BioMed Research International
Volume 2020 (2020), Article ID 6046501, 8 pages
https://doi.org/10.1155/2020/6046501

Research Article

Size-Specific Dose Estimates of Radiation Based on Body Weight and Body Mass Index for Chest and Abdomen-Pelvic CTs

Jian Xu,¹ Xiangquan Wang,¹ Panfeng Yang,¹ Kuangnan Luo,² and Xiaolong He³

¹Department of Radiology, Zhejiang Provincial People’s Hospital, People’s Hospital of Hangzhou Medical College, Hangzhou, Zhejiang, China
²Shenshi Technology, Co., Ltd, Hangzhou, Zhejiang, China
³Department of Radiology, Quzhou People’s Hospital, Quzhou, Zhejiang, China

Correspondence should be addressed to Xiaolong He; qzhxl_2007@163.com

Received 18 April 2020; Revised 2 June 2020; Accepted 18 June 2020; Published 11 July 2020

Academic Editor: Arnaldo Scardapane

Copyright © 2020 Jian Xu et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

Abstract.

Background. To correlate body weight, body mass index (BMI), and water-equivalent diameter () and to assess size-specific dose estimates (SSDEs) based on body weight and BMI for chest and abdomen-pelvic CT examinations. Methods. An in-house program was used to calculate , size-dependent conversion factor (), and SSDE for 1178 consecutive patients undergoing chest and abdomen-pelvic CT examinations. Associations among body weight, BMI, and were determined, and linear equations were generated using linear regression analysis of the first 50% of the patient population. SSDEs (SSDEweight and SSDEBMI) were calculated based on body weight and BMI as surrogates on the second 50% of the patient population. Mean root-mean-square errors of SSDEweight and SSDEBMI were computed with SSDE from the axial images as reference values. Results. Both body weight and BMI correlated strongly with for the chest (, 0.87, all ) and abdomen-pelvis (, 0.86, all ). Mean values of SSDEweight and SSDEBMI based on the linear equations for body weight, BMI, and were in close agreement with SSDE from the axial images, with overall mean root-mean-square errors of 0.62 mGy (6.10%) and 0.57 mGy (5.65%), for chest, and 0.76 mGy (5.61%) and 0.71 mGy (5.22%), for abdomen-pelvis, respectively. Conclusions. Both body weight and BMI, serving as surrogates, can be used to calculate SSDEs in the chest and abdomen-pelvis CT examinations, providing values comparable to SSDEs from the axial images, with an overall mean root-mean-square error of less than 0.76 mGy or 6.10%.

1. Introduction

The increased risk of radiation-induced cancer from CT scans is a major concern for the medical community [1–4]. Compared with other radiologic imaging modalities, CT scanning is associated with a higher radiation dosage because of the scanner type, operation condition, scan protocol, and diagnostic reliability [2, 5]. CT examinations account for up to 60% of the total medical radiation dose; however, CT examinations only contribute to approximately 6% of radiological procedures [6, 7]. Thus, it is necessary to execute the scan protocol in agreement with the ALARA principle. Unfortunately, the implementation of the ALARA principle is challenged because the dose metrics routinely used in the clinical setting, such as volume CT dose index (CTDIvol) and dose length product (DLP), cannot provide the exact radiation dose absorbed by the patient [8, 9]. CTDIvol is susceptive to scanning parameters (e.g., kVp, mAs, pitch, and filter) and is a standardized dose metric derived from a cylindrical acrylic polymethyl methacrylate phantom with a diameter of 16 or 32 cm, whereas DLP is derived by multiplying CTDIvol by the scan length. Owing to the fact that the parameter does not account for factors related to patients, CTDIvol only reflects the average output level of the CT scanner for certain scan settings and is appropriately adapted to its purposes for the comparison of different CT scanners and scanning protocols.

The concept of size-specific dose estimate (SSDE) was introduced by the American Association of Physicists in Medicine (AAPM) to refine CTDIvol on the basis of patient body size [10, 11]. In the radiation dose structured report, the CTDIvol is normalized to SSDE using size-dependent conversion factor () based on patient size expressed through an effective diameter () or water-equivalent diameter (). Measured in centimeters, can be computed in combination with the anterior-posterior (AP) and lateral (LAT) dimensions. Measurements of AP and LAT are performed using the axial images or CT scanogram, as recommended by AAPM [10], whereas requires delineating the region of interest and measuring CT attenuation on the axial images slice by slice to accurately calculate SSDE [11], which is time-consuming and tedious. Recently, a series of studies focused on simplifying the calculation of SSDE via correlations between and weight or body mass index (BMI) [12–16]. Because of the limitation of , which only represents the geometric size of the patient, the accuracy of SSDE is inferior to that calculated using . Consequently, a number of efforts [17–19] have been made to investigate the relationship between and with the aims of using as the surrogate and achieving an accurate SSDE, comparable to that derived from the of the axial images. However, the methodologies still required measuring on the axial images, and a prescan SSDE could not be obtained. In addition, until recently, CT scanner manufacturers have not included SSDE on the CT scanner console display. Therefore, optimization of CT protocols on the basis of SSDE is impractical, and radiologists cannot use SSDE to assess the risk versus benefits of a patient undergoing a CT examination prior to imaging.

