The Scientific World Journal

Volume 2014 (2014), Article ID 537260, 7 pages

http://dx.doi.org/10.1155/2014/537260

Review Article

Extracorporeal Ultrasound-Guided High Intensity Focused Ultrasound: Implications from the Present Clinical Trials

Tinghe Yu and Xiao Fu

Key Medical Laboratory of Obstetrics and Gynecology, The Second Affiliated Hospital, Chongqing Medical University, Chongqing 400010, China

Received 13 August 2013; Accepted 16 November 2013; Published 3 April 2014

Academic Editors: M. Altaf, J. Hofmann, and S. Mocellin

Copyright © 2014 Tinghe Yu and Xiao Fu. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

Abstract.

Extracorporeal ultrasound-guided high intensity focused ultrasound (HIFU) has been clinically used for 15 years, and over 36000 cases have been reported. However, there yet lacked a consensus in the clinical values, suggesting the necessity of checking clinical findings. Clinical trials were searched and data reevaluated. HIFU was hardly performed alone; almost all present anticancer means have been applied during an HIFU treatment, and a specific regimen varied between trials; there were heterogeneity and disagreement between trials. The complexity made it difficult to distinguish the effect of HIFU. Based upon evaluable data, the efficacy of HIFU was similar to that of radio frequency, chemoembolization, chemotherapy, radiotherapy, or hormone therapy; a combined therapy did not improve the efficacy. The survival rate of HIFU plus radiotherapy was lower than that of radical surgery in liver cancers. Adverse events had no downtrend in the past years. HIFU was not a standardized procedure where the intensity and insonation mode were modified constantly throughout a treatment, limiting an evaluation from the perspective of ultrasonics. These implied that HIFU should be applied as an alternative at most occasions. The present clinical trials had defects making against the understating of HIFU.

1. Introduction

High intensity focused ultrasound (HIFU) can be focused on the preselected volume within the body without harming overlying tissues, thereby inducing heat and cavitation affecting the target area. HIFU treatment is guided by ultrasound or magnetic resonance image, which is even considered as the future of surgery for the noninvasive essence [1].

Extracorporeal ultrasound-guided HIFU (USgFU) has been clinically used for 15 years. This modality has been applied to cancers of liver, pancreas, and kidney in Europe and Asia and the feasibility has been demonstrated [2, 3]. HIFU has been used to manage many types of benign and malignant diseases in China and some advocates state that it is safe and effective [4, 5]. However, those data are usually published in Chinese and only few trials are released in English, which makes it difficult for scientists outside China to catch HIFU. Over 36000 cases have been reported till 2012. Theoretically, such large quantities of data provide comprehensive information. However, there still lacks a consensus in the clinical values. Indeed, the efficacy in bone tumors has been queried [6]. These suggest that the present clinical findings should be checked.

In the present study, clinical trials were searched and reevaluated and some limitations outlined. Those defects have been limiting the understating of HIFU and should be avoided in following trials. HIFU should be an alternative for a subset of patients in some disease types.

2. Methods

2.1. Searching Clinical Trials

Clinical trials were searched in the Chinese Scientific and Technical Periodicals Database and the China National Knowledge Infrastructure.

2.2. Statistics

Data were extracted and reevaluated. The rates were compared with the chi-square test and multitest with the bootstrap. Statistics were performed with the software SAS (SAS Inst., Cary, NC).

3. Results and Discussion

3.1. General

36454 cases were reported in 846 papers. Most trials were case-series. 66 controlled trials were used for the reevaluation for adopting the response evaluation criteria in solid tumors.

3.2. Therapeutic Regimens

A benign lesion was usually treated with single HIFU. Combined therapies included the use of mifepristone, iohexol, or artery embolization in uterine fibroid and finasteride in prostate hyperplasia. Benign diseases can be managed by partial ablation. An adjuvant was applied as a pre-HIFU means increasing the ablation rate via inducing the shrinkage and devasculation of a lesion, or as a post-HIFU strategy reducing the relapse [7].

