Reliable Lifespan Evaluation of a Remote Environment Monitoring System Based on Wireless Sensor Networks and Global System for Mobile Communications

The use of wireless sensor networks (WSN) for monitoring physical and chemical variables in large areas allows density and frequency measurements which have been unavailable to date in classical measurement systems. To fully take advantage of this technology in a particular application, besides an accurate design and selection of all the components involved in its operation, it is essential to dispose of reliable lifetime estimation prior to deployment. This paper presents an experimental approach to determine the actual lifetime of such battery-operated systems,making use of a customWSNarchitecture, and for different batteries technologies. To render a reliable evaluation, the energy consumption of the sensor nodes under their different operation modes, in correlation with the battery characteristics and the voltage regulation system, is jointly considered. The result is a complete and practical lifetime model, whose appropriate performance has been validated in a real deployment scenario.


Introduction
The fast development of wireless communications systems for the past two decades has favored the emergence of many previously unthinkable applications.An emerging example is that of wireless sensor networks (WSN), composed of a large number of sensor nodes capable of monitoring different physical and chemical magnitudes from the environment in which they are distributed and where the transmission of data is done through a suitable RF module.These systems are mostly based on the Low Rate Wireless Personal Area Network (LR-WPAN) IEEE 802.15.4 communication standard.This standard is designed to be compliant with the requirement of low-power consumption, thus allowing being fed by small batteries, with a lifetime of months or even years.
The limited range of the IEEE 802.15.4 RF modules (usually a few hundred meters) requires the use of techniques of multihop data transmission to cover larger monitoring areas.In multihop transmission, some sensor nodes act as cell stations for data provided by other nodes.Thus, it is possible to achieve greater transmission distances.The information collected by a WSN should be finally transmitted to a central node (usually connected to a PC), which will process and represent the received data in order to get straightforwardly interpretable results.
This work presents the development and characterization of a sensor node platform, validated through real WSN deployment and test.Since nowadays WSN energy efficiency remains a primary challenge for the success of these networks, one key part of the work is the study and analysis of the power consumption with the objective to obtain a simple but reliable experimental lifetime estimation model that prevents a battery network failure.
The paper is structured as follows: Section 2 shortly presents the state of the art about environmental monitoring based on wireless sensor networks.Section 3 provides a brief description of the sensor-router nodes including hardware, software, communication characteristics, and power management techniques.Section 4 addresses the proposed methodology for modeling the battery lifetime of a basic sensor node within the custom WSN platform.Section 5 extends this analysis to the coordinator nodes.To complete the network description, Section 6 is focused on the central node, where the data collected by the coordinator nodes are received and properly represented and processed.Section 7 shows the network test, performed to evaluate both the complete system operation and the battery lifetime prediction model.Finally, Section 8 draws the conclusions.

