BioMed Research International
Volume 2020 (2020), Article ID 5639651, 8 pages
https://doi.org/10.1155/2020/5639651

Research Article

2-Methacryloyloxyethyl Phosphorylcholine Polymer Coating Inhibits Bacterial Adhesion and Biofilm Formation on a Suture: An In Vitro and In Vivo Study

Taizo Kaneko,¹ Taku Saito,¹ Takeo Shobuike,² Hiroshi Miyamoto,² Junpei Matsuda,³ Kyoko Fukazawa,⁴ Kazuhiko Ishihara,⁴ Sakae Tanaka,¹ and Toru Moro1,5

¹Sensory & Motor System Medicine, Graduate School of Medicine, The University of Tokyo, 7-3-1 Hongo, Bunkyo-ku, Tokyo 113-8655, Japan
²Sensory Department of Pathology and Microbiology, Faculty of Medicine, Saga University, Saga 849-8501, Japan
³Project Management Department, KEISEI Medical Industrial Corporation Limited, 3-19-6, Hongo, Bunkyo-ku, Tokyo 113-0033, Japan
⁴Department of Materials Engineering, School of Engineering, The University of Tokyo, 7-3-1 Hongo, Bunkyo-ku, Tokyo 113-8656, Japan
⁵Division of Science for Joint Reconstruction, Graduate School of Medicine, The University of Tokyo, 7-3-1 Hongo, Bunkyo-ku, Tokyo 113-8655, Japan

Correspondence should be addressed to Toru Moro; moro-ort@h.u-tokyo.ac.jp

Received 4 June 2020; Accepted 24 August 2020; Published 1 October 2020

Academic Editor: Silvia Todros

Copyright © 2020 Taizo Kaneko et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

Abstract.

Initial bacterial adhesion to medical devices and subsequent biofilm formation are known as the leading causes of surgical site infection (SSI). Therefore, inhibition of bacterial adhesion and biofilm formation on the surface of medical devices can reduce the risk of SSIs. In this study, a highly hydrophilic, antibiofouling surface was prepared by coating the bioabsorbable suture surface with poly(2-methacryloyloxyethyl phosphorylcholine (MPC)-co-n-butyl methacrylate) (PMB). The PMB-coated and noncoated sutures exhibited similar mechanical strength and surface morphology. The effectiveness of the PMB coating on the suture to suppress adhesion and biofilm formation of methicillin-resistant Staphylococcus aureus and methicillin-susceptible Staphylococcus aureus was investigated both in vitro and in vivo. The bacterial adhesion test revealed that PMB coating significantly reduced the number of adherent bacteria, with no difference in the number of planktonic bacteria. Moreover, fluorescence microscopy and scanning electron microscopy observations of adherent bacteria on the suture surface after contact with bacterial suspension confirmed PMB coating-mediated inhibition of biofilm formation. Additionally, we found that the PMB-coated sutures exhibited significant antibiofouling effects in vivo. In conclusion, PMB-coated sutures demonstrated bacteriostatic effects associated with a highly hydrophilic, antibiofouling surface and inhibited bacterial adhesion and biofilm formation. Therefore, PMB-coated sutures could be a new alternative to reduce the risk of SSIs.

1. Introduction

Surgical site infections (SSIs) are the most common surgical complications, accounting for approximately 40% of all healthcare-associated infections [1] and 5% of all surgical complications [2]. SSIs result in further surgical intervention, prolonged treatment period, and unexpected high medical costs [3, 4]. Several factors are believed to be involved in the onset of SSIs, including implanted surgical sutures [5].

It is widely accepted that initial bacterial adhesion to biomaterial surfaces and biofilm formation secondary to bacterial adhesion are primary causes of SSI [6, 7]. Once a biofilm has formed, it protects bacteria from the host’s immune system as well as systemically and locally applied antibiotics [7], thus rendering the SSI intractable [4, 7]. Therefore, inhibition of the initial bacterial adherence and biofilm formation on the surface of implanted sutures could be an effective countermeasure to reduce the incidence of SSIs.

