Stem Cells International
Volume 2018 (2018), Article ID 8207071, 11 pages
https://doi.org/10.1155/2018/8207071

Research Article

Tissue Engineering of Large Full-Size Meniscus Defects by a Polyurethane Scaffold: Accelerated Regeneration by Mesenchymal Stromal Cells

Matthias Koch,¹ Felix P. Achatz,¹ Siegmund Lang,¹ Christian G. Pfeifer,¹ Girish Pattappa,1,2 Richard Kujat,1,2 Michael Nerlich,¹ Peter Angele,1,3 and Johannes Zellner¹

¹Department of Trauma Surgery, University Medical Centre Regensburg, Franz-Josef-Strauss-Allee 11, 93053 Regensburg, Germany
²Laboratory of Experimental Trauma Surgery, Department of Trauma Surgery, University Regensburg Medical Centre, Regensburg, Germany
³Sporthopaedicum Regensburg/Straubing, Hildegard-von-Bingen-Str. 1, 93053 Regensburg, Germany

Correspondence should be addressed to Johannes Zellner; johannes.zellner@ukr.de

Received 24 November 2017; Revised 6 March 2018; Accepted 4 April 2018; Published 7 May 2018

Academic Editor: Mikel Sánchez

Copyright © 2018 Matthias Koch et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

Abstract.

The endogenous healing potential of avascular meniscal lesions is poor. Up to now, partial meniscectomy is still the treatment of choice for meniscal lesions within the avascular area. However, the large loss of meniscus substance predisposes the knee for osteoarthritic changes. Tissue engineering techniques for the replacement of such lesions could be a promising alternative treatment option. Thus, a polyurethane scaffold, which is already in clinical use, loaded with mesenchymal stromal cells, was analyzed for the repair of critical meniscus defects in the avascular zone. Large, approximately 7 mm broad meniscus lesions affecting both the avascular and vascular area of the lateral rabbit meniscus were treated with polyurethane scaffolds either loaded or unloaded with mesenchymal stromal cells. Menisci were harvested at 6 and 12 weeks after initial surgery. Both cell-free and cell-loaded approaches led to well-integrated and stable meniscus-like repair tissue. However, an accelerated healing was achieved by the application of mesenchymal stromal cells. Dense vascularization was detected throughout the repair tissue of both treatment groups. Overall, the polyurethane scaffold seems to promote the vessel ingrowth. The application of mesenchymal stromal cells has the potential to speed up the healing process.

1. Introduction

Lesions of the meniscus are amongst the most frequent knee injuries in orthopedic surgery [1, 2]. In many cases, partial meniscectomy has to be performed due to the poor endogenous healing capacity of avascular parts of the meniscus [2–4]. However, the loss of meniscus continuity predisposes for the development of osteoarthritic changes, which correlates with the amount of resected meniscus substance [5–7].

The meniscus has a decisive functional and biomechanical relevance for an intact knee joint [8, 9]. The knee menisci provide essential qualities in load bearing and shock absorption as well as in stabilization, lubrication, and proprioception of the knee joint [10–14]. (Partial) Meniscectomy causes severe changes in the biomechanics of the knee joints. The effect is directly proportional to the amount of lost tissue [15] and results in drastically increased contact pressure [16]. Therefore, it is of importance to restore as much meniscus tissue as possible.

Successful repair strategies for lesions in the vascular zone of the meniscus like suturing have already been developed [17, 18]. However, up to now, there is still no established curative therapy for lesions within the avascular parts of the meniscus in clinical practice [19–21].

According to the current literature, mesenchymal stromal cells (MSC) are the focus of attention in newly developed approaches for meniscus repair [2, 21–27]. As these cells have both the potential to differentiate into fibrochondrocytes and the ability to secrete repair-promoting growth factors, they seem to be an ideal tool for meniscus repair [22, 28, 29]. Preclinical studies have already shown the repair potential of mesenchymal stromal cells in combination with a scaffold in the treatment of small tears, punch defects in the avascular zone of the meniscus, and full-size meniscal defects [2, 12, 23, 24, 27].

There are also a few clinical studies showing promising results after application of MSCs for meniscus regeneration in humans [21, 30].

However, clinical translation of these repair approaches has not been achieved. The polyurethane scaffold (Actifit®, Orteq, London, UK) used in this study has successfully been clinically applied in a cell-free approach in recent years [1, 31–39]. Therefore, the aim of this study was to investigate the effect of a polyurethane scaffold (Actifit, Orteq, London, UK) loaded with mesenchymal stromal cells concerning the effectiveness of a combination of mesenchymal stromal cells and this scaffold for the treatment of large, critical size combined vascular and avascular meniscus defects. Special emphasis was given on vascularity as an important factor for the integration of the repair tissue in the native surrounding meniscus.

2. Material and Methods

The local government’s animal rights protection authorities approved this study in accordance with the National Institutes of Health guidelines for the use of laboratory animals.

2.1. Study Design

14 male New Zealand white rabbits aged 12 to 14 weeks and weighing 2.8 to 3.2 kg were used. Each animal received the study treatment on the right knee joint and the control treatment on the left knee joint. Depending on the study period of 6 or 12 weeks, two groups were differed from each group consisting of 7 animals. No randomization or matching was done to allocate the animals to the experimental groups. All animals were kept in single animal cages with free access to food and water.

2.2. Bone Marrow-Derived Mesenchymal Stromal Cells: Harvest and Culture

For the preparation of the study treatment, bone marrow-derived MSCs were harvested 4 weeks before the index surgery. The bone marrow harvest and MSC cell isolation was performed as previously described [22, 40]. The New Zealand white rabbits were anesthetized using a combined intramusculary application of 0.6 ml/kg of ketamine 10% and xylazin 2%. Bone marrow-derived mesenchymal stromal cells were harvested by puncturing the iliac crest of the rabbits on both sides by a small incision and penetrating the bone cortex with an 18-gauge needle. The bone marrow was collected into a heparinized syringe. Dulbecco’s modified Eagle’s medium (DMEM), low glucose concentration, with 10% fetal bovine serum, 1% penicillin, and 1% HEPES, was added to the aspirate. Nucleated cells (20 × 10⁶) were plated in 75 cm² culture dishes and cultivated at 37°C. Plastic adhesion distinguishes MSCs from other cell populations as previously described [41–43]. The medium, containing DMEM, low glucose concentration, 10% fetal bovine serum, 1% penicillin, 1% HEPES, and 0.01% fibroblast growths factor, was changed after one week of adhesion twice a week until the adherent cells (MSCs) reached 80% confluence [3].

