Sleep Disorders

Volume 2013 (2013), Article ID 840723, 6 pages

http://dx.doi.org/10.1155/2013/840723

Clinical Study

The SomnuSeal Oral Mask Is Reasonably Tolerated by Otherwise CPAP Noncompliant Patients with OSA

N. Katz,¹ Y. Adir,² T. Etzioni,3,4,5 E. Kurtz,³ and G. Pillar4,5

¹Sleep Laboratory, Assuta Medical Services and Wolfson Hospital, 69710 Holon, Israel

²Pulmonary Unit, Carmel Hospital and Clalit Health Care, 34362 Haifa, Israel

³Sleep Clinic, Clalit Health Care, 34362 Haifa, Israel

⁴Sleep Laboratory, Rambam Health Care Campus, P.O. Box 9602, 31096 Haifa, Israel

⁵Department of Pediatrics, Carmel Hospital and Faculty of Medicine, Technion-Israel Institute of Technology, 34362 Haifa, Israel

Received 5 June 2013; Revised 31 August 2013; Accepted 14 September 2013

Academic Editor: Liborio Parrino

Copyright © 2013 N. Katz et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

Abstract.

Compliance with CPAP is the major limiting factor in treating patients with OSA. The novel SomnuSeal mask is an oral self-adaptable mask located between the teeth and the lips ensuring that there are no air leaks or skin abrasions. Fifty patients with , who failed previous CPAP trials, were asked to sleep with the mask for one month. In all patients, the mask was connected to an AutoPAP machine with a heated humidifier. Efficacy, convenience, and compliance (average usage for 4 or more hours per night) were monitored. Fifty patients (41 m and 9 f, mean age years, BMI  kg/m², and AHI /h) participated. Eleven were classified as compliant (average mask usage of 26 nights, 4.7 hours per night), five were only partially compliant (average usage of 13 nights, 2.9 hours per night), and 34 could not comply with it. In all patients who slept with it, the efficacy (assessed by residual AHI derived from the CPAP device) was good with an AHI of less than 8/hour. Interestingly, the required optimal pressure decreased from an average of 9.3 cmH₂O to 4.6 cmH₂O. The SomnuSeal oral interface is effective and may result in converting noncompliant untreated patients with OSA into well-treated ones.

1. Introduction

Obstructive sleep apnea (OSA) is a common disorder characterized by recurrent hypoxemia, hypercapnia, and arousal from sleep and is associated with adverse neurocognitive and cardiovascular sequelae [1–6]. Application of continuous positive airway pressure (CPAP) leads to improvements in many of these adverse parameters [7–9], although residual sleep disordered breathing may still persist [10, 11].

The major limiting factor of CPAP treatment is compliance [12–14].

Some of the most important factors that have been reported as limiting compliance are skin abrasions or eruptions due to the pressure exerted by the mask, mask pressure on the ridge of the nose, claustrophobia, aerophagia, air leaks (eye irritation), dry mouth, dry nose, nasal stuffiness, epistaxis, sinusitis, facial pain or a noisy device, or pressure intolerance [15–21]. Other factors that have been identified as affecting compliance consist of disease severity, daytime sleepiness, motivation, age, socioeconomic status, education, race, marital status, spouse support, and copayment [12–31]. Even with the advanced and newer devices (such as the “C-Flex” CPAP device, BPAP, or automatic CPAP), data are not convincing for improved compliance [32–35].