State of the Art
Wireless sensor networks (WSN) have become a key environmental technology, with applications in intelligent agriculture [1,2], diffuse greenhouse gas emission detection, smart composting monitoring and control [3], prevention and early detection of forest fires, or protection of critical infrastructure [4], among others.Nevertheless, currently available commercial WSN generic platforms need considerable improvements to meet the stringent requirements desirable for real open nature deployments, mainly in terms of complexity, cost, lifetime, and maintenance.As a result, the development of custom nodes to replace commercial ones to enhance performance is gaining strength.In this attempt, most results remain at laboratory test level, being rather hazardous to find implementations validated in an actual deployment scenario.In fact, focusing on the implementation and test of complete full-custom WSN dedicated to monitoring environmental parameters, only a few solutions can be found in the open literature [5][6][7][8][9].However, the use of sophisticated and powerconsuming sensor devices, communication technologies, and protocols results in cost per node, energy requirements, transmit data rate, and required infrastructure which render these solutions unsuitable for battery-operated nodes deployed in natural areas [5][6][7].In [5], IP video cameras are used for smoke and fire detection, which require high computational power, increasing the cost of both software and hardware; in [6], a Digital Enhanced Cordless Telecommunications (DECT) subnet is connected to a Global System Mobile (GSM) coordinator which communicates with a PC control center; in [7] a combination is proposed between ZigBee, Wi-Fi, and satellite communications.Solutions in [8,9] focus the power optimization on efficient data communications: in [8], a wireless sensor network for microenvironmental application achieving efficient node data transmission is presented, showing a web-based tool for monitoring the deployed network, controlling data traffic, data processing and presentation, and remote node control and monitoring; in [9], the authors propose a network with a duty cycle management able to address stability in network connections and coverage at the same time.By synchronizing disjoint sensor subsets, the system can detect events which could be omitted by classical communication architectures, achieving a tradeoff between power consumption and event detection.This is useful in applications where event lost are not allowed, as defense tasks.
In this scenario, with power, node size, and cost per node being the restrictive conditions, specific novel designs are desirable to come across all the main challenges encountered in environmental projects.Accordingly, a custom low-power low-cost reliable WSN platform has been developed by authors for use in remote areas, which can be easily adapted to other monitoring systems.Based on a first prototype [10,11], a number of modifications have been included in order to reach an enhanced WSN implementation mainly concerning the following: (i) the reduction of power consumption at both software and hardware levels; (ii) communication developments (different communication protocols, such as GSM/GPRS and IEEE 802.15.4,can be used, adding more versatility to the proposal); and (iii) time synchronization stability (a real-time clock fed by a supercapacitor is added for data integrity purposes).
Making use of this WSN architecture, this paper is focused on finding a reliable estimation of the lifetime; this is an essential information before deployment, especially in systems targeting remote environmental monitoring, which are designed for long-term unattended operation.Despite its importance, this key metric is usually parameterized through predictions that often use high level descriptions and/or ideal or basic battery models, simplistic assumptions which overestimate lifetime.Recent works progressively include more complete analysis but still exhibit limitations since they are mainly based on emulation methods, usually developed for particular platforms, which ultimately fail to consider real effects.The simulator mTOSSIM [12] evaluates lifetime in WSN but is restricted to platforms embedding the TinyOS operative system; power consumption is estimated at microcontroller instruction level, varying different network parameters (duty cycle, transmission frequency, and idle or sleep transceiver modes) and using battery models based on technical specifications.Similarly [13] is an emulatedbased method using MSPSim, which incorporates battery models with nonideal effects and a low-level description of the node hardware.In [14], authors consider the average current consumption over the different operation processes (transmission, reception, and idle or sleep states) and the corresponding consumption of the different platform elements in these states, but the battery model does not consider the influence of self-discharge, temperature, and recovery effects over the battery capacity.In addition, sensors are not included and the communication is only made between an end device and the coordinator, which are not real operation conditions.Following an experimental approach, in [15] authors use LR06 batteries of different brands and capacity in order to evaluate their influence on a node lifetime; however, the study is based on a node in listen state, considering neither transmission nor sensor measurement processes and thus not describing a real network behavior.
From the above discussion, it is clear that an accurate lifespan prediction model must consider the energy consumption of the sensor nodes under their different operation modes, in correlation with the real battery characteristics and the voltage regulation system, that is, including all hardware and software constraints.To do so, we consider that an experimental methodology is the best suitable choice.This is the goal of this paper: to attain a complete and practical lifetime model, based on the remaining energy battery level as standardized metric through an experimental methodology: using real nodes under real operating conditions, from the measured power consumption profile and the measured battery discharge pattern, the battery discharge rate is inferred.Preliminary results were introduced in [16 different battery technologies, and validates results through a real outdoor deployment.

WSN Description
3.1.Sensor Node Architecture.The implemented basic node consists of the following core elements (Figure 1): (i) An 8-bit low-power CMOS ATmega1281 microcontroller from Atmel that manages and synchronizes the node operation has been used.
(ii) An XBeePro transceiver operating in the 2.4 GHz Industrial-Scientific-Medical (ISM) free RF band [17] has been included.By properly configuring the firmware parameters that control the transceiver operation, it is possible to drive all the network nodes to a low-power sleep state, thus providing a multihop energy-optimized communications protocol.
(iii) A real-time clock (RTC) model PCF2123 from NXP has been included.This component has been included to (i) enable arrangement of the collected data and (ii) detect errors in data transmission from the sequential information that it provides.This clock is programmed and activated before the network is deployed.It is firstly powered by a preloaded supercapacitor which stores the energy required for 5 days of RTC operation prior to its deployment, thus saving battery resources.In addition, powering the RTC by a supercapacitor allows replacement of node batteries without losing date information.
(iv) A power supply system is used consisting of a battery connected to a SEPIC DC-DC TPS61131 converter from Texas Instruments, which supplies the node components with the required energy and regulated voltage level (3.3 V).Also, a 100 mF supercapacitor feeds the RTC for a further 7 days even when the node battery is fully discharged, thus avoiding the need to reprogram the clock.
(v) A 2.4 GHz 7  and 5 dBi dipole vertical polarization antenna has been used which allows a maximum outdoor range of about 1600 m.
(vi) An IP65 standard protection box is used.

