Journal of Spectroscopy
Volume 2016 (2016), Article ID 1596845, 7 pages
http://dx.doi.org/10.1155/2016/1596845

Research Article

Discrimination of Nasopharyngeal Carcinoma from Noncancerous Ex Vivo Tissue Using Reflectance Spectroscopy

Zhihong Xu,¹ Wei Huang,1,2 Duo Lin,1,3 Shanshan Wu,¹ Maowen Chen,¹ Xiaosong Ge,¹ Xueliang Lin,¹ and Liqing Sun⁴

¹Key Laboratory of Optoelectronic Science and Technology for Medicine, Ministry of Education, Fujian Normal University, Fuzhou 350007, China
²Fujian Metrology Institute, Fuzhou 350003, China
³College of Integrated Traditional Chinese and Western Medicine, Fujian University of Traditional Chinese Medicine, Fuzhou, Fujian 350122, China
⁴Affiliated Fuzhou First Hospital of Fujian Medical University, Fuzhou 350009, China

Received 2 January 2016; Revised 16 March 2016; Accepted 28 March 2016

Academic Editor: Rickson C. Mesquita

Copyright © 2016 Zhihong Xu et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

Abstract.

Reflectance spectroscopy is a low-cost, nondestructive, and noninvasive method for detection of neoplastic lesions of mucosal tissue. This study aims to evaluate the capability of reflectance spectroscopy system under white light (400–700 nm) with a multivariate statistical analysis for distinguishing nasopharyngeal carcinoma (NPC) from nasopharyngeal benign ex vivo tissues. High quality reflectance spectra were acquired from nasopharyngeal ex vivo tissues belonging to 18 noncancerous and 19 cancerous subjects, and the combination of principal component analysis-linear discriminant analysis (PCA-LDA) along with leave-one-spectrum-out cross-validation (LOOCV) diagnostic algorithm was subsequently employed to classify different types of tissue group, achieving a diagnostic sensitivity of 73.7% and a specificity of 72.2%. Furthermore, in order to distinguish NPC from nasopharyngeal benign ex vivo tissues based on reflectance spectra simply, spectral intensity ratios of oxyhemoglobin (540/576) were used as an indicator of the carcinogenesis associated transformation in the hemoglobin oxygenation. This tentative work demonstrated the potential of reflectance spectroscopy for NPC detection using ex vivo tissue and has significant experimental and clinical value for further in vivo NPC detection in the future.

1. Introduction

Nasopharyngeal carcinoma (NPC) is a prominent malignant tumor that occurs in the top and the side wall of nasopharyngeal cavity with no obvious early symptom, which directly causes harm to human health [1]. The incidence of NPC has remained high in southeast Asia, northern Africa, parts of the Middle East, and so forth. We should be concerned about the fact that approximately 80% of NPC cases were reported from China, especially in southern provinces, such as Guangdong and Fujian [2–4]. NPC was defined as a special disease with a remarkable racial and geographical distribution worldwide [5]. According to the epidemiological investigation, NPC has various factors to its etiology, and the main relevant risk factors include inherited genetic factors, the dormant infection of Epstein-Barr virus (EBV), and natural environment factor [6]. The clinical reports indicated that NPC patients treated in Stages I and Stages II have 5-year survival rates of approximately 90% and 75%, respectively. However, early stage of NPC has no obvious symptoms until the middle and advanced stages of disease [2, 6, 7]. Therefore, early diagnosis is the effective way to improve the survival rates of patient and promote the quality of their life.

Currently common detection and screening method for NPC is dependent on clinicians’ experience, which easily causes possible diagnostic errors owing to lack of the biochemical information of suspicious lesions. Although the histopathological examination of biopsy is still the golden standard for NPC detection [8], it involves the destruction and invasion of tissue from suspected lesions. Diagnostic techniques based on optical spectroscopy, in contrast, can provide quantitative information in real time from tissues, which are excellent in the field of precancerous tumor localization and pathology analysis.