HIFU was hardly performed alone in cancers. Almost all present anticancer means have been used during an HIFU treatment (Table 1). Moreover, a regimen varied between trials, even a specific strategy for a specific disease type, such as transarterial chemoembolization (TACE) and chemotherapy. TACE was actually a combined treatment (cutoff of the blood supply, arterial infusion, and systemic chemotherapy), which was a poor reference for drastic variance in the operative technique and ingredients applied [8]; thus, a clinical trial using TACE as the reference cannot provide convincing evidence to support the benefit of the candidate modality. HIFU was not a standardized procedure (the intensity and insonation mode were modified constantly throughout a treatment) [9]. The combination undoubtedly resulted in an intricacy. Thus, it was not surprising that there was disagreement between trials. The complexity made it difficult to distinguish the efficacy of HIFU and to merge data to perform a standard meta-analysis. The reason for the prevalence of TACE in an HIFU treatment remained unclear. That HIFU was performed several weeks after TACE indicated that TACE had caused cell death or irreversible damage before HIFU, suggesting HIFU an enhancer of TACE. Drugs varied in chemotherapy, even for a specific disease type. The more adjuvant a therapy required, the weaker potency the modality itself was. Multiple therapeutic strategies suggested that HIFU should be an alternative at most occasions. These demerits should be considered when designing an HIFU trial. Single therapy with confirmed clinical benefit should be set as the reference.

Table 1: List of the therapeutic strategies applied during HIFU for cancers.

Disease and treatment
Regimen

Liver

    TACE
ADM

BLM

DDP

5-FU + ADM

5-FU + DDP

5-FU + EPI

5-FU + GEM

GEM + Oxa

5-FU + ADM + CBP

5-FU + ADM + MMC

5-FU + CBP + MMC

5-FU + DDP + EPI

5-FU + DDP + MMC

5-FU + EPI + Oxa

ADM + FUDR + MMC

FUDR + Oxa + THP

5-FU + (ADM/THP) + (DDP/Oxa)