Although antimicrobial agent-coated (e.g., triclosan [8]) suture surfaces reportedly suppress bacterial adhesion and biofilm formation, these antibacterial agents exert potential toxicity and side effects, including cytotoxicity, toxic side products, and antibiotic resistance [9, 10]. Due to these defects, the development of alternative suture materials with adequate antibiofouling ability and excellent biocompatibility has clinical value.

Several polymers can be used for surface modification of medical devices to prevent undesired biological responses [11–14]. Among them, zwitterionic phosphorylcholine-based polymers are the most attractive owing to their wide applicability to modify medical devices [15, 16]. For instance, 2-methacryloyloxyethyl phosphorylcholine (MPC) polymers are commonly used to provide highly hydrophilic and biocompatible surfaces [17–20]. Many medical devices, such as intravascular stents, ventricular assist devices, contact lenses, and hip acetabular liners, have been developed using MPC polymers and are clinically widespread [20–24]. Besides, the medical applicability and safety of MPC polymers are well established [20]. Thus, we expected that MPC polymer coating could be an ideal surface modification material with potential secondary benefits related to the inhibition of bacterial infection on the suture surface [25]. Poly(MPC-co-n-butyl methacrylate) (PMB) is one of the MPC polymers most widely used for surface modification of medical devices. Therefore, this study aimed to investigate the effects of PMB-coated sutures on initial bacterial adhesion and subsequent biofilm formation, both in vitro and in vivo.

2. Materials and Methods

2.1. Materials

The MPC polymer, PMB (Lipidure®-CM5206, NOF, Tokyo, Japan) and absorbable polyglactin sutures (KRAYON Plus®, KEISEI Medical Industrial Ltd., Tokyo, Japan) were used in this study. Other reagents, solvents, and analysis kits are commercially available and were used as obtained.

2.2. Surface Coating of Sutures with PMB

PMB was dissolved in 100 mL ethanol at a concentration of 1.0 wt%. The suture was immersed in the solution for several minutes using a dedicated coating machine and thoroughly air-dried for 10 min, followed by drying for 30 min at 50°C in a hot air oven. The dried suture was cut to 10 mm in length and wrapped in aluminum pouches for sterilization, respectively. All prepared sutures were sterilized with ethylene oxide gas under sterilization conditions according to ISO 11135:2014, prior to the experiment.

2.3. Characterization of Coated Sutures

2.3.1. Confirmation of the Presence of PMB Coating by Various Methods

The surface of PMB-coated sutures was examined by X-ray photoelectron spectroscopy (XPS) (AXIS-HSi, Shimadzu/KRATOS, Kyoto, Japan) using Mg Kαsources. High-resolution scans for C1s and P2p were acquired at a photoelectron takeoff angle of 90°. The energies of all spectra were corrected using the C1s energy calibration peak at 285.0 eV. We visualized the phosphorylcholine groups on PMB’s side chains by rhodamine 6G staining (Wako Pure Chemical Industries, Ltd, Tokyo, Japan) [19]. All samples were immersed in the rhodamine 6G solution for 30 s, washed twice with distilled water for 30 s, and let dry at room temperature. After staining, suture samples were observed under a fluorescence microscope (BZ-X700; Keyence, Japan). Besides, the absolute amount of phosphorus atoms on suture surfaces was measured by the molybdenum method [26].

2.3.2. Mechanical Property Measurement

The diameter and weight of noncoated and coated sutures were measured at five different positions along the length of the suture. The average diameter was calculated using an electronic thickness gauge and expressed as millimeter. The weight was determined using a precision electronic balance and expressed as milligram. The tensile strength of coated and noncoated sutures was assessed by straight-pull tests, using an Instron 5960 Universal testing system (Instron, Canton, MA, USA) under the procedures outlined in the United States Pharmacopeia standard. Ten specimens were tested, and the results are expressed as .