2.3. Polyurethane Scaffolds

Commercially available polyurethane scaffolds Actifit (Orteq, London, UK) were used for this study. The material was cut in small parts approximately fitting the planned meniscus defects. Before use, these scaffolds were sterilized with 25 kGy beta rays (Beta-Gamma-Service GmbH, Saal, Germany).

2.4. Loading of the Polyurethane Scaffolds

Depending on the group, the fitted polyurethane scaffolds were loaded with either 200 μl of the MSC cell suspension or 200 μl of cell-free chondrogenic medium.

The cell suspension contained 2 × 10⁴ nucleated cells/μl resuspended in chondrogenic medium. The chondrogenic medium consisted of the same ingredients like the cell-free chondrogenic medium except for the additional nucleated cells. The cell-free chondrogenic medium contained DMEM, high glucose concentration (4500 mg/l), with 10% pyruvate, 10% ITS, 10% dexamethasone, 10% TGFβ1, and 10% ascorbic acid as previously described [25].

The loading process of the scaffolds was achieved using a rotary valve vacuum pump (Vacuubrand GmbH, Wertheim, Germany) as already described by Achatz et al. [44].

Subsequently, the loaded scaffolds were incubated at 37°C for 1 hour and then kept in chondrogenic media overnight to allow cell adherence before being implanted the following day. Pretests showed a successful loading of the scaffolds.

2.5. Surgical Procedure for Meniscus Defects

In vivo experiments were performed bilaterally in 14 New Zealand white rabbits. One blinded experienced orthopedic surgeon performed all surgical procedures.

The New Zealand white rabbits were anesthetized using a combined intramuscular application of 0.6 ml/kg of ketamine 10% and xylazin 2%. For the dissection of the lateral menisci, the lateral knee joint compartment was opened by lateral parapatellar arthrotomy. The lateral meniscus was luxated anteriorly by a limited soft tissue release. In the pars intermedia, a partial meniscectomy sized 7 × 3 mm was performed. The created combined vascular and avascular defect presented a complete discontinuance of the meniscus circumference. The meniscus defects were filled with Actifit scaffolds, on the one site loaded with mesenchymal stromal cells and on the contralateral site with a cell-free polyurethane scaffold. Preparation of the scaffolds was performed as described above. Subsequently, the scaffolds were attached site to site to the crus anterior and posterior of the lateral meniscus with a resorbable 4-0 suture.

Wound healing was checked daily. Postoperative pain control was achieved by subcutaneous application of carprofen 5 mg/kg. There were no limitations concerning movement and weight bearing postoperative. The animals were euthanized at 6 or 12 weeks by an intravenous application of an overdose of narcoren (0.5 g/kg). The New Zealand white rabbits were anesthetized using a combined intramuscular application of 0.6 ml/kg of ketamine 10% and xylazin 2% beforehand.

2.6. Gross Assessment of Joint Morphology

The gross assessment of joint morphology was conducted as described previously by Zellner et al. [24]. Rabbits were euthanized in deep narcosis with an overdose of narcoren intravenously depending on the group 6 or 12 weeks after the implantation of the scaffolds. Knee joints were exposed, and the menisci were harvested. Afterwards, the macroscopic morphology of the meniscus and the joint compartment were evaluated and photographed. The correct anatomic location of the menisci, the macroscopic integration of the repair tissue, the state of the meniscus surface, and the color changes were evaluated. Eventual signs of degenerative changes in terms of osteophyte development and cartilage deterioration were documented. All menisci and knee joints were analyzed by two experienced and blinded scorers, and the results were collected for an established scoring system. Afterwards, the lateral menisci were harvested for further histological examination and photographic documentation.

2.7. Histology

The lateral menisci were fixed in a solution containing 4% paraformaldehyde and 15% picric acid, embedded in Tissue-Tek OCT (Sakura Finetek, Tokyo, Japan) and frozen in liquid nitrogen. All samples were cut in 10 μm transversal sections, and every 10th of them was stained with dimethylmethylen blue (DMMB). The content of proteoglycan was determined according to the percentage of the filling of the porous area of the scaffolds by stained extracellular matrix.

Overall, two blinded scorers, both experienced in the knee anatomy of rabbits and in histological assessment, analyzed the sections according to the established scoring system. [13]

2.8. Immunohistochemistry

2.8.1. Type II Collagen

The frozen sections, embedded in Tissue-Tek OCT (Sakura Finetek, Tokyo, Japan), were washed in a phosphate-buffered saline and digested with 0.1% pepsin at pH 3.5 for 15 minutes to facilitate antibody access to the target epitopes. Type II collagen was immunolocalized by the immunoperoxidase ABC technique (Vector, Burlingame, CA, USA). As primary antibodies, anticollagen II (clone II-4C11; Calbiochem Merck, Schwalbach Germany) was used. The antibody dilution was 1 : 100. After staining with biotin conjugated polyclonal goat anti-mouse IgG secondary antibody (Jackson, West Grove, PA, USA), positive signals were visualized by nickel and cobalt-enhanced 3,3-diaminobenzidine (DAB) [13, 23].

Concerning the evaluation of the content of collagen type II, the percentage of collagen type II stained area within the extracellular matrix in the porous area of the scaffolds was assessed in comparison to the extracellular matrix in the DMMB-stained samples.

2.8.2. CD 31

For the evaluation of vascularization, CD 31 immunohistochemistry was performed almost consistent to type II collagen immunohistochemistry (described above).

The procedure differed in two facts. No pepsin digestion was performed, and primary monoclonal CD31 mouse anti-rabbit antibodies (clone JC-70A IgG1 light chain-type kappa; Abcam, Cambridge, UK) were used with a dilution of 1 : 50.

2.9. Meniscus Scoring System

For standardized evaluation and comparison of the meniscus repair, an established scoring system was applied. It has previously been published by Zellner et al. [23, 24] for the evaluation of the repair of meniscal punch defects and meniscal tears.

Scoring items of the macroscopical analysis were stability and defect filling with repair tissue. The quality of the surface area, the integration of the repair tissue in the native meniscus, cellularity, cell morphology and the content of proteoglycan were subgroups of the histological assessment. The expression of collagen II was analyzed by immunohistochemistry.

That way, 8 individual scoring subgroups were formed, each receiving a scoring value ranging from 0 (no repair) to 3 (meniscus-like tissue). The values of these items were summed up, consequently reaching a combined score from 0 (no repair) to 24 (complete reconstitution of the meniscus) (Table 1). Two experienced blinded scorers conducted the data collection. A high internal consistency has been attributed to this scoring system (Cronbach’s alpha = 0.88) [24]. By this, validated scoring system results can be easily interpreted and compared to other results.

Table 1: Meniscus scoring system for the evaluation of meniscus repair tissue [23, 24].

