Journal of Spectroscopy
Volume 2018 (2018), Article ID 9765806, 5 pages
https://doi.org/10.1155/2018/9765806

Research Article

Photoacoustic Measurement of Ethane with Near-Infrared DFB Diode Laser

Gang Cheng,1,2,3 Yuan Cao,1,2 Kun Liu,¹ Gongdong Zhu,¹ Guishi Wang,¹ and Xiaoming Gao¹

¹Anhui Institute of Optics and Fine Mechanics, Chinese Academy of Sciences, Hefei 230031, China
²University of Science and Technology of China, Hefei 230031, China
³Anhui University of Science and Technology, Huainan 232001, China

Correspondence should be addressed to Kun Liu; liukun@aiofm.ac.cn

Received 27 February 2018; Accepted 17 April 2018; Published 9 May 2018

Academic Editor: Angelo Sampaolo

Copyright © 2018 Gang Cheng et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

Abstract.

A compact resonant photoacoustic sensor based on a near-infrared distributed-feedback (DFB) diode laser was developed for detection of ethane (C₂H₆). A DFB laser emission at 5937.25 cm⁻¹ with a power of ∼5 mW was used as an excitation light source for generating the photoacoustic signal. Wavelength modulation and second harmonic detection scheme were employed. Modulation frequency and modulation amplitude were optimized for getting optimum performance. Performance evaluation based on the linearity response of the PAS sensor system with respect to C₂H₆ concentration levels was performed, and a good linear dependence of the PAS signal on the C₂H₆ concentration was obtained. A minimum detectable concentration of 9 ppmv was achieved for detection of C₂H₆ with a lock-in time constant of 10 ms.

1. Introduction

Ethane (C₂H₆) is the second largest component among natural gases; therefore, detection of C₂H₆ was an effective method for discrimination of CH₄ leakage between natural gas pipe line and biogenic sources of landfills or wetlands. Detection of C₂H₆ is also important in environmental or atmosphere monitoring as C₂H₆ strongly affects both atmosphere chemistry and climate as a most abundant nonmethane hydrocarbon in the atmosphere [1]. In addition, C₂H₆ detection has found applications in breath analysis as a noninvasive method to monitor and identify diseases, such as lung cancer and asthma [2]. Therefore, developing the C₂H₆ sensor has a strong need in industrial, environmental, or medical applications.

Laser spectroscopy-based gas sensors is an effective tool for real time, selective, and sensitive detection of trace gases, which has a wide range of potential applications in environmental monitoring, medical diagnostics, and industrial process control [3, 4]. Photoacoustic spectroscopy, based on the photoacoustic effect which was first discovered by Alexander Graham Bell in 1880 [5], is a sensitive, selective, and well-established method for sensing trace gases that has been successfully employed in numerous applications [6, 7]. The PAS technique offers several practically important advantages, such as high trace gas detection sensitivity with a small sensing module and wide dynamic range and no need for an optical detector such as a photodiode. Unlike spectroscopic techniques, such as integrated cavity output spectroscopy (ICOS) [8], cavity ring down spectroscopy [9], or absorption spectroscopy with multipass cell [10], the PAS is a “zero base line” method. In addition, the amplitude of the photoacoustic signal depends on the power of the exciting light beam. Therefore, the PAS can benefit from using continuous wave infrared laser sources capable of emitting high optical power [11]. To date, there are several PAS techniques including traditional resonant PAS with wide-band microphones [12], cantilever-based PAS [13], and quartz-enhanced PAS (QEPAS) [14].

In the present work, we report on our recent efforts in the development of a compact PAS sensor for detection of C₂H₆ by using a distributed-feedback (DFB) diode laser operating at 5937.2 cm⁻¹ with a power of ∼5 mW and an excitation light source to generate the photoacoustic signal. The near-infrared DFB diode lasers that are commercially available offer a robust, relatively inexpensive, and attractive alternative. This low-cost, compact, and sensitive PAS sensor of C₂H₆ is very suitable for various field applications, in particular, for discrimination of CH₄ leakage between natural gas pipe line and biogenic sources of landfills or wetlands [15].