2.3.3. Contact Angle Measurement

Static water contact angle measurements on the suture surface were carried out by the half-angle method using a goniometer (Model CA-S Micro2; Kyowa Interface Science Co., Ltd., Saitama, Japan). In brief, the suture attached to the measurement device jig was sprayed with ultrapure water, and the water contact angles were directly measured after 1.0 min using a contact angle meter. Fifteen specimens were tested, and the results are expressed as .

2.4. Bacterial Strain Culture

Two different types of gram-positive bacteria strains were used in this study, including methicillin-resistant Staphylococcus aureus (MRSA) UOEH6 (University of Occupational and Environmental Health Hospital, Fukuoka, Japan), which was isolated from blood samples of a septic patient, and S. aureus (MSSA) NBRC12732 (Biological Resource Center, National Institute of Technology and Evaluation, Chiba, Japan). They were stored at -80°C in a glycerol stock. Bacteria were cultured overnight in Tryptic Soy Broth (TSB; Eiken Chemical, Tokyo, Japan) at 37°C. After incubation, bacterial cells were harvested by centrifugation and resuspended in PBS at a density of bacteria/mL.

2.5. Evaluation of Bacterial Adhesion

To allow adhesion to the suture samples, the suture samples were exposed to bacterial cells in 24-well plates and incubated for 1.0 h at 37°C. Subsequently, suture samples were placed in 1.5 mL tubes filled with 1.0 mL PBS, followed by vortexing and ultrasonication to scrape off firmly adhered bacteria from suture surfaces. The solutions were collected after ultrasonication, then serially diluted 10-folds, and plated on lysogeny broth plates supplemented with 7.5% sodium chloride. The CFU numbers for adherent and planktonic bacteria were counted after incubation at 37°C for 1 day. For nucleic acid staining in bacteria, the suture samples were stained with SYTO® 9 green fluorescent nucleic acid stain (FilmTracer™ LIVE/DEAD Biofilm Viability kit; Molecular Probes, Eugene, OR, USA) at room temperature for 30 min. After staining, suture samples were observed under a fluorescence microscope (BZ-X700; Keyence, Japan). All quantifications were performed at 40x magnification. Random areas (eight per suture) were analyzed, and the coverage area was estimated using the image analysis software program ImageJ (National Institute of Health, Bethesda, MD, USA). The mean coverage area of 24 images from three sutures was calculated. Further, sutures were fixed with 2.0% glutaraldehyde for 2.0 h and dehydrated in graded ethanol. The well-dehydrated suture samples were observed under a scanning electron microscopy (SEM; JCM-6000, JEOL Ltd., Tokyo, Japan) at an accelerating voltage of 15 kV after platinum deposition. The number of adherent bacteria was estimated using the field of view at 1000x magnification. Random areas (eight per suture) were recorded, and visibly adherent bacteria were counted using ImageJ. The mean number of adherent bacteria of 24 images from three sutures was calculated.

2.6. Evaluation of Biofilm Formation

To generate biofilms on the suture surfaces, bacterial cells were prepared to a density of bacteria/mL in TSB supplemented with 0.25% glucose. Next, the suture samples were exposed to bacterial cells in 24-well plates and incubated for 24 h at 37°C while shaking at 50 rpm. Suture samples were rinsed three times with 1.0 mL PBS and observed by fluorescence microscopy and SEM.

2.7. In Vivo Evaluation

2.7.1. Animals

All experimental procedures were performed under a protocol approved by the Animal Care and Use Committee of the University of Tokyo. In brief, twelve 8-week-old male C57BL/6J mice from SLC (Shizuoka, Japan) were used in this study. All mice were housed in plastic cages with bedding material in a specific pathogen-free facility, with 12 h light-dark cycles.