5-FU + AMD + DDP + MMC

5-FU + DDP + EPI + MMC

5-FU + MMC + THP + CF

5-FU, (DDP/Oxa), EPI, GEM

5-FU, DDP, EPI, HCPT, MMC

    Chemotherapy
Capecitabine

5-FU + CBP

DDP + GEM

5-FU + ADR + DDP

5-FU + CTX + VCR

ADR + CTX + DDP

5-FU + Oxa + CF

    Radiotherapy
3DCRT

Stereotactic

    Biotherapy
Thymosin α1

Tumor necrosis factor

DCCIK

    Surgery

    Ethanol

    Others
Chlorin e6

Thalidomide

Pancreas

    Chemotherapy
5-FU

Capecitabine

DDP

GEM

5-FU + CF

5-FU + GEM

GEM + Capecitabine

GEM + DDP

GEM + Oxa

5-FU + GEM + CF

    Arterial infusion
GEM

5-FU + GEM

5-FU + DDP + EPI

5-FU + GEM + Oxa

DDP + GEM + Interferon

5-FU + EPI + MMC + CF

    Radiotherapy
3DCRT

    Celiac ganglia destruction

    Biotherapy
Mycobacterium phlei F.U.36

Octreotide

Ovary

    Chemotherapy
BLM + DDP + VCR

CTX + DDP + VCR

    Radiotherapy

Uterine cervix

    Chemotherapy
5-FU + DDP + CF

CTX + BLM + DDP

5-FU + DDP + PTX + CF

    Radiotherapy

Vagina

    Radiotherapy

Bone

    Arterial infusion
DDP

ADM, DDP, MTX, IFO

    Chemotherapy
ADM + DDP

HDMTX + VCR

ADM + MTX + IFO

ADM, DDP, MTX, IFO

    Radiotherapy

Breast

    Chemotherapy
PTX + ADM

PTX + EPI

5-FU + ADM + CTX

5-FU + CTX + EPI

5-FU + CTX + MTX

CTX + EPI + Tegafur

    Hormone therapy
Tamoxifen

    Radiotherapy

    Endoscopic axillary node dissection

Soft tissues

    Chemotherapy
ADM + DDP

DDP + IFO

CBP + VP-16

ADM + DTIC + IFO

CBP + EPI + VCR

    Arterial infusion
DDP

ADM + DTIC + IFO

    Radiotherapy

    Surgery

Retroperitoneal lesions

    Radiotherapy
3DCRT

    Chemotherapy
N/A

    Surgery

Esophagus

    Chemotherapy
DDP + PTX

5-FU + DDP + CF

Stomach

    Chemotherapy
5-FU + DDP + PTX

5-FU + Oxa + CF

Colorectum

    Chemotherapy
5-FU + ADM + MTX

5-FU + Oxa + CF

    Radiotherapy

Kidney

    Biotherapy
Interferon

Prostate

    Hormone therapy
Orchidectomy

Flutamide/bicalutamide

Leuprorelin/goserelin

    Radiotherapy
125I

External beam

    Transurethral resection

Bladder

    Infusion
ADM

HCPT

MMC

    Radiotherapy

5-FU: 5-fluorouracil; ADM: adriamycin; BLM: bleomycin; CBP: carboplatin; CF: calcium folinate; CTX: cyclophosphamide; DDP: cisplatin; DTIC: dacarbazine; EPI: epirubicin; FUDR: floxuridine; GEM: gemcitabine; HCPT: hydroxycamptothecin; IFO: ifosfamide; MMC: mitomycin C; MTX: methotrexate; Oxa: oxaliplatin; PTX: paclitaxel; THP: pirarubicin, VCR: vincristine; VP-16: etoposide.

3.3. Efficacy

Most trials were case-series. The efficacy can be objectively assessed in controlled trials. Based upon evaluable controlled trials, the benefit of HIFU was similar to that of surgery or drugs in benign disease types and to radio frequency, TACE, chemotherapy, radiotherapy, or hormone therapy in cancers (Table 2). HIFU therefore should be restricted to a subgroup. A guideline should be developed to determine those cases, and the ratios of harm-benefit and cost-benefit should be assessed.

Table 2: The efficacy of HIFU in controlled trials.

Disease and treatment (number of cases)
Response
Reference

Uterine fibroid

    HIFU (72)

    Myomectomy (74)
87.5% versus 94.6% (P = 0.1282)
[10]

    HIFU (49)

    Mifepristone (53)
Tumor shrinkage

95.9% versus 90.6% (P = 0.2770)

Symptom relief

93.9% versus 96.2% (P = 0.5824)
[11]

    Mifepristone (20)

    HIFU (20)

    HIFU + mifepristone (20)
85.0% versus 90.0% versus 95.0% (P = 0.5606)
[12]

Ectopic pregnancy

    HIFU (20)

    Mifepristone + methotrexate (20)
80.0% versus 85.0% (P = 0.6769)

[13]

Chyluria

    HIFU (25)

    Lymphatic disconnection (30)
84.0% versus 83.3% (P = 0.9469)

Relapse

14.3% versus 16.0% (P = 0.8717)
[14]

Liver cancer

    HIFU (20)

    Radio frequency (20)
3-, 6-, 9-, and 12-month survival

80.0%, 61.1%, 42.9%, and 33.3% versus

85.0%, 58.8%, 46.7%, and 36.4%

(P = 0.6769, 1.00, 1.00, 1.00)
[15]

    HIFU (30)

    TACE (30)
0.5-, 1-, and 2-year survival

83.3%, 63.3%, and 40.0% versus

66.7%, 43.3%, and 23.3%

(P = 0.1331, 0.1192, 0.1634)
[16]

    HIFU (40)

    γ-knife (38)
Early response

20.0% versus 39.5% (P = 0.0580)

PVTT

52.5% versus 47.4% (P = 0.6504)

MST

10 versus 11 months* (NS)
[17]

Pancreas cancer

    HIFU (20)

    Chemotherapy (20)
Early response

50.0% versus 30.0% (P = 0.1949)

6-month survival

70.0% versus 50.0% (P = 0.1949)
[18]

    HIFU + 3DCRT (22)

    3DCRT (29)

    HIFU (22)
Early response

63.6% versus 44.8% versus 40.9% (P = 0.2611)

0.5-, 1-, and 2-year survival

95.5%, 59.1%, and 50.0% versus

93.1%, 41.4%, and 24.1% versus

95.5%, 40.9%, and 22.7%

(P = 0.9127, 0.3704, 0.0891)

MST

17.6 versus 12.4 versus 12.3 months
[19]

Prostate cancer

    HIFU + emasculation (21)

    Emasculation (19)
5-year survival

83.3% versus 66.7% (P = 0.4430)

Bone metastasis

27.8% versus 50.0% (P = 0.3053)
[20]

PVTT: portal vein tumor thrombosis; MST: median survival time.