Sensor Types and Connectivity.
To attain a versatile configuration, a custom low-voltage plug-and-play programmable electronic interface suitable for connecting both active and passive analog sensors has been developed.Programmability allows achieving an optimum reading performance for every connected sensor.

Network Communications.
Depending on the WSN location, communication between the sensor nodes and the central node can be carried out in different ways: if the WSN is deployed in a region with mains infrastructure (such as indoor applications), nodes could be powered directly from the mains.In this case, energy-saving restrictions can be relaxed.However, in most cases sensor networks are not installed in regions with these infrastructures.The nodes therefore send the information to the coordinators using radiofrequency (RF) low-power communication protocols, most based on the IEEE 802.15.4 standard.Among these, some protocols [18] can drive the sensor nodes to a lowenergy mode but keep the router devices awake.These algorithms have several disadvantages: first, sensor distribution must be carefully selected to cover the transmission area of all the sensor nodes.Also, network scalability and selforganization are limited because of the need of performing the configuration of the nodes as end devices or routers before being installed.In addition, the operating life of the sensor network is clearly limited by the batteries of these router nodes.These limitations are overcome by using the XBeePro transceiver including the DM-24 firmware, as will be shown next.Furthermore, this transceiver is fully pinout compatible with the WiFly 802.11 b/g transceiver from Roving Networks [19] that allows the use of the same hardware node with high network versatility.

Power Management.
The difficulty of battery replacement for sensors deployed in natural environments, joined to the unfavorable cost and size of most of battery recharge systems, makes lifespan a critical issue.Thus, in order to increase the network lifetime, the nodes use a Dynamic Power Management (DPM) technique.The microcontroller sets the digital components of the node (transceiver, smart sensors, and the microcontroller itself) into their corresponding lowpower modes when they are not active.Also, the polarization of analog components (like analog sensors, amplifiers, and filters) is controlled through an ADG701 digital switch from Analog Devices.Thus, when analog components are not in use, the microcontroller switches off the corresponding bias path.The node transceiver has five controllable power levels (Table 1) that allow optimizing the power consumption of the device.Finally, the IEEE 802.15.4 firmware programmed into the transceivers allows all the nodes to operate as Full Function Devices (FFD).In this way, the network router nodes are enabled to be in sleep mode without losing the network architecture, unlike the ZigBee specification [20,21].Note that a router which is permanently in the awake mode means a continuous current consumption of 67 mA (due mainly to the RF transceiver).So, if fed by a 2 Ah battery like the one finally used in this work, the operating life will be limited to only 30 hours.
Other power management widespread techniques are dynamic voltage scaling and dynamic frequency scaling.Dynamic voltage scaling (DVS) [22,23] allows reducing the power consumption in low activity node modes.However, its implementation requires the use of additional electronics to provide the suitable different voltage supply values for the node components, increasing the final cost.In our case, due to the narrow voltage operation range of the XBee transceivers, whose bias voltage is limited between 2.8 V and 3.4 V, the nodes include a SEPIC DC-DC converter that holds the voltage level between these limits.Reducing the bias voltage below 2.8 V resets the transceiver, thereby hardly limiting the DVS efficiency.In addition, in applications with a low duty cycle such as environmental monitoring, the required linear voltage converter needed for this method increases the power consumption in the low-power operation mode, degrading the node lifetime.On the other hand, the use of dynamic frequency scaling (DFS) [24,25] allows reducing power consumption by dropping the system frequency according to the node operation requirements, though this technique is only suitable for microcontrollers that include specific integrated components.In some C without these specifications, such as the one used in this work, it is possible to implement DFS using an additional RC circuit oscillator, which generates an output frequency value controlled by an iterative algorithm, but at a high cost in computational time and energy which drastically reduces its performance, and thus has been disregarded.