Since CPAP treatment has a dramatic beneficial impact on patients [7–9, 23, 36–39], it is of great importance to seek interfaces that can improve compliance. The novel SomnuSeal mask (Figure 1) is an oral self-adaptable mask located between the teeth and the lips, ensuring that there are no air leaks or skin abrasions. It is more comfortable, adjusts better to the patient’s specific anatomical structure, and potentially reduces rejection by claustrophobic patients. In a series of preliminary studies (published as abstracts [40, 41]), it has been shown to potentially improve compliance in struggling or otherwise CPAP noncompliant patients. These preliminary studies were conducted on a relatively small number of patients and for relatively short periods of time (one night up to one week of use). However, the results were encouraging, indicating that up to 40% of patients with moderate-severe OSA may comply and tolerate the SomnuSeal mask. We speculated that the oral mask will be a second and not a first line of treatment, and therefore this study was planned to examine a longer period of efficacy and compliance with the SomnuSeal interface (one month of treatment), only in CPAP noncompliant patients with moderate-severe OSA.

Figure 1: SomnuSeal mask.

2. Methods

2.1. Participants

Fifty patients were recruited with moderate-severe OSA who were otherwise untreated. All patients were established as noncompliant, having tried at least one CPAP mask previously and failed to comply with it or with any previous CPAP mask they had tried. Inclusion criteria consisted of a previous diagnosis of OSA with an AHI > 20/h, age above 18 years (males or females), failure of at least one CPAP mask in the past, not considering any other treatment (i.e., patients who are otherwise untreated), and consent to participate. All patients were recruited from the patients’ registry (archives) of our sleep clinic and gave written informed consent prior to participation. The study was approved by the Rambam Health Care Campus Institutional Review Board (RMB-0010-11).

Exclusion criteria consisted of any unstable medical condition, active malignancy, age below 18 years, pregnant or lactating women, treatment for OSA other than CPAP (i.e., patients who use dental appliance or consider surgery), and periodontal diseases or mouth lesions according to the investigator’s judgment.

2.2. Baseline Polysomnographic Study

All participants had a baseline full night sleep study using electroencephalogram, electrooculogram, submental electromyogram, bilateral anterior tibialis electromyography, electrocardiogram (ECG), nasal-oral airflow (thermistors and nasal pressure), chest and abdominal wall motion (piezo- or impedance belts), body position, and arterial oxygen saturation (Embla, Broomfield, USA). Sleep staging and respiratory indices were scored by a trained technician. Apnea was defined as a ≥90% decrease in airflow persisting for at least 10 sec, while hypopnea was defined as a ≥50% decrease in the airflow amplitude (relative to baseline, persisting for at least 10 sec) with an associated ≥3% oxygen desaturation or an arousal. The apnea hypopnea index (AHI) was calculated as the number of respiratory events (apnea and hypopnea) divided by the total sleep time.

2.3. CPAP Devices and Interface

All participants were fitted in the clinic with CPAP and a SomnuSeal oral mask and were instructed to sleep with it on a nightly basis at home. In the clinic visit a trained sleep technician/respiratory therapist fitted them with the equipment. They were given two sizes of oral masks and were allowed to decide which one was more comfortable.

The SomnuSeal is an intraoral CPAP interface that provides the needed seal of the oral cavity from outside air (atmosphere) without the shortcomings of other intra- or extraoral masks (Figure 1). The mask is composed of a central part that delivers the compressed air directly to the oral cavity without impingement on intraoral tissues. This central part is surrounded by a special soft silicone part that can engage gently the delicate intraoral tissues in such a way as as to create a peripheral seal so that the intra-oral cavity is secluded from outside atmosphere. The interface is held in place by engagement of the central part with the lingual side of the lips. A nasal peg was not needed. After fitting the mask to a patient, it was connected to an AutoPAP machine (Winemann WM 27460/S). This is a CPAP with a heated humidifier and a usage meter which was reviewed after one month of participation (see below). In addition to hours of use, the device recorded the provided pressure, residual sleep disordered breathing events, and air leaks.

2.4. Study Design

The study consisted of two clinic visits and three telephone visits. In the first clinic visit, the selected patients reported to the clinic and signed an informed consent. A brief physical examination focusing on their oral cavity was then performed by a sleep physician. They were then fitted with the SomnuSeal mask, and, for about 30 minutes, they exercised breathing with it whilst being awake. They were trained for correct mask usage, how to connect it to the AutoPAP machine themselves, how to handle the mask and device in home, and how to wash the mask. They then took the equipment home and were instructed to use it every night for a one-month period.

