ISRN Chromatography

Volume 2013 (2013), Article ID 484592, 7 pages

http://dx.doi.org/10.1155/2013/484592

Research Article

Determination of Ranitidine in Human Plasma by SPE and ESI-LC-MS/MS for Use in Bioequivalence Studies

Karini B. Bellorio,1,2 Maria Isabel R. Alves,¹ and Nelson R. Antoniosi Filho¹

¹Laboratório de Métodos de Extração e Separação (LAMES), Instituto de Química, Universidade Federal de Goiás, Campus Samambaia, CP 131, 74001-970 Goiânia, GO, Brazil

²Núcleo Integrado de Farmacocinética da Universidade Federal de Goiás e Instituto Melon de Estudos e Pesquisas/Instituto de Ciências Farmacêuticas, CP 131, 74001-970 Goiânia, GO, Brazil

Received 9 November 2012; Accepted 16 December 2012

Academic Editors: J.-F. Jen, J. Millership, and A. Sanches Silva

Copyright © 2013 Karini B. Bellorio et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

Abstract.

A method for determining ranitidine in human plasma by ESI-LC-MS/MS was validated, using propranolol as internal standard. The extraction method used was solid phase extraction (SPE). Chromatographic separation was performed in a Chromolith C18 (50 mm × 4.6 mm i.d.) analytical column, which provided good separation of ranitidine and propranolol peaks with an analysis time of 2.5 minutes. Extraction yields of 94.4% for ranitidine and 89.4% for the internal standard were obtained. The lower limit of quantification (LLOQ) was 3.00 ng/mL, and limit of detection (LOD) was 0.05 ng/mL, with linearity ranging from 3.00 to 500 ng/mL. The results, thus, showed that this method is suitable for application in bioequivalence studies of ranitidine in human plasma.

1. Introduction

The chemical structure of ranitidine hydrochloride (Figure 1) consists of a five-membered furan heterocyclic ring and a nitroethenodiaminic group. The molecular mass of ranitidine hydrochloride is 350.9 daltons (g/mol), and its molecular formula is C₁₃H₂₂N₄O₃SHCl corresponding to hydrochloride N[2-[[[5-[(dimethylamine)methyl)-2-furan]methyl]thio]ethyl]-N′-methyl-nitro2-1,1-ethenodiamine. Ranitidine hydrochloride appears as a crystalline powder, practically odorless and white to light yellow. It is very soluble in water, somewhat soluble in alcohol, and slightly soluble in chloroform. Furthermore, it should be stored in hermetically sealed containers and protected from light [1].

Figure 1: Chemical structure of ranitidine hydrochloride.

Ranitidine is widely used in the treatment of gastric pathologies, has a chemical structure that partially resembles histamine, and acts primarily as a competitive histamine inhibitor through the H₂ receptors [2, 3].

Since around two thousand analyses are conducted before bioequivalence studies can be concluded, the search for quick and simple methods to determine drugs in biological matrices has been intense. Thus, several methods for determining ranitidine in human plasma have been reported. Zendelovska and Stafilov [4] developed a method for determining ranitidine and cimetidine in human plasma using HPLC with a diode array detector. In this study, the analysis time was 9 minutes, with a 50.0 ng/mL limit of detection for ranitidine. Famotidine was used as internal standard. In another study using the same detector, an analysis time of 6.5 minutes was obtained in a C18 column (300 mm 4.6 mm id, 5 μm), with a limit of quantification of 20.0 ng/mL [5].

Methods for analyzing ranitidine in plasma by LC-MS/MS have been reported in some studies [6–8]. The limits of quantification ranged from 1.0 ng/mL to 12.2 ng/mL. However, these methods use the protein precipitation procedure. This has the disadvantage of reducing the useful life of the column due to the presence of compounds that are not precipitates or that use liquid-liquid extraction, which requires large volumes of sample and has a higher limit of quantification.

Thus, the object of this paper was to develop an analytical methodology for determining ranitidine in human plasma using the solid phase extraction (SPE) technique and high performance liquid chromatography coupled to mass spectrometry (HPLC-MS/MS) in order to develop a fast, highly specific, and repeatable methodology for use in pharmaceutical bioequivalence studies.

2. Experimental

2.1. Chemicals and Reagents

Ultrapure water generated by the Millipore Milli-Q Gradient 10 ultrapurification system was used in all processes. HPLC-grade reagents were supplied by Merck. We used the (United Stated Pharmacopeia) USP reference standard for ranitidine hydrochloride (lot G) and the Brazilian Pharmacopoeia primary standard for propranolol hydrochloride (lot 1005).

