160

The combined gold standard test comprising PCR assays and microscopy.

bThese two samples with false-negative test results were diagnosed as negative by microscopy but Cryptosporidium-positive by the reference PCR assay (group-2).

cThese three samples with false-positive test results were PCR negative for Giardia and Cryptosporidium and gave no reactions when individually tested by the two subsequent rapid tests.

Table 2: The diagnostic performance of the ELISA kit for Cryptosporidium and Giardia.

Genus
The diagnostic performance parameters

Sensitivity
Specificity
PPV
NPV

SE%
95% CI
SP%
95% CI
PPV%
95% CI
NPV%
95% CI

Giardia
100%
0.9-1.0
95.71%
0.8-0.9
94.34%
0.8-0.9
100%
0.9-1.0

Cryptosporidium
95%
0.8-0.9
95.71%
0.8-0.9
92.68%
0.8-0.9
97%
0.8-0.9

3.3. The Diagnostic Performance of RIDA Quick Giardia

The kit successfully identified the target antigens of all Giardia-positive control faecal samples apart from one (i.e., 49/50). Importantly, this sample was Giardia PCR positive but negative by the microscope. No coproantigen for Giardia was detected for all negative control samples (). Equally important, no coproantigen for Giardia was detected when Cryptosporidium-positive control samples () were examined by the kit. As seen in Table 3, the RIDA Quick Giardia kit showed SE, SP, PPV, and NPV of 97.5% (95% CI: from 0.86 to 0.99), 100% (95% CI: from 0.9 to 1.0), 100% (95% CI: from 0.9 to 1.0), and 98.5% (95% CI: from 0.9 to 1. 0), respectively.

Table 3: The diagnostic performance of the two rapid tests.

RIDA Quick
The diagnostic performance parameters

Sensitivity
Specificity
PPV
NPV

SE%
95% CI
SP%
95% CI
PPV%
95% CI
NPV%
95% CI

Giardia
97.5%
0.8-0.9
100%
0.9-1.0
100%
0.9-1.0
98.5%
0.9-1.0

Cryptosporidium
95%
0.8-0.9
100%
0.9-1.0
100%
0.9-1.0
97%
0.9-1.0

3.4. The Diagnostic Performance of RIDA Quick Cryptosporidium Test

The kit successfully identified the target antigens of all Cryptosporidium-positive control faecal samples apart from two (i.e., 38/40). Interestingly, these two samples were those diagnosed as false negatives by the ELISA screening kit. No coproantigen for Cryptosporidium was detected for all negative control samples (). Equally important, no coproantigen for Cryptosporidium was detected when Giardia-positive control samples () were examined by the kit. As shown in Table 3, the RIDA Quick Cryptosporidium kit showed SE, SP, PPV, and NPV of 95% (95% CI: from 0.8 to 0.9), 100% (95% CI: from 0.9 to 1.0), 100% (95% CI: from 0.9 to 1.0), and 97% (95% CI: from 0.9 to 1. 0), respectively.

3.5. Validation of the Algorithm by Random Samples

The screening ELISA kit detected Giardia and/or Cryptosporidium coproantigens in 76 samples. By the second discriminatory kits, Cryptosporidium was identified in 24 samples (i.e., 9.6%) and Giardia was identified in 42 samples (i.e., 16.8%). Neither Cryptosporidium nor Giardia coproantigen was identified in the remaining ten samples (4%). Neither Cryptosporidium nor Giardia copro-DNA was detected when these samples were subjected to diagnosis by the two reference PCR tests. By reviewing the hospital laboratory records regarding these 250 randomly collected samples, Cryptosporidium oocysts were identified in 13 samples by microscopic examination of mZN stained wet mounts prepared from fresh unconcentrated specimens. Giardia trophozoite and/or cyst stages were detected in 33 samples using microscopic examination of iodine-stained wet mounts prepared directly from fresh faecal specimens. On the other hand, the reference PCR assays accurately detected Cryptosporidium DNA in 26 faecal samples and Giardia DNA in 42 samples (Table 4). Taking PCR tests’ results as a gold standard, the combined immunoassay-based algorithm demonstrated SE, SP, PPV, and NPV of 97.06% (66 of 68; 95% CI: from 0.89 to 0.99), 100.00% (66 of 68; 95% CI: from 0.97 to 1.0), 100.00% (66 of 66; 95% CI: from 0.94 to 1.0), and 98.91% (182 of 184; 95% CI: from 0.96 to 0.99), respectively.

