ISRN Chromatography

Volume 2013 (2013), Article ID 738397, 10 pages

http://dx.doi.org/10.1155/2013/738397

Research Article

Quality by Design Approach for the Development and Validation of Glipizide, an Antidiabetic Drug, by RP-UPLC with Application to Formulated Forms and Urine

Cijo M. Xavier, Kanakapura Basavaiah, K. B. Vinay, and N. Swamy

Department of Studies in Chemistry, University of Mysore, Manasagangothri, Mysore Karnataka 570006, India

Received 30 November 2012; Accepted 10 January 2013

Academic Editors: M. C. Bruzzoniti, M. Hassan, and A. I. Suarez

Copyright © 2013 Cijo M. Xavier et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

Abstract.

Quality by design (QbD) refers to the achievement of certain predictable quality with desired and predetermined specifications. The objective of this study was to develop and demonstrate an integrated multivariate approach to develop and quantify the constituent concentrations of glipizide (GPZ) drug in its pure and tablet forms. The method was developed using Zorbax Extend C-18 (50 mm × 4.6 mm × 1.8 μm) column with mobile phase consisting of a mixture of phosphate buffer of pH 3.5 and acetonitrile (60 : 40 v/v). The method fulfilled validation criteria and was shown to be sensitive, with limits of detection (LOD) and quantitation (LOQ) of 0.001 and 0.005 μg mL⁻¹, respectively. The percentage relative standard deviations for robustness and ruggedness were observed within the range of 0.1 and 0.99. The calibration graph was linear in the range of 0.005–300 μg mL⁻¹. The applicability of the method was shown by the analysis of formulated drug and spiked urine samples. The proposed method can be used for routine analysis in quality control laboratories for its bulk and formulated product, and this is the first UPLC method reported for the assay of GPZ in bulk, formulated form and urine.

1. Introduction

Quality by design (QbD) is a systematic approach to development that begins with predefined objectives and emphasizes product and process understanding and process control, based on sound science and quality risk management [1, 2]. The objective of the QbD initiative is to demonstrate both understanding and control of pharmaceutical processes to deliver high quality pharmaceutical products while affording opportunities for continuous improvement. QbD delivers a better understanding of method capabilities and limitations and ensures a superior chance of successful downstream method validation and transfer. It has become an important paradigm in the pharmaceutical industry since its introduction by the US Food and Drug Administration [3–8]. The QbD concept can be extended to analytical methods [9–16]. In addition, all international drug administration agencies endorse the QbD approach because it is expected that such performance-based routine methods can be changed within the analytical target profile without regulatory resubmission and approval.

Glipizide (GPZ), chemically known as N-[2-[4-[[[(Cyclohexylamino) carbonyl] amino] sulfonyl]phenyl]ethyl]-5-methylpyrazine carboxamide] (Figure 1) is an oral antihyperglycemic agent [17] used in the treatment of noninsulin-dependent diabetes mellitus [18]. It lowers the blood glucose level in humans by stimulating the release of insulin from the pancreas and helping the body to use insulin efficiently [19]. Among its other qualities, it has been shown (i) to stimulate insulin action through extrapancreatic effects; (ii) to favorably influence the principal pathophysiologic abnormalities, defective secretory dynamics, and target-cell resistance to insulin observed in noninsulin-dependent diabetes; (iii) to improve control of blood glucose; and (iv) to lower the level of plasma glucose and to maintain this effect despite a short half-life [20].

Figure 1: Structure of glipizide.

GPZ is included in the United States Pharmacopoeia [21] and European Pharmacopoeia [22]. The former recommends HPLC method for its assay and the latter recommends titrimetry. In the literature, several HPLC methods are available for the determination of GPZ in plasma [18, 19, 23–30] dosage forms [31–38] and urine [30, 39, 40]. Reported methods for the determination of GPZ in pharmaceuticals include UV spectrophotometry [41–44], HPLC, and capillary electrophoresis [45].

According to literature survey, there are quite a few publications on UPLC method development [46, 47] strategy. The reported methods either lack instability indicating study or used huge costly mass spectra for the determination. However, method development approach with RP-UPLC specifically focused on pharmaceutical development in a QbD environment for GPZ that has not been reported anywhere. Therefore, there is an unmet need to investigate a systematic UPLC method development approach for pharmaceutical development using QbD principles to ensure the quality of the method throughout the product lifecycle.

The primary objective of this study was to implement QbD approach to develop and validate an RP-UPLC method that could separate drug in the bulk, formulated forms and human urine with in-depth understanding of the method and build in the quality during the method development to ensure optimum method performance over the lifetime of the product with a suitable degradation data.