Commercial WSN Platforms
Comparison.This basic sensor node component has been designed for flexibility, allowing for easy configuration as a sensor, sensor-router, or coordinator node.Table 2 shows a comparison of the proposed node with similar commercial devices.Crossbow nodes present lower energy requirements, but with 5 to 10 times lower transmission range.Therefore, for applications where sensors may be located several hundreds of meters apart, the need would arise to deploy additional router nodes, thereby increasing the network cost and complicating maintenance.On the other hand, the Waspmote shows similar outdoor and indoor transmission ranges.However, improved energy management is achieved in the proposed nodes, resulting in better overall performance.In addition, the use of a SEPIC enables biasing the nodes with a wider input voltage range.

Lifetime Model
Depending on what is under consideration, several different battery modeling methods can be found in the literature [26,27]: physical, abstract, empirical, and mixed models, based on a combination of the previous modeling techniques.Physical modeling is the most accurate method, but at the cost of a high complexity that makes its practical application difficult.Abstract modeling describes battery operation based on electronic models, and it becomes highly suitable when the full system is to be simulated as a circuit.Finally, empirical modeling provides the simplest models, based on a numerical approach to describe the battery behavior.In this work, an empirical battery model has been developed.Following a systematic methodology, the proposed modeling considers the effect of the battery voltage reduction during the discharge process in real operation and thereby permits simple but realistic lifetime estimation.Besides, as several characteristics differ between rechargeable and nonrechargeable batteries, lifetime analysis has been performed using both battery type examples: a rechargeable 3.7 V-2000 mAh Lithium Polymer (LiPo) battery and 2 × 1.5 V alkaline LR06 batteries with 2200 mAh of charge.
The energy consumption of the basic sensor node is first analyzed to obtain the behavior in the different states over a working cycle.To perform this analysis, the node was configured as follows: (i) both analogue and smart digital sensors were connected: ambient parameters within the housing box are measured using two low-cost resistive analog sensors, NTC for temperature and Sencera H25K5A for relative humidity, while external parameters were measured using two digital smart sensors, Sensirion SHT11 to measure the relative humidity and Intersema MSB5540B for barometric pressure and temperature; (ii) a power transmission level of 18 dBm (worst case) is selected; and (iii) the work cycle is set to a measuring time of 1.746 s every 900 s (15 min), that is, a duty cycle of 0.18%, a suitable choice for our target application.With this node configuration, sampling the power every 2 ms using a 2602A System SourceMeter (SMU) from Keithley Instruments connected to a PC through a USB-GPIB adapter, the power consumption profile of the full node presents three different levels: the first one (( 1 ), 898 s) corresponds to the system in sleep mode; the second one (( 2 ), 448 ms) corresponds to the parameters' measurement time, with microcontroller, transceiver, and sensors switched on; lastly, in the third one (( 3 ), 1.316 s), both sensors and microcontroller are switched off, whereas the transceiver remains on to complete the data transmission.In measurement (2) and transmission (3) stages, the power consumption is constant ( 2 = 220 mW and  3 = 235 mW, resp.), whereas, in sleep mode (1), the current remains almost constant over all the battery operation range, with an average value below  1 = 12 A.
Next, the battery behavior is characterized.The two selected alternatives, a rechargeable 3.7 V, 2000 mAh LiPo and a 2 × 1.5 V LR06 2200 mAh alkaline battery, were   discharged through the SEPIC converter at a constant power of 225 mW to match our working conditions.Figure 2 shows the resulting battery voltage drop versus time measured using an Agilent 34410A Multimeter connected to a PC through a USB-GPIB interface, sampling the battery voltage every 60 s.In the case of LiPo, for battery voltage ranges from 4.2 to 3.6 V, the behavior can be modeled according to the approximated expression in (1), and once the battery voltage falls below 3.6 V, its value quickly drops to 2.6 V.For the alkaline batteries, operation fails when the voltage across both batteries drops to 1.8 V, that is, the minimum operation voltage of the SEPIC converter.In this case, the related behavior model is given by (2): Based on the results on Figure 2, Figure 3 shows the measured and modeled remaining energy versus battery voltage V() for a maximum initial battery charge of 2 Ah (120,000 mAmin) for the LiPo battery and 2.2 Ah (132,000 mAmin) for the alkaline one.In this paper, capacity  (usually expressed in mAh) and energy (expressed in Joules) are used interchangeably.
Then, it is possible to determine the battery lifetime by evaluating the remaining charge as a function of the energy consumed by the full-node electronics.By assuming a 15minute work cycle and taking into account the three different power consumption profiles present over cycle ( 1 ,  2 , and  3 ), the battery discharge for a work cycle can be estimated according to where  0 - 1 are the limits of the sleep mode timespan, where  1 is set to 15 A to ensure a conservative estimation of the lifetime and partially compensate possible effects not considered in the model that could worsen the battery operation,  1 and  2 define the limits of the measurement timespan, where  2 =  2 /V(), and  2 and  3 define the transmission timespan, where  3 =  3 /V().Using this onecycle model, the evolution in the battery charge is iteratively calculated over time.Figure 4 shows the discharge process for the two considered battery types connected to a wireless node in full operation under our work conditions in view of different effects: ideal discharge (from (3)), considering maximum power transmission spikes and considering selfdischarge effects.In both cases, the discharge curve is highly linear.We next analyze power transmission spikes and selfdischarge dependences.Discharge spikes have been measured using an Agilent Mixed Signal Oscilloscope 9409A connected to an Agilent N2783A current probe.Despite these discharge spikesdue to transmission operations-the operation lifetime is not significantly affected.For this reason, the blue and red lines are overlapped in both graphics in Figure 4.In fact, battery lifetime can be just slightly increased by selecting the suitable power transmission level (Table 1): in the maximum transmission level #4 the power transmission spikes are 1 W high.This value is reduced by 165 mW per transmission spike for each power level reduction.Thus, assuming an ideal data transmission process for all data frames (without retries), the lifetime difference between the highest (4) and lowest (0) power level modes is less than 30 hours using the proposed duty cycle.
However, the repercussion of the self-discharge effect significantly influences lifetime, especially with rechargeable batteries.In the case of LiPo used in this work, it is 20% [27].Thus, depending on the application requirements it is very important to make a careful choice of battery.Considering all the main effects for LiPo and the alkaline batteries, the equations that represent lifetime are given by  () = −16.447+ 120210 [mAmin] , (4) respectively.The results obtained using (3) render a lifetime of around 670 days for a LiPo battery in ideal conditions.However, by considering the self-discharge effect this value, according to (4), falls to 300 days.Therefore, the use of this battery type requires a recharge system if longer operation time is expected.The same estimation is made with the alkaline batteries.In this case, the ideal lifetime is 500 days, less than the LiPo battery.In contrast, the lifetime, adding the effect of self-discharge, is reduced by only 20 days, giving an operating time of 480 days.