0
1
2
3

Defect-filling
No fill
<25%
25–75%
>75%

Surface
No surface
Ruptured
Fissured/fibrillated
Meniscus-like

Integration
No integration
Partial, unilateral integration
Bilateral partial or unilateral complete integration
Bilateral complete integration

Cellularity
No cells
>10 cell cluster/slide
No cell cluster/slide, cell-ECM
Meniscus-like cell-ECM ratio

Cell morphology
No cells
<25% meniscus-like cells
25–75% meniscus-like cells
>75% meniscus-like cells

Content of proteoglycan
No staining for proteoglycan
<25%
25–75%
>75%

Content of collagen II
No staining for collagen II
<25%
25–75%
>75%

Stability
No stability
Weak
Stable in shape
Stable to pressure and pulling stress

ECM = extracellular matrix.

2.10. Statistical Analysis

Statistical analysis was performed using the SPSS software version 23.0 (SPSS, Chicago, IL, USA) to determine relationships between variables. To determine whether data followed a Gaussian distribution, a Kolmogorov-Smirnov test was conducted. For comparison of normal distributed data, a paired -test was used. For nonnormal distributed data, the Mann–Whitney test was used. A probability value of less than 0.05 was set as the level of statistical significance for all evaluations.

3. Results

3.1. Cell-Free Polyurethane Scaffold: 6 Weeks

The evaluation of the menisci of the New Zealand white rabbits after meniscus defect therapy with cell-free polyurethane scaffolds and a study period of 6 weeks showed macroscopically distinguishable meniscus defects. In one animal, almost a complete filling of the meniscal defect was observed. The remaining menisci showed a partial filling of the meniscus defects. Regarding the meniscus surface, in four cases, the meniscus surface was fissured, whereas in three menisci, complete ruptures were present. Overall, one meniscus showed a complete integration of the polyurethane scaffold to the surrounding meniscus tissue. In three cases, bilateral partial or unilateral complete integration was observed. In the remaining three menisci, only unilateral partial integration was found. When evaluating the cellularity of the meniscus scaffold in comparison to the native meniscus tissue, in all cases, the cell number within the meniscal repair tissue was higher than that within the native meniscus tissue, with less than a quarter of cells registered as meniscus-like cells found. The repair tissue generally showed a lower content of proteoglycans and type II collagen in comparison to the surrounding meniscus tissue. The stability testing showed only a low stability and the developed tissue appeared soft and ruptured. Signs of inflammation or foreign body reaction were found in none of the animals (see Figure 1).

Figure 1: Left: macroscopic view on a lateral meniscus (b). The defect is filled with a cell-free polyurethane scaffold (a) 6 weeks after implantation. Right: microscopic view; DMMB staining of the cell-free polyurethane scaffold (a) and surrounding meniscus tissue (b = avascular part; c = vascular zone) 6 weeks after implantation. Unilateral integration on the left side and no integration on the right side. Benchmark for both figures: .

3.2. Mesenchymal Stromal Cell-Loaded Polyurethane Scaffold: 6 Weeks

In the group of meniscal defects treated with a mesenchymal stromal cell-loaded polyurethane scaffold in four animals. an almost complete defect filling was seen. In the remaining three cases, over half of the defect was filled. In all menisci, the surface of the repair tissue was fissured without any complete ruptures. Regarding the integration, in three menisci, a bilateral complete integration of the cell-loaded scaffolds to the surrounding native meniscus tissue was observed. In four cases, only unilateral complete or bilateral partial integration was reached. The cellularity of the developed tissue was more than a quarter higher than the native surrounding meniscus tissue. Whereas the type II collagen content appeared generally low, the content of proteoglycans ranged between 25% and 75% in comparison to the native meniscus tissue. The new developed tissue of six animals was stable in shape. In one animal, the repair tissue was of poor quality and ruptured during stability testing. Signs of inflammation or foreign body reaction were found in none of the animals (see Figure 2).

Figure 2: Left: macroscopic view on a lateral meniscus (b = avascular part; c = vascular part). The defect is filled with a mesenchymal stromal cell-loaded polyurethane scaffold (a) completely surrounded by native meniscus tissue 6 weeks after implantation. Right: microscopic view; DMMB staining of the mesenchymal stromal cell-loaded polyurethane scaffold (a) and surrounding meniscus tissue (b = avascular part; c = vascular zone) 6 weeks after implantation. Bilateral integration into the surrounding meniscus tissue. Benchmark for both figures: .

3.3. Cell-Free Polyurethane Scaffold: 12 Weeks

Regarding cell-free polyurethane scaffolds after a study period of 12 weeks , two menisci showed a complete defect filling. A partial defect filling was observed in 4 menisci, and in one case, just a marginal filling was seen. In three cases, the surface of the repair tissue appeared to be meniscus-like. In three cases, it seemed to be fissured, and in one case, it completely ruptured. In two menisci, the full integration of polyurethane scaffolds into the native meniscus was achieved. In five cases, the repair tissue integrated partially bilateral or completely unilateral. The cellularity of the new formed tissue was higher than that in the surrounding native meniscus tissue. In five cases, approximately half of the cells were of meniscus-like morphology, whereas in two cases, less than a quarter of the cells showed meniscus-like morphology.

The proteoglycan content was below 25% in two menisci and ranged between 25% and 75% in five cases. The content of type II collagen was less than 25% in six menisci and ranged between 25% and 75% in one case. In the stability testing, six menisci were in shape, and in one case, it appeared soft and ruptured. None of the animals showed signs of inflammation or foreign body reaction (see Figure 3).

Figure 3: Left: macroscopic view on a lateral meniscus (b = avascular part; c = vascular part). The defect is filled with a cell-free polyurethane scaffold (a) completely surrounded by native meniscus tissue 12 weeks after implantation. Right: microscopic view; DMMB staining of the cell-free polyurethane scaffold (a) and surrounding meniscus tissue (b = avascular part; c = vascular zone) 12 weeks after implantation. Bilateral integration into the surrounding meniscus tissue. Benchmark for both figures: .

3.4. Mesenchymal Stromal Cell-Loaded Polyurethane Scaffold: 12 Weeks

12 weeks after meniscus defect treatment with mesenchymal stromal cell-loaded polyurethane scaffolds , an almost complete defect filling was achieved in two cases, while in five cases, the defect filling was more than half. Complete ruptures did not occur. In three menisci, the surface of the repair tissue appeared to be meniscus-like, whereas in four cases, the surface seemed to be fissured. A complete integration into the surrounding native meniscus was reached in two cases. A unilateral complete or bilateral partial integration was observed in four menisci. Only one animal presented with partial unilateral integration. The cellularity of the new formed tissue was higher than that in native meniscus tissue, and in almost all cases, between 25% and 75% of the cells showed meniscus-like morphology. While in almost all animals, the content of proteoglycan ranged between 25% and 75% and only two animals reached a content of type II collagen higher than 25%. The repair tissue was stable and in shape in five cases and appeared weak and ruptured in another two cases. None of the animals showed signs of inflammation or foreign body reaction (see Figure 4).