Table 4: Diagnostic tests’ results on 250 random stool samples.

Test result
Microscopy
The algorithma
The reference PCRsb

Cryptosporidium-positive
13
24
26

Giardia-positive
33
42
42

Crypto/Giardia-negative
204
184
182

Total
250
250
250

The algorithm picked 20 positive samples more than the staining methods adopted at the Microbiology Laboratory at King Faisal Hospital. These samples were also positive for protozoan DNA by the matching reference PCR assay. bTwo Cryptosporidium DNA positive samples were diagnosed by PCR and missed by the algorithm. cCrypto stands for Cryptosporidium.

4. Discussion

To the best of my knowledge, none of the three coproantigen detection kits has been previously evaluated in Saudi Arabia. In this study, an immunoassay-based algorithm, intended to be used as screening test of Cryptosporidium/Giardia in stool samples, was appropriately evaluated and shown to meet the defined performance targets. It has to be clear that the differences of a test performance among studies are attributed, to a large extent, to the study methodology followed and population targeted. The prevalence of infection, which differs substantially among populations, affects the predictive value of any diagnostic test. Equally important, the gold standard test used to estimate the diagnostic performance of a diagnostic test varies among studies. A gold standard test with inadequate performance can overestimate false-positive and/or false-negative results of a diagnostic test, under evaluation.

To address the performance characteristics of the algorithm, an assembled reference standard test comprising PCR and microscopy was assembled. Samples were subjected to the combined gold standard test and the results were interpreted as follows. First, stool sample that tested as positive with microscopy was assumed to contain high parasite loads. Second, samples that were negative by microscopy but diagnosed as positive by a reference PCR test were likely to have low to moderate parasite loads. This assumption was taken relying on the previously reported sensitivities of the two diagnostic methods [9, 11, 13, 14, 27]. The TechLab ELISA kit, under evaluation, offered sensitivity towards Cryptosporidium of 95%. The kit failed to identify two Cryptosporidium-positive specimens. These two samples were microscopically negative but positive with the nested PCR. Interestingly, the optical density readings at 450 nm for the false-negative results were very close to the manufacturer-defined assay cutoff value (0.15). The Cryptosporidium oocysts surface antigens being present in these two specimens below the detection limit of the assay may be an explanation to this reduced sensitivity, especially when we know that Cryptosporidium oocysts could not be seen microscopically in the mZN stained smears. Chalmers and her colleagues reported sensitivity of 93.4% of the same kit towards Cryptosporidium control samples from United Kingdom [28]. In the same study, Remel ProSpect Giardia/Cryptosporidium and IVD Research Giardia/Cryptosporidium ELISA kits offered sensitivities of 91.4% and 92.8%, respectively. In many previous studies the kit proved to be 100% specific; however, in this study, it exhibited a slightly lower specificity of 95.71% [28, 29]. This reduced specificity of the TechLab ELISA kit could not be explained. Although the final results of the algorithm were not affected as the samples were ultimately reported as negatives, these false-positive samples required unnecessary retesting.