Coordinator Node Architecture
5.1.Preliminary Considerations.When using a GSM/GPRS coordinator, there are several options for transmitting data from a WSN to a central node.First, communication technology must be selected (the Internet mobile, phone call, short message service, etc.).Some authors propose the use of Internet mobile technology [28,29].However, this choice requires high infrastructure and communication bandwidth, which may be impractical in many cases.Alternatively, the short message service (SMS) requires much less infrastructure and bandwidth, making it suitable for such applications.In addition, all providers of mobile phone services provide SMS as a basic service without special subscription.Furthermore, this solution can be easily adopted using a portable satellite phone service and can therefore be used worldwide.
For an efficient network sensor management, the optimum maximum network size must be estimated.Although the number of motes connected to a single coordinator node in a WSN is theoretically unlimited, energy constraints and cost make it unfeasible.For most applications, networks consisting of 100-200 nodes connected to a single coordinator are adequate for monitoring large areas.If more nodes are to be connected, further coordinator nodes can be added to form subnetworks to send the information from the new items to the central node.
The data transmission protocol to the central node poses an important restriction: hardware limitations in GSM modules will limit the data size to be sent through an SMS.The module used in this work is a GM862 from Telit [30,31], whose bias voltage ranges from 3.4 to 4.2 V with maximum current peaks of 2 A. Assuming that each sensor node collects the information from 16 sensors and a sensor measurement is represented by 2 bytes, the sensor data transmitted by a node to its coordinator are stored in 32 bytes.Also, a sensor node sends 7 bytes corresponding to the time and date information provided by the RTC module.Therefore, including other additional parameters such as node address, the total information forwarded from a sensor node to its coordinator is sent in 50 bytes.In addition, the GSM module only transmits printable characters; thus, the data bytes received, represented as hexadecimal values, are converted to ASCII characters.Then, the coordinator node concatenates the information before the transmission through the GSM.According to the data size, information is split into the required number of SMS frames and sent consecutively.Figure 5 shows an SMS complete data frame.Because the data received by the network coordinator could require several concatenated SMSs, the first bytes of each SMS identify the position of the SMS in the full SMSs thread.The next bytes indicate the length of the total information sent by the network coordinator, while the following information corresponds to the length of the information sent by one node.The rest of the SMS frame contains the information collected by the corresponding sensor nodes, including its identification and the sensor readings.The last byte is an end of transmission (EOT) code.Two different ways exist to store the data received from the sensor nodes in the coordinator before forwarding it to the central node: by using either the memory of the node microcontroller or that of the GSM module.Although the data size is similar in both cases, the program required to send data via SMS from the microcontroller memory is simpler.Therefore, to simplify the coordinator setup and ensure correct operation of the system, the data are stored directly in the microcontroller memory, thereby limiting the number of sensor nodes to 100 (<90 SMS per data transfer).