2. Methods

2.1. Materials and Reagents

Pure active ingredient sample of GPZ was kindly supplied by Bal Pharma, Bangalore, India, as gift. GPZ-containing tablets, Dibizide (5 mg) (Micro Labs Limited, Hosur, India) and Glynase-5 (5 mg) (USV Limited, Aurangabad), were procured from the local market. HPLC grade acetonitrile was purchased from Merck Specialities, Mumbai, India. Potassium dihydrogen orthophosphate, benzene, and orthophosphoric acid were from Qualigens, India. Doubly distilled water was used throughout the investigation.

2.2. Chromatographic Conditions and Equipment and Instrumental Parameters

Waters (Waters Chromatography Division, Milford, MA, USA) AQUITY UPLC system with a tunable UV detector was used for the determination of GPZ. Empower 2 software was used to record and evaluate the data collected during and following chromatographic analysis. Shimadzu Pharmaspec 1700 UV/Visible spectrophotometer was used for the initial absorbance measurement. The chromatographic separation was achieved on a Zorbax Extend C-18 () column using a mobile phase consisting of acetonitrile buffer (0.025 M KH₂PO₄ of pH 3.5) with 40 : 60 v/v ratio at a flow rate of 0.2 mL/min. The mobile phase was filtered through 0.22 μm nylon-66 filter prior to use. The eluent was monitored using UV detection at a wavelength of 220 nm. The column was maintained at ambient temperature (25°C) and an injection volume of 2 μL was used. The run time was fixed for 5 minutes and the diluent used was mobile phase.

2.3. Stress Study

In order to establish whether the analytical method and the assay were stability indicating, pure active pharmaceutical ingredient (API) of GPZ was subjected to stress under various conditions to conduct forced degradation studies [48, 49]. As the drug is freely soluble and stable in mobile phase, the same was used as a cosolvent in all forced degradation studies. All solutions used in forced degradation studies were prepared by dissolving GPZ in small volume of diluent and later diluted with aqueous hydrogen peroxide (5%), distilled water, aqueous hydrochloric acid (1N), and aqueous sodium hydroxide (1N) for 2 h. For photolytic degradation studies, pure drug in solid state was exposed to 1.2 million lux hours in a photostability chamber [49]. Additionally, the drug powder was exposed to dry heat at 105°C for 2 h. After the degradation these solutions were diluted with mobile phase to yield starting concentration of 200 μg mL⁻¹ and subjected to UPLC analysis.

2.4. Preparation of Stock Solution

Stock solution of GPZ (1 mg/mL) was prepared in diluent and it was stored at 5°C until use.

2.5. Procedures

2.5.1. Procedure for Preparation of Calibration Curve

Working solutions containing 0.005–300 μg mL⁻¹ GPZ were prepared by serial dilutions of aliquots of the stock solution. Aliquots of 2 μL were injected (six injections) and eluted with the mobile phase under the reported chromatographic conditions. The mean peak areas were plotted against the corresponding concentrations of the pure drug to obtain the calibration graphs and corresponding regression equation was also computed.

2.5.2. Preparation of Tablet Extracts and Assay Procedure

Fifty numbers, each of Glynase and Dibizide tablets (each tablet contained 5.0 mg GPZ), were weighed and powdered. Tablet powder equivalent to 20 mg of GPZ was transferred into 100 mL volumetric flasks and 60 mL of the mobile phase were added. The solution was sonicated for 20 min to achieve complete dissolution of GPZ, made up to the mark with mobile phase and then filtered through 0.22 μm nylon membrane filter. The resultant solution (200 μg mL⁻¹ in GPZ) obtained was analyzed through the UPLC system.

2.5.3. Preparation of Urine Sample

A 1.00 mL of 0.05 M hydrochloric acid was added to 0.50 mL of urine, resulting in a pH of 3. The mixture was extracted with 3.00 mL of benzene in a 12 mL glass tube, which was shaken gently for 15 min. After centrifugation for 5 min, the organic phase was transferred to a conical tube for evaporation to dryness under a stream of a well-ventilated fume chambers. The residue was redissolved in mobile phase and an aliquot of 2 μL was injected into the chromatograph.

2.5.4. Procedure for Method Validation

Accuracy and Precision. To determine the accuracy and intra-day precision, pure GPZ solutions at three different concentrations were analyzed in seven replicates during the same day. Mobile phase was injected as blank solution before sample injection and the RSD (%) values of peak area and retention time were calculated.

Limits of Detection (LOD) and Quantification (LOQ). LOQ and LOD determinations were performed on samples containing very low concentrations of analyte under the ICH guidelines. By applying the signal to noise ratio (S/N) method, LOD was expressed by establishing the minimum level at which the analyte can be reliably detected. LOQ was considered as the lowest concentration of analytes in standards that can be reproducibly measured with acceptable accuracy and precision. Precision study was performed at LOQ. LOQ solution was injected seven times () and calculated the % RSD values for the obtained peak area and retention time.