On the other hand, the RIDA Quick Giardia rapid test demonstrated sensitivity and specificity of 97.5% and 100%, a performance which is much better than it has achieved in previous study. Authors at the Institute of Tropical Medicine in Berlin, Germany, in 2006, have reported sensitivity and specificity figures of 80% and 99.4%, respectively [30]. Similarly, RIDA Quick Cryptosporidium lateral flow test, used in the study, showed higher performance than it offered previously. Equally important, the RIDA Quick Cryptosporidium did not cross-react with stool samples containing Giardia and RIDA Quick Giardia kit did not cross-react with Cryptosporidium-positive stool samples. Previous studies have reported false-positive Cryptosporidium infections with other rapid tests [31, 32]. Others have demonstrated higher specificity of 99.0%–100.0, closely similar to the rapid kits, evaluated in this study [33, 34].

Further evaluation of the algorithm was carried with a fairly large number of random stool samples with blinded microscopic test results as generated by routine testing procedures at the Clinical Microbiology Laboratory. Based on tests’ results on randomly collected samples, the immunoassay-based algorithm performance was comparable to two PCR tests’ results and much better than the two traditional staining methods used for Giardia/Cryptosporidium diagnosis. This performance rates achieved by the algorithm were higher than those reported for the individual kits, under evaluation, [28, 30] and others immunoassay-based kits [9, 11, 13, 14, 27]. Cryptosporidium coproantigen was confirmed in 24 samples (9.6%) and Giardia coproantigen was confirmed in 42 (16.8%). These prevalence rates of infections do not reflect the true prevalence of these two protozoa in Taif city because samples were collected within a short period. Ten samples were positive by the ELISA kit but were negative for Giardia or Cryptosporidium coproantigens by the two rapid tests. None of these samples was positive for Giardia or Cryptosporidium by the reference PCR assays and microscopy. It is clear that the high specificities offered by the two rapid tests counterbalanced the unwanted false-positive results of the ELISA kit and, as a result, the overall algorithm test results were not affected.

5. Conclusion

The immunoassay-based algorithm performed well with the collected stool samples. It was proved to be a very useful screening tool for the target specific protozoan antigens in stool specimen. The algorithm showed diagnostic performance higher than those achieved by the classical staining methods. Its results were comparable to those given by the reference PCR assays used in the study. Bearing in mind the large number of samples that can be screened, the ease of use, the less hands-on time, and its amenability for future semiautomation, the introduced algorithm has a real potential to be adopted in diagnostic laboratories, especially those having a large number of specimens likely to be tested.

Conflict of Interests

The author declares that there is no conflict of interests regarding the publication of this paper.

Acknowledgment

The author would like to thank Dr. Younes M. (King Faisal Hospital Taif, Saudi Arabia) for helping him in collecting clinical faecal specimens and reviewing their laboratory tests’ results.

References

J. Aranda-Michel and R. A. Giannella, “Acute diarrhea: a practical review,” American Journal of Medicine, vol. 106, no. 6, pp. 670–676, 1999.

S. M. Cacciò, R. C. A. Thompson, J. McLauchlin, and H. V. Smith, “Unravelling Cryptosporidium and Giardia epidemiology,” Trends in Parasitology, vol. 21, no. 9, pp. 430–437, 2005.

S. Baldursson and P. Karanis, “Waterborne transmission of protozoan parasites: review of worldwide outbreaks—an update 2004–2010,” Water Research, vol. 45, no. 20, pp. 6603–6614, 2011.

W. L. Current and L. S. Garcia, “Cryptosporidiosis,” Clinical Microbiology Reviews, vol. 4, no. 3, pp. 325–358, 1991.

M. J. G. Farthing, “Diarrhoeal disease: current concepts and future challenges. Pathogenesis of giardiasis,” Transactions of the Royal Society of Tropical Medicine and Hygiene, vol. 87, no. 3, pp. 17–21, 1993.

L. S. Garcia, D. A. Bruckner, T. C. Brewer, and R. Y. Shimizu, “Techniques for the recovery and identification of Cryptosporidium oocysts from stool specimens,” Journal of Clinical Microbiology, vol. 18, no. 1, pp. 185–190, 1983.