Hardware Design.
The core architecture of a coordinator node is the same as that of a sensor node (see Figure 1).However, the connection of the sensor device to a GSM module requires certain changes in the power management system and the software programmed into the microcontroller.As the value of the GSM bias voltage is higher than that of the rest of the node components, the 3.7 V Lithium Polymer battery is selected as the energy source.It feeds the GSM module directly, whereas the rest of the electronics are powered through the SEPIC DC-DC converter that regulates the voltage to the suitable 3.3 V.
The node microcontroller is responsible for arranging the data provided by the nodes in the network to be sent in the SMSs, starting the GSM module, and sending the messages.Communication between the node microcontroller and the GSM module is performed through a serial protocol, using a Universal Asynchronous Receiver-Transmitter (UART) port.Because of the difference in the bias voltage of the node microcontroller (3.3 V) and the GSM module (from 3.4 to 4.2 V), signal accommodation is required: the microcontroller receiving port is able to accurately read the data sent by the communications module; however, the receiving port of the GSM module needs voltage attenuation in the logic signals for a suitable interface.

Microcontroller Software.
Control of the GSM module operation is performed by the coordinator node microcontroller by means of a Finite State Machine (FSM).This FSM controls the system evolution through the different defined states, from the start pulse of 2 s, which turns on the device, until the same pulse is sent for shutdown.The FSM also allows monitoring the correct transmission of the GSM AT commands, thus avoiding undesired states of the system that could drive the communications module to a malfunction.Accordingly, the FSM evolves to the next state only when the GSM module confirms that the command has been successfully processed.The FSM flowchart is shown  in Figure 6.In a measurement operation, once the wake-up signal is given by the transceiver to the node components, the node microcontroller first checks if there are data to be sent to the network controller via SMS.In this case, the C sends a wake-up pulse to the GSM device.Data transmission from the C to the GSM module starts when the measurements of the corresponding node sensors are fully collected by the node coordinator.Parameters of the GSM connection to the service provider (e.g., pin number) are sequentially sent to the module from the node microcontroller, verifying the proper reception by the corresponding acknowledgment (ACK) signals.When the module configuration is completed, the microcontroller sends the data to be transmitted to the GSM, which then starts the corresponding transmission.If the process is successfully executed, the GSM module is switched off until a new data transmission is to be started.Parallel to this supervision method a timer controls the confirmation timeout of the communications module.If this time is exceeded, the system restarts the FSM state and repeats the process of sending the command.This time must be long enough to avoid the loss of the acknowledgement responses sent by the GSM module when a command is properly received and processed.We have found that a timeout between 1 and 2 s is adequate.A module reset has also been implemented, which the microcontroller activates when the module stops answering; however, this GSM reset must be avoided whenever possible as it does not send a disconnection notice to the GSM network.

