Mathematical Problems in Engineering
Volume 2020 (2020), Article ID 8942096, 8 pages
https://doi.org/10.1155/2020/8942096

Research Article

Open-Ended Coaxial Cable Selection for Measurement of Liquid Dielectric Properties via the Reflection Method

Jingchen Wang, Eng Gee Lim, Mark Paul Leach, Zhao Wang, and Ka Lok Man

School of Advanced Technology, Xi’an Jiaotong-Liverpool University, Suzhou 215123, China

Correspondence should be addressed to Eng Gee Lim; enggee.lim@xjtlu.edu.cn

Received 30 July 2020; Accepted 24 August 2020; Published 6 October 2020

Guest Editor: Mohamed Nayel

Copyright © 2020 Jingchen Wang et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

Abstract.

An open-ended coaxial cable is used to measure the dielectric properties of lossy liquid. The method which is based on the measurement of the reflection coefficient of the open-ended cable makes it easy to operate and postprocess. To meet the accuracy requirements, the dimensions of the coaxial cable need to be taken into consideration; therefore, it is necessary to select an appropriate coaxial cable for the measurement. This paper investigates the influence of cable dimensions on dielectric measurement accuracy. With careful choice of the coaxial cable, the relative error of calculated results can be less than 0.1%.

1. Introduction

Obtaining information regarding material dielectric properties is an important part of material analysis. Multiple approaches are available to acquire this information as discussed in [1]. In medical dosimetry, biological tissue properties such as dielectric constant, conductivity, and loss factor are of significance [2, 3]. Considering the increasing interest in dielectric properties of biomedical tissues due to advances in medical applications and for research, it is necessary to find an effective solution for the measurement of these dielectric properties. A commonly used method to determine the complex permittivity of materials at radio and microwave frequencies [4–7] is based on the open-ended coaxial cable due to its relative simplicity and accuracy [8]. The calibration of an open-ended coaxial probe for dielectric measurements is proposed in [9, 10].

Open-ended coaxial lines are particularly attractive for in vivo or in vitro measurements of biological material [11] as they offer merits including accuracy, the foundation of a simplified equivalent circuit with the form of capacitive impedance [12], and negligible radiation losses [5].

The capacitive nature of the cable changes with cable size; thus, there is an optimum cable size at a given frequency and permittivity, which offers the minimum measurement uncertainty [13]. Using a reference liquid, with well-known dielectric properties, the cable can be optimized to provide high measurement accuracy. Measurements are made of an unknown liquid using a comparison technique to a known reference liquid. In this work, liquid properties relating to the human body are of primary interest, though the technique could be extended to measure the dielectric properties of any liquid over a wide frequency range. In the literature, the complex dielectric structure of the body is represented by one set of properties: relative permittivity εr = 56 and conductivity σ = 0.8 S/m at 403 MHz. Being able to reproduce liquids with the same properties is a common problem, for which the process described here could provide a reliable, rapid, and cost-effective solution.

In Section 2, the equivalent circuit for the measurement is illustrated. From the circuit, the complex permittivity of the object under test is analysed. In Section 3, investigations into the properties of the coaxial cable are performed. With the knowledge that the cable radius and length are related to the reflection coefficient and conductivity, analyses of how the accuracy of the measurements is influenced by the coaxial cable’s dimensions are provided. Finally, conclusions are drawn in Section 4.

2. Equivalent Circuit

Several conventional methods are applied in permittivity measurements, such as the lumped capacitance method [13], short monopole antenna [7], and quasi-static analysis [14]. The method used in this paper is the open-ended coaxial cable reflection method.

The open-ended coaxial cable reflection method is based on the measurement of the complex reflection coefficient at a single network port. From this measurement, the complex dielectric constant of the object under test can be calculated. The conductivity of the object under test can also be found based on the relationship between the measured reflection parameters and the dielectric properties [15]. The coaxial cable has an inner conductor surrounded by a dielectric material (Teflon) and an outer conductor [16]. The configuration of the coaxial cable and the measurement system is shown in Figure 1.

(a)

(b)

Figure 1: (a) The configuration of a coaxial cable. (b) Configuration of a permittivity measurement system using reflection methods [17].