Linearity. Linearity solutions were prepared from LOQ level to 150% of the actual sample concentration (200 μg mL⁻¹ GPZ). A total of seven concentrations of the solutions were made separately and injected (LOQ: 50, 100, 150, 200, 250, and 300 μg mL⁻¹ levels).

Robustness and Ruggedness. To determine the robustness of the method the experimental conditions were deliberately changed. The flow rate of the mobile phase ( mL min⁻¹), column oven temperature (°C), mobile phase composition ratio (35 : 65, 45 : 55, acetonitrile : buffer), and detection wavelength ( nm) were the varied parameters. In each case, the % RSD values were calculated for the obtained peak area and retention time. The number of theoretical plates and tailing factors were compared with those obtained under the optimized conditions. Three different columns of same dimensions were used for the analyses. The studies were performed on the same day (intra day) and on three different days (inter day) by three different analysts for three different concentrations of GPZ (triplicate injections). The area obtained from each concentration was compared with that of the optimized one. The relative standard deviation values were evaluated for each concentration.

Solution Stability and Mobile Phase Stability. The stability of GPZ in solution was determined by leaving test solutions of the sample and reference standard in tightly capped volumetric flasks at room temperature for 48 h during which they were assayed at 12 h intervals. Stability of mobile phase was determined by analysing freshly prepared sample solutions at 12 h intervals for 48 h and comparing the results with those obtained from freshly prepared reference standard solutions. The mobile phase was prepared at the beginning of the study period and was not changed during the experiment. The % assay of the results was calculated for both the mobile phase and solution-stability experiments.

3. Results

The method development consisted of two phases: the first is screening and the second is optimisation.

3.1. Screening

Solubility of the drug is the main criteria for any method to develop and then the UV response of the product at varying wavelengths. GPZ was found soluble in a mixture of acetonitrile and phosphate buffer. A 200 μgml⁻¹ solution showed excellent response at 220 nm when it was scanned from 400 to 200 nm in a UV spectrophotometer (Figure 2). In UPLC, variables such as mobile phase, buffer, temperature, sample concentration, and pH were varied by keeping one parameter (stationary phase) constant. Summarized observations are made as per the Table 1.

Table 1: Screening: summary of method selection*.

S. no.
Column
Observations
Remarks

1
Acquity BEH C8 (100 × 2.1) mm, 1.7 µm
Asymmetrical peak with fronting and tailing
Not satisfactory

2
Acquity BEH C18 (100 × 2.1) mm, 1.7 µm
Asymmetrical peak with tailing
Not satisfactory

3
Acquity HSS cyano (50 × 2.1) mm, 1.7 µm
Split peak
Not satisfactory

4
Acquity HSS BEH shield RP18 (50 × 2.1) mm, 1.7 µm
Asymmetrical peak
Not satisfactory

5
Zorbax Extend C-18 (50 × 4.6) mm, 1.8 µm
Symmetrical peak
satisfactory

 *By keeping one column constant, the mobile phase, temperature, buffer, and sample concentration parameters were changed.

Figure 2: UV absorption spectra of glipizide.

3.2. Optimisation

Zorbax Extend C-18 () mm, 1.8 μm column, was found suitable and the fine tuning of the method was performed with suitable adjustment of pH. All the trials are as shown in Table 2 and chromatograms are as shown in Figure 3.

Table 2: Final optimization of pH with Zorbax Extend C-18 () mm, 1.8 µm column.

S. no.
Trails taken
Observations
Remarks

1
ACN : buffer (pH 3.5) (40 : 60% v/v)
Peaks found symmetrical
Satisfactory

2
ACN : buffer (pH 4) (40 : 60% v/v)
Peak eluted before 1 min with less theoretical plates
Not satisfactory

3
ACN : buffer (pH 3) (40 : 60% v/v)
Broad peak
Not satisfactory

4
ACN : buffer (pH 5) (40 : 60% v/v)
Broad peak and late elution
Not satisfactory

(a) ACN: buffer (pH 3.5) (40 : 60% v/v)

(b) ACN: buffer (pH 4.0) (20 : 80% v/v)

(c) ACN: buffer (pH 2.2) (20 : 80% v/v)

(d) ACN: buffer (pH 5.5) (20 : 80% v/v)

Figure 3: Method optimization at different pH conditions.

3.2.1. Final Method Conditions

 Column: Zorbax Extend C-18 () mm, 1.8 μm. Oven temp.: 25°C. Mobile phase: ACN: buffer (pH 3.5) (40 : 60% v/v). Run time: 5 min. Flow rate: 0.2 mL/min. Diluent: mobile phase. Injection volume: 2 μL. Blank: diluent. Wavelength: 220 nm.

3.2.2. Validation of the Method

The described method for the assay of GPZ was validated as per the current ICH Q2 (R1) guidelines.