*The raw data were not described.

HIFU plus 3-dimensional conformal radiotherapy (3DCRT) was compared with the radical surgery in resectable liver cancer. The response rate was 97.1% versus 100% () but the surgery had a higher rate of complete response (70.6% versus 94.1%, ). The 1-, 2-, and 5-year survival rates were 91.2%, 85.3%, and 58.8% in the combination and 97.1%, 94.1%, and 82.4% in surgery (, 0.2244, and 0.0314) [21]. The clinical benefit of 3DCRT has been confirmed [22]. HIFU alone therefore may produce a poorer outcome and should be curtailed in patients with a surgical opportunity.

It was difficult to outline the efficacy of a combined therapy because of the disagreement between trials and the variance of a specific therapeutic regimen between trials. Several therapeutic strategies can be evaluated. HIFU was combined with 3DCRT to treat advanced cancer; response rates were 37.5% and 22.7% () in the combination and 3DCRT, respectively [23]. In prostate cancer, the 5-year survival rate of HIFU plus emasculation was similar to that of emasculation (83.3% versus 66.7%, ) [20]. Data can be merged in pancreatic cancer; the response rates were 30.8% in HIFU combined with chemotherapy (gemcitabine plus cisplatin) and 27.1% in chemotherapy (); HIFU plus 3DCRT did not improve the response rate compared with 3DCRT (38.3% versus 24.6%, ). These findings did not demonstrate that HIFU enhanced other regimens. A combined therapy, therefore, should not be recommended at most occasions. HIFU directly ablated tissues, but drugs and radiation deactivated cells via a series of intracellular processes. Cancer type and drug were the determents of the interaction, and HIFU cavitation can decrease the potency of a drug [24, 25]. Thus, only a specific regimen can be introduced during HIFU for a specific case. The efficacy of a combined therapy including the sequence effect should be explored. The improper coadministration of an adjuvant led to unexpected events—drug or radiation induced predamages to noncancerous tissues thereby increasing the risk of untoward effects due to HIFU [26].

3.4. Heterogeneity

HIFU plus 3DCRT was compared with 3DCRT in the management of retroperitoneal metastasis in 2 trials. Response rates of the combination versus 3DCRT were 48.0% versus 28.0% () and 86.4% versus 60.0% (), respectively [27, 28]. Noticeably, the evaluation with merged data indicated that the combination improved the efficacy (66.0% versus 42.2%, ).

The illogicality demonstrated the heterogeneity of present clinical trials. A conventional meta-analysis, therefore, cannot be applied to HIFU. Aforementioned event may be only the tip of an iceberg, considering the diversity of therapeutic strategies. The heterogeneity was related to the cases involved. HIFU was usually applied to patients with an advanced cancer or failure to other treatments (i.e., inhomogeneity between individuals), and the insonation mode needs to be constantly modified throughout a treatment (i.e., inherent inhomogeneity) [9]. Thus, the quality of clinical trials of HIFU was unsatisfactory and it was difficult to eliminate the heterogeneity. The heterogeneity made it difficult to generalize the clinical benefit of HIFU despite 15 years of clinical experience and >36000 reported cases.

The heterogeneity decreased the feasibility of a randomized controlled trial (RCT). Indeed, several recent papers published in English cannot be classed as RCT [29, 30]. This limited the understanding of clinical benefit of HIFU. To debase the heterogeneity was a great challenge for an HIFU clinical trial.