Reflectance spectroscopy has attracted wide attention in human tissue study including oral, breast, lung, bladder, cervical, esophagus, kidney, and prostate tissue [9–14], due to its simple operation and low investment cost of equipment. In reflectance, broadband beam interaction with the biological tissue surface and two forms of cases can occur, that is, multiple elastic scattering and absorption phenomenon [15]. In this process, part of broadband beam returns as reflectance spectral signals, which are able to provide the optical properties of the tissue [16]. The reflectance spectral signatures are dependent on biological tissue morphology, surface characteristics, and structural components [13, 15]. Hence, reflectance spectra have certain advantages for explaining major tissue features within early cancerous transform, for example, the cellular metabolism, capillary distribution, and hemoglobin oxygenation [15, 17–19].

Zonios et al. reported a diffuse reflectance spectra study on human adenomatous colon polyps, showing that differences can be ascribed to hemoglobin concentration and effective scatterer size between normal and adenomatous tissue sites [15]. Utzinger and coworkers have investigated an exploratory pilot study to measure reflectance spectra at different source-detector separations from normal and neoplastic ovarian tissue and found the changes of intensities of the reflectance spectrum between 550 and 580 nm, which are dramatically attributed to the absorption of blood oxygenation [20]. Subhash and coworkers have characterized reflectance spectral intensity ratio of oxygenated hemoglobin absorption bands (540/570) associated with different types of lesions and histopathological conditions, and results showed that the 540/570 ratio for malignant lesions was always lower than that for normal areas [13]. Li et al. used the reflectance ratio 540/575 method to distinguish malignant from nonmalignant sites in human ex vivo gastric epithelial tissues and found some pronounced differences in various pathological types [21]. Zhao et al. reported a study about oxygenated hemoglobin absorption band ratios of esophageal epithelial tissues at different heat treatment temperature, showing the capability of employing oxygenated hemoglobin band ratios 540/570 as indicators of malignancy [22]. Recent results have shown that tissue absorption in the short wavelength region and visible region is assigned to hemoglobin, with the oxygenation forms exhibiting different absorption spectral characteristic [23]. These reports demonstrated that reflectance spectroscopy could obtain spectral information about the pathological changes of tissue, which can serve as important supplement information for clinical diagnosis. Nevertheless, as far as we are concerned, still few reports have been found using reflectance spectroscopy for NPC detection.

In this preliminary study, we investigated reflectance spectral signals from human nasopharyngeal ex vivo tissue with different pathological types. Reflectance spectra were subjected to multivariate statistical tools, including principal component analysis-linear discriminant analysis (PCA-LDA) algorithm together with the leave-one-spectrum-out cross-validation (LOOCV). To validate the feasibility of reflectance spectroscopy as a tool to differentiate noncancerous ex vivo tissues from NPC, the receiver operating characteristic (ROC) curve was employed. Empirical method using reflectance spectral intensity ratios of oxyhemoglobin bands (540/576) was used to distinguish cancer from benign ex vivo tissue.

2. Materials and Methods

2.1. Patients and Sample Preparations

Nasopharyngeal ex vivo tissues for this study were collected from 37 patients who were recruited for biopsy from nasopharyngeal cavity lesions with written informed consent. This procedure was authorized by the Ethics Committee of Affiliated Fuzhou First Hospital of Fujian Medical University. All nasopharyngeal tissues were divided into two groups with clinical diagnosis (19 NPC samples and 18 benign tissue samples), as shown in Table 1. The pathological types of NPC patients contain 17 undifferentiated nonkeratinizing carcinomas and 2 differentiated nonkeratinizing carcinomas. The pathological types of benign tissues consist of 15 chronic inflammations, 2 lymphomas, and 1 lymphoid tissues hyperplasia. These fresh tissue samples were immediately stored at 4°C until being sent to the laboratory moments later, where they were stored in a minus 80°C refrigerator until experimental measurements. These targets were placed on an aluminum plate and thawed at room temperature immediately before spectral measurement (Figure 1).