An equivalent circuit for the measurement system is shown in Figure 2. The capacitance Cf represents the electric field concentration, which is inside the filled dielectric material part of the coaxial cable. The capacitance C(ε) is the fringing field concentration in the dielectric, and the conductance G(ε) is radiation into the dielectric surrounding the cable. The value of the conductance (G) is frequency (f) dependent. The equivalent admittance of the open-ended coaxial cable Y can be written as [18]

where C₀ represents the capacitance of the air-filled parallel plate capacitance, ε represents the permittivity of the object under test, and G₀ represents the external radiation conductance of a coaxial cable. From transmission line theory [7], terminal admittance Y based on the reflection coefficient Г can be defined as follows:

where Z₀ represents the characteristic impedance of the coaxial line (50 Ω). The external radiation conductance of a coaxial cable is very small. As a result, it can be ignored (). Hence, the relationship between the measured complex permittivity of the object under test and the reflection coefficient Г is as follows:

Figure 2: Cable equivalent circuit.

The relative dielectric constant ε′ and the loss factor ε″ are calculated using (4) and (5):

where Γ and φ are the modulus and phase of the input reflection coefficient, respectively. The complex permittivity ε of the object under test and the relationship between loss factor ε″ and conductivity σ can be expressed as follows:

The relationship between the sample capacitance measured by this method and the dielectric constant is linear, which can be found in [19]. In equations (4) and (5), the values of Cf and C₀ are calculated using equations (8) and (9) and the dielectric properties of the reference materials used, such as the deionized water or methanol.

The complex permittivity of the object under test can be obtained using (4) and (5) and the measured reflection coefficient from the object under test. Finally, the dielectric constant and conductivity of the object under test can be calculated from the complex permittivity.

3. Permittivity Measurements

3.1. The Influence of Coaxial Cable Radius

A previous work has shown that the use of a reference liquid with similar dielectric properties to that of the test subject significantly improves measurement accuracy [18]. The dielectric properties of a body-equivalent phantom liquid (used to represent the average dielectric properties of the entire human body) are εr = 56 and σ = 0.8 S/m at 403 MHz. To measure the body-equivalent phantom liquid, for the purposes of simulation and investigation, a reference liquid with dielectric properties of εr = 54 and σ = 0.6 S/m is selected (close to the desired values). To observe the effect of the various cable radii, three radius (R) sizes of the cable are used. Each of the cables has an impedance of 50 Ω and a dielectric constant of 2.17, and they all have a fixed length of 100 mm.

The cable is located within the reference liquid, as shown in Figure 3. Commercial software (CST Studio Suite) has been used to simulate the measurement procedure and calculate the complex permittivity. For each radius of the coaxial cable, the modulus and phase of the reflection coefficient against frequency are shown in Figure 4. The modulus of the reflection coefficient shows a gentle increase with frequency for each cable. The phase of the reflection coefficient covers more than one cycle over the frequency range.

Figure 3: Cross section of the measurement model.

(a)

(b)

Figure 4: (a) The modulus and (b) phase of the reflection coefficient in the reference liquid for coaxial cables with different radii.

From these results, values of Cf and C₀ can be calculated using equations (6) and (7), and they are shown in Figure 5. It can be seen that the values of capacitance are different for each cable. Furthermore, the capacitances have a peak value within the frequency range. The peak value occurs at the frequency where the phase of the reflection coefficient is equal to −180° and can be seen from Figure 5.

(a)

(b)

Figure 5: (a) Calculated value of C₀ and (b) calculated value of Cf for coaxial cables with different radii when cables are located in the reference liquid.

The same process is then followed using the liquid under test. The modulus and phase of the reflection coefficient against frequency obtained from the new simulation for each of the coaxial cables are shown in Figure 6.

(a)

(b)

Figure 6: (a) The modulus and (b) phase of the reflection coefficient for the liquid under test with different cable radii.

As would be expected, the modulus of the reflection coefficient decreases with respect to the results obtained for the reference liquid due to the increase in liquid permittivity and conductivity. The dielectric constant and conductivity of the liquid under test can be predicted from the reflection coefficient using the calculated complex permittivity, as shown in Figure 7.

(a)

(b)

Figure 7: (a) Calculated dielectric constant εr and (b) calculated conductivity σ for the liquid under test for each coaxial cable.

From Figure 7, it can be seen that cable B with a radius of 2.5 mm offers the best accuracy in comparison with the actual values (εr = 56, σ = 0.8 S/m). The cable with a radius of 1 mm does not function suitably; therefore, it could not be used as a dielectric probe. It is clear from the results that the radius of the cable has a significant influence on the accuracy of the measurement. There is significant variation in the simulation for all 3 cables over the frequency range 200–600 MHz; this is caused by reaching the resonant frequency of the probe in the liquid. This cannot be predicted without knowing the dielectric properties of the liquid in advance; hence, measurements are made over a wide band of frequencies, and the dielectric characteristics are calculated at frequencies between resonances in the region where the calculated dielectric properties appear approximately linear.