2.2. Instrumentation and Chromatographic Conditions

Chromatographic analyses were performed on a Shimadzu AD Vp HPLC (Shimadzu, Japan) coupled to a Quattro LC (MS-MS) triple quadrupole mass spectrometer (Micromass, UK) equipped with Z-spray operated in electrospray positive (ES +) mode, with argon as the collision gas. The system was controlled by MassLynx software version 3.4 for Windows NT. The chromatographic conditions were determined using a Merck Chromolith C18 analytical column (50 mm 4.6 mm i.d.) with a flow rate of 0.5 mL/min for a mobile phase consisting of aqueous 0.1% formic acid solution : methanol (50 : 50 v/v ) pumped by a VDP-10A LC pump (Shimadzu, Japan), using a 1 : 2 postcolumn split. Solvents were first placed in ultrasound for 3 minutes and then put through a DGU-14A degassing system (Shimadzu, Japan). The system operated at pressures between 100 and 120 bar. The column and autosampler were kept at a temperature of 20°C, and the injection volume was 20 μL.

The Quattro LC mass spectrometer (MS-MS) operated in positive electrospray ionization mode (ES +) and MRM dual-channel mode. The following parameters were used for the electrospray source: capillary voltage 3.00 kV, cone voltage 25.00 V for ranitidine and 30.00 V for propranolol, extractor voltage 3.00 V, RF lens voltage 0.2 V, source temperature 150°C, cone gas flow 0.0 L/h, and desolvation gas flow 439 L/h. The following conditions were established for the quadrupoles of the Quattro LC mass spectrometer: LM1 resolution of 13.0, HM1 resolution of 13.0, ion 1 energy 0.2, second hexapole input 5.0 V, collision energy at 25.00 eV for ranitidine and 17.00 eV for propranolol, 5.0 output, LM2 resolution of 9.0, HM2 resolution 9.0, ion 2 energy 2.5, multiplier at 650 V, 0.5 s dwell time, and 0.1 s delay time.

2.3. Preparation of the Stock and Standard Solutions

The stock solutions of ranitidine and propranolol were prepared at a concentration of 0.5 mg/mL in methanol : water (50 : 50 v/v). These solutions were used to prepare working solutions of ranitidine at concentrations of 30.00, 50.00, 100.00, 500.00, 1000.00, 2000.00, 4000.00, and 5000.00 ng/mL. The working solutions of propranolol were also prepared from stock solutions (0.5 mg/mL) at a concentration of 2000 ng/mL. All solutions were placed in Falcon-type tubes. The tubes with the ranitidine solutions were protected from light with aluminum foil and stored in a freezer at −20°C. The other solutions were stored in a refrigerator at +4°C and replaced daily.

2.4. Sample Preparation

Plasma samples were thawed at room temperature, homogenized by vortexing and centrifuged at 3400 rpm for 4 minutes to precipitate materials suspended in the plasma. Waters Oasis 1-cc HLB solid phase extraction (SPE) cartridges and 30 mg of adsorbent were used. These cartridges were previously conditioned with 1 mL of methanol and 1 mL of ultrapure water for activation of the drug binding sites.

More ever, 250 μL of human plasma, 100 μL of propranolol, and 500 μL of ultrapure water were added to the SPE. The cartridge was then placed inside a clean test tube, and the cartridge-test tube combination was centrifuged at 3400 rpm for 4 minutes for plasma elution and retention of the drug in the cartridge. Subsequently, the eluate was discarded, and the analytes were eluted with 1000 μL of water : methanol solution (20 : 80 v/v) with 0.1% of formic acid.

2.5. Analytical Curve

The analytical curve was prepared at concentration levels of 3.00, 5.00, 10.00, 50.00, 100.00, 200.00, 400.00, and 500.00 ng/mL of ranitidine. Also 250 μL of human plasma, 100 μL of each of the ranitidine solutions, 100 μL of propranolol, and 400 μL of ultrapure water were added to the SPE. Then, the sample preparation procedure was carried out. The sample quality controls (QCs) were prepared at concentrations of 3.00, 5.00, 200.0, and 400.00 ng/mL by spiking of blank plasma and the subsequent extraction process.

2.6. Extraction Recovery

The extraction method’s recovery was determined by comparing ranitidine peak areas in samples that did not undergo the extraction process considering the extraction yield to be 100% and ranitidine peak areas in extracted samples at concentrations of 3.00, 5.00, 10.00, 50.00, 100.00, 200.00, 400.00, and 500.00 ng/mL. For the internal standard, the extraction yield of replicates at a concentration of 200.00 ng/mL was determined.