Table 1: Sample characteristic.

Sample type
Pathologic types
Number

Nasopharyngeal cancerous (NPC) patients ()
Undifferentiated
nonkeratinizing carcinomas
17

Differentiated
nonkeratinizing carcinomas
2

Nasopharyngeal noncancerous patients ()
Chronic inflammations
15

Lymphoma
2

Lymphoid tissues hyperplasia
1

Figure 1: Nasopharyngeal ex vivo tissue placed on the pure aluminum plate. Pure aluminum and sample are pointed out by the black arrow.

2.2. Laboratory Instruments

All the reflectance spectra of nasopharyngeal ex vivo tissues were recorded using a special measurement system (Verisante Technology, Inc., Vancouver, Canada); more details were presented in previous report [24]. In brief, this detection system included a light source (a xenon arc lamp), a spectrometer (USB2000, Ocean Optics), and a PC. A band-pass filter (400–700 nm) was installed at the reflectance spectrograph entrance to obstruct the reflected unwanted light to improve reflectance signals detection. The exposure time of the spectrometer is 200 ms. An optical fiber is used to collect reflectance spectra, of which the entrance corresponds to a spot size of 1 mm diameter on the sample. The dimensions of the samples are generally between 1.8 and 2.5 mm. Three reflectance spectra were obtained from different positions for each ex vivo tissue and averaged to a mean spectrum so that we could acquire representative spectra for each sample. This detection system was calibrated by a certified reflectance standard (WS-1-SL, Labsphere) before each measurement to eliminate interference by the status of the system.

2.3. Data Processing and Analysis

To eliminate the interference of slow-moving backgrounds, the raw spectra were smoothed by using Savitzky-Golay method with points of window of 50 and polynomial order of 5. Reflectance spectra in the 400–700 nm range were subjected to principal component analysis (PCA) and linear discriminant analysis (LDA) combined with LOOCV by using SPSS statistical software (SPSS Inc., Chicago). PCA-LDA was employed to achieve a diagnostic algorithm for classification between different groups within the dataset. Briefly, PCA is a statistical method that is used on the dataset to reduce the dimensions of the data, while finding a series of the linear combination of the original variables, so-called principal components (PCs) [25, 26]. An independent-samples -test was used to determine significant PCs ( value < 0.05) [25, 27]. The findings, which represent most of the significant variance in the dataset, are selected as input data for the development of LDA model for tissue classification [28]. LDA is a supervision algorithm that maximizes the covariance in the data between different groups and minimizes the variances within group [27]. The classification efficiency of the PCA-LDA diagnostic models for correctly predicting the sample groups (i.e., normal versus malignancy) was evaluated by LOOCV method so that to eliminate the risk of overfitting [29, 30]. The receiver operating characteristic (ROC) curves were plotted to evaluate the performance of the above algorithm [26].

3. Results

3.1. Analysis of Reflectance Spectra

Figure 2 shows the comparison of mean ex vivo reflectance spectra with standard deviations (SD) of the biopsy lesions from malignant sites and benign mucosa sites with the same detection system setup. Prominent valleys of reflectance spectra were observed in 412 nm, 540 nm, and 576 nm which can be attributed to absorption from hemoglobin bands [31, 32]. For all the spectra measured, the absorption peak intensity from NPC group was slightly weaker than that of noncancerous group in these absorption valleys. Notable absorption variations in reflectance intensity are more significant at green light region (around 540 nm and 576 nm bands), indicating an increase of microvasculature and hemoglobin [12, 13]. Significant differences were found in the spectral valleys at 540 nm ( value = 0.006 < 0.05) and 576 nm ( value = 1.6 × 10⁻⁴ < 0.05) between the two groups by using the analysis of -test.

Figure 2: Mean reflectance spectra from NPC (red line, ) and noncancerous ex vivo tissue (blue line,  ). Asterisks indicated significant difference ( value < 0.05) after -test, in which the values are 0.006 and 1.6 × 10⁻⁴ for 540 nm and 576 nm, respectively. Areas (shaded) illustrate the respective standard deviations of the means spectra.