3.2.3. Analytical Parameters

Linearity. A calibration curve was obtained for GPZ from LOQ to 150% of its stock solution. The response of the drug was found to be linear in the investigation concentration range from 0.005 to 300 μg mL⁻¹ and the linear regression equation was with correlation coefficient of 0.9999, where Y is the mean peak area. The LOD and LOQ values and their standard deviations were evaluated and presented in Table 3. These results confirm the linear relation between the mean peak area and concentration as well as the sensitivity of the method.

Table 3: Linearity and regression parameters with precision data.

Parameter
Value

Linear range, µg mL⁻¹
0.005–300

Limits of quantification (LOQ), µg mL⁻¹
0.005

Limits of detection (LOD), µg mL⁻¹
0.001

Regression equation

    Slope ()
11701.20

    Intercept ()
5660.74

    Correlation coefficient ()
0.99999

Accuracy and Precision. The percentage relative error which is an index of accuracy is ≤1.56 and is indicative of high accuracy. The calculated percentage relative standard deviation (%RSD) can be considered to be satisfactory. The peak-area-based and retention-time-based RSD values were found <1. The results obtained for the evaluation of precision and accuracy of the method is compiled in Tables 4 and 5.

Table 4: Results of accuracy study ().

Concentration of GPZ injected, µg mL⁻¹
Intraday
Interday

Concentration of GPZ found, µg mL⁻¹
REa, %
Concentration of GPZ found, µg mL⁻¹
RE, %

100
101.34
1.34
99.47
0.53

200
203.12
1.56
202.21
1.11

300
302.12
0.71
303.45
1.15

 aRelative error.

Table 5: Results of precision study.

Concentration injected (µg mL⁻¹)
Mean area ± SD
RSDa
Mean Rt ± SD
RSDb

Intraday precision ()

100
1165237 ± 2333
0.20
2.13 ± 0.006
0.26

200
2347961 ± 16728
0.71
2.13 ± 0.006
0.28

300
3482365 ± 23172
0.67
2.13 ± 0.002
0.10

Concentration injected (µg mL⁻¹)
Mean area ± SD
RSDa
Mean RT ± SD
RSDb

Interday precision ()

100
1161904 ± 3535
0.30
2.14 ± 0.018
0.82

200
2347961 ± 16728
0.71
2.13 ± 0.006
0.29

300
3488820 ± 7850
0.23
2.14 ± 0.020
0.91

aRelative standard deviation based on peak area. bRelative standard deviation based on retention time.

Robustness and Ruggedness. Analytical methods need to be robust so that they can be used routinely without problems and can be easily transferred for use in another laboratory, if necessary. The intrinsic robustness of the UPLC method depends on a range of variables related to the method parameters, such as preparation of wavelength ( nm), mobile phase composition (actual ±10%), temperature (°C), flow rate ( mL), and so forth. No significant effect was observed on system suitability parameters such as theoretical plates, tailing factor, capacity factor, and % RSD of PGZ, when small but deliberate changes were made to chromatographic conditions. The RSD values ranged from 0.1 to 0.9% resume the robustness of the proposed method. In method ruggedness, different columns (different lots with the same manufacturer), days, and analysts () were performed. The results were summarized in Table 6.

Table 6: Results of method robustness and ruggedness study.

Condition
Modification
Mean peak area ± SD
% RSD
Mean Rt ± SD
% RSD
Theoretical plates ± SD
% RSD
Tailing factor ± SD
% RSD

Actual
—
2351828 ± 11571
0.49
2.130 ± 0.005
0.24
2968 ± 5.16
0.17
1.01 ± 0.005
0.47

Temperature
24°C
2346161 ± 14060
0.60
2.131 ± 0.004
0.19
2962 ± 20.00
0.68
1.10 ± 0.008
0.76

26°C
2345161 ± 16987
0.72
2.131 ± 0.004
0.20
2963 ± 16.02
0.54
1.10 ± 0.01
0.50

Mobile phase composition (acetonitrile : buffer)
35 : 65
2350328 ± 13836
0.59
2.129 ± 0.005
0.24
2960 ± 24.01
0.81
1.11 ± 0.005
0.47

45 : 55
2351828 ± 11572
0.49
2.131 ± 0.004
0.19
2970 ± 4.08
0.14
1.10 ± 0.004
0.370

Flow rate
0.22 mL/min
2346828 ± 14139
0.60
2.133 ± 0.005
0.25
2965 ± 12.11
0.41
1.14 ± 0.008
0.72

0.18 mL/min
2343495 ± 15290
0.65
2.132 ± 0.006
0.26
2968 ± 4.90
0.17
1.11 ± 0.006
0.57

Wavelength
219 nm
2348495 ± 13198
0.56
2.132 ± 0.005
0.24
2970 ± 4.00
0.13
1.10 ± 0.006
0.57