2.7.2. Surgical Technique

All mice were intraperitoneally injected with a mixture of anesthetic agents (0.38 mg/kg medetomidine, 2.0 mg/kg midazolam, and 2.5 mg/kg butorphanol). Hair was removed using clippers. Subcutaneous pockets were created on both sides of the back of mice, and each suture was aseptically implanted into the subcutaneous pockets: a PMB-coated suture was implanted on the right side of the mouse and a noncoated suture on the left side. MRSA suspension ( bacteria/mL) was inoculated onto the suture before implantation. Incisions were closed using a 5-0 nylon suture. No antibiotics were used following the operation. After 1, 3, and 7 days, the animals were euthanized, and the sutures removed. All samples were observed by fluorescence microscopy and SEM. All operations were performed by one surgeon under the same experimental conditions.

2.8. Statistical Analysis

All data are expressed as . Statistical analyses were conducted with the unpaired two-tailed Student’s -test using BellCurve for Excel. The value < 0.05 was considered significant.

3. Results

3.1. Characterization of PMB-Coated Sutures

The XPS peaks of the phosphorus atom region (P2p), detected at 133 eV, corresponding to the phosphate group, were observed exclusively on PMB-coated sutures (Figure 1(a)). In addition, strong fluorescence was observed on the suture surface treated with PMB (Figure 1(b)), confirming that the suture was covered with sufficient amounts of PMB. The quantity of phosphorus atoms on the surface of noncoated and PMB-coated sutures was 0.09 μg/cm and 6.2 μg/cm (), respectively (Figure 1(c)).

(a)

(b)

(c)

Figure 1: Analysis of the surface composition of biomaterials. (a) XPS spectra of sutures, C1s, and P2p. (b) Representative fluorescence micrographs of noncoated and PMB-coated sutures stained with rhodamine 6G. The scale bar indicates 100 μm. (c) The absolute amount of phosphorus atoms on suture surfaces. .

PMB-coated sutures showed sustained integrity and uniformly covered the tested surfaces. There was no difference in diameter and weight between noncoated and coated sutures (Table 1). The maximum tensile forces of noncoated control and coated sutures were measured as 170 N (Table 1). The static water contact angles on the surface of noncoated and PMB-coated sutures were 38° and 31° (), respectively (Table 1).

Table 1: Diameter, weight, maximum tensile strength, and static water contact angle of sutures.

Suture
Diameter (mm)
Weight (mg)
Maximum tensile strength (N)
Static water contact angle (degree)

Noncoated

PMB-coated

Data are expressed as . .

3.2. Bacterial Adhesion and Growth on PMB-Coated Surfaces

The initial bacterial adhesion of MRSA and MSSA on PMB-coated suture surfaces demonstrated drastic reductions in bacterial adhesion compared to those observed on noncoated surfaces (93 and 79% reduction, respectively) (Figure 2(a)). However, there were no significant differences in the mean relative abundance of planktonic bacteria (Figure 2(b)). SYTO® 9 nucleic acid staining exhibits green fluorescence on live bacteria, which was used to measure adherent bacteria by fluorescence microscopy. The noncoated suture surface had a higher number of live bacteria than the PMB-coated suture (Figure 3(a)). Specifically, the area of adherent MRSA and MSSA was significantly lower on the PMB-coated suture (72 and 74% reduction, respectively) (Figure 3(b)). In SEM images (Figure 3(c)), similar trend of initial bacterial adhesion was observed, in which many bacterial colonies grew on uncoated suture surfaces. By contrast, very few bacterial colonies adhered to the PMB-coated suture surface (Figure 3(d)). Thus, the PMB-coated suture inhibited initial bacterial adhesion in all evaluation. For both bacterial strains, significant biofilm formation was observed on the noncoated suture. By contrast, very few biofilms were observed on the PMB-coated suture surface (Figure 4).

(a)

(b)

Figure 2: (a) Adherent and (b) planktonic bacteria on sutures after 1 h of incubation. Data are expressed as . .