3.2. Multivariate Statistical Analysis

Multivariate analyses including PCA and LDA were explored to evaluate the feasibility of classifying cancerous samples from noncancerous ones by employing the recorded reflectance dataset. Figure 3 shows this multivariate statistical analysis in a flowchart. In the first step the dataset underwent dimension reduction using unsupervised PCA to extract the principal components (PCs) which represent the measured spectra by a weighted linear combination. The PCs were tested by independent-samples -test. PC3 and PC6 have the significant variance in the dataset ( value < 0.05) and were selected to discriminate the two pathological groups, in which the values of are 0.022 and 0.007 for PC3 and PC6, respectively. The selected PCs (PC3 and PC6) were subsequently imported into LDA model with LOOCV method to evaluate classification efficiency.

Figure 3: The work flowchart of multivariate statistical analysis.

The scatter plot of the diagnostically significant PC3 versus PC6 scores is displayed in Figure 4. Also displayed is the diagnosis equation (0.738 PC3 + 0.836 PC6 + 0.017 = 0) that separates the two groups of ex vivo tissue according to the result of statistical analysis. The diagnosis equation is defined as , where and are the standardized canonical discriminant function coefficients of PC3 and PC6 (0.738 and 0.836, resp.). This decision line is controlled by choosing the coordinate parameters of and . Figure 5 shows the posterior probability generated from PCA-LDA algorithm belonging to noncancerous and NPC subjects, with a diagnostic sensitivity of 73.7% (14/19) and a specificity of 72.2% (13/18). The dotted lines were set to 0.5, where the total number of misclassified tissues was minimized [33], which represented the sensitivities and specificities for separating NPC from nasopharyngeal noncancerous ex vivo tissue. The sensitivity and specificity were defined as [34]

Figure 4: A two-dimension scattered point diagram of the PCA result for NPC group (red triangles ▲) and noncancerous group (blue circle ●). The dotted lines (0.738 PC3 + 0.836 PC6 + 0.017 = 0) served as a diagnostic method for NPC sample and noncancerous sample differentiation.

Figure 5: Scatter plots of the posterior probability belonging to NPC and nasopharyngeal noncancerous subjects (▲—NPC group; ●—noncancerous group), generated from PCA-LDA diagnosis algorithms along with LOOCV method. Error bars represent the mean and standard deviation of NPC (magenta) and nasopharyngeal noncancerous (green) ex vivo tissue posterior probability. The dotted lines represented the sensitivities and specificities for separating NPC from nasopharyngeal noncancerous tissue.

Figure 6 is the ROC curve, which was generated from the posterior probability plot at different threshold levels to evaluate the proposed diagnostic algorithm. The integration area under ROC curve (AUC) is 0.833, which further demonstrates that using reflectance spectra with multivariate statistical algorithms may be viable for discriminating NPC group from benign ex vivo tissue group.

Figure 6: ROC curve for discriminating NPC from nasopharyngeal noncancerous ex vivo tissue for the training dataset by employing PCA-LDA diagnosis algorithms combined with LOOCV method. The AUC is 0.833.

3.3. Analysis of Hemoglobin Absorption Ratio 540/576

Significant differences were found in the spectral valleys at 540 nm ( value = 0.006 < 0.05) and 576 nm ( value = 1.6 × 10⁻⁴ < 0.05) between the two groups by using the analysis of -test. The ratios of intensity of 540 nm and 576 nm were calculated for each original reflectance spectrum. Then all the ratios were examined by -test, and the value is 1.89 × 10⁻⁷. The ratios of 540/576 were calculated by the related original data, and then these original ratios were averaged. Finally, the mean ratios were used to draw the boxplots (Figure 7). Figure 7 shows the comparison of intensities and standard deviations of the mean 540/576 ratios between nasopharyngeal noncancerous (blue box) and NPC (red box) ex vivo tissues. In the case of noncancerous tissue, the mean 540/576 ratio was found to be higher than that of nasopharyngeal malignancy group (1.06 versus 1.02).