During the month of the study, telephone visits were performed on a weekly basis. After one, two, and three weeks of use, a research assistant called each participant for an update. In the call, the participants reported their level of satisfaction and subjective tolerance of the mask. In cases when they needed some extra assistance with the interface or CPAP, the study technician discussed it with them, and upon their request they were also offered an actual meeting with him.

After one month of use, the second clinic visit (end of study visit) took place, in which another brief physical examination of the subjects was conducted (again focusing on the oral cavity). At that time the subjects were asked to complete a questionnaire evaluating their use of the mask, and the CPAP meter and output were downloaded and examined by the researcher. The outcome measures included the data retrieved from the CPAP objective usage meter so the actual time it was used by the patient was objectively quantified. In addition, efficacy was quantified based on the CPAP internal recorder (i.e., optimal pressure, residual AHI, and leaks). A subjective satisfactory questionnaire was completed by each participant at that visit. Data were collected regarding the convenience of using the mask, difficult/ease of using and handling it, changes in these parameters with time (from the first to last week of usage), potential side effects, self-assessment of usage, and free text of subjective judgment of the mask.

Data from the patients’ charts, demographic data, usage meter data driven from the CPAP machine, and data from the satisfaction questionnaires were all collected in a data management sheet (Excel) for statistical analyses.

3. Results

Fifty patients participated in the study. Their mean age was years. Their average AHI was /hour, and their average BMI (body mass index) was  kg/m². Forty-one patients were males and 9 were females.

The patients were categorized into three subgroups according to their compliance with CPAP: compliant patients (usage of the CPAP for more than 70% of the nights, for four hours per night or more), struggling or partially compliant patients (who used it intermittently but in less than required to be considered as a compliant patient), and noncompliant patients (who could not tolerate the device). These results are summarized in Table 1.

Table 1: Data categorized according to compliance with CPAP.

Compliant
Strugglers
Noncompliant
All

(% of all)
11 (22)
5 (10)
34 (68)
50 (100)

Number of males (%)
8 (73)
5 (100)
28 (82)
41 (82)

Age

BMI

Pretreatment AHI (/h)

Pretreatment minimal O2 Sat

Days of usage

Average usage per night (h)

Residual obstructive AHI (/h)

n/a
n/a

Average heart rate

Total sleep time (min)

Sleep efficiency

Stage 2 (% of TST)

Stage 3 (% of TST)

REM sleep (% of TST)

The major results obtained from the satisfaction questionnaire are summarized in Table 2 (same classification as in Table 1). As can be seen (and expected), patients who complied with the SomnuSeal mask expressed it in their satisfaction, complained less of excess salivation, and indicated a high index of willingness to obtain the mask.

Table 2: Results from the satisfaction questionnaire. In all questions, scale runs between 1 (weak, bad) and 5 (strong, excellent).

Compliant
Strugglers
Noncompliant
All

Air seal with the SomnuSeal

Quality of airflow

Excess salivation

Comparison with previous tried masks (higher is better)

Willingness to purchase (higher is better)

The two most importantly reported side effects were excess salivation and inconvenience of the mask in the mouth due to pressure on the lips. However, it should be stated that, in the physical examination at the end of the study, none had any ulcers or bruises on their lips or gingival.

Of note, the automated supplied air pressure provided by the autoPAP dramatically decreased with the SomnuSeal compared to the nasal mask. While with the nasal mask the average required pressure was  cmH₂O, the pressure required with the SomnuSeal mask was  cmH₂O.

4. Discussion

The major finding of our study is that 22% of patients with moderate-severe OSA, who failed any other treatment and were noncompliant to at least one CPAP mask which they tried and who otherwise remained untreated, managed to comply with the SomnuSeal oral mask for at least one month of use. They used it on average for 26 nights, with an average of 4.7 hours per night. This can potentially convert them from untreated to reasonably treated patients.