(a)

(b)

(c)

(d)

Figure 3: Evaluation of bacterial adhesion. (a) Representative fluorescence microscopy images of sutures after bacterial culture for 1 h in vitro. The scale bar indicates 20 μm. (b) Box-and-whisker plots of the area of adherent bacteria on suture surfaces after bacterial culture for 1 h in vitro. . (c) Representative SEM images of bacterial adherence after bacterial culture for 1 h in vitro. The scale bar indicates 10 μm. Insets are magnified images (scale bar indicates 2 μm). (d) Box-and-whisker plots of adherent bacteria on suture surfaces after bacterial culture for 1 h in vitro. and .

(a)

(b)

(c)

(d)

Figure 4: Evaluation of biofilm formation. (a) Representative fluorescence microscopy images of sutures after bacterial culture for 24 h in vitro. The scale bar indicates 20 μm. (b) Box-and-whisker plots of the area of adherent bacteria on sutures after bacterial culture for 24 h in vitro. . (c) Representative SEM images of bacterial adherence after bacterial culture for 24 h in vitro. The scale bar indicates 10 μm. Insets are magnified images (scale bar indicates 2 μm). (d) Box-and-whisker plots of adherent bacteria on suture surfaces after bacterial culture for 24 h in vitro. .

3.3. In Vivo Effects of PMB Coating on Biofilm Formation

Although biofilm formation increased over time, no mice exhibited skin disorders, poor wound healing, or died during the study. Similar to the in vitro findings, bacterial attachment and biofilm formation were suppressed on PMB-coated sutures (Figure 5).

(a)

(b)

Figure 5: Representative images of bacterial adherence and biofilm formation after implantation for 1, 3, and 7 days in vivo. (a) Fluorescence microscopy images. The scale bar indicates 10 μm. (b) SEM images. The scale bar indicates 10 μm. Insets are magnified images (scale bar indicates 2 μm).

4. Discussion

MPC contains a zwitterionic and hydrophilic phosphorylcholine group and a polymerizable methacryloyl group in its molecular structure [17, 27]. As MPC has suitable polymerization and copolymerization abilities with other vinyl monomers, through conventional radical polymerization processes, the molecular design of the MPC polymer is flexible and can be tailored to obtain desired antifouling properties. When the MPC polymer binds to a substrate by physical adsorption and adhesion, the modified surface can inhibit protein absorption and cell adhesion sufficiently [19]. Due to these beneficial properties, many medical devices have been developed using MPC polymers for clinical implementation. Since the initial adherence of microbes and adsorbed proteins to biomaterial surfaces is critical for the pathogenesis of foreign body infections [28], MPC polymers are optimal tools to prevent biofilm formation. In this study, we coated the suture surface with the MPC polymer layer to prevent suture-associated infections, which led to significant antibiofouling effects both in vitro and in vivo.

One of the MPC polymers, PMB, was designed given its solubility and thin film formation ability on the substrate. The PMB is not hydrosoluble due to its high-molecular weight and 70% hydrophobic BMA content. Due to the stable coating with PMB, in this study, we chose a 30/70-unit mole ratio of MPC/BMA. Of course, the MPC content is important to reduce bacterial adhesion. When the MPC content is smaller, the adhesion of bacteria increases. On the other hand, when the MPC content exceeds 40-unit mol%, the polymer became too hydrophilic to stably attach to the substrate. Therefore, in this and related previous studies, a 30/70-unit mole ratio of MPC/BMA simultaneously and effectively forms a coating layer and inhibits bacteria adhesion. Immediately after the coating, van der Waals forces are established between PMB and suture substrate, while hydrophobic interactions are formed upon suture contact with an aqueous medium. The molecular weight of PMB is controlled above to generate a stable thin film that firmly attaches to the substrate. Therefore, we used PMB to prevent SSIs based on the reduction of surface biofilm formation.