Figure 7: Mean reflectance ratios (540/576) of NPC (red box) versus those of noncancerous group (blue box).

4. Discussion

In this work, we investigated reflectance spectra obtained from benign and malignant ex vivo tissues in the human nasopharynx. The results of this tentative study show that there were some differences in reflectance spectra of ex vivo tissue between NPC and noncancerous subjects, demonstrating potential of reflectance spectroscopy for NPC detection and preliminary precancer screening. Our spectral measurement (Figure 2) shows that reflectance intensity curves of both ex vivo benign and malignant mucosa tissues are similar at 400–700 nm. However, the differences of reflectance intensity exist in curve valleys at around 412, 540, and 576 nm region owing to the absorption from oxyhemoglobin bands [31, 32]. These spectral absorption valleys are much more significant in the malignant tumor spectrum at 500–600 nm bands. The spectral absorption characteristics at 540 nm and 576 nm in ex vivo nasopharyngeal mucosa may arise from the absorption from oxyhemoglobin associated with high demand of oxygen by cancerous tissues, leading to a decrease in the content of oxygen levels in the blood [13, 23]. Therefore, the spectral absorption of cancerous group is weaker than that of noncancerous group at 540 nm and 576 nm (Figure 2). In the case of malignant lesion tissues, reflectance spectral intensities at 412 nm, 540 nm, and 576 nm bands were slightly stronger than that of noncancerous tissues. One possible explanation for the special spectral intensity differences is that cancerous tissues are often characterized by abnormally histological structure and microvascular volume [12, 13, 15]. In addition, collagen is more densely distributed in the submucosa [15], which is usually able to affect reflectance property of tissue surface.

These preliminary results may indicate the carcinogenesis process of nasopharyngeal carcinoma, since the cancer cells’ growth and proliferation relate intimately on the capillary proliferation and nutrient requirements, which caused the metabolic disorders during this period. The tissular capillary will continue to spread, which may lead to the anomalies of histological structure. Furthermore, the cancerous tissue clonal expansion and evolution cause a significant decrease in oxygenated hemoglobin in tumor tissue during the long process of malignant cancer [13, 15, 35]. So, malignant tissues are of low blood-oxygen content owing to metabolic disorders [15]; that is, oxygenated hemoglobin content will be greatly reduced in cancerous tissues.

To further evaluate the capability of reflectance spectroscopy for differentiating nasopharyngeal malignant tumor from benign ex vivo tissue, multivariate statistical analysis (PCA-LDA) was performed on the spectral analysis. PCA-LDA yields a sensitivity of 73.7% (14/19) and a specificity of 72.2% (13/18), which demonstrates the availability of the diagnostic algorithm for distinguishing NPC from noncancerous tissue samples by using reflectance spectra.

At the same time, we used a simple empirical method; that is, reflectance spectral intensity ratios (540/576) were used as an indicator of neoplastic associated changes in the hemoglobin oxygenation. The mean intensity of 540/576 ratio of oxyhemoglobin for malignant and benign tissue was 1.02 and 1.06, respectively. This computation coincides with the results of Subhash et al. [13]. The mean reflectance spectral intensity ratio (540/576) of oxyhemoglobin was weaker for NPC compared to noncancerous ex vivo tissue, which is probably for the reason that malignant tissues are much hungry for oxygen.

Finally, we have compared the diagnostic result of the 540/576 method with that of the PCA-LDA algorithm. The 540/576 method generated a sensitivity of 78.9% and specificity of 69.8% for differentiation between ex vivo malignant and noncancerous tissues. The PCA-LDA diagnostic algorithm has a sensitivity of 73.7% and a specificity of 72.2%. As PCA-LDA diagnostic algorithm used entire ex vivo range of reflectance spectra for classification by the algorithm, the PCA-LDA method gives a more comprehensive analysis for the dataset. Therefore, the diagnostic specificity of the PCA-LDA algorithm is slightly better than that of the 540/576 method. So, the 540/576 method is used as a kind of simple empirical methods. The discrimination results of PCA-LDA are more comprehensive and reliable. Therefore, reflectance spectroscopy could be one of the screening methods for nasopharyngeal carcinoma detection.