3.2. Influence of Coaxial Cable Length

The influence of the cable length (L) on measurement accuracy is now investigated using the cable with 2.5 mm radius from the first investigation with lengths of 100 mm, 50 mm, and 15 mm. The modulus and phase of the reflection coefficient obtained from the simulation for each cable length in the reference liquid are shown in Figure 8.

(a)

(b)

Figure 8: (a) The modulus and (b) phase of the reflection coefficient for the reference liquid with varying cable length.

It can be seen from Figure 8 that the length of the cable has little influence on the modulus of the reflection coefficient; however, it has a significant influence on its phase. The shorter cable has a smaller phase range, whereas the longer cable has a larger phase range.

The values of Cf and C₀ are again calculated using equations (6) and (7), as shown in Figure 9. It is seen that the cables with lengths of 100 mm and 50 mm have a peak value of capacitance at the frequency where the phase is equal to −180°. The 15 mm long cable has a stable value of capacitance over the whole simulation frequency range. This results from the phase range being smaller than 180°.

(a)

(b)

Figure 9: Calculated values of (a) C₀ and (b) Cf with varying cable length.

Similarly, the simulated modulus and phase of the reflection coefficient for the liquid under test for each cable length are shown in Figure 10. Similar trends in the modulus and phase of the reflection coefficient are observed as that in the simulation using the reference liquid. The reflection coefficient modulus increases from 0.64 to 0.77 for each of the cables over the simulation frequency range. However, the change in phase is affected dramatically by cable length. The 15 mm cable has less than 180° of reflection coefficient phase variation, whereas the 50 mm and 100 mm extend far beyond that.

(a)

(b)

Figure 10: (a) Modulus and (b) phase of the reflection coefficient for the liquid under test with varying cable length.

The complex permittivity is again used to find the dielectric constant and conductivity of the liquid under test, as shown in Figure 11. It can be seen that when the length of the coaxial cable is 15 mm, the average of the calculated results (εr = 56.55, σ = 0.83 S/m) are the closest to the true dielectric constant and conductivity of the liquid under test (εr = 56, σ = 0.8 S/m). Over the simulation frequency range, the predicted values of dielectric constant and conductivity for the liquid under test are generally constant when using the 15 mm cable. Conversely, there is a big change in both the dielectric constant and the conductivity when using the cables with lengths of 50 mm and 100 mm. Therefore, cable length has a significant impact on measurement accuracy.

(a)

(b)

Figure 11: (a) Dielectric constant εr and (b) conductivity σ of object under test with varying cable length.

3.3. Analysis and Discussion

Through simulation, the relationships between the coaxial cable radius and length with conductivity and reflection coefficient prediction accuracy have been investigated and the importance of radius/length is emphasized in this regard.

The radius of the probe is related to the equivalent capacitance Cf and C₀, so the optimum cable radius can guarantee the accuracy of the calculated results. The most effective choice of cable radius was found to be 2.5 mm in this case. The length of the probe is related to the equivalent length of the coaxial cavity in the equivalent circuit model and is also related to frequency. A shorter probe has a longer period, which can lead to more stable results over a restricted frequency range. The optimal value of the cable length in this work was found to be 15 mm.

4. Conclusion

This paper investigates the influence of the dimensions of a coaxial cable on the accuracy of measured liquid dielectric properties from two aspects: radius and length. It was found that the accuracy of the cable has a strong dependence on the radius, with very small radii offering no accuracy. It was also found that decreasing cable length improved accuracy. The measured results can be close to the actual value over a restricted frequency range after the dimensions of cables are appropriately selected. It can be seen that, for the coaxial cable, selection of cable size and measurement frequency range are important factors when measuring the dielectric properties of a liquid using the return loss method.

Data Availability

The data used to support the findings of this study are included within the article.

Conflicts of Interest

The authors declare that they have no conflicts of interest.

Acknowledgments

This work was partially supported by the Natural Science Foundation of Jiangsu Province (grant no. BK20131183) and Suzhou Science and Technology Bureau (grant no. SYG201211) and partially supported by the AI University Research Centre (AI-URC) through XJTLU Key Programme Special Fund (KSF-P-02) and XJTLU Research Development Fund (PGRS-13-03-06, RDF-14-03-24, RDF-14-02-48, and PGRS2006007). The authors would like to express their sincere gratitude to CST AG for providing the CST STUDIO SUITE® electromagnetic simulation software package under the China Key University Promotion Program and Suzhou Municipal Key Lab for Wireless Broadband Access Technologies in the Department of Electrical and Electronics Engineering, Xi’an Jiaotong-Liverpool University, for research facilities.