We first analyzed the surface composition of the biomaterials to determine whether PMB coating was successful. The PMB coating was confirmed by XPS, which showed the atomic detection of distinctive phosphorus and carbon peaks (Figure 1(a)). Rhodamine 6G is a cationic fluorescent dye and is known to selectively adsorb to phospholipids, which aids the visualization of phospholipid-based polymers [29]. Intense fluorescence staining was observed on the surface treated with PMB, confirming the successful coating of PMB on the whole suture surface (Figure 1(b)). Additionally, the absolute amount of phosphorus atoms on PMB-coated sutures was significantly higher than that on noncoated sutures (Figure 1(c)). From these results, we concluded that PMB properly coated the suture surfaces.

Consistent with previous studies [18, 19, 30], the water wettability of the PMB-modified suture surfaces was higher than that of noncoated sutures (Table 1), without reducing mechanical strength. Since MPC polymers have an extremely hydrophilic nature, the PMB-modified suture surface showed a hydrated layer in aqueous environments [19]. Hence, we successfully influenced bacterial adhesion and biofilm formation by treating the suture surfaces with PMB.

Expectedly, PMB-coated sutures limited bacterial adherence without affecting planktonic bacteria (Figure 2), indicating that PMB treatment has bacteriostatic but not antibacterial effects [18, 19]. In other words, PMB treatment only suppresses the initial bacterial adhesion without killing bacteria, which also prevents biofilm formation. This mechanism of action indicated that PMB poses no health-related risks.

Biofilms are microbial communities that become attached and embedded in an implanted material [31]. Biofilm formation protects bacteria against the host’s immune system and antibiotics, resulting in SSIs. Here, fluorescence microscopy and SEM images revealed that PMB coating significantly inhibited bacterial adhesion and biofilm formation both in vitro and in vivo (Figures 3–5).

Our study, however, has several limitations. First, only MRSA and MSSA were examined, without considering other pertinent strains that can cause infection, including S. epidermidis, Enterococcus faecalis, and Escherichia coli. However, we speculate that PMB-coated sutures will be effective against all strains because the PMB-coated suture surface suppresses the first step of infection, which is protein adsorption. Second, the SYTO9 green fluorescent nucleic acid stain was used to evaluate the biofilm coverage rates. This stain reacts with both the biofilm and debris of inflammatory tissues in vivo. Finally, although potential resistance to acute infection was evaluated over 7 days in vivo, these findings do not apply to antibiofilm activity in chronic infections. Further investigations, including analysis of bacterial adhesion and complex physicochemical interactions in various environments, are needed to confirm the efficacy of PMB-coated sutures against bacterial adhesion and biofilm formation.

5. Conclusions

Here, we developed an antibiofouling surface by coating suture surfaces with PMB. The PMB-coated sutures exhibited mechanical strength and surface morphology equal to those of noncoated sutures. Besides, significant inhibition of MRSA and MSSA initial adhesion and biofilm formation were validated, both in vitro and in vivo. This study indicates significant bacteriostatic effects of PMB-coated sutures, implying their potential applications to reduce the risk of suture-related SSIs.

Data Availability

The data used to support the findings of this study are included within the article.

Conflicts of Interest

Co-author (M.J.) is employed by KEISEI Medical Industrial Corporation Limited.

Acknowledgments

We thank Ms. Aiko Noguchi of KEISEI Medical Industrial Corporation Limited for technical assistance.

References

R. S. Jones, C. Brown, and F. Opelka, “Surgeon compensation: “Pay for performance,” the American College of Surgeons National Surgical Quality Improvement Program, the Surgical Care Improvement Program, and other considerations,” Surgery, vol. 138, no. 5, pp. 829–836, 2005.

I. Mattavelli, P. Rebora, G. Doglietto et al., “Multi-center randomized controlled trial on the effect of triclosan-coated sutures on surgical site infection after colorectal surgery,” Surgical Infections, vol. 16, no. 3, pp. 226–235, 2015.