L. S. Garcia, Practical Guide to Diagnostic Parasitology, ASM Press, Washington, DC, USA, 5th edition, 2007.

E. Mohr and I. Mohr, “Statistical analysis of the incidence of positives in the examination of parasitological specimens,” Journal of Clinical Microbiology, vol. 30, no. 6, pp. 1572–1574, 1992.

L. S. Garcia and R. Y. Shimizu, “Evaluation of nine immunoassay kits (enzyme immunoassay and direct fluorescence) for detection of Giardia lamblia and Cryptosporidium parvum in human fecal specimens,” Journal of Clinical Microbiology, vol. 35, no. 6, pp. 1526–1529, 1997.

L. S. Garcia, A. C. Shum, and D. A. Bruckner, “Evaluation of a new monoclonal antibody combination reagent for direct fluorescence detection of Giardia cysts and Cryptosporidium oocysts in human fecal specimens,” Journal of Clinical Microbiology, vol. 30, no. 12, pp. 3255–3257, 1992.

K. S. C. Kehl, H. Cicirello, and P. L. Havens, “Comparison of four different methods for detection of Cryptosporidium species,” Journal of Clinical Microbiology, vol. 33, no. 2, pp. 416–418, 1995.

E. H. Scheffler and L. L. Van Etta, “Evaluation of rapid commercial enzyme immunoassay for detection of Giardia lamblia in formalin-preserved stool specimens,” Journal of Clinical Microbiology, vol. 32, no. 7, pp. 1807–1808, 1994.

S. K. Zimmerman and C. A. Needham, “Comparison of conventional stool concentration and preserved-smear methods with merifluor Cryptosporidium/Giardia Direct Immunofluorescence Assay and ProspecT Giardia EZ microplate assay for detection of Giardia lamblia,” Journal of Clinical Microbiology, vol. 33, no. 7, pp. 1942–1943, 1995.

G. H. Tee, A. H. Moody, A. H. Cooke, and P. L. Chiodini, “Comparison of techniques for detecting antigens of Giardia lamblia and Cryptosporidium parvum in faeces,” Journal of Clinical Pathology, vol. 46, no. 6, pp. 555–558, 1993.

L. S. Garcia and R. Y. Shimizu, “Detection of Giardia lamblia and Cryptosporidium parvum antigens in human fecal specimens using the ColorPAC combination rapid solid-phase qualitative immunochromatographic assay,” Journal of Clinical Microbiology, vol. 38, no. 3, pp. 1267–1268, 2000.

R. Chan, J. Chen, M. K. York et al., “Evaluation of a combination rapid immunoassay for detection of Giardia and Cryptosporidium antigens,” Journal of Clinical Microbiology, vol. 38, no. 1, pp. 393–394, 2000.

Z. H. Khan, S. S. Namnyak, A. A. Al Jama, and I. Madan, “Prevalence of cryptosporidiosis in Dammam and Alkhobar, Saudi Arabia,” Annals of Tropical Paediatrics, vol. 8, no. 3, pp. 170–172, 1988.

A. S. Bolbol, “Cryptosporidiosis in young children suffering from diarrhoea in Riyadh, Saudi Arabia,” Journal of Hygiene Epidemiology Microbiology and Immunology, vol. 36, no. 4, pp. 396–400, 1992.

A. Johargy, H. Ghazi, and A. Mumenah, “Frequency of viral, bacterial and parasitic enteropathogens among young children with acute diarrhoea in Saudi Arabia,” Journal of the Pakistan Medical Association, vol. 60, no. 6, pp. 456–459, 2010.

S. M. El-Sheikh and S. M. El-Assouli, “Prevalence of viral, bacterial and parasitic enteropathogens among young children with acute diarrhoea in Jeddah, Saudi Arabia,” Journal of Health Population and Nutrition, vol. 19, no. 1, pp. 25–30, 2001.