2.7. Precision and Accuracy

The accuracy and precision of the method were determined by the extraction of eight samples containing the internal standard and the ranitidine analyte at concentrations 3.00, 5.00, 200.00, and 400.00 ng/mL, which were analyzed in replicates. This procedure was carried out intraday and between-day. The approval criterion for relative standard deviation (RSD) was 15%.

2.8. Stability

Stability studies of ranitidine in plasma were performed under three conditions. First, stability was evaluated in three freezing cycles over 24 h at −20°C, followed by thawing. Subsequently, we evaluated the stability of ranitidine which was kept at room temperature for eight hours. Long-term stability was evaluated over a forty-day storage period. The acceptance criteria were a deviation from nominal concentration that was less than or equal to 15% and precision and accuracy less than or equal to 15%.

The long-term stability of the standard drug solutions and the internal standard in the biological liquid at room temperature was evaluated after 12 h of preparation. These solutions were kept frozen at −20°C for seven days before the analyses were carried out.

2.9. Biological Samples

The samples for bioequivalence tests were collected at the Cardiology Institute of Anápolis in the state of Goiás, Brazil. The study involved 24 volunteers and was conducted after approval of the study protocol by the Research Ethics Committee of the Universidade Federal de Goiás. Ranitidine was assessed for bioequivalence in 150 mg tablet form.

Blood samples were collected with a total volume of 10 mL. The blood was centrifuged and the plasma removed. Plasma samples were frozen at −20°C. The collection times were 15 min, 30 min, 45 min, 1:00 h, 1:15 h, 1:30 h, 2:00 h, 2:30 h, 3:00 h, 4:00 h, 5:00 h, 6:00 h, 8:00 h, and 12:00 h. All samples collected were placed in test tubes containing 2 drops of heparin and then sent to the hospital analysis laboratory for plasma extraction.

3. Results and Discussion

3.1. Analyses by HPLC-MS-MS

The chromatographic method which was used allowed the ranitidine and the internal standard (propranolol) to elute with a resolution of 1.23, with a retention time of 1.66 min for ranitidine and 2.13 min for the internal standard (Figure 2). The retention time obtained was similar to that found in a study by Zhang et al. [7] and was lower than that reported by Sun et al. [8] who obtained an elution time of 4.5 minutes for ranitidine.

(a)

(b)

Figure 2: Total ion chromatogram (TIC) for ranitidine and propranolol.

Ranitidine and propranolol were identified through analysis of mass spectra obtained during the first and second ionizations of these analytes, as illustrated in Figure 3. Ranitidine underwent protonation in the first mass spectrometer, generating a quasimolecular ion [M+H]⁺   of 315.1. Through collision with argon gas, this generates the fragment (daughter ion) 129.9 determined in the second mass spectrometer. These mass spectra coincide with those obtained by Schellen et al. [9] and Kataoka et al. [10].

(a)

(b)

(c)

(d)

Figure 3: Mass spectrum of ranitidine and propranolol. (a) The first mass spectrometer. (b) The second mass spectrometer.

The quasimolecular ion generated by the internal standard (propranolol) [M+H]⁺ has of 260.3 which produces the fragment (daughter ion) 116.1 (Figure 3). These spectra coincide with those obtained by Schellen et al. [9] and Xia et al. [11].

4. Method Validation

4.1. Selectivity/Specificity

Figure 4 shows chromatograms of normal blank plasma for ranitidine and the internal standard (propranolol) after the blank plasma samples were tested using the same procedure and under the same conditions as for sample analysis. As can be seen, no metabolites or interferents significant for the retention time of the drug or of the internal standard were found.

(a)

(b)

Figure 4: Chromatograms of normal blank plasma monitoring ranitidine and propranolol.

4.2. Analytical Curve

The mean determination coefficient of the calibration curve for ranitidine was 0.9989. Linearity varied from 3.00 to 500.00 ng/mL. Relative error values ranged from −6.60 to 8.66%, which indicates an analytical curve of excellent linearity, broad linear dynamic range and adequate accuracy. This provides the quantification process with analytical reliability.