It is worth mentioning that ex vivo reflectance spectral intensities and absorption bands intensities of oxyhemoglobin in both normal and malignant tissues decreased over time [13]. Subhash et al. observed that there are some spectral differences between ex vivo and in vivo normal tissues [13]. To be a more effective diagnosis method, the features of in vivo nasopharyngeal carcinoma from that of noncancerous one need further investigation with noninvasive in vivo detection in the future.

5. Conclusions

In summary, we studied the characteristics of reflectance spectra from benign and malignant ex vivo tissues in the human nasopharynx for the first time. Mean reflectance spectra of benign and malignant ex vivo tissue showed significant differences in the absorption bands from oxyhemoglobin bands. Using PCA-LDA diagnostic algorithms, we got a sensitivity of 73.7% and a specificity of 72.2% for NPC detection, indicating the potential of reflectance spectroscopy for ex vivo diagnosis of nasopharyngeal malignant tumor. In order to differentiate NPC from noncancerous ex vivo tissue based on reflectance spectra simply, mean spectral intensity ratios of oxyhemoglobin (540/576) were employed as an indicator of cancerization associated transformation in the hemoglobin oxygenation. These results demonstrated that reflectance spectroscopy combined with multivariate statistical tools can be used to indicate the inherent information like blood hemoglobin content, which has the potential to detect nasopharyngeal cancerous tissues quickly in real time, presenting a significant experimental and clinical value for further nasopharyngeal cancer detection in vivo in the future.

Competing Interests

The authors declare that they have no competing interests.

Acknowledgments

This work was supported by the Program for Changjiang Scholars and Innovative Research Team in University (no. IRT15R10) and the National Natural Science Foundation of China (nos. 61210016, 61178090, and 61405036) and the Project of Science Foundation of Ministry of Health and United Fujian Provincial Health and Education Project for Tackling the Key Research (no. WKJ-FJ-01). The authors thank Professor Haishan Zeng (British Columbia Cancer Research Centre, Vancouver, BC, Canada) for technical support and helpful advice and thank Verisante Technology, Inc. (Vancouver, Canada), for instrumental support.

References

J. H. C. Ho, “An epidemiologic and clinical study of nasopharyngeal carcinoma,” International Journal of Radiation Oncology, Biology, Physics, vol. 4, no. 3-4, pp. 183–198, 1978.

T. Liu, “Issues in the management of nasopharyngeal carcinoma,” Critical Reviews in Oncology/Hematology, vol. 31, no. 1, pp. 55–69, 1999.

K. W. Lo, K. F. To, and D. P. Huang, “Focus on nasopharyngeal carcinoma,” Cancer Cell, vol. 5, no. 5, pp. 423–428, 2004.

W. I. Wei and J. S. T. Sham, “Nasopharyngeal carcinoma,” The Lancet, vol. 365, no. 9476, pp. 2041–2054, 2005.

W.-H. Jia, Q.-H. Huang, J. Liao et al., “Trends in incidence and mortality of nasopharyngeal carcinoma over a 20–25 year period (1978/1983–2002) in Sihui and Cangwu counties in southern China,” BMC Cancer, vol. 6, article 178, 2006.

D. T. T. Chua, J. S. T. Sham, D. L. W. Kwong, D. T. K. Choy, G. K. H. Au, and P. M. Wu, “Prognostic value of paranasopharyngeal extension of nasopharyngeal carcinoma: a significant factor in local control and distant metastasis,” Cancer, vol. 78, no. 2, pp. 202–210, 1996.