Figure 4: Left: macroscopic view on a lateral meniscus (b = avascular part; c = vascular part). The defect is filled with a mesenchymal stromal cell-loaded polyurethane scaffold (a) completely surrounded by native meniscus tissue 12 weeks after implantation. Right: microscopic view; DMMB staining of the mesenchymal stromal cell-loaded polyurethane scaffold (a) and surrounding meniscus tissue (b = avascular part; c = vascular zone) 12 weeks after implantation. Bilateral integration into the surrounding meniscus tissue. Benchmark for both figures: .

3.5. Meniscus Scoring

After six weeks, the meniscus scoring showed overall significant higher values in the group of mesenchymal stromal cell-loaded polyurethane scaffolds (mean: 15.6, SD: ±1.3) in comparison to the group of cell-free polyurethane scaffolds (mean: 10.4, SD: ±0.7) (Figure 5).

Figure 5: Mean meniscus scoring value: comparison of all meniscus score results of cell-free polyurethane scaffolds (= cell free) and mesenchymal stromal cell-loaded scaffolds (= MSC) after 6 and 12 weeks. () describes significant differences .

Moreover, the histological evaluation showed a significantly higher content of proteoglycans in the group of mesenchymal stromal cell-loaded polyurethane scaffolds (mean: 2.0, SD: ±0.1) as observed in the group of cell-free polyurethane scaffolds (mean: 1.1, SD: ±0.4) after 6 weeks. The significant advantage of the mesenchymal stromal cell loading is lost after 12 weeks (Figure 6).

Figure 6: Mean meniscus scoring value concerning the subcategory “content of proteoglycans”: comparison of the meniscus score results of cell-free polyurethane scaffolds (= cell free) and mesenchymal stromal cell-loaded scaffolds (= MSC) after 6 weeks. () describes significant differences .

Regarding the integration of the polyurethane scaffolds into the surrounding meniscus tissue, a significantly better integration could be obtained by mesenchymal stromal cell loading (MSC-loaded polyurethane scaffolds 2.4 SD ± 0.5 versus cell-free polyurethane scaffolds 1.6 SD ± 0.6; ) (Figure 7).

Figure 7: Mean meniscus scoring value concerning the subcategory “integration of the scaffolds in the surrounding meniscus tissue”: comparison of the meniscus score results of cell-free polyurethane scaffolds (= cell free) and mesenchymal stromal cell-loaded scaffolds (= MSC) after 6 weeks. () describes significant differences .

Overall, after 12 weeks, no significantly better meniscus scoring results were obtained for the treatment of combined vascular and avascular meniscus defects by mesenchymal stromal cell loading of polyurethane scaffolds in comparison to cell-free polyurethane scaffolds (mean overall score 15.3SD ± 14 versus 14.5 SD ± 1.7; ) (Figure 5). Additionally, no significant differences were detected in the meniscus scoring subcategories.

3.6. Meniscus Vascularization

In addition to the regenerative potential, vascularization of the repair tissue of all harvested menisci ( menisci of 14 animals) was assessed by CD31 immunohistochemistry. Consistent results were obtained for all groups. No differences between mesenchymal stromal cell-loaded polyurethane scaffolds and cell-free polyurethane scaffolds were observed. Also, the study period of 6 or 12 weeks showed no influence on the results. Overall dense vascularization was observed in each histological cut showing an abundance of various-sized transverse and longitudinal vessel cuts (Figure 8). Vessels were detected both in the part of the repair tissue corresponding to the peripheral, vascular area of the native meniscus, and in the repair tissue corresponding to the avascular zone of the native meniscus.

Figure 8: Left: microscopic view on transversal cut meniscus sections; CD 31 immunohistochemistry staining for the evaluation of vessel ingrowth. Vascular endothelial cells are stained black. The polyurethane scaffold (a) is completely surrounded by native meniscus tissue (b). Vascular endothelial cells are stained black and are shown by the vessels cuts (c), which were found in the peripheral as well as central parts of the scaffolds; benchmark for the left picture: . Right: microscopica view, CD 31 immunohistochemistry staining for the evaluation of vessel ingrowth. Vascular endothelial cells are stained black. (a) demonstrates the polyurethane scaffold, and (b) presents examples of vessel cuts; benchmark for the right picture: . Both pictures refer to MSC-loaded scaffold group after 6 weeks.

4. Discussion

In this study, it was shown that an accelerated repair of a combined avascular and vascular meniscus defect can be achieved by using a mesenchymal stromal cell-loaded porous polyurethane scaffold in a New Zealand white rabbit model. After six weeks, the stromal cell-loaded approach showed significant advantages in proteoglycan content, integration into the surrounding native meniscus, and overall scoring compared to the cell-free approach.

However, the treatment of injured meniscus, especially of lesions located in the avascular zone, remains challenging due to its poor endogenous regenerative capacity. Partial meniscectomy often remains the only treatment option, although it is well known that the loss of meniscus tissue predisposes for the onset of osteoarthritic changes in the knee joint [3, 5, 7, 12, 45]. So tissue engineering offers new treatment modalities for meniscus repair or even for meniscus replacement [46].

An accelerated meniscus repair enables patients’ earlier mobilization and load bearing, thus leading to shorter rehabilitation periods. Moffet et al. emphasized the importance of early rehabilitation after meniscus treatment to achieve a good functional outcome [47]. Furthermore, Eriksson and Häggmark observed the rapid atrophy of the quadriceps femoris muscle, after immobilization of the knee joint caused by surgical procedures [48]. Its rebuilding can be problematic especially for older patients [49, 50]. However, regaining full quadriceps strength before returning to extensive stress exposure is important both in athletes and for the normal population [51]. Therefore, studies generally recommend shortened immobilization periods [52, 53]. Importantly, an accelerated meniscus repair would also mean a drastically lightened burden for patients with an earlier return to full weight bearing, to sport activities, and to work.

The mesenchymal stromal cells seem to make a major contribution to the accelerated regeneration. Due to their properties to differentiate to fibrochondrocytes, act on the vessel growths, and secrete different growth factors [54], they are suitable for cell-based tissue engineering approaches. Many studies already showed their feasibility for meniscus regeneration [28, 40, 55]. The application of MSCs alone and in combination with growth factors and scaffolds has been tested in several in vivo and in vitro studies. In the last decades, stem cell-based therapies have been translated into clinical settings [56–58]. For example, Zellner et al. achieved the repair of meniscal tears in rabbits with the combination of a hyaluronic acid gelatin scaffold and mesenchymal stem cells [24].