The lower limit of quantification (LLOQ) was 3.00 ng/mL, with precision of 4.61% and accuracy of 100.62%. The limit of detection (LOD) was 0.05 ng/mL, a concentration whose signal-to-noise ratio was 3.98. The limit of quantification was lower than the LLOQ obtained by Zhang et al. [7] and Sun et al. [8] and near that obtained by Wang et al. [6] who used the protein precipitation method and obtained an LLOQ of 1 ng/mL. Thus, the precision values are adequate and provide analytical reliability for LLOQ determination and validation.

4.3. Method Recovery

The absolute recovery of ranitidine was evaluated for eight concentrations (). The recoveries were % for a nominal concentration of 3.00 ng/mL; % for 5.00 ng/mL; % for 10.00 ng/mL; % for 50.00 ng/mL; % for 100.00 ng/mL; % for 200.00 ng/mL; % for 400.00 ng/mL; % for 500 ng/mL. The recovery of the internal standard was %, and the mean recovery of ranitidine in plasma by SPE was 94.41%, a result that demonstrates the effectiveness of the method in bioequivalence studies.

4.4. Accuracy and Precision

The intra-day and between-day accuracy and precision results for the determination of ranitidine in human plasma are presented in Table 1. The results indicate that the proposed method provides intra-day and between-day accuracy and precision within the parameters desirable for an analytical method for the study of drug bioequivalence.

Table 1: Intraday and between-day precision and accuracy for the determination of ranitidine in plasma.

Nominal
Concentration (μg/mL)

Intraday
Between-days

Determined
RSD (%)
Accuracy
Determined
RSD (%)
Accuracy

3.00
2.79
9.75
93.2
2.92
4.04
97.3

5.00
4.67
2.99
93.5
4.64
0.99
92.8

200.00
197.19
3.29
98.6
202.43
2.41
101.2

400.00
406.26
3.37
101.6
419.38
2.73
104.8

4.5. Stability

Table 2 presents data on the stability of ranitidine and propranolol under different conditions. The acceptance criteria were a deviation from the nominal concentration and precision less than or equal to 15%. Neither ranitidine nor propranolol was degraded by the influence of various storage temperature fluctuations, which ensures that these samples may undergo several cycles of freezing and thawing with no loss in analyte concentration. In addition, the drugs are not degraded during the period in which the sample remains in the equipment to be analyzed. The long-term stability assessment has shown that the samples remain stable over the forty-day period.

Table 2: Stability of ranitidine and propranolol under different conditions ().

Conditions
Concentration (ng/mL)

Drug
Nominal
Determined mean

Freezing and thawing cycles
Ranitidine
5.06
5.46 1.39%

407.13
429.99 1.17%

Room temperature for 8 h
Ranitidine
4.97
5.05 2.63%

411.44
423.99 2.01%

Propranolol
214
195 7.06%

Storage at −20°C for 40 days
Ranitidine
4.8
5.28 3.23%

390.79
400.33 4.05%

Standard solutions for 12 h
Ranitidine
5.00
5.15 0.87%

404.66
429.78 1.89%

Propanolol
185.2
181.2 7.06

4.6. Pharmacokinetic Study

The method was effectively applied in the bioequivalence study. Figure 5 displays mean plasma concentration of ranitidine versus time, comparing the test drug (T) to the reference (R), for the 24 volunteers who participated in the study. The test ranitidine showed a profile similar to that of the reference drug, with a maximum plasma concentration () of 90.0 ng/mL occurring at 1.8 h.

Figure 5: Mean plasma concentration of ranitidine versus time for test (T) and reference (R) products.

For some volunteers, there were two plasma concentration peaks for ranitidine. Several other studies have also reported this [12–14]. Some authors attribute this finding to the mechanism of enterohepatic recirculation [15], while others attribute it to the existence of multiple drug binding sites along the gastrointestinal tract [16].

5. Conclusions

The analytical method for determining ranitidine in human plasma using SPE and LC-MS/MS offers high recovery, excellent linearity, accuracy, and precision, and a lower limit of quantification that is adequate for checking the remaining plasma concentration of ranitidine. In addition, the procedure offers fast execution and chromatographic analysis, with excellent resolution.

This methodology is appropriate for assessing the stability of ranitidine under various conditions. Ranitidine was stable in all the processes used to assess the stability of these drugs in biological matrices. Analysis of these drugs is therefore possible after long periods of adequate storage at −20°C.

Thus, the method is fully applicable to bioequivalence and bioavailability studies of ranitidine.

Conflict of Interests

The authors declare that they do not have a direct financial relation with Merck and Millipore (Milli-Q).

Acknowledgments

The authors acknowledge MCT, FINEP, FUNAPE, and CNPq for the financial support, and CAPES CNPq for the productivity fellowship to N. R. A. Filho (Process no. 309832/2010-1).