The advantages of CPAP treatment in patients with OSA are well documented. CPAP has been shown to reduce insulin resistance, improve blood pressure, reduce stroke, improve endothelial function, reduce health care utilization, reduce road accidents, improve cognition, improve mood, and improve quality of life [7–9, 23, 36–39]. Despite all of these factors, about 40% of patients who need CPAP remain untreated. The leading cause for CPAP failure is that the device is not tolerated by patients. There are many reported reasons for this noncompliance [15–21, 42]. Since the SomnuSeal mask is placed in the mouth and not on the nose, it eliminates potential limiting factors such as skin abrasions or eruptions due to pressure exerted by nasal masks, mask pressure on the ridge of the nose, eye irritation due to air leaks, dry nose, nasal stuffiness, epistaxis, and potentially sinusitis. It is less cumbersome so even claustrophobia may not be as dramatic as it is with nasal mask. Thus, it is plausible that, in the 22% of patients who complied with the oral SomnuSeal mask and not with nasal mask, these were the limiting factors. Obviously, 22% of patients are not enough, but it should be kept in mind that the participants in this study were otherwise not treated at all! Thus, 22% is a substantially positive number.

The finding that a lower pressure is required for the SomnuSeal mask compared to the nasal mask is very interesting and not intuitive. Potential explanations for this finding consist of stabilizing respiration, and maybe even more relevant, a forward displacement of the lower jaw as occurs with oral appliances [43]. In the current study we did not perform PSG studies on the treated patients and did not assess hypercapnia or respiratory control variables. As a result, we can only speculate that respiratory stabilization had occurred. In addition, we have not assessed anatomical changes of jaw position, and thus the potential anterior mandibular displacement with the SomnuSeal mask is just a speculation. Regardless of the reason for improvement in respiration with relatively low pressure, it may have beneficial effects on the cardiovascular system. Bradley et al. [44], in a study of CPAP in congestive heart failure, showed that, at a CPAP of 5 cmH₂O, the cardiac index and stroke volume indices were increased in the subgroup with poor baseline hemodynamics and higher LV diastolic pressures [44]. Clearly, long-term studies to show any difference in cardiac function with differing CPAP pressures are needed, but it is reasonable to believe that reduction in the required optimal CPAP pressure as observed with the SomnuSeal mask may be beneficial for patients with OSA, especially those with cardiac dysfunction.

Our study has several limitations. Firstly, from the efficacy point of view, our study was a home and not a lab study. Although not tested with oral masks, but with nasal masks, there are very strong data indicating that the automatic algorithm of the CPAP is very accurate and that most patients sleeping with autoPAP have an AHI of less than 10/h. In order to reduce costs, we conducted our study in home and relied on the residual respiratory event counter of the CPAP itself. Obviously, this is a limitation, and future studies will need to be conducted in the lab. Secondly, we did not quantify outcome measures such as vigilance, mood, cognitive function, or medical outcome such as blood pressure or glucose control. This was beyond the scope of our study. The primary aim was to examine the tolerability and compliance of treated patients with this oral mask. Future studies will have to deal with behavioral medical and cognitive outcome with this interface. Thirdly, as with many previous CPAP compliance studies, this was a specific clinical study and not a field study. It is plausible that some of our patients may have made an extra effort to tolerate the SomnuSeal and it does not guarantee that they would have complied with this device in a real-life field setting. Such a study would be possible only if patients start using the SomnuSeal interface on a clinical basis. Finally, the of this study is not huge. For a CPAP compliance study, more than 11 users (of 50 potential users) would be needed. We consider this study as a preliminary study with encouraging results.

In conclusion, despite these limitations, this prospective open study of 50 noncompliant patients demonstrated that a new self-adapting mask has encouraging results in this challenging group of patients. The finding of lower than average CPAP pressures may confer important long-term cardiovascular benefits.