Power Management.
The GSM module power consumption constitutes the main reduction factor on the coordinator node battery.It includes a standby mode in which it is possible to receive SMS.However, if the time between two consecutive data transmissions is set over the minute, the power consumption in this standby mode will be higher than the consumption for the process of switching on the module, connecting to the GSM network, transmitting the data, and switching off.Typically, in environmental applications, the time communication between two consecutive data transmissions is usually of several minutes, and thus we chose this second option of operation for the GSM module.Estimating the consumption in an SMS transmission cycle by averaging the measured consumptions over the different states from the module startup to switch off, with the same sensors as the previously reported sensor node, the coordinator node reaches 3.11 mA consumption per cycle due to the presence of the GSM module.If the node is powered our LiPo selected battery (2000 mAh) lifetime is below one month; for a 6000 mAh battery, similar in size to those in the sensor nodes, the lifetime will be below 3 months.Therefore an additional energy harvesting system (Figure 1) is required to extend it.

Central Node Architecture
6.1.GSM Data Receiver Module.For the central node, where data are received, consumption is not a main issue.In this case, the module is connected to a PC, which subsequently processes the information.Therefore, an energy source must be available for the PC and, hence, for the GSM module, that can be switched on permanently.
The GSM receiver hardware has been implemented following the manufacturer's specifications and using only the relevant features for this application, thus minimizing the design requirements.The power supply complies with the requirements of the module: an output voltage between 3.4 and 4.2 V, providing a maximum peak current of 2 A.

PC Control.
A software application has been specifically developed for SMS reception, data unthreading, and information processing and display.This software has been implemented using the Graphical User Interface Development Environment (GUIDE) from MATLAB.The application can interpret the information received by the coordinator node independently of the communication technology; namely, it can process the data irrespective of whether it is received by IEEE 802.15.4,SMS, or WLAN.It allows selecting the type of transmission protocol, making it possible to both receive or send data and change the network settings.Moreover, the application can arrange the data collected by various sensor types and locations (nodes) using the accurate time stamp provided by the RTC chip found in the hardware of the sensor nodes.Making the desired selection, this interface allows data from different nodes of the sensor network to be shown.It is also possible to choose the parameters to be monitored (relative humidity inside or outside the node housing box, temperature, etc.).In addition, a collection of historic data from a specific node or a group of them can be displayed.Finally, the interface allows changing different network parameters (measurement frequency, maximum network hops, mesh retries, node reset, power level, node disconnection, etc.), thus adapting the network architecture to the evolution of the environmental conditions.