References

B. Neves, “Permittivity/dielectric constant: do the math,” Circuitree, vol. 116, 1996.

G. Brix, M. Reinl, and G. Brinker, “Sampling and evaluation of specific absorption rates during patient examinations performed on 1.5 Tesla MR systems,” Magnetic Resonance Imaging, vol. 19, no. 6, pp. 769–779, 2001.

P. A. Bottomley, R. W. Redington, W. A. Edelstein, and J. F. Schenck, “Estimating radiofrequency power deposition in body NMR imaging,” Magnetic Resonance in Medicine, vol. 2, no. 4, pp. 336–349, 1985.

T. P. Marsland and S. Evans, “Dielectric measurements with an opened coaxial probe,” IEE Proceedings, vol. H134, no. 4, pp. 341–349, 1987.

G. S. Smith and J. D. Nordgard, “Measurement of the electrical constitutive parameters of materials using antennas,” IEEE Transactions on Antennas & Propagation, vol. AP-33, pp. 783–792, 1985.

M. M. Brady, S. A. Symons, and S. S. Stuchly, “Dielectric behaviour of selected animal tissues in vitro at frequencies from 2 to 4 GHz,” IEEE Transactions on Biomedical Engineering, vol. BME-28, pp. 305–307, 1981.

E. C. Burdette, F. L. Cain, and J. Seals, “In vivo probe measurement technique for determining dielectric properties at VHF through microwave frequencies,” IEEE Transactions on Microwave Theory and Techniques, vol. 28, pp. 414–427, 1980.

H. L. Gong, S. D. Chen, and D. J. Qian, “Rebuilt and modified model for complex permittivity measurement using the coaxial line,” Information and Electronic Engineering, vol. 6, no. 6, pp. 448–451, 2008.

G. P. Otto and W. C. Chew, “Improved calibration of a large open-ended coaxial probe for dielectric measurements,” IEEE Transactions on Instrumentation and Measurement, vol. 40, no. 4, pp. 742–746, 1991.

L. Zhang, P. G. Liu, X. T. Shi et al., “A comparative study of a calibration method for measuring the dielectric properties of biological tissues on electrically small open-ended coaxial probe,” in Proceedings of the 2012 International Conference on Biomedical Engineering and Biotechnology, Macao, China, May 2012.

T. W. Athey and M. A. Stuchly, “Measurement of radio frequency permittivity of biological tissues with an open-ended coaxial line (parts 1 and 2),” IEEE Transactions on Microwave Theory and Techniques, vol. MTT-30, pp. 82–92, 1982.

M. A. Stuchly, M. M. Brady, S. S. Stuchly, and G. Gajda, “Equivalent circuit of an open-ended coaxial line in a lossy dielectric,” IEEE Transactions on Instruction and Measurement, vol. IM-31, no. 2, pp. 116–119, 1982.

M. A. Stuchly and S. S. Stuchly, “Coaxial line reflection methods for measuring dielectric properties of biological substances at radio and microwave frequencies—a review,” IEEE Transactions on Instrumentation and Measurement, vol. IM-29, no. 3, pp. 176–183, 1980.

D. K. Misra, “A quasi-static analysis of open-ended coaxial lines,” IEEE Transactions on Microwave Theory and Techniques, vol. MTT-35, no. 10, pp. 925–928, 1987.

L. Zhang, “Study on measurement of biological tissue dielectric properties in high frequency using reflection method on the open-ended coaxial,” National University of Defense Science and Technology, Changsha, China, 2011, Master thesis.

T. T. Khan, “A method for measuring dielectric properties of non-magnetic liquids and predicting their contamination level,” in Proceedings of the 2012 IEEE/OSA/IAPR International Conference on Informatics, Electronics & Vision, pp. 445–450, Dhaka, Bangladesh, 2012.

J. Wang, M. Leach, E. G. Lim, and Z. Wang, “An open-ended coaxial cable reflection method for measuring dielectric properties of biological liquid,” in Proceedings of the 7th International Conference on Information, System and Convergence Applications, Ho Chi Minh, Vietnam, February 2020.

D. Berube, F. M. Ghannouchic, and P. Savard, “A comparative study of four open-ended coaxial probe models for permittivity measurements of lossy dielectric biological materials at microwave frequencies,” IEEE Transactions on Microwave Theory and Techniques, vol. 44, no. 10, pp. 192-193, 1996.

V. Hartwig, G. Giovannetti, N. Vanello et al., “An electrodeless system for measurement of liquid sample dielectric properties in radio frequency band,” in Proceedings of the 2006 International Conference of the IEEE Engineering in Medicine and Biology Society, New York, NY, USA, August 2006.