Conflict of Interests

G. Pillar was a consultant for DMD. The other authors have no financial conflict of interests.

Acknowledgment

This study was supported by a nonrestricted Grant from DMD (Discover Medical Devices), the manufacturer of the SomnuSeal mask.

References

J. E. Remmers, W. J. deGroot, E. K. Sauerland, and A. M. Anch, “Pathogenesis of upper airway occlusion during sleep,” Journal of Applied Physiology Respiratory Environmental and Exercise Physiology, vol. 44, no. 6, pp. 931–938, 1978.

P. Lavie, P. Herer, and V. Hoffstein, “Obstructive sleep apnoea syndrome as a risk factor for hypertension: population study,” British Medical Journal, vol. 320, no. 7233, pp. 479–482, 2000.

P. E. Peppard, T. Young, M. Palta, and J. Skatrud, “Prospective study of the association between sleep-disordered breathing and hypertension,” The New England Journal of Medicine, vol. 342, no. 19, pp. 1378–1384, 2000.

J. Hung, E. G. Whitford, R. W. Parsons, and D. R. Hillman, “Association of sleep apnoea with myocardial infarction in men,” The Lancet, vol. 336, no. 8710, pp. 261–264, 1990.

M. E. Dyken, V. K. Somers, T. Yamada, Z. Ren, and M. B. Zimmerman, “Investigating the relationship between stroke and obstructive sleep apnea,” Stroke, vol. 27, no. 3, pp. 401–407, 1996.

U. Leuenberger, E. Jacob, L. Sweer, N. Waravdekar, C. Zwillich, and L. Sinoway, “Surges of muscle sympathetic nerve activity during obstructive apnea are linked to hypoxemia,” Journal of Applied Physiology, vol. 79, no. 2, pp. 581–588, 1995.

J. Hedner, B. Darpo, H. Ejnell, J. Carlson, and K. Caidahl, “Reduction in sympathetic activity after long-term CPAP treatment in sleep apnoea: cardiovascular implications,” European Respiratory Journal, vol. 8, no. 2, pp. 222–229, 1995.

K. Narkiewicz, M. Kato, B. G. Phillips, C. A. Pesek, D. E. Davison, and V. K. Somers, “Nocturnal continuous positive airway pressure decreases daytime sympathetic traffic in obstructive sleep apnea,” Circulation, vol. 100, no. 23, pp. 2332–2335, 1999.

J. M. Marin, S. J. Carrizo, E. Vicente, and A. G. N. Agusti, “Long-term cardiovascular outcomes in men with obstructive sleep apnoea-hypopnoea with or without treatment with continuous positive airway pressure: an observational study,” The Lancet, vol. 365, no. 9464, pp. 1046–1053, 2005.

D. P. White and T. J. Gibb, “Evaluation of the Healthdyne NightWatch system to titrate CPAP in the home,” Sleep, vol. 21, no. 2, pp. 198–204, 1998.

A. I. Pack, J. E. Black, J. R. L. Schwartz, and J. K. Matheson, “Modafinil as adjunct therapy for daytime sleepiness in obstructive sleep apnea,” American Journal of Respiratory and Critical Care Medicine, vol. 164, no. 9, pp. 1675–1681, 2001.

M. R. DiMatteo, “Variations in patients' adherence to medical recommendations: a quantitative review of 50 years of research,” Medical Care, vol. 42, no. 3, pp. 200–209, 2004.

N. B. Kribbs, A. I. Pack, L. R. Kline et al., “Objective measurement of patterns of nasal CPAP use by patients with obstructive sleep apnea,” American Review of Respiratory Disease, vol. 147, no. 4, pp. 887–895, 1993.

T. E. Weaver, N. B. Kribbs, A. I. Pack et al., “Night-to-night variability in CPAP use over the first three months of treatment,” Sleep, vol. 20, no. 4, pp. 278–282, 1997.