Network Test
To validate the WSN operation, a network prototype consisting of 11 sensor nodes was deployed for a 6-month period at the Faculty of Science at the University of Zaragoza.The node distribution is shown in Figure 7. Green thumbtacks represent nodes located inside the building, whereas yellow thumbtacks are nodes located on the roof.The network coordinator is represented by a red thumbtack.Table 3 shows the complete deployment network characteristics and information on the nodes' spatial distribution (location, height, line or non-line of sight, power transmission, and role).All match the previously performed analysis; that is, each sensor node includes an NTC thermistor and an analog Sencera H25K5A sensor for monitoring temperature and relative humidity in the sensor housing box; external parameters are monitored through a Sensirion SHT11 smart relative humidity and temperature sensor and an Intersema MS5540B smart pressure sensor.The duty cycle is of 900 s (15 min), with 1.746 s awake time (including measurement and transmission operations), and power supply is provided by a 3.7 V LiPo battery through a SEPIC DC-DC TPS61131 with fixed 3.3 V output.
Before deploying the network, sensor nodes are configured to assure a fast network connection: maximum transmission power level (18 dBm), sleep support mode (which allows setting all the nodes to sleep mode), and low sleep period.After deploying the nodes, the central node starts a searching process.Once a node is associated to the network, its parameters are set to their final values (sleep and wake periods, maximum network hops, etc.).
The tested network revealed that the success frame transmission ratio is highly dependent on the suitable configuration of the transceiver firmware.The inclusion in the transmitted data of the timing information provided by the RTC included in the node electronics enabled both time arrangement of the received frames and the determination of the missing messages.As the date of every node transmission is made known through its RTC, the actual date provided by the PC allows organizing the node transmissions in time.Also, as the central node knows the number of nodes in the network, it is possible to discover any unsuccessful transmissions by comparing the number of transmissions received from other nodes following the time corresponding to the maximum number of transmission retries per node (mesh retries).Table 4 shows the average Package Error Ratio (PER) for the presented configuration with mesh retries limited to three and the maximum network hops to six.
The environmental monitoring results obtained in node #1 over two days are shown in Figure 8. Figure 8(a) corresponds to the temperature measured in the environment (blue) and inside the container (red).Figure 8(b) is the relative humidity in the environment (blue) and inside the container.Finally, Figure 8(c) shows the barometric pressure.As shown in the figures, the inside relative humidity is almost constant.This indicates that the prototype is adequately isolated from the environment, which is the desired effect using an IP65 box container.In the same way, the temperature evolution presents similar values inside and outside, the difference given by the slower inertia presented by the inbox sensor at temperature variations.
The suitability of the battery discharge model developed in this work was tested by monitoring the battery energy evolution in nodes #1, #6, #7, #8, and #11 (Figure 9) for one month.Compared to measurements, the developed model (including transmission spikes and battery self-discharge) shows an average error of below 5% for outdoor nodes and 10% for indoor nodes.Differences between data and discharge model are due to not only the model errors but also the data sending retries, not included in the model.In fact, note that indoor node batteries show less adequate model fitting.This difference is mainly due to the number of data frames lost.In this case, the node retries required to send lost frames which are set by the mesh retries transceiver parameter consume more battery energy, thereby reducing its operating life.
Different tests have been carried out to check the effects of inclusion of new nodes in the network already deployed, working and achieving successful results.The network reconfiguration due to the changes in the topology uses a work cycle; that is, during the first cycle after deployment of the new nodes, the system registers their inclusion; then data begins to be sent to the coordinator in the next cycle.Additionally, the effects on battery life for a node acting as a router for up to 10 sensor nodes have been analyzed.Using the previously specified network parameters, the measurementtransmission energy profile presented by a router is similar to that of a sensor node but presents additional power peaks whose influence on the average power consumption is below 1.5% per routed node.Therefore, more powerful batteries must be considered with mesh topologies where nodes close to the coordinator are acting as routers for several sensor devices.

Conclusions
This paper presents a general approach to accomplish a simple but reliable lifetime prediction of a battery-powered wireless sensor node.For this, an experimental based methodology including all hardware and software constraints has been developed, using a custom low-power low-cost reliable WSN platform.Thanks to it, prior to deployment in the field, it is possible to estimate the application's lifetime, while adopting strategies (adjustment of parameters such as power transmission level and duty cycle) to increase the network lifetime.The analysis of different kind of batteries allows comparing the best selection for each network deployment, depending on the application.Finally, through a real deployment, we have been able to test the complete system, validating the lifetime predicting model, while the WSN has proved to be robust for environmental monitoring.One shortcoming that has emerged is that, due to the high energy consumption of the GSM modules included in the coordinator units, their operating life is rather restricted.Thus, present studies are  oriented towards the development of an efficient energy harvesting system for the coordinator node in order to increase network autonomy.

Figure 2 :
Figure 2: Experimental battery voltage (V) versus time (min) for a constant 220 mW discharge process.

Figure 5 :
Figure 5: Data collection format sent by the GSM coordinator through Telit GM862 module.

Figure 7 :
Figure 7: (a) Test sensor network deployment.Yellow thumbtacks correspond with outdoor nodes and green thumbtacks with indoor nodes.The red thumbtack is the network coordinator (scale: 1 : 2500, © 2013 Google); (b) node deployment detail; and (c) sensor node architecture.

Figure 9 :
Figure 9: Lifetime with ideal self-discharge model, comparing with lifetime measures for indoor and outdoor nodes.
], for a LiPo battery.This work deepens and completes the study, considering two
* Measures include sensors and RTC.