K. B. Kirkland, J. P. Briggs, S. L. Trivette, W. E. Wilkinson, and D. J. Sexton, “The impact of surgical-site infections in the 1990s: attributable mortality, excess length of hospitalization, and extra costs,” Infection Control and Hospital Epidemiology, vol. 20, no. 11, pp. 725–730, 1999.

C. D. Owens and K. Stoessel, “Surgical site infections: epidemiology, microbiology and prevention,” The Journal of Hospital Infection, vol. 70, pp. 3–10, 2008.

U. K. Ercan, F. İbiş, C. Dikyol et al., “Prevention of bacterial colonization on non-thermal atmospheric plasma treated surgical sutures for control and prevention of surgical site infections,” PLoS One, vol. 13, no. 9, article e0202703, 2018.

R. A. Mendoza, J.-C. Hsieh, and D. R. Galiano, “The impact of biofilm formation on wound healing,” in Wound Healing - Current Perspectives, IntechOpen, London, UK, 2019.

A. Omar, J. Wright, G. Schultz, R. Burrell, and P. Nadworny, “Microbial biofilms and chronic wounds,” Microorganisms, vol. 5, no. 1, article e9, p. 9, 2017.

M. L. Storch, S. J. Rothenburger, and G. Jacinto, “Experimental efficacy study of coated VICRYL plus antibacterial suture in guinea pigs challenged with Staphylococcus aureus,” Surgical Infections, vol. 5, no. 3, pp. 281–288, 2004.

G. Bedoux, B. Roig, O. Thomas, V. Dupont, and B. le Bot, “Occurrence and toxicity of antimicrobial triclosan and by-products in the environment,” Environmental Science and Pollution Research International, vol. 19, no. 4, pp. 1044–1065, 2012.

S. P. Yazdankhah, A. A. Scheie, E. A. Høiby et al., “Triclosan and antimicrobial resistance in bacteria: an overview,” Microbial Drug Resistance, vol. 12, no. 2, pp. 83–90, 2006.

B. W. Hanak, C. Y. Hsieh, W. Donaldson, S. R. Browd, K. K. S. Lau, and W. Shain, “Reduced cell attachment to poly(2-hydroxyethyl methacrylate)-coated ventricular catheters in vitro,” Journal of Biomedical Materials Research. Part B, Applied Biomaterials, vol. 106, no. 3, pp. 1268–1279, 2018.

C. Weis, E. K. Odermatt, J. Kressler, Z. Funke, T. Wehner, and D. Freytag, “Poly(vinyl alcohol) membranes for adhesion prevention,” Journal of Biomedical Materials Research. Part B, Applied Biomaterials, vol. 70B, no. 2, pp. 191–202, 2004.

A. D. Malkin, S. H. Ye, E. J. Lee et al., “Development of zwitterionic sulfobetaine block copolymer conjugation strategies for reduced platelet deposition in respiratory assist devices,” Journal of Biomedical Materials Research. Part B, Applied Biomaterials, vol. 106, no. 7, pp. 2681–2692, 2018.

G. Cheng, G. Li, H. Xue, S. Chen, J. D. Bryers, and S. Jiang, “Zwitterionic carboxybetaine polymer surfaces and their resistance to long-term biofilm formation,” Biomaterials, vol. 30, no. 28, pp. 5234–5240, 2009.

J. Fang, S. H. Ye, V. Shankarraman, Y. Huang, X. Mo, and W. R. Wagner, “Biodegradable poly(ester urethane)urea elastomers with variable amino content for subsequent functionalization with phosphorylcholine,” Acta Biomaterialia, vol. 10, no. 11, pp. 4639–4649, 2014.

S. Monge, B. Canniccioni, A. Graillot, and J. J. Robin, “Phosphorus-containing polymers: a great opportunity for the biomedical field,” Biomacromolecules, vol. 12, no. 6, pp. 1973–1982, 2011.