2. C₂H₆ PAS Sensor Configuration

A schematic diagram of the developed C₂H₆ PAS sensor platform is shown in Figure 1(a). A PAS cell built out of aluminum with a central cylinder tube of radius r = 5 mm and length of L = 100 mm acting as an acoustic resonator was designed. A buffer volume with a radius of 25 mm and length of 50 mm was set at both sides of the cylinder resonator, and the total length of the PAS cell was 200 mm (Figure 1(b)). C₂H₆ shows an unresolvable vibration-rotation absorption band near 1.68 μm [16, 17]. Therefore, a fiber-coupled distributed-feedback (DFB) diode laser (NLK1E5GAAA, NEL) operating at 1.684 μm was used as the excitation light source for generating the photoacoustic signal. The laser current and temperature were controlled by a commercial diode laser controller (ILX Lightwave LDC-3724). Coarse and fine wavelength tuning was performed by changing the laser temperature and current, respectively. Wavelength modulation and second harmonic detection are used in this work. A dual channel function generator was used to provide a voltage ramp for scanning the laser wavelength and a sine wave for modulating the laser wavelength. The voltage ramp and the sine wave were combined with a home-made adder before being fed to the laser driver. The laser current was the sine wave modulated at half the resonant frequency f₀ of the PAS cell. The laser beam was collimated with a fiber-coupled collimator (f ∼4.8 mm) and subsequently focused into the PAS cell by using a lens L with a 30 mm focal length. A preamplifier (EG&G, Model 5113) with a gain of 25 and 6 dB bandpass filter (300 Hz–30 kHz) was used for signal amplification prior to demodulation at f₀ by a lock-in amplifier (Stanford Research Systems, Model SR 830 DSP). The time constant of the lock-in amplifier was set at 10 ms in combination with an 18 dB/octave slope filter (leading to a detection bandwidth of Δf = 9.375 Hz). The demodulated signal was subsequently digitalized with a DAQ card (NI-USB-6212) and displayed on a laptop via a LabVIEW interface.

(a)

(b)

Figure 1: (a) Schematic diagram of the developed C₂H₆ PAS experimental setup and (b) geometric dimension of the PAS cell.

3. Results and Discussion

Direct absorption spectroscopy of C₂H₆ was first measured for determining absorption peak. Figure 2 shows a 3000 ppm C₂H₆ absorption spectroscopy that was measured by using a compact multipass absorption cell with an optical path length of 26.4 m [10] at normal atmosphere pressure. The absorption peak was found to be at 5937.21 cm⁻¹ and the corresponding laser temperature was 31°C. Therefore, the laser temperature was controlled at 31°C during following measurement.

Figure 2: Direct absorption spectroscopy of C₂H₆.

Characteristics of the acoustic resonator were experimentally investigated to determine the specific resonant frequency of the resonator in an atmospheric environment. Measurements of the C₂H₆ PAS signal with constant concentration of 5000 ppm at different modulation frequencies were performed for this study. Figure 3 shows the square of the signal amplitude as a function of frequency. The data were fitted to a Lorentz contour, which describes power in a classical driven oscillator as a function of frequency. From the results shown in Figure 3, optimum resonant frequencies f₁₀ = 1580 Hz were found for the resonator. This value corresponds to the first longitudinal resonant mode of the resonator [18]. Therefore, the laser wavelength was modulated at a frequency of 790 Hz.

Figure 3: Square of the PAS signals amplitude as a function of frequency. The data are fitted with a Lorentz profile.

The C₂H₆ PAS sensor system was designed to operate in a wavelength modulation mode, which can effectively suppress the background noise. However, wavelength modulation of the PAS signal is dependent on modulation amplitude. Therefore, the laser frequency modulation amplitude was also to be optimized to get an optimum PAS signal. Figure 4 shows the measured PAS signal as a function of the modulation amplitude. The measurements were carried out under normal atmospheric conditions. As can be seen in Figure 4, the optimum modulation amplitude of the sine wave for PAS detection of C₂H₆ was found to be 0.25 V which was then used in the following measurements.

Figure 4: C₂H₆ PAS signals as a function of modulation amplitude.

A performance evaluation based on the linearity response with optimized parameters of the C₂H₆ PAS sensor system with respect to C₂H₆ concentration levels was performed. A calibrated standard gas mixture consisting of 10,000 ppm C₂H₆:N₂ was employed and diluted with pure N₂ by using a commercial gas dilution system (Environics, S4000). As expected, a linear dependence of the PAS signal on the C₂H₆ concentration was found. The measured PAS signals and the fitted straight line are shown in Figure 5.

Figure 5: Linear response of the PAS signals to C₂H₆ concentration.

As an example, Figure 6 depicts the PAS 2f-signals of C₂H₆ with a concentration of 100 ppmv, 1000 ppmv, and 5000 ppmv, respectively. The corresponding signal (the peak value at the dotted line) was 0.12 mV, 0.83 mV, and 3.96 mV, respectively. For evaluation of the minimum detection limit of the C₂H₆ PAS sensor, the PAS signal in air, as the noise level, was measured. Figure 7 shows the PAS noise obtained in air and the 100 ppmv C₂H₆ PAS signal for comparison. The noise level (1σ) defined as standard deviation was found to be 0.028 mV. Taking into account the PAS signal of 0.33 mV obtained with 100 ppmv C₂H₆, the signal to noise ratio was then found to be 11. Therefore, the minimum detection limit was estimated to be 9 ppmv for C₂H₆ detection.

Figure 6: C₂H₆ PAS signals obtained with 100 ppmv, 1000 ppmv, and 5000 ppmv C₂H₆ samples, respectively. The dotted line position corresponds to the absorption peak of Figure 2.

Figure 7: PAS signal of 100 ppmv C₂H₆ and noise obtained in air.