3.5. Adverse Events

Adverse events were described detailedly in few controlled trials. In uterine fibroid, the rates of adverse events were 75.0% in HIFU and 9.5% in surgery () and were 36.0% in radio frequency and 16.0% in HIFU () [10, 31]. The rate in HIFU was lower than that in chemotherapy (20.0% versus 55.0%, ) in pancreatic cancer [18]. The present data cannot support the allegation that HIFU had fewer untoward effects. Adverse events can be evaluated in RCT. The combined regimen made it difficult to identify HIFU-related toxicities, so there lacked consentaneous data yet. This should be considered in following trials.

Adverse events in 18596 reported cases during 2000–2012 were generalized. There was no downtrend in either benign or malignant disease types (Figure 1). A previous trial indicated that the rate of adverse events varied between disease types (up to 280%), and the location of a lesion and the HIFU device were the determinants of side effects; severe toxicities, such as lung embolism, occlusion of the superior mesenteric artery, intrahepatic metastasis, and tumor rupture, occurred; some (metastasis and artery occlusion) were inconsistent with the verdicts established in preclinical studies [26]. Empirically, only a fraction of undesired events were reported [32]. The actual incidence of untoward events will be much higher. Theoretically, adverse events decreased with the development of HIFU devices and the accumulation of clinical experiences. The assumption was not supported with the present data, suggesting that HIFU was still at an early stage.

Figure 1: Chronological analysis of the rate of adverse events of HIFU during 2000–2012. There was no downtrend in either malignant or benign disease types.

It was believed that tissue response was monitored in real time during a USgFU treatment, suggesting fewer adverse events. Indeed, this was not the fact. Because diagnostic ultrasound must be suspended during the release of therapeutic ultrasound, there was a short delay (milliseconds) to acquire tissue information [9]. Consequently, only the insonation outcome was viewed but the tissue changes during insonation were actually invisible, which was a bug of USgFU. Ultrasound with a frequency of 1–5 MHz corresponded to a wavelength of 1.5–0.3 mm (far larger than the cell diameter). The therapeutic precision of HIFU therefore was lower than that of radiation (wavelength of <10 nm), which was decreased further by the nonlinear behavior of ultrasound in tissues. HIFU may destruct tissues outside the focus, leading to untoward damages. The only approach to improve the safety was to understand the tissue-ultrasound interactions. The low precision may be one of the reasons that the survival time in radiotherapy was longer than that in HIFU in pancreatic cancer [19].

3.6. Operative Manner and Parameters

Only the acoustic power was described in most trials, but the intensity, exposure duration, and insonation mode were not described. HIFU utilized heat and cavitation—the intensity was the leading physical parameter for both therapeutic effect and toxicity [9]. Thus, the present clinical data cannot be generalized to evaluate the benefits from the perspective of ultrasonics. HIFU was not a standardized procedure; the insonation manner needed constant modifications throughout a treatment according to tissue responses—a process depending on the experience of physicians. Thus, HIFU was at a high risk of misplay; underdose-insonation reduced the efficacy and overdose led to unexpected tissue damages. The diversity of insonation manner made it difficult to outline the “dose-effect” and “dose-toxicity” relationships of HIFU therapy, even in a specific disease type. As an example, the required ultrasonic energy to necrotize 1 cm³ of human uterine fibroid had a 77× disparity in trials [9].

As a complex medical device, to operate USgFU properly was not easy. USgFU was the integration of diagnostic and therapeutic ultrasound (i.e., relating to both medicine and physics). The manipulation of a USgFU device placed high demands on the operator. Too many man-reliant processes were a critical root of human blunders [32]. This device therefore was in connection with a high risk of improper manipulation leading to poor therapeutic outcomes and side effects. However, no concerns have been given over this plight.

3.7. Summary

The present clinical data did not provide convincing data to support the alleged advantage of HIFU, including the therapeutic efficacy and adverse events. HIFU should be applied as an alternative for a subset of patients, in most disease types. The present clinical trials had flaws thereby limiting the understating of clinical benefits of HIFU. These should be considered in following trials to develop this emerging therapeutic modality.

Conflict of Interests

The authors declare that there is no conflict of interests.

Acknowledgments

This work was supported by Grants from the Natural Science Foundation of China (11174376) and the State Ministry of Education (SRFDP 20135503130002).