Yu et al. summarized that bone marrow-derived mesenchymal stem cells are the most commonly used cell source in tissue engineering [27].

However, also further cell sources, such as synoviocytes, meniscus cells, or adipose tissue-derived cells, have been tested concerning their relevance for meniscus regeneration [12]. At least, the therapeutic capacity of synoviocytes has been described to be inferior to the capacity of bone marrow-derived MSCs [59, 60]. Concerning the use of meniscus-derived MSCs as well as adipose tissue-derived MSCs, a regeneration-promoting effect in vivo has been reported [61–63]. However, harvesting and cell isolation as well as the application protocols of these cells are not as well validated as for bone marrow-derived MSCs, which advocates the use of bone marrow-derived MSCs. Nevertheless, up to now, there is no consensus concerning the best cell source for meniscus regenerative cell-based tissue engineering approaches.

The polyurethane scaffold (Actifit, Orteq, London, UK) used in this study previously showed good results and statistically significant improvements in clinical outcome as well as improved macroscopic and histological meniscus healing in both clinical trials [31, 32] as well as in vivo experiments [64]. Additionally, our own previous studies showed the suitability and advantages of this polyurethane scaffold for a cell-based tissue engineering approach in comparison to another hyaluronic acid gelatin scaffold [44]. Therefore, it was also considered to be suitable for the implementation of this study.

While the present study showed a significant benefit for the cell-based approach after 6 weeks regarding proteoglycan production and integration, the results after 12 weeks did not differ significantly between the mesenchymal stromal cell-loaded and cell-free approaches. It has been shown that the progenitor cell concentration in the synovial fluid is elevated after a meniscal injury. This is presumably due to progenitor cell release from meniscal tissue [65, 66]. The existence of these cells and their integration into the polyurethane scaffold is a possible explanation for the increase of proteoglycans and type II collagen content in the group of cell-free scaffold after 12 weeks. Furthermore, the increased proteoglycan production and integration of the cell-free scaffold into the surrounding native meniscus could be explained by the cell ingrowth due to neovascularization.

However, in a similar meniscus defect model, Angele et al. were able to show that only a fibrous muted healing response can be observed both in untreated menisci and in menisci treated with an empty hyaluronic acid-based scaffold [67]. Verdonk et al. published a case series with 52 partial meniscectomies that were treated with a cell-free polyurethane scaffold [1]. In the two-year follow-up, they demonstrated a significantly improved clinical outcome with a reduced pain level, better function, and healthier cartilage state. Other clinical studies also confirmed the safety and improved clinical outcome using a cell-free polyurethane scaffold for the treatment of both lateral and medial meniscus defects [31, 32, 68]. Although cell-free scaffolds are currently used in clinics with promising results [34, 69], our results suggest that this approach could possibly be further enhanced by the addition of autologous mesenchymal stem cells [13]. Especially in an early phase of regeneration, loading with mesenchymal stromal cells showed significant advantages in comparison to the cell-free scaffold, as shown in the results after 6 weeks.

In line with our findings, several studies have shown statistically significant benefits that resulted in loading a scaffold with mesenchymal stem cells compared to cell-free scaffolds [25, 70]. Most likely, this benefit originates from the differentiation capacity, the ability of paracrine secretion of bioactive substances, and immunoregulatory properties of mesenchymal stem cells [40, 71, 72].

In the present study, we found extensive vascularization in the repair tissue both in the cell-free and in the mesenchymal stromal cell-based study groups. These findings are in line with results from a previous study, where we showed that preconditioning of meniscus cells/mesenchymal stem cell-based tissue engineering products influences the angiogenic potential of tissue engineering products [3]. A dense abundance of vessels was present throughout the repair tissue, remarkably even in parts that correspond to the nonvascularized part of the native meniscus. Apparently, the polyurethane scaffold seems to promote the ingrowth of vessels. These results correlate to the results of a clinical study by Verdonk et al. [39], which also suggested vessel ingrowth into the repair tissue after three months. However, it is yet to be determined if this neovascularization has a positive or negative effect on the meniscus repair. On the one hand, Petersen et al. [73] found that the vascular endothelial growth factor (VEGF) coating of a meniscus suture has a negative effect on meniscus healing. Ahsraf et al. [74] described a positive correlation between osteoarthritic changes in the knee and meniscus neovascularization; hence, they proposed that neovascularization and nerve ingrowth might be a reason for pain genesis in osteoarthritic knees. On the other hand, several clinical studies with the same polyurethane scaffold that was used in the present study rather described a significant reduction in pain levels [1, 31, 32, 68]. Furthermore, needling, rasping, or trephination to promote vessel ingrowth into damaged meniscus by puncturing healthy tissue is frequently and successfully used in meniscus surgery [75, 76]. It is also generally accepted that the vascular part of the meniscus has a far greater healing capacity than the avascular part [7, 24].

Nevertheless, there are a few limitations standing in contrast to the strength of the present study. Although we confirmed the isolation of the stromal cell population by plastic adherence, there was no determination of stem cell markers or an assessment concerning the multilineage potential of the harvested MSCs. A further limitation can be seen in the limited transferability to human meniscus, as the small size and low weight of New Zealand rabbits lead to different surgery circumstances and acting forces in the knee compared to human knees. Furthermore, the biomechanics of the scaffold have not been assessed in the present study, as biomechanics and biocompatibility are essential properties of any biomaterial. However, this animal model has been established and extensively tested in our study group before. So several other authors have considered it to be a suitable model for meniscus defects [77–79]. Additionally, bone marrow-derived MSCs were used, as they are an easily accessible cell source both preclinical and clinical. Also, the tested polyurethane scaffold has found its way into the everyday clinical application. Thus, this setting is well suited to represent the actual clinical situation.

In further studies, scaffolds loaded with different specified cells have to be investigated concerning the best cell source for tissue engineering-related meniscus regeneration. As biomechanics are essential properties of any biomaterial and have not been assessed within this study, they have to be examined in further studies. Additionally, longer-term studies are required since no complete meniscus healing was achieved within the current study period of 12 weeks.

5. Conclusion

The present study showed for the first time the accelerated healing process of tissue engineering products for meniscus regeneration due to a MSC-based approach. Previous studies indicated the relevance of vascularization in context to meniscus regeneration. Vascularization was also documented in the investigated polyurethane scaffolds, which confirms previous findings concerning a vascularization-promoting effect of MSCs in combination with polymer scaffolds.

Conflicts of Interest

The authors declare that they have no conflicts of interest.

Acknowledgments

The authors thank Daniela Drenkard for her excellent technical assistance.