References

United States Pharmacopeial Convention, U.S. Pharmacopeia & National Formulary, National Publishing, Philadelphia, Pa, USA, 24th edition, 2000.

M. A. Rocha Júnior, “Histamina e Anti-histamínicos,” in Farmacologia, P. Silva, Ed., p. 54, Guanabara Koogan, Rio de Janeiro, Brazil, 5th edition, 1998.

J. Dawson, D. A. Richards, and R. Stables, “Ranitidine—pharmacology and clinical use,” Journal of Clinical and Hospital Pharmacy, vol. 8, no. 1, pp. 1–13, 1983.

D. Zendelovska and T. Stafilov, “Development of an HPLC method for the determination of ranitidine and cimetidine in human plasma following SPE,” Journal of Pharmaceutical and Biomedical Analysis, vol. 33, no. 2, pp. 165–173, 2003.

T. G. Do Nascimento, E. De Jesus Oliveira, and R. O. Macêdo, “Simultaneous determination of ranitidine and metronidazole in human plasma using high performance liquid chromatography with diode array detection,” Journal of Pharmaceutical and Biomedical Analysis, vol. 37, no. 4, pp. 777–783, 2005.

L. Wang, L.-M. Zhao, Y.-T. Sun, and J.-K. Gu, “Rapid determination of ranitidine in human plasma by liquid chromatography-tandem mass spectrometry and its application to a clinical pharmacokinetic study,” Chemical Research in Chinese Universities, vol. 26, no. 6, pp. 910–914, 2010.

Y. Zhang, N. Mehrotra, N. R. Budha, M. L. Christensen, and B. Meibohm, “A tandem mass spectrometry assay for the simultaneous determination of acetaminophen, caffeine, phenytoin, ranitidine, and theophylline in small volume pediatric plasma specimens,” Clinica Chimica Acta, vol. 398, no. 1-2, pp. 105–112, 2008.

X. Sun, Y. Tian, Z. Zhang, and Y. Chen, “A single LC-tandem mass spectrometry method for the simultaneous determination of four H₂ antagonists in human plasma,” Journal of Chromatography B, vol. 877, no. 31, pp. 3953–3959, 2009.

A. Schellen, B. Ooms, D. Van De Lagemaat, R. Vreeken, and W. D. Van Dongen, “Generic solid phase extraction-liquid chromatography-tandem mass spectrometry method for fast determination of drugs in biological fluids,” Journal of Chromatography B, vol. 788, no. 2, pp. 251–259, 2003.

H. Kataoka, H. L. Lord, and J. Pawliszyn, “Automated in-tube solid-phase microextraction-liquid chromatography-electrospray ionization mass spectrometry for the determination of ranitidine,” Journal of Chromatography B, vol. 731, no. 2, pp. 353–359, 1999.

Y. Q. Xia, R. Bakhtiar, and R. B. Franklin, “Automated online dual-column extraction coupled with teicoplanin stationary phase for simultaneous determination of (R)- and (S)-propranolol in rat plasma using liquid chromatography-tandem mass spectrometry,” Journal of Chromatography B, vol. 788, no. 2, pp. 317–329, 2003.

M. A. Campanero, A. Lopez-Ocariz, E. García-Quetglás, B. Sádaba, and A. De La Maza, “Rapid determination of ranitidine in human plasma by high-performance liquid chromatography,” Chromatographia, vol. 47, no. 7-8, pp. 391–395, 1998.

G. Mullersman and H. Derendorf, “Rapid analysis of ranitidine in biological fluids and determination of its erythrocyte partitioning,” Journal of Chromatography, vol. 381, no. 2, pp. 385–391, 1986.

K. S. Reynolds, M. H. Song, W. D. Heizer, C. B. Burns, D. A. Sica, and K. L. R. Brouwer, “Effect of pancreatico-biliary secretions and GI transit time on the absorption and pharmacokinetic profile of ranitidine in humans,” Pharmaceutical Research, vol. 15, no. 8, pp. 1281–1285, 1998.

R. Miller, “Pharmacokinetics and bioavailability of ranitidine in humans,” Journal of Pharmaceutical Sciences, vol. 73, no. 10, pp. 1376–1379, 1984.

P. Schaiquevich, A. Niselman, and M. Rubio, “Comparison of two compartmental models for describing ranitidine's plasmatic profiles,” Pharmacological Research, vol. 45, no. 5, pp. 399–405, 2002.