H. M. Engleman, N. Asgari-Jirhandeh, A. L. McLeod, C. F. Ramsay, I. J. Deary, and N. J. Douglas, “Self-reported use of CPAP and benefits of CPAP therapy: a patient survey,” Chest, vol. 109, no. 6, pp. 1470–1476, 1996.

J. Strollo Jr., M. H. Sanders, and C. W. Atwood, “Positive pressure therapy,” Clinics in Chest Medicine, vol. 19, no. 1, pp. 55–68, 1998.

J. L. Pepin, P. Leger, D. Veale, B. Langevin, D. Robert, and P. Levy, “Side effects of nasal continuous positive airway pressure in sleep apnea syndrome: study of 193 patients in two French sleep centers,” Chest, vol. 107, no. 2, pp. 375–381, 1995.

G. N. Richards, P. A. Cistulli, R. G. Ungar, M. Berthon-Jones, and C. E. Sullivan, “Mouth leak with nasal continuous positive airway pressure increases nasal airway resistance,” American Journal of Respiratory and Critical Care Medicine, vol. 154, no. 1, pp. 182–186, 1996.

S. A. Herrejon, A. I. Inchaurraga, and M. Gonzalez, “Spontaneous pneumothorax associated with the use of nighttime BiPAP with a nasal mask,” Archivos de Bronconeumología, vol. 34, no. 10, p. 512, 1998.

R. Alvarez-Sala, S. Diaz, C. Prados, C. Villasante, and J. Villamor, “Increase of intraocular pressure during nasal CPAP,” Chest, vol. 101, no. 5, p. 1477, 1992.

N. N. Jarjour and P. Wilson, “Pneumocephalus associated with nasal continuous positive airway pressure in a patient with sleep apnea syndrome,” Chest, vol. 96, no. 6, pp. 1425–1426, 1989.

N. McArdle, G. Devereux, H. Heidarnejad, H. M. Engleman, T. W. Mackay, and N. J. Douglas, “Long-term use of CPAP therapy for sleep apnea/hypopnea syndrome,” American Journal of Respiratory and Critical Care Medicine, vol. 159, no. 4 I, pp. 1108–1114, 1999.

A. Tarasiuk, G. Reznor, S. Greenberg-Dotan, and H. Reuveni, “Financial incentive increases CPAP acceptance in patients from low socioeconomic background,” PLoS ONE, vol. 7, no. 3, Article ID e33178, 2012.

F. Campos-Rodriguez, N. Peña-Griñan, N. Reyes-Nuñez et al., “Mortality in obstructive sleep apnea-hypopnea patients treated with positive airway pressure,” Chest, vol. 128, no. 2, pp. 624–633, 2005.

T. E. Weaver, “How much is enough CPAP?” Sleep Medicine, vol. 4, article S52, 2003.

L. E. Burke and I. S. Ockene, Compliance in Healthcare and Research, Futura Publishing, Armonk, NY, USA, 2001.

H. M. Engleman, S. E. Martin, and N. J. Douglas, “Compliance with CPAP therapy in patients with the sleep apnoea/hypopnoea syndrome,” Thorax, vol. 49, no. 3, pp. 263–266, 1994.

T. Simon-Tuval, H. Reuveni, S. Greenberg-Dotan, A. Oksenberg, A. Tal, and A. Tarasiuk, “Low socioeconomic status is a risk factor for CPAP acceptance among adult OSAS patients requiring treatment,” Sleep, vol. 32, no. 4, pp. 545–552, 2009.

C. A. Massie, R. W. Hart, K. Peralez, and G. N. Richards, “Effects of humidification on nasal symptoms and compliance in sleep apnea patients using continuous positive airway pressure,” Chest, vol. 116, no. 2, pp. 403–408, 1999.

M. E. Billings, D. Auckley, R. Benca et al., “Race and residential socioeconomics as predictors of CPAP adherence,” Sleep, vol. 34, no. 12, pp. 1653–1658, 2011.