References

P. Verdonk, P. Beaufils, J. Bellemans et al., “Successful treatment of painful irreparable partial meniscal defects with a polyurethane scaffold: two-year safety and clinical outcomes,” American Journal of Sports Medicine, vol. 40, no. 4, pp. 844–853, 2012.

P. Angele, R. Kujat, M. Koch, and J. Zellner, “Role of mesenchymal stem cells in meniscal repair,” Journal of Experimental Orthopaedics, vol. 1, no. 1, p. 12, 2014.

M. Koch, T. Ehrenreich, G. Koehl et al., “Do cell based tissue engineering products for meniscus regeneration influence vascularization?” Clinical Hemorheology and Microcirculation, vol. 67, no. 2, pp. 125–140, 2017.

W. Petersen, T. Pufe, C. Stärke et al., “Locally applied angiogenic factors—a new therapeutic tool for meniscal repair,” Annals of Anatomy = Anatomischer Anzeiger, vol. 187, no. 5-6, pp. 509–519, 2005.

C. A. Petty and J. H. Lubowitz, “Does arthroscopic partial meniscectomy result in knee osteoarthritis? A systematic review with a minimum of 8 years’ follow-up,” Arthroscopy, vol. 27, no. 3, pp. 419–424, 2011.

C. Liu, I. C. Toma, M. Mastrogiacomo, C. Krettek, G. von Lewinski, and M. Jagodzinski, “Meniscus reconstruction: today’s achievements and premises for the future,” Archives of Orthopaedic and Trauma Surgery, vol. 133, no. 1, pp. 95–109, 2013.

T. Brindle, J. Nyland, and D. L. Johnson, “The meniscus: review of basic principles with application to surgery and rehabilitation,” Journal of Athletic Training, vol. 36, no. 2, pp. 160–169, 2001.

S. C. Mordecai, N. Al-Hadithy, H. E. Ware, and C. M. Gupte, “Treatment of meniscal tears: an evidence based approach,” World Journal of Orthopedics, vol. 5, no. 3, pp. 233–241, 2014.

A. L. McNulty and F. Guilak, “Mechanobiology of the meniscus,” Journal of Biomechanics, vol. 48, no. 8, pp. 1469–1478, 2015.

L. C. Wei, S. G. Gao, M. Xu, W. Jiang, J. Tian, and G. H. Lei, “A novel hypothesis: the application of platelet-rich plasma can promote the clinical healing of white-white meniscal tears,” Medical Science Monitor, vol. 18, no. 8, pp. HY47–HY50, 2012.

S. A. Taylor and S. A. Rodeo, “Augmentation techniques for isolated meniscal tears,” Current Reviews in Musculoskeletal Medicine, vol. 6, no. 2, pp. 95–101, 2013.

W. Niu, W. Guo, S. Han, Y. Zhu, S. Liu, and Q. Guo, “Cell-based strategies for meniscus tissue engineering,” Stem Cells International, vol. 2016, Article ID 4717184, 10 pages, 2016.

F. Achatz, R. Kujat, C. Pfeifer et al., “In vitro testing of scaffolds for mesenchymal stem cell-based meniscus tissue engineering—introducing a new biocompatibility scoring system,” Materials, vol. 9, no. 4, p. 276, 2016.

K. H. Yoon and K. H. Park, “Meniscal repair,” Knee Surgery & Related Research, vol. 26, no. 2, pp. 68–76, 2014.

A. M. Ahmed and D. L. Burke, “In-vitro measurement of static pressure distribution in synovial joints—part I: Tibial surface of the knee,” Journal of Biomechanical Engineering, vol. 105, no. 3, pp. 216–225, 1983.

A. C. T. Vrancken, P. Buma, and T. G. van Tienen, “Synthetic meniscus replacement: a review,” International Orthopaedics, vol. 37, no. 2, pp. 291–299, 2013.

P. Beaufils and N. Pujol, “Meniscal repair: technique,” Orthopaedics & Traumatology, Surgery & Research, vol. 104, no. 1, pp. S137–S145, 2018.

J. R. Steadman, L. M. Matheny, S. B. Singleton et al., “Meniscus suture repair: minimum 10-year outcomes in patients younger than 40 years compared with patients 40 and older,” American Journal of Sports Medicine, vol. 43, no. 9, pp. 2222–2227, 2015.

M. Kimura, K. Shirakura, A. Hasegawa, Y. Kobuna, and M. Niijima, “Second look arthroscopy after meniscal repair: factors affecting the healing rate,” Clinical Orthopaedics and Related Research, vol. 314, no. 314, pp. 185–191, 1995.

Z. Zhang, W. Guo, S. Gao et al., “Native tissue-based strategies for meniscus repair and regeneration,” Cell and Tissue Research, 2018.

J. Pak, J. H. Lee, K. S. Park, J. H. Jeon, and S. H. Lee, “Potential use of mesenchymal stem cells in human meniscal repair: current insights,” Open Access Journal of Sports Medicine, vol. 8, pp. 33–38, 2017.

J. Zellner, C. D. Taeger, M. Schaffer et al., “Are applied growth factors able to mimic the positive effects of mesenchymal stem cells on the regeneration of meniscus in the avascular zone?” Biomed Research International, vol. 2014, Article ID 537686, 10 pages, 2014.

J. Zellner, M. Mueller, A. Berner et al., “Role of mesenchymal stem cells in tissue engineering of meniscus,” Journal of Biomedical Materials Research Part A, vol. 94, no. 4, pp. 1150–1161, 2010.

J. Zellner, K. Hierl, M. Mueller et al., “Stem cell-based tissue-engineering for treatment of meniscal tears in the avascular zone,” Journal of Biomedical Materials Research Part B, Applied Biomaterials, vol. 101, no. 7, pp. 1133–1142, 2013.

P. Angele, B. Johnstone, R. Kujat et al., “Stem cell based tissue engineering for meniscus repair,” Journal of Biomedical Materials Research Part A, vol. 85, no. 2, pp. 445–455, 2008.

J. Zellner, G. Pattappa, M. Koch et al., “Autologous mesenchymal stem cells or meniscal cells: what is the best cell source for regenerative meniscus treatment in an early osteoarthritis situation?” Stem Cell Research & Therapy, vol. 8, no. 1, p. 225, 2017.

H. Yu, A. B. Adesida, and N. M. Jomha, “Meniscus repair using mesenchymal stem cells—a comprehensive review,” Stem Cell Research & Therapy, vol. 6, no. 1, p. 86, 2015.

A. I. Caplan and J. E. Dennis, “Mesenchymal stem cells as trophic mediators,” Journal of Cellular Biochemistry, vol. 98, no. 5, pp. 1076–1084, 2006.

P. R. Baraniak and T. C. McDevitt, “Stem cell paracrine actions and tissue regeneration,” Regenerative Medicine, vol. 5, no. 1, pp. 121–143, 2010.