4. Conclusion

A PAS sensor for detection of C₂H₆ was developed. Using wavelength modulation and second harmonic detection scheme, combined with optimization of modulation frequency and modulation amplitude, good performance was achieved for detection of C₂H₆. Good linear dependence of the PAS signal on the C₂H₆ concentration was obtained. A minimum detectable concentration of 9 ppmv was achieved for detection of C₂H₆ with a lock-in time constant of 10 ms and optical power of 5 mW. The developed C₂H₆ PAS sensor could be applied in monitoring the natural gas pipe line leakage or industrial process control.

Data Availability

The data used to support the findings of this study are available from the corresponding author upon request.

Conflicts of Interest

The authors declare that they have no conflicts of interest.

Acknowledgments

This work was financially supported by the National Natural Science Foundation of China (nos. 41475023, 41730103, and 41575030) and the National Key Research and Development Program of China (nos. 2016YFC0303900 and2017YFC0209700).

References

B. Franco, W. Bader, G. C. Toon et al., “Retrieval of ethane from ground-based FTIR solar spectra using improved spectroscopy: recent burden increase above Jungfraujoch,” Journal of Quantitative Spectroscopy and Radiative Transfer, vol. 160, pp. 36–49, 2015.

K. R. Parameswaran, D. I. Rosen, M. G. Allen, A. M. Ganz, and T. H. Risby, “Off-axis integrated cavity output spectroscopy with a mid-infrared interband cascade laser for real-time breath ethane measurements,” Applied Optics, vol. 48, pp. B73–B79, 2009.

M. Jahjah, W. Ren, P. Stefański et al., “A compact QCL based methane and nitrous oxide sensor for environmental and medical applications,” Analyst, vol. 139, pp. 2065–2069, 2014.

L. Tao, K. Sun, M. A. Khan, D. J. Miller, and M. A. Zondlo, “Compact and portable open-path sensor for simultaneous measurements of atmospheric N₂O and CO using a quantum cascade laser,” Optics Express, vol. 20, no. 27, pp. 28106–28118, 2012.

A. G. Bell, “On the production and reproduction of sound by light,” American Journal of Science, vol. 20, pp. 305–324, 1880.

C. K. N. Patel, “Laser photoacoustic spectroscopy helps fight terrorism: high sensitivity detection of chemical Warfare Agent and explosives,” European Physical Journal Special Topics, vol. 153, no. 1, pp. 1–18, 2008.

C. Haisch, P. Menzenbach, H. Bladt, and R. Niessner, “A wide spectral range photoacoustic aerosol absorption spectrometer,” Analytical Chemistry, vol. 84, no. 21, pp. 8941–8945, 2012.

W. Zhao, X. Gao, W. Chen et al., “Wavelength modulated off-axis integrated cavity output spectroscopy in the near infrared,” Applied Physics B, vol. 86, no. 2, pp. 353–359, 2007.

J. J. Scherer, J. B. Paul, A. O’Keefe, and R. J. Saykally, “Cavity ringdown laser absorption spectroscopy: history, development, and application to pulsed molecular beams,” Chemical Reviews, vol. 97, no. 1, pp. 25–51, 1997.

K. Liu, L. Wang, T. Tan et al., “Highly sensitive detection of methane by near-infrared laser absorption spectroscopy using a compact dense-pattern multipass cell,” Sensors and Actuators B: Chemical, vol. 220, pp. 1000–1005, 2015.

R. F. Curl, F. Capasso, C. Gmachl et al., “Quantum cascade lasers in chemical physics,” Chemical Physics Letters, vol. 487, no. 1–3, pp. 1–18, 2010.

K. Liu, J. Mei, W. Zhang, W. Chen, and X. Gao, “Multi-resonator photoacoustic spectroscopy,” Sensors and Actuators B: Chemical, vol. 251, pp. 632–636, 2017.

T. Kuusela and J. Kauppinen, “Photoacoustic gas analysis using interferometric cantilever microphone,” Applied Spectroscopy Reviews, vol. 42, pp. 443–474, 2007.

K. Liu, X. Guo, H. Yi, W. Chen, W. Zhang, and X. Gao, “Off-beam quartz-enhanced photoacoustic spectroscopy,” Optics Letters, vol. 34, no. 10, pp. 1594–1596, 2009.

W. Ye, C. Li, C. Zheng et al., “Mid-infrared dual-gas sensor for simultaneous detection of methane and ethane using a single continuous-wave interband cascade laser,” Optics Express, vol. 24, no. 15, pp. 16973–16985, 2016.

M. Hepp and M. Herman, “Vibration-rotation bands in ethane,” Molecular Physics, vol. 98, no. 1, pp. 57–61, 2000.

L. G. Smith, “The infrared spectrum of C₂H₆,” Journal of Chemical Physics, vol. 17, no. 2, pp. 139–167, 1949.

A. Miklos, P. Hess, and Z. Bozoki, “Application of acoustic resonators in photoacoustic trace gas analysis and metrology,” Review of Scientific Instruments, vol. 72, no. 4, pp. 1937–1955, 2001.