References

G. Malietzis, L. Monzon, J. Hand et al., “High-intensity focused ultrasound: advances in technology and experimental trials support enhanced utility of focused ultrasound surgery in oncology,” British Journal of Radiology, vol. 86, Article ID 20130044, 2013.

R. W. Ritchie, T. Leslie, R. Phillips et al., “Extracorporeal high intensity focused ultrasound for renal tumours: a 3-year follow-up,” BJU International, vol. 106, no. 7, pp. 1004–1009, 2010.

H. Y. Sung, S. E. Jung, S. H. Cho et al., “Long-term outcome of high-intensity focused ultrasound in advanced pancreatic cancer,” Pancreas, vol. 40, no. 7, pp. 1080–1086, 2011.

F. Orsi, P. Arnone, W. Chen, and L. Zhang, “High intensity focused ultrasound ablation: a new therapeutic option for solid tumors,” Journal of Cancer Research and Therapeutics, vol. 6, no. 4, pp. 414–420, 2010.

L. Zhang and Z.-B. Wang, “High-intensity focused ultrasound tumor ablation: review of ten years of clinical experience,” Frontiers of Medicine in China, vol. 4, no. 3, pp. 294–302, 2010.

S. S. Bielack, N. Marina, and M. Bernstein, “High-intensity focused ultrasound (HIFU) is not indicated for treatment of primary bone sarcomas,” Cancer, vol. 117, no. 12, p. 2822, 2011.

T. Yu, S. Zhou, and J. Zhang, “Ultrasonic therapy for gynecologic tumors,” Journal of Minimally Invasive Gynecology, vol. 15, no. 6, pp. 667–672, 2008.

K. Stuart, “Chemoembolization in the management of liver tumors,” Oncologist, vol. 8, no. 5, pp. 425–437, 2003.

T. Yu and P. Huang, “Extracorporeal ultrasound-guided high intensity focused ultrasound therapy: present limitations,” African Journal of Pharmacy and Pharmacology, vol. 5, no. 12, pp. 1501–1507, 2011.

Q. Ke, S. Zhou, C. Chen, D. Jiang, and J. Yang, “Treatment of 72 cases of uterine fibroid with high intensity focused ultrasound,” Zhonghua Linchuang Yishi Zazhi, vol. 1, pp. 61–63, 2007.

Q. Jiang, Q. Tang, Y. Wu et al., “A clinical study of the treatment of the uterine leiomyoma with high-intensity focused ultrasound,” Guangxi Yike Daxue Xuebao, vol. 19, pp. 344–345, 2002.

C. Li, “Treatment of uterine fibroid with high intensity focused ultrasound and mifepristone,” Hebei Yiyao, vol. 30, pp. 460–461, 2008.

G. B. He, X. D. Zhou, X. L. Ren et al., “High-intensity focused ultrasound therapy in patients with ectopic pregnancy,” Xibei Guofang Yixue Zazhi, vol. 28, pp. 425–426, 2007.

W. Zeng, K. Yi, J. Xie, J. Li, and W. Song, “Treatment of chyluria with high intensity focused ultrasound or lymphorenal disconnection,” Shiyong Yiyao Zazhi, vol. 27, pp. 595–596, 2010.

H. J. Zhai, L. S. Jiang, and N. Li, “Effect of high intensity focused ultrasound on immunity of patients with advanced primary liver cancer,” Zhonguo Puwai Jichu Yu Linchuang Zazhi, vol. 13, pp. 178–180, 2006.

Y. Huang, W. Zhang, H. Peng, and Y. Wei, “Effect of high intensity focused ultrasound and transarterial chemoembilization on immunity of patients with primary liver cancer,” Zhejiang Linchuang Yixue, vol. 12, pp. 337–339, 2010.

Y. Z. Dang, X. C. Zhang, W. L. Lu et al., “Body gamma knife and high intensity focused ultrasound (HIFU) for the treatment of portal vein tumor thrombosis of hepatocellular carcinoma,” Xindai Zhongliu Yixue, vol. 18, pp. 114–117, 2010.