L. Xu, J. Pan, J. Wu et al., “Factors associated with overall survival in 1706 patients with nasopharyngeal carcinoma: significance of intensive neoadjuvant chemotherapy and radiation break,” Radiotherapy and Oncology, vol. 96, no. 1, pp. 94–99, 2010.

N. Stone, C. Kendall, N. Shepherd, P. Crow, and H. Barr, “Near-infrared Raman spectroscopy for the classification of epithelial pre-cancers and cancers,” Journal of Raman Spectroscopy, vol. 33, no. 7, pp. 564–573, 2002.

F. Koenig, R. Larne, H. Enquist et al., “Spectroscopic measurement of diffuse reflectance for enhanced detection of bladder carcinoma,” Urology, vol. 51, no. 2, pp. 342–345, 1998.

I. Georgakoudi, B. C. Jacobson, J. Van Dam et al., “Fluorescence, reflectance, and light-scattering spectroscopy for evaluating dysplasia in patients with Barrett's esophagus,” Gastroenterology, vol. 120, no. 7, pp. 1620–1629, 2001.

S. K. Chang, Y. N. Mirabal, E. N. Atkinson et al., “Combined reflectance and fluorescence spectroscopy for in vivo detection of cervical pre-cancer,” Journal of Biomedical Optics, vol. 10, no. 2, Article ID 024031, 2005.

Y. S. Fawzy, M. Petek, M. Tercelj, and H. Zeng, “In vivo assessment and evaluation of lung tissue morphologic and physiological changes from non-contact endoscopic reflectance spectroscopy for improving lung cancer detection,” Journal of Biomedical Optics, vol. 11, no. 4, Article ID 044003, 2006.

N. Subhash, J. R. Mallia, S. S. Thomas, A. Mathews, P. Sebastian, and J. Madhavan, “Oral cancer detection using diffuse reflectance spectral ratio R540/R575 of oxygenated hemoglobin bands,” Journal of Biomedical Optics, vol. 11, no. 1, Article ID 014018, 2006.

Z. Volynskaya, A. S. Haka, K. L. Bechtel et al., “Diagnosing breast cancer using diffuse reflectance spectroscopy and intrinsic fluorescence spectroscopy,” Journal of Biomedical Optics, vol. 13, no. 2, Article ID 024012, 2008.

G. Zonios, L. T. Perelman, V. Backman et al., “Diffuse reflectance spectroscopy of human adenomatous colon polyps in vivo,” Applied Optics, vol. 38, no. 31, pp. 6628–6637, 1999.

H. Zeng, A. McWilliams, and S. Lam, “Optical spectroscopy and imaging for early lung cancer detection: a review,” Photodiagnosis and Photodynamic Therapy, vol. 1, no. 2, pp. 111–122, 2004.

R. K. Jain, “Determinants of tumor blood flow: a review,” Cancer Research, vol. 48, no. 10, pp. 2641–2658, 1988.

L. T. Perelman, V. Backman, M. Wallace et al., “Observation of periodic fine structure in reflectance from biological tissue: a new technique for measuring nuclear size distribution,” Physical Review Letters, vol. 80, article 627, 1998.

K. Bensalah, D. Peswani, A. Tuncel et al., “Optical reflectance spectroscopy to differentiate benign from malignant renal tumors at surgery,” Urology, vol. 73, no. 1, pp. 178–181, 2009.

U. Utzinger, M. Brewer, E. Silva et al., “Reflectance spectroscopy for in vivo characterization of ovarian tissue,” Lasers in Surgery and Medicine, vol. 28, no. 1, pp. 56–66, 2001.

L. Q. Li, H. J. Wei, Z. Y. Guo et al., “Oxygenated hemoglobin diffuse reflectance ratio for in vitro detection of human gastric pre-cancer,” Laser Physics, vol. 20, no. 7, pp. 1667–1672, 2010.