E. Chew, R. Prakash, and W. Khan, “Mesenchymal stem cells in human meniscal regeneration: a systematic review,” Annals of Medicine & Surgery, vol. 24, pp. 3–7, 2017.

H. Bouyarmane, P. Beaufils, N. Pujol et al., “Polyurethane scaffold in lateral meniscus segmental defects: clinical outcomes at 24 months follow-up,” Orthopaedics & Traumatology: Surgery & Research, vol. 100, no. 1, pp. 153–157, 2014.

T. Efe, A. Getgood, M. D. Schofer et al., “The safety and short-term efficacy of a novel polyurethane meniscal scaffold for the treatment of segmental medial meniscus deficiency,” Knee Surgery, Sports Traumatology, Arthroscopy, vol. 20, no. 9, pp. 1822–1830, 2012.

G. Filardo, E. Kon, F. Perdisa et al., “Polyurethane-based cell-free scaffold for the treatment of painful partial meniscus loss,” Knee Surgery, Sports Traumatology, Arthroscopy, vol. 25, no. 2, pp. 459–467, 2017.

A. Dhollander, P. Verdonk, and R. Verdonk, “Treatment of painful, irreparable partial meniscal defects with a polyurethane scaffold: midterm clinical outcomes and survival analysis,” American Journal of Sports Medicine, vol. 44, no. 10, pp. 2615–2621, 2016.

K. F. Schüttler, F. Haberhauer, M. Gesslein et al., “Midterm follow-up after implantation of a polyurethane meniscal scaffold for segmental medial meniscus loss: maintenance of good clinical and MRI outcome,” Knee Surgery, Sports Traumatology, Arthroscopy, vol. 24, no. 5, pp. 1478–1484, 2016.

E. Kon, G. Filardo, S. Zaffagnini et al., “Biodegradable polyurethane meniscal scaffold for isolated partial lesions or as combined procedure for knees with multiple comorbidities: clinical results at 2 years,” Knee Surgery, Sports Traumatology, Arthroscopy, vol. 22, no. 1, pp. 128–134, 2014.

P. Bulgheroni, E. Bulgheroni, G. Regazzola, and C. Mazzola, “Polyurethane scaffold for the treatment of partial meniscal tears: clinical results with a minimum two-year follow-up,” Joints, vol. 01, no. 4, pp. 161–166, 2017.

S. J. Spencer, A. Saithna, M. R. Carmont, M. S. Dhillon, P. Thompson, and T. Spalding, “Meniscal scaffolds: early experience and review of the literature,” Knee, vol. 19, no. 6, pp. 760–765, 2012.

R. Verdonk, P. Verdonk, W. Huysse, R. Forsyth, and E. L. Heinrichs, “Tissue ingrowth after implantation of a novel, biodegradable polyurethane scaffold for treatment of partial meniscal lesions,” American Journal of Sports Medicine, vol. 39, no. 4, pp. 774–782, 2011.

B. Johnstone, T. M. Hering, A. I. Caplan, V. M. Goldberg, and J. U. Yoo, “In vitro chondrogenesis of bone marrow-derived mesenchymal progenitor cells,” Experimental Cell Research, vol. 238, no. 1, pp. 265–272, 1998.

H. Li, R. Ghazanfari, D. Zacharaki, H. C. Lim, and S. Scheding, “Isolation and characterization of primary bone marrow mesenchymal stromal cells,” Annals of the New York Academy of Sciences, vol. 1370, no. 1, pp. 109–118, 2016.

M. Dominici, K. le Blanc, I. Mueller et al., “Minimal criteria for defining multipotent mesenchymal stromal cells: the International Society for Cellular Therapy position statement,” Cytotherapy, vol. 8, no. 4, pp. 315–317, 2006.

E. M. Horwitz, K. le Blanc, M. Dominici et al., “Clarification of the nomenclature for MSC: the International Society for Cellular Therapy position statement,” Cytotherapy, vol. 7, no. 5, pp. 393–395, 2005.

F. P. Achatz, R. Kujat, C. G. Pfeifer et al., “In vitro testing of scaffolds for mesenchymal stem cell-based meniscus tissue engineering-introducing a new biocompatibility scoring system,” Materials, vol. 9, no. 4, 2016.

W. Hamlet, S. H. Liu, and R. Yang, “Destruction of a cyropreserved meniscal allograft: a case for acute rejection,” Arthroscopy, vol. 13, no. 4, pp. 517–521, 1997.

P. Buma, N. N. Ramrattan, T. G. van Tienen, and R. P. H. Veth, “Tissue engineering of the meniscus,” Biomaterials, vol. 25, no. 9, pp. 1523–1532, 2004.

H. Moffet, C. L. Richards, F. Malouin, G. Bravo, and G. Paradis, “Early and intensive physiotherapy accelerates recovery postarthroscopic meniscectomy: results of a randomized controlled study,” Archives of Physical Medicine and Rehabilitation, vol. 75, no. 4, pp. 415–426, 1994.

E. Eriksson and T. Häggmark, “Comparison of isometric muscle training and electrical stimulation supplementing isometric muscle training in the recovery after major knee ligament surgery. A preliminary report,” American Journal of Sports Medicine, vol. 7, no. 3, pp. 169–171, 2016.

H. Magne, I. Savary-Auzeloux, E. Vazeille et al., “Lack of muscle recovery after immobilization in old rats does not result from a defect in normalization of the ubiquitin-proteasome and the caspase-dependent apoptotic pathways,” Journal of Physiology, vol. 589, no. 3, pp. 511–524, 2011.

C. Suetta, L. G. Hvid, L. Justesen et al., “Effects of aging on human skeletal muscle after immobilization and retraining,” Journal of Applied Physiology, vol. 107, no. 4, pp. 1172–1180, 2009.

A. Frizziero, R. Ferrari, E. Giannotti, C. Ferroni, P. Poli, and S. Masiero, “The meniscus tear: state of the art of rehabilitation protocols related to surgical procedures,” Muscles, Ligaments and Tendons Journal, vol. 2, no. 4, pp. 295–301, 2012.

G. A. Hanks, T. M. Gause, W. J. Sebastianelli, C. S. O’Donnell, and A. Kalenak, “Repair of peripheral meniscal tears: open versus arthroscopic technique,” Arthroscopy, vol. 7, no. 1, pp. 72–77, 1991.

R. P. Jakob, H. U. Staubli, K. Zuber, and M. Esser, “The arthroscopic meniscal repair: techniques and clinical experience,” American Journal of Sports Medicine, vol. 16, no. 2, pp. 137–142, 1988.