K. Ishihara, N. P. Ziats, B. P. Tierney, N. Nakabayashi, and J. M. Anderson, “Protein adsorption from human plasma is reduced on phospholipid polymers,” Journal of Biomedical Materials Research, vol. 25, no. 11, pp. 1397–1407, 1991.

S. Kang, M. Lee, M. Kang et al., “Development of anti-biofouling interface on hydroxyapatite surface by coating zwitterionic MPC polymer containing calcium-binding moieties to prevent oral bacterial adhesion,” Acta Biomaterialia, vol. 40, pp. 70–77, 2016.

M. Kyomoto, T. Shobuike, T. Moro et al., “Prevention of bacterial adhesion and biofilm formation on a vitamin E-blended, cross-linked polyethylene surface with a poly(2-methacryloyloxyethyl phosphorylcholine) layer,” Acta Biomaterialia, vol. 24, pp. 24–34, 2015.

K. Ishihara, “Blood-compatible surfaces with Phosphorylcholine-based polymers for cardiovascular medical devices,” Langmuir, vol. 35, no. 5, pp. 1778–1787, 2019.

A. L. Lewis, L. A. Tolhurst, and P. W. Stratford, “Analysis of a phosphorylcholine-based polymer coating on a coronary stent pre- and post-implantation,” Biomaterials, vol. 23, no. 7, pp. 1697–1706, 2002.

T. Moro, Y. Takatori, K. Ishihara et al., “Surface grafting of artificial joints with a biocompatible polymer for preventing periprosthetic osteolysis,” Nature Materials, vol. 3, no. 11, pp. 829–836, 2004.

T. Moro, Y. Takatori, S. Tanaka et al., “Clinical safety and wear resistance of the phospholipid polymer-grafted highly cross-linked polyethylene liner,” Journal of Orthopaedic Research, vol. 35, no. 9, pp. 2007–2016, 2017.

Y. Iwasaki and K. Ishihara, “Cell membrane-inspired phospholipid polymers for developing medical devices with excellent biointerfaces,” Science and Technology of Advanced Materials, vol. 13, no. 6, article 064101, 2012.

K. Ishihara, “Revolutionary advances in 2-methacryloyloxyethyl phosphorylcholine polymers as biomaterials,” Journal of Biomedical Materials Research. Part A, vol. 107, no. 5, pp. 933–943, 2019.

S. R. Crouch and H. V. Malmstadt, “Mechanistic investigation of molybdenum blue method for determination of phosphate,” Analytical Chemistry, vol. 39, no. 10, pp. 1084–1089, 1967.

K. Ishihara, H. Nomura, T. Mihara, K. Kurita, Y. Iwasaki, and N. Nakabayashi, “Why do phospholipid polymers reduce protein adsorption?” Journal of Biomedical Materials Research, vol. 39, no. 2, pp. 323–330, 1998.

A. G. Gristina, C. D. Hobgood, L. X. Webb, and Q. N. Myrvik, “Adhesive colonization of biomaterials and antibiotic resistance,” Biomaterials, vol. 8, no. 6, pp. 423–426, 1987.

J. H. Wang, J. D. Bartlett, A. C. Dunn et al., “The use of rhodamine 6G and fluorescence microscopy in the evaluation of phospholipid-based polymeric biomaterials,” Journal of Microscopy, vol. 217, no. 3, pp. 216–224, 2005.

M. Kyomoto, T. Moro, S. Yamane, Y. Takatori, S. Tanaka, and K. Ishihara, “A hydrated phospholipid polymer-grafted layer prevents lipid-related oxidative degradation of cross-linked polyethylene,” Biomaterials, vol. 112, pp. 122–132, 2017.

C. Wilson, R. Lukowicz, S. Merchant et al., “Quantitative and qualitative assessment methods for biofilm growth: a mini-review,” Research and Reviews Journal of Engineering and Technology, vol. 6, no. 4, 2017.