221 nm
2351802 ± 11535
0.49
2.129 ± 0.005
0.23
2960 ± 24.42
0.82
1.10 ± 0.004
0.37

Analyst 1, column 1,

day 1
2350541 ± 11350
0.48
2.129 ± 0.005
0.23
2961 ± 14.95
0.50
1.11 ± 0.008
0.76

Analyst, column, day
Analyst 2, column 2,

day 2
2349495 ± 10475
0.45
2.129 ± 0.005
0.24
2964 ± 15.23
0.51
1.10 ± 0.005
0.47

Analyst 3, column 3,

day 3
2353178 ± 12027
0.51
2.135 ± 0.010
0.47
2968 ± 6.98
0.24
1.11 ± 0.010
0.99

Stability of the Solution. Stability of sample solution was established by storage of sample solution at ambient temperature for 24 h. PGZ sample solution was reanalyzed after 12 and 24 h time intervals and assay was determined and compared against fresh sample. Sample solution did not show any appreciable change in assay value when stored at ambient temperature up to 24 h. At the specified time interval, % RSD values for the peak area obtained from drug solution stability and mobile phase stability were within 1%. This shows no significant change in the elution of the peak and its system suitability criteria (% RSD, tailing factor, theoretical plates).

Selectivity. Selectivity of the method was evaluated by injecting the mobile phase, placebo blank, pure drug solution, and tablet extract. No peaks were observed for mobile phase and placebo blank and no extra peaks were observed for tablet extracts (Figures 4(a) and 4(b)).

(a) Placebo blank

(b) Tablet extract

Figure 4: Chromatograms obtained for placebo blank and tablet extract.

3.3. Application to Tablet Analysis and Urine

A 200 μg mL⁻¹ solution of tablets was prepared as per “preparation of tablet extracts and assay procedure” and was injected in triplicate to the UPLC system. From the mean peak area, the concentration and hence mg/tablet were computed, and the results were compared with those of a reference method [22]. The reference method involved the titration of the sample in dimethyl formamide against 0.1 M lithium methoxide. The accuracy and precision of the proposed method were further evaluated by applying Student’s -test (<2.7) and variance ratio -test (<6.4), respectively. The - and -values at 95% confidence level did not exceed the tabulated values and this further confirms that there is no significant difference between the reference and proposed methods with respect to accuracy and precision. Table 7 illustrates the results obtained from this study.

Table 7: Results of determination of GPZ in formulations and statistical comparison with the reference method.

Formulation brand namea
Nominal amount, mg
% GPZ foundc  ± SD
value
value

Reference method
Proposed method

Dibizidea
5.0
99.58 ± 0.78
98.88 ± 0.68
1.28
1.32

Glynaseb
5.0
100.2 ± 0.57
99.94 ± 0.97
0.53
2.9

 aMarketed by Micro Labs Limited, Hosur, India. bMarketed by USV Limited, Aurangabad, India. cMean value of five determinations. Tabulated value at 95% confidence level is 2.78; tabulated value at 95% confidence level is 6.39.

3.4. Recovery Study

The proposed method was successfully applied to the determination of GPZ in spiked urine sample with mean percentage recovery in the range of 103.1–104.5% as shown in Table 8. A standard addition procedure was followed to evaluate the accuracy of the method. The sample is analyzed for the analyte of interest by adding a specified amount of this analyte to the sample, thus increasing its concentration. The analysis is then repeated and the resulting increase in peak area due to addition of the standard amount is noted. Hence, the concentration of the analyte in the original sample was calculated. The percentage recovery of GPZ from pharmaceutical dosage forms ranged from 100.4 to 101.3%. Detailed results presented in Table 9 reveal good accuracy of the proposed method.

Table 8: Results of determination of GPZ in spiked urine sample.

Spiked concentration µg mL⁻¹
Founda  ± SD
% Recovery ± RSDb

150.23
155.46 ± 3.16
103.48 ± 1.01

200.22
209.24 ± 1.28
104.51 ± 1.02

250.32
258.12 ± 2.45
103.12 ± 0.98

 aMean value of five determinations. bRSD is relative standard deviation.

Table 9: Results of recovery study by standard addition method.