A. M. Sandokji, K. R. Murshid, A. A. El-Badry, K. H. Al-Ali, and S. A. Shalaby, “Infectious nosocomial diarrhea in the surgical wards: role of parasites and microbes imply stool analysis,” Journal of Taibah University Medical Sciences, vol. 4, no. 1, pp. 73–81, 2009.

N. F. Abbasa, K. A. El-Shaikhb, and M. S. Almohammady, “Prevalence of Giardia lamblia in diarrhoeic children in Almadinah Almunawarh, KSA,” Journal of Taibah University Medical Sciences, vol. 5, pp. 25–30, 2011.

TechLab, Giardia/Cryptosporidium Quik Chek Package Insert, TechLab, Blacksburg, Va, USA, 2011.

L. Xiao, C. Bern, J. Limor et al., “Identification of 5 types of Cryptosporidium parasites in children in Lima, Peru,” Journal of Infectious Diseases, vol. 183, no. 3, pp. 492–497, 2001.

R. M. Hopkins, B. P. Meloni, D. M. Groth, J. D. Wetherall, J. A. Reynoldson, and R. C. A. Thompson, “Ribosomal RNA sequencing reveals differences between the genotypes of Giardia isolates recovered from humans and dogs living in the same locality,” Journal of Parasitology, vol. 83, no. 1, pp. 44–51, 1997.

R. S. Galen and S. R. Gambino, Beyond Normality: The Predictive Value and Efficiency of Medical Diagnosis, John Wiley, New York, NY, USA, 1st edition, 1975.

T. G. Mank, J. O. M. Zaat, A. M. Deelder, J. T. M. Van Eijk, and A. M. Polderman, “Sensitivity of microscopy versus enzyme immunoassay in the laboratory diagnosis of giardiasis,” European Journal of Clinical Microbiology and Infectious Diseases, vol. 16, no. 8, pp. 615–619, 1997.

R. M. Chalmers, B. M. Campbell, N. Crouch, A. Charlett, and A. P. Davies, “Comparison of diagnostic sensitivity and specificity of seven Cryptosporidium assays used in the UK,” Journal of Medical Microbiology, vol. 60, no. 11, pp. 1598–1604, 2011.

S. Youn, M. Kabir, R. Haque, and W. A. Petri Jr., “Evaluation of a screening test for detection of Giardia and Cryptosporidium parasites,” Journal of Clinical Microbiology, vol. 47, no. 2, pp. 451–452, 2009.

T. Weitzel, S. Dittrich, I. Möhl, E. Adusu, and T. Jelinek, “Evaluation of seven commercial antigen detection tests for Giardia and Cryptosporidium in stool samples,” Clinical Microbiology and Infection, vol. 12, no. 7, pp. 656–659, 2006.

T. J. Robinson, E. A. Cebelinski, C. Taylor, and K. E. Smith, “Evaluation of the positive predictive value of rapid assays used by clinical laboratories in Minnesota for the diagnosis of cryptosporidiosis,” Clinical Infectious Diseases, vol. 50, no. 8, pp. e53–e55, 2010.

A. Cronquist, M. J. Beach, S. P. Johnston, and A. da Silva, “Manufacturer's recall of rapid cartridge assay kits on the basis of false-positive Cryptosporidium antigen tests-Colorado,” Morbidity and Mortality Weekly Report, vol. 53, p. 198, 2004.

S. P. Johnston, M. M. Ballard, M. J. Beach, L. Causer, and P. P. Wilkins, “Evaluation of three commercial assays for detection of Giardia and Cryptosporidium organisms in fecal specimens,” Journal of Clinical Microbiology, vol. 41, no. 2, pp. 623–626, 2003.

L. S. Garcia, R. Y. Shimizu, S. Novak, M. Carroll, and F. Chan, “Commercial assay for detection of Giardia lamblia and Cryptosporidium parvum antigens in human fecal specimens by rapid solid-phase qualitative immunochromatography,” Journal of Clinical Microbiology, vol. 41, no. 1, pp. 209–212, 2003.