W. H. Wang, X. L. Lei, H. Y. Wang, and H. Z. Pu, “Comparative study of HIFU and GP regimen in treatment of advanced pancreatic cancer,” Zhongguo Yixue Chuanxin, vol. 7, no. 2, pp. 44–46, 2010.

J. Chen, Z. Xu, F. Pei, N. He, X. Kong, and Q. Qu, “Clinical analysis of 206 patients with advanced pancreatic cancer treated with nonsurgical therapy,” Chinese Journal of Clinical Oncology, vol. 38, no. 23, pp. 1453–1457, 2011.

G. Hu, S. Wu, L. Zhu, H. Shao, F. Dai, and X. You, “Treatment of prostate cancer with high intentiey focused ultrasound combined with emasculation,” Bengbu Yixueyuan Xuebao, vol. 32, pp. 729–730, 2007.

Q. H. Ke, X. Y. Su, S. Q. Zhou, Z. Liu, W. T. Zhang, and J. Y. Yang, “High intensity focused ultrasound combined with 3-dimensional conformal radiotherapy for primary hepatic carcinoma,” Shiyong Aizheng Zazhi, vol. 26, pp. 184–187, 2011.

C. H. Rim, D. S. Yang, Y. J. Park, W. S. Yoon, J. A. Lee, and C. Y. Kim, “Effectiveness of high-dose three-dimensional conformal radiotherapy in hepatocellular carcinoma with portal vein thrombosis,” Japanese Journal of Clinical Oncology, vol. 42, no. 8, pp. 721–729, 2012.

G. L. Sun, Z. H. Ren, and L. F. Li, “Clinical study of thermotherapy of HIFU in combination with 3D-CRT treating advanced primary liver cancer,” Jilin Yiyue, vol. 29, pp. 542–543, 2008.

H. He, T. Yu, and Y. Zhang, “The Interaction between a drug and ultrasound in sonochemotherapy against ovarian cancers,” Ultraschall in der Medizin, vol. 33, no. 3, pp. 275–282, 2012.

T. Yu, Y. Zhang, H. He, S. Zhou, Y. Liu, and P. Huang, “Anticancer potency of cytotoxic drugs after exposure to high-intensity focused ultrasound in the presence of microbubbles and hematoporphyrin,” Molecular Pharmaceutics, vol. 8, no. 4, pp. 1408–1415, 2011.

T. Yu and J. Luo, “Adverse events of extracorporeal ultrasound-guided high intensity focused ultrasound therapy,” PLoS ONE, vol. 6, no. 12, Article ID e26110, 2011.

Y. Lu, D. Chen, A. Han, Z. Zhao, and R. Wang, “Clinical efficacy of treatment of retroperitoneal metastasis with three-dimensional conformal radiotherapy combined high intensity focused ultrasound,” Shiyong Aizheng Zazhi, vol. 23, pp. 313–314, 2010.

F. Kong, “Treatment of retroperitoneal tumors with three-dimensional conformal radiotherapy combined high intensity focused ultrasound,” Shandong Yiyao, vol. 49, no. 38, pp. 79–80, 2009.

C. Li, W. Zhang, R. Zhang, L. Zhang, P. Wu, and F. Zhang, “Therapeutic effects and prognostic factors in high-intensity focused ultrasound combined with chemoembolisation for larger hepatocellular carcinoma,” European Journal of Cancer, vol. 46, no. 13, pp. 2513–2521, 2010.

C. Jin, H. Zhu, Z. Wang et al., “High-intensity focused ultrasound combined with transarterial chemoembolization for unresectable hepatocellular carcinoma: long-term follow-up and clinical analysis,” European Journal of Radiology, vol. 80, no. 3, pp. 662–669, 2011.

X. Meng, J. P. Li, M. Yu et al., “A comparative study of ablation rates of uterine fibroid with different size using radio frequency and high-intensity focused ultrasound,” Zhongguo Yixue Chaosheng Zazhi, vol. 9, pp. 1040–1044, 2012.

U. Wagner, “Risks in the application of medical devices: human factors in the medical environment,” Quality Management in Health Care, vol. 19, no. 4, pp. 304–311, 2010.