A. J. Melchiorri, B.-N. B. Nguyen, and J. P. Fisher, “Mesenchymal stem cells: roles and relationships in vascularization,” Tissue Engineering Part B, Reviews, vol. 20, no. 3, pp. 218–228, 2014.

P. Bianco, P. G. Robey, and P. J. Simmons, “Mesenchymal stem cells: revisiting history, concepts, and assays,” Cell Stem Cell, vol. 2, no. 4, pp. 313–319, 2008.

C. T. Vangsness Jr., J. Farr II, J. Boyd, D. T. Dellaero, C. R. Mills, and M. LeRoux-Williams, “Adult human mesenchymal stem cells delivered via intra-articular injection to the knee following partial medial meniscectomy: a randomized, double-blind, controlled study,” Journal of Bone & Joint Surgery, vol. 96, no. 2, pp. 90–98, 2014.

E. M. Horwitz, D. J. Prockop, L. A. Fitzpatrick et al., “Transplantability and therapeutic effects of bone marrow-derived mesenchymal cells in children with osteogenesis imperfecta,” Nature Medicine, vol. 5, no. 3, pp. 309–313, 1999.

R. Peçanha, L. L. Bagno, M. B. Ribeiro et al., “Adipose-derived stem-cell treatment of skeletal muscle injury,” Journal of Bone & Joint Surgery, vol. 94, no. 7, pp. 609–617, 2012.

Y. Moriguchi, K. Tateishi, W. Ando et al., “Repair of meniscal lesions using a scaffold-free tissue-engineered construct derived from allogenic synovial MSCs in a miniature swine model,” Biomaterials, vol. 34, no. 9, pp. 2185–2193, 2013.

D. Hatsushika, T. Muneta, M. Horie, H. Koga, K. Tsuji, and I. Sekiya, “Intraarticular injection of synovial stem cells promotes meniscal regeneration in a rabbit massive meniscal defect model,” Journal of Orthopaedic Research, vol. 31, no. 9, pp. 1354–1359, 2013.

W. Shen, J. Chen, T. Zhu et al., “Intra-articular injection of human meniscus stem/progenitor cells promotes meniscus regeneration and ameliorates osteoarthritis through stromal cell-derived factor-1/CXCR4-mediated homing,” Stem Cells Translational Medicine, vol. 3, no. 3, pp. 387–394, 2014.

W. Shen, J. Chen, T. Zhu et al., “Osteoarthritis prevention through meniscal regeneration induced by intra-articular injection of meniscus stem cells,” Stem Cells and Development, vol. 22, no. 14, pp. 2071–2082, 2013.

J. Pak, J. H. Lee, and S. H. Lee, “Regenerative repair of damaged meniscus with autologous adipose tissue-derived stem cells,” BioMed Research International, vol. 2014, Article ID 436029, 10 pages, 2014.

S. A. Maher, S. A. Rodeo, H. G. Potter, L. J. Bonassar, T. M. Wright, and R. F. Warren, “A pre-clinical test platform for the functional evaluation of scaffolds for musculoskeletal defects: the meniscus,” HSS Journal, vol. 7, no. 2, pp. 157–163, 2011.

D. Seol, C. Zhou, M. J. Brouillette et al., “Characteristics of meniscus progenitor cells migrated from injured meniscus,” Journal of Orthopaedic Research, vol. 35, no. 9, pp. 1966–1972, 2017.

R. L. Mauck, G. J. Martinez-Diaz, X. Yuan, and R. S. Tuan, “Regional multilineage differentiation potential of meniscal fibrochondrocytes: implications for meniscus repair,” Anatomical Record: Advances in Integrative Anatomy and Evolutionary Biology, vol. 290, no. 1, pp. 48–58, 2007.

P. Angele, B. Johnstone, R. Kujat et al., “Stem cell based tissue engineering for meniscus repair,” Journal of Biomedical Materials Research. Part A, vol. 85A, no. 2, pp. 445–455, 2008.

K. F. Schüttler, S. Pöttgen, A. Getgood et al., “Improvement in outcomes after implantation of a novel polyurethane meniscal scaffold for the treatment of medial meniscus deficiency,” Knee Surgery, Sports Traumatology, Arthroscopy, vol. 23, no. 7, pp. 1929–1935, 2015.

G. Filardo, L. Andriolo, E. Kon, F. de Caro, and M. Marcacci, “Meniscal scaffolds: results and indications: a systematic literature review,” International Orthopaedics, vol. 39, no. 1, pp. 35–46, 2015.

J. Zellner, M. Mueller, A. Berner et al., “Role of mesenchymal stem cells in tissue engineering of meniscus,” Journal of Biomedical Materials Research. Part A, vol. 94A, pp. 1150–1161, 2010.

N. G. Singer and A. I. Caplan, “Mesenchymal stem cells: mechanisms of inflammation,” Annual Review of Pathology: Mechanisms of Disease, vol. 6, no. 1, pp. 457–478, 2011.

A. I. Caplan, “Adult mesenchymal stem cells for tissue engineering versus regenerative medicine,” Journal of Cellular Physiology, vol. 213, no. 2, pp. 341–347, 2007.

W. Petersen, T. Pufe, C. Stärke et al., “The effect of locally applied vascular endothelial growth factor on meniscus healing: gross and histological findings,” Archives of Orthopaedic and Trauma Surgery, vol. 127, no. 4, pp. 235–240, 2007.

S. Ashraf, H. Wibberley, P. I. Mapp, R. Hill, D. Wilson, and D. A. Walsh, “Increased vascular penetration and nerve growth in the meniscus: a potential source of pain in osteoarthritis,” Annals of the Rheumatic Diseases, vol. 70, no. 3, pp. 523–529, 2011.

A. J. S. Fox, A. Bedi, and S. A. Rodeo, “The basic science of human knee menisci: structure, composition, and function,” Sports Health, vol. 4, no. 4, pp. 340–351, 2012.

Z. Zhang, J. A. Arnold, T. Williams, and B. McCann, “Repairs by trephination and suturing of longitudinal injuries in the avascular area of the meniscus in goats,” American Journal of Sports Medicine, vol. 23, no. 1, pp. 35–41, 1995.

F. N. Ghadially, I. Thomas, N. Yong, and J. M. Lalonde, “Ultrastructure of rabbit semilunar cartilages,” Journal of Anatomy, vol. 125, Part 3, pp. 499–517, 1978.

S. P. Arnoczky and R. F. Warren, “Microvasculature of the human meniscus,” American Journal of Sports Medicine, vol. 10, no. 2, pp. 90–95, 1982.

A. P. Newman, D. R. Anderson, A. U. Daniels, and M. C. Dales, “Mechanics of the healed meniscus in a canine model,” American Journal of Sports Medicine, vol. 17, no. 2, pp. 164–175, 2016.