Tablet studied
GPZ µg mL⁻¹, tablet
GPZ µg mL⁻¹, pure
Total GPZ found, µg mL⁻¹
Percent recovery of pure GPZ,

(% GPZ ± SD)

Dibizide
49.21
50
100.23
101.03 ± 0.36

49.21
150
199.93
100.36 ± 0.45

49.21
250
301.87
100.89 ± 0.89

Glynase
48.67
50
99.89
101.24 ± 1.12

48.67
150
201.23
101.29 ± 0.1.02

48.67
250
302.23
101.19 ± 0.98

3.5. Stress Study

The degradation was performed as per the current practice of ICH guidelines [49]. GPZ was found to be sensitive towards oxidation and alkali hydrolysis. The drug degraded up to 25% under oxidation (5% H₂O₂) and 36% with base (0.1N NaOH). The drug was found to be more stable under acidic (0.1N HCl), thermal (105°C for 3 hours), neutral degradation (water), and photolytic (1200 K lux hours) conditions rather than under alkali stress conditions. No significant changes (<1%) were observed for the chromatographic responses for the solutions analyzed with other stress conditions. Following are the % degradation pattern of GPZ in acidic, thermal, neutral, and photolytic conditions: 0.21, 0.01, 0.02, and 0.13. Figure 5 shows the degradation chromatograms of GPZ with the corresponding solvent as blank.

(a) 5% H₂O₂ degradation

(b) 0.1N HCl degradation

(c) 0.1N NaOH degradation

(d) H₂O degradation

(e) Thermal degradation

(f) Photolytic degradation

Figure 5: Chromatograms obtained for forced degradation.

4. Conclusion

A gradient RP-UPLC method was successfully developed for the estimation of PGZ in pharmaceutical dosage form and urine. The QbD project aims to encourage debate about quality in the complete development of the drug in a systematic manner. From a solid-state point of view, QbD implementation involves the design of manufacturing processes based on a thorough scientific understanding of the properties and stability of the components of the drug at critical points throughout the development. The experimental design describes the scouting of the key UPLC method components including column, temperature, pH, and mobile phase. The interrelationships are studied and the preliminary optimized conditions are obtained for each combination. Moreover, this approach ensures better design of products with fewer problems in development, reduces the number of trials required for postmarket changes, relies more on process and understanding and mitigation of risk, allows implementation of new technology to improve manufacturing without regulatory scrutiny, and enables possible reduction in overall costs of manufacturing resulting in less waste. The validated method is specific, linear, precise, accurate, robust, rugged, and stable for 24 hours and can be applied for the determination in formulated form. The drug is stable in acidic, thermal, photolytic, and hydrolytic conditions and degrades in basic, oxidative conditions. The potential of this QbD approach lies for the simultaneous determination of multiple forms of GPZ like pure drug, dosage forms, and in urine, and thus it should be implemented.

Acknowledgments

The authors thank Bal Pharma, Bangalore, India, for gifting pure glipizide. They are also thankful to the University of Mysore for providing the permission and facilities to do research work. Two of the authors (Cijo M. Xavier and K. B. Vinay) are grateful to Jubilant Life Sciences, Nanjangud, Mysore, India, for permission to pursue Ph.D. degree programme.

References

Renu lal, “New drug quality,” FDA/CDER Small Business Chronicle, 2012

“International conference on harmonisation of technical requirements for registration of pharmaceuticals for human use,” Step 4 ICH Q8, (R) step-2, 2009

Y. Tang, “Quality by design approaches to analytical methods—FDA perspective,” http://www.fda.gov/downloads/AboutFDA/CentersOffices/OfficeofMedicalProducts_and_Tobacco/CDER/UCM301056.pdf

“International Conference on Harmonization of technical requirements for registration of pharmaceuticals for human use,” ICH harmonized tripartite guideline, Draft Step 4. Pharmaceutical Development Q8(R1), 2008

L. Yu, “Pharmaceutical quality by design: product and process development, understanding and control,” Pharmaceutical Research, vol. 25, no. 4, pp. 781–791, 2008.

S. L. Lee, A. S. Raw, and L. Yu, “Significance of drug substance physiochemical properties in regulatory quality by design,” in Preformulation in Solid Dosage Form Development, pp. 571–583, Informa Healthcare, London, UK, 2008.

R. Mhatre and A. S. Rathore, “Quality by design: an overview of the basic concepts,” in Quality by Design for Biopharmaceuticals, chapter 1, Wiley, New York, NY, USA, 2009.

F. G. Vogt and A. S. Kord, “Development of quality-by-design analytical methods,” Journal of Pharmaceutical Sciences, vol. 100, no. 3, pp. 797–812, 2011.

Y. Li, G. J. Terfloth, and A. S. Kord, “A systematic approach to RP-HPLC method development in a pharmaceutical QbD environment,” American Pharmaceutical Review, vol. 12, no. 4, 2009.

L. Zhou, J. M. Socha, F. G. Vogt, S. Chen, and A. S. Kord, “A systematic method development strategy for water determinations in drug substance using Karl Fischer titrations,” American Pharmaceutical Review, vol. 13, no. 1, pp. 74–84, 2010.

M. Sun, D. Q. Liu, and A. S. Kord, “A systematic method development strategy for determination of pharmaceutical genotoxic impurities,” Organic Process Research and Development, vol. 14, no. 4, pp. 977–985, 2010.