Q. L. Zhao, Z. Y. Guo, J. L. Si et al., “Heat treatment of human esophageal tissues: effect on esophageal cancer detection using oxygenated hemoglobin diffuse reflectance ratio,” Laser Physics, vol. 21, no. 3, pp. 559–565, 2011.

S.-H. Tseng, C.-K. Hsu, J. Y.-Y. Lee, S.-Y. Tzeng, W.-R. Chen, and Y.-K. Liaw, “Noninvasive evaluation of collagen and hemoglobin contents and scattering property of in vivo keloid scars and normal skin using diffuse reflectance spectroscopy: pilot study,” Journal of Biomedical Optics, vol. 17, no. 7, Article ID 077005, 2012.

H. Zeng, M. Petek, M. T. Zorman, A. McWilliams, B. Palcic, and S. Lam, “Integrated endoscopy system for simultaneous imaging and spectroscopy for early lung cancer detection,” Optics Letters, vol. 29, no. 6, pp. 587–589, 2004.

R. Shaikh, T. K. Dora, S. Chopra et al., “In vivo Raman spectroscopy of human uterine cervix: exploring the utility of vagina as an internal control,” Journal of Biomedical Optics, vol. 19, no. 8, Article ID 087001, 2014.

S. Duraipandian, W. Zheng, J. Ng, J. J. H. Low, A. Ilancheran, and Z. Huang, “Simultaneous fingerprint and high-wavenumber confocal Raman spectroscopy enhances early detection of cervical precancer in vivo,” Analytical Chemistry, vol. 84, no. 14, pp. 5913–5919, 2012.

J. Mo, W. Zheng, J. J. H. Low, J. Ng, A. Ilancheran, and Z. Huang, “High wavenumber raman spectroscopy for in vivo detection of cervical dysplasia,” Analytical Chemistry, vol. 81, no. 21, pp. 8908–8915, 2009.

Z. Huang, M. S. Bergholt, W. Zheng et al., “In vivo early diagnosis of gastric dysplasia using narrow-band image-guided Raman endoscopy,” Journal of Biomedical Optics, vol. 15, no. 3, Article ID 037017, 2010.

D. Lin, G. Chen, S. Feng et al., “Development of a rapid macro-Raman spectroscopy system for nasopharyngeal cancer detection based on surface-enhanced Raman spectroscopy,” Applied Physics Letters, vol. 106, no. 1, Article ID 013701, 2015.

M. Jermyn, K. Mok, J. Mercier et al., “Intraoperative brain cancer detection with Raman spectroscopy in humans,” Science Translational Medicine, vol. 7, no. 274, Article ID 274ra19, 2015.

P. Chikezie, A. Akuwudike, and F. Chilaka, “Absorption spectra of normal adults and patients with sickle cell anaemia haemoglobins treated with hydrogen peroxide at two pH values,” Iranian Journal of Blood & Cancer, vol. 5, no. 4, pp. 129–135, 2013.

W. G. Zijlstra and A. Buursma, “Spectrophotometry of hemoglobin: absorption spectra of bovine oxyhemoglobin, deoxyhemoglobin, carboxyhemoglobin, and methemoglobin,” Comparative Biochemistry and Physiology Part B: Biochemistry and Molecular Biology, vol. 118, no. 4, pp. 743–749, 1997.

J. Y. Qu, H. Chang, and S. Xiong, “Fluorescence spectral imaging for characterization of tissue based on multivariate statistical analysis,” Journal of the Optical Society of America A, vol. 19, no. 9, pp. 1823–1831, 2002.

J. Y. Qu, P. Wing, Z. Huang et al., “Preliminary study of in vivo autofluorescence of nasopharyngeal carcinoma and normal tissue,” Lasers in Surgery and Medicine, vol. 26, no. 5, pp. 432–440, 2000.

R. Mallia, S. S. Thomas, A. Mathews et al., “Oxygenated hemoglobin diffuse reflectance ratio for in vivo detection of oral pre-cancer,” Journal of Biomedical Optics, vol. 13, no. 4, Article ID 041306, 2008.