L. Ye, A. I. Pack, G. Maislin et al., “Predictors of continuous positive airway pressure use during the first week of treatment,” Journal of Sleep Research, vol. 21, no. 4, pp. 419–426, 2012.

J. Bakker, A. Campbell, and A. Neill, “Randomized controlled trial comparing flexible and continuous positive airway pressure delivery: effects on compliance, objective and subjective sleepiness and vigilance,” Sleep, vol. 33, no. 4, pp. 523–529, 2010.

J. Pépin, J. Muir, T. Gentina et al., “Pressure reduction during exhalation in sleep apnea patients treated by continuous positive airway pressure,” Chest, vol. 136, no. 2, pp. 490–497, 2009.

D. C. Dolan, R. Okonkwo, F. Gfullner, R. J. Hansbrough, R. J. Strobel, and L. Rosenthal, “Longitudinal comparison study of pressure relief (C-Flex) versus CPAP in OSA patients,” Sleep and Breathing, vol. 13, no. 1, pp. 73–77, 2009.

N. S. Marshall, A. M. Neill, and A. J. Campbell, “Randomised trial of compliance with flexible (C-Flex) and standard continuous positive airway pressure for severe obstructive sleep apnea,” Sleep and Breathing, vol. 12, no. 4, pp. 393–396, 2008.

S. K. Sharma, S. Agrawal, D. Damodaran et al., “CPAP for the metabolic syndrome in patients with obstructive sleep apnea,” The New England Journal of Medicine, vol. 365, no. 24, pp. 2277–2286, 2011.

I. H. Iftikhar, M. F. Khan, A. Das, and U. J. Magalang, “Meta-analysis: continuous positive airway pressure improves insulin resistance in patients with sleep apnea without diabetes,” Annals of the American Thoracic Society, vol. 10, no. 2, pp. 115–120, 2013.

M. A. Martínez-García, F. Campos-Rodríguez, P. Catalán-Serra et al., “Cardiovascular mortality in obstructive sleep apnea in the elderly: role of long-term continuous positive airway pressure treatment: a prospective observational study,” American Journal of Respiratory and Critical Care Medicine, vol. 186, no. 9, pp. 909–916, 2012.

T. G. Weinstock, X. Wang, M. Rueschman et al., “A controlled trial of CPAP therapy on metabolic control in individuals with impaired glucose tolerance and sleep apnea,” Sleep, vol. 35, no. 5, pp. 617–625, 2012.

G. Pillar, S. Suraiya, A. Segev, and M. Majdob, “A novel oral mask improve convenience in CPAP treated OSA patients,” Sleep, vol. 33, article A168, 2010.

G. Pillar, A. Segev, and E. Kurtz, “The SomnuSeal oral mask is reasonably tolerated by otherwise CPAP non compliant patients with OSA,” Sleep, vol. 35, article A164, 2012.

J. C. Borel, R. Tamisier, S. Dias-Domingos et al., “Type of mask may impact on continuous positive airway pressure adherence in apneic patients,” PLoS ONE, vol. 8, no. 5, Article ID e64382, 2013.

V. K. Somers, D. P. White, R. Amin et al., “Sleep Apnea and Cardiovascular Disease: an American Heart Association/American College of Cardiology Foundation scientific statement from the American Heart Association Council for High Blood Pressure Research Professional Education Committee, Council on Clinical Cardiology, Stroke Council, and Council on Cardiovascular Nursing,” Circulation, vol. 118, no. 10, pp. 1080–1111, 2008.

T. D. Bradley, R. M. Holloway, P. R. McLaughlin, B. L. Ross, J. Walters, and P. P. Liu, “Cardiac output response to continuous positive airway pressure in congestive heart failure,” American Review of Respiratory Disease, vol. 145, no. 2, part 1, pp. 377–382, 1992.