P. F. Gavin and B. A. Olsen, “A quality by design approach to impurity method development for atomoxetine hydrochloride,” Journal of Pharmaceutical and Biomedical Analysis, vol. 46, no. 3, pp. 431–441, 2008.

M. G. Schweitzer, M. Pohl, M. H. Brown et al., “Implications and opportunities of applying the principles of QbD to analytical measurements,” Pharmaceutical Technology, vol. 22, no. 2, pp. 29–36, 2010.

P. Borman, P. Nethercote, M. Chatfield, D. Thompson, and K. Truman, “The application of quality by design to analytical methods,” Pharmaceutical Technology, vol. 31, pp. 142–152, 2007.

I. Krull, M. Swartz, J. Turpin, P. H. Lukulay, and R. Verseput, “A quality-by-design methodology for rapid LC method development, part-I,” LC-GC North America, vol. 26, pp. 1190–1197, 2008.

I. Krull, M. Swartz, J. Turpin, P. H. Lukulay, and R. Verseput, “A quality-by-design methodology for rapid LC method development, part II,” LC-GC North America, vol. 27, pp. 48–61, 2009.

Merck Research Laboratories, The Merck Index, Merck, Whitehouse Station, NJ, USA, 13th edition, 2001.

S. R. Lahoti, P. K. Puranik, A. A. Heda, and R. B. Navale, “Development and validation of RP-HPLC method for analysis of glipizide in guinea pig plasma and its application to pharmacokinetic study,” International Journal of PharmTech Research, vol. 2, no. 3, pp. 1649–1654, 2010.

Z. J. Lin, D. Desai-Krieger, and L. Shum, “Simultaneous determination of glipizide and rosiglitazone unbound drug concentrations in plasma by equilibrium dialysis and liquid chromatography- tandem mass spectrometry,” Journal of Chromatography B, vol. 801, no. 2, pp. 265–272, 2004.

C. R. Shuman, “Glipizide: an overview,” The American Journal of Medicine, vol. 75, no. 5, part 2, pp. 55–59, 1983.

“The United States pharmacopoeia 32,” the National Formulary Rockville, USP Convention, pp. 2499, 2009

“European pharmacopoeia,” EDQM, version 6, pp. 1977–1979, 2008

L. Narasimham and Y. S. Barhate D, “Simultaneous quantification of metformin and glipizide in human plasma by high-performance liquid chromatography-tandem mass spectrometry and its application to a pharmacokinetic study,” Journal of Pharmacy Research, vol. 3, no. 12, pp. 3081–3087, 2010.

N. Li, Y. Deng, F. Qin, J. Yu, and F. Li, “Simultaneous quantification of metformin and glipizide in human plasma by high-performance liquid chromatography-tandem mass spectrometry and its application to a pharmacokinetic study,” Biomedical Chromatography, vol. 27, no. 2, pp. 191–196, 2013.

V. Rayanam, L. Rao, and M. V. Ramana, “Development and validation of LC method for the estimation of glipizide in pharmaceutical dosage form and serum,” International Journal of Research in Pharmacy and Chemistry, vol. 1, no. 1, pp. 50–54, 2011.

K. S. Lakshmi and T. Rajesh, “Separation and quantification of eight antidiabetic drugs on a high-performance liquid chromatography: its application to human plasma assay,” ISRN Pharmaceutics, vol. 2011, Article ID 521353, 7 pages, 2011.

N. Sultana, M. S. Arayne, S. N. Ali, and M. H. Zuberi, “Simultaneous determination of glipizide and glimepride by Rp-Hplc in dosage formulations and in human serum,” Medicinal Chemistry Research, vol. 21, no. 9, pp. 2443–2448, 2011.

K. S. Lakshmi and T. T. Rajesh, “Development and validation of RP-HPLC method for simultaneous determination of glipizide,rosiglitazone, pioglitazone, glibenclamide and glimepiride in pharmaceutical dosage forms andhuman plasma,” Journal of the Iranian Chemical Society, vol. 8, no. 1, pp. 31–37, 2011.

S. J. Rahila, P. P. Manisha, M. R. Prabha, K. Asif, and Y. G. Pramod, “Sensitive and selective analytical method for the quantification of glipizide in human plasma,” International Research Journal of Pharmacy, vol. 1, no. 1, pp. 378–383, 2010.

H. Emilsson, “High-performance liquid chromatographic determination of glipizide in human plasma and urine,” Journal of Chromatography, vol. 421, no. 2, pp. 319–326, 1987.

J. Shaodong, W. J. Lee, J. W. Ee, J. H. Park, S. W. Kwon, and J. Lee, “Comparison of ultraviolet detection, evaporative light scattering detection and charged aerosol detection methods for liquid-chromatographic determination of anti-diabetic drugs,” Journal of Pharmaceutical and Biomedical Analysis, vol. 51, no. 4, pp. 973–978, 2010.

H. Liping, X. Yan, H. Jiaxiu et al., “Determination of glipizide in sustained release tablets by RP-HPLC,” Zhonghua Yixue Yanjiu Zazhi, vol. 11, no. 6, pp. 337–339, 2011.

M. U. Reddy, P. Y. Reddy, V. Reddy, P. Somasekhar, and B. Varaprasad, “Single and high resolution RP-HPLC method for the determination of 6 antidiabetic drug products,” Journal of Pharmacy Research, vol. 4, no. 4, pp. 1209–1212, 2011.

S. Rahila and K. Asif, “Reverse phase high performance liquid chromatographic method for the analysis of glipizide in pharmaceutical dosage forms,” International Journal of Research in Ayurveda & Pharmacy, vol. 1, no. 2, pp. 455–458, 2010.

A. Gumieniczek, A. Berecka, and Ł. Komsta, “Stability-indicating validated HPLC method for simultaneous determination of oral antidiabetic drugs from thiazolidinedione and sulfonylurea groups in combined dosage forms,” Journal of AOAC International, vol. 93, no. 4, pp. 1086–1092, 2010.

M. Mangesh and S. Shanmukhappa, “A validated RP-HPLC method for estimation of Glipizide in it's pure and pharmaceutical dosage form (tablets),” Material Science Research India, vol. 6, no. 1, pp. 223–226, 2009.

S. V. Saradhi, V. S. Kiran, V. H. Bindu, and G. D. Rao, “High pressure liquid chromatography estimation of glipizide in pharmaceutical dosage forms,” Asian Journal of Chemistry, vol. 18, no. 2, pp. 1309–1312, 2006.

A. Dubey and I. C. Shukla, “Simultaneous determination of Glipizide and Metformin hydrochloride in pharmaceutical preparation by HPLC,” Journal of the Indian Chemical Society, vol. 81, no. 1, pp. 84–86, 2004.

L. Yang, H. Lijuan, W. Jide et al., “Simultaneous determination of three diabetic drugs in urine by high performance liquid chromatography,” Fenxi Ceshi Xuebao, vol. 30, no. 3, pp. 293–297, 2011.

L. Adhikari, S. Jagadev, S. Sahoo, P. N. Murthy, and U. S. Mishra, “Devlopement and validation of UV-visible spectrophotometric method for simultaneous determination of Pioglitazone Hydrochloride, Metformin Hydrochloride and glipizide in its bulk and pharmaceutical dosage form (tablet),” International Journal of ChemTech Research, vol. 4, no. 2, pp. 625–630, 2012.

C. T. Thomas, R. Y. Padmanabha, and N. Devanna, “Simultaneous spectrophotometric estimation of Metformin Hydrochloride and Glipizide in tablet dosage forms,” International Journal of PharmTech Research, vol. 3, no. 4, pp. 2064–2067, 2011.

R. M. Abd-Elgawad and E. H. Shimaa, “Voltammetric and spectrophotometric studies on the inclusion complex of glipizide with β-cyclodextrin,” Eurasian Journal of Analytical Chemistry, vol. 6, no. 1, pp. 13–21, 2011.

M. Mangesh and A. Shanmukhappa, “UV spectrophotometric method for determination of glipizide in its pure and pharmaceutical dosage form (tablets),” Journal of the Institution of Chemists, vol. 81, no. 3, pp. 93–96, 2009.

A. Aruna and K. Nancey, “Simultaneous estimation of metformin HCL and glipizide in solid dosage forms by ultraviolet spectrophotometry,” Indian Drugs, vol. 37, no. 11, pp. 533–536, 2000.

P. Rita, C. Barbara, A. Cinzia et al., “Comparison of capillary electrophoresis with HPLC for diagnosis of factitious hypoglycaemia,” Clinical Chemistry, vol. 46, no. 11, pp. 1773–1780, 2000.

N. Li, M. Cui, X. Lu, F. Qin, K. Jiang, and F. Li, “A rapid and reliable UPLC-MS/MS method for the identification and quantification of fourteen synthetic anti-diabetic drugs in adulterated Chinese proprietary medicines and dietary supplements,” Biomedical Chromatography, vol. 24, no. 11, pp. 1255–1261, 2010.

H. K. Trivedi, N. Kshtri, V. Patel, and V. Roa, “Development and validation of an uplc method for in-vitro study of glipizide extended release tablets,” International Journal of Pharmaceutical Science and Research, vol. 3, no. 9, pp. 3317–3322, 2012.

“Photostability testing of new active substances and medicinal products,” ICH Q1, (B), 1998

“Stability testing of new drug substances and products,” ICH Q1A (R2), 2003