Sarmento et al.’s study reported that is a function of patient weight, suggesting that the calculation of SSDEs could be simplified using weight [20, 21]. To the best of our knowledge, there have been no previous studies on the calculation of SSDE from the chest and abdomen-pelvic CT examinations based on the correlations of body weight, BMI, and . Hence, this study is aimed at determining the correlations between two biometric indicators of weight, BMI with and verifying the accuracy of an SSDE predicted using weight and BMI as surrogate body size metrics for for the chest and abdomen-pelvic CT examinations of adult patients.

2. Materials and Methods

2.1. Patient Population

This retrospective study was executed with approval from Zhejiang Provincial People’s Hospital Ethics Committee. The requirement of written informed consent was waived in accordance with hospital policies for clinical retrospective studies.

A Picture Archive and Communication System (PACS, Greenlander version 6.0, Mindray Healthcare, Shenzhen, China) terminal was used to extract axial images and radiation dose structured reports. For the period between January and August 2019, the records of consecutive inpatients undergoing chest and abdomen-pelvic CT examinations were electronically retrieved. Patients with nondiagnostic images, truncated images, severe motion artifacts, and in-body or in-skin metal objects were excluded from this study. Patients with renal insufficiency ( mg/dL) or known allergic reactions to iodinated contrast medium for contrast-enhanced CT were excluded from the study. The final study cohort was composed of a total of 1178 patients, including those that underwent chest CT () and abdomen-pelvic CT (). Among the patients, there were 782 males and 396 females with a mean age of years (range, 18.00-90.00), mean weight of  kg (range, 35.00-110.00), and mean BMI of  kg/m² (range, 14.10-36.84). The first 50% of patients in chronological order, 308 and 281 individuals that received chest and abdomen-pelvic CT examinations, respectively, served as a model sample to generate regression equations of body weight, BMI, and . The second 50% of patients, referred to as the verified sample, were used to confirm the feasibility of calculating SSDE based on body weight and BMI.

In accordance with our hospital’s clinical practice, the weight and height of inpatients had been documented in the electronic medical records and displayed in the electronic application form for CT examination. BMI was calculated as weight divided by height squared (kg/m²). The body habitus was divided into four types according to the redefined World Health Organization (WHO) criteria for the Asia-Pacific Region [22]: B1: underweight, (); B2: normal weight, (); B3: overweight, (); and B4: obese, ().

2.2. CT Acquisition

CT examinations were performed using a 40-slice scanner and a 128-slice scanner (Definition AS and Definition AS+, Siemens AG, Healthcare Sector, Forchheim, Germany). The scan parameters of both chest and abdomen-pelvic CT examinations were 100 or 120 kVp tube voltage, 0.5-second rotation-time, and a pitch factor of 1.35. Automatic tube current modulation (ATCM) was activated to reduce the radiation dose for all scans. ATCM is a combination of applications that reduce dose exposure in four dimensions (CARE Dose 4D), created by the manufacturer Siemens. Axial images were reconstructed with a standard tissue algorithm (B30f), a matrix,  mm FOV, and 5/5 mm slice thickness and increment. The scan range for the chest CT was from the lung apices to the costophrenic angle, and the scan range for abdomen-pelvis CT included diaphragmatic top and pubic symphysis. The aforementioned scan parameters were set according to standard department CT protocols for adults.

2.3. Calculation of Size-Specific Dose Estimates

An in-house program based on MATLAB (MathWorks, Natick, Mass) was developed to automatically calculate , , and SSDE slice by slice and provide mean values across all axial images in accordance with the methodologies detailed in the AAPM report 220 [11]. The automatic calculation process included several steps. The first step was a threshold-based segmentation with a threshold of -500 HU, and axial images were converted from grayscale to binary using an image threshold. The second step was removal of the patient table from the binary image using the largest connected area method. The third step was to label the entire region of interest with the hole-filled technique. The last step was to delineate the object boundary and calculate , , and SSDE, including a single value slice by slice and mean values across all slices. The automatic calculation of , , and SSDE required the following formulas [11]:

where CTDIvol(s) is the slice CTDIvol, mAs(s) is the actual mAs per slice, mAs(a) is the averaged mAs displayed in the radiation dose structured report, CTDIvol is the volume CT dose index displayed in the radiation dose structured report, is the water-equivalent diameter, CTROI is the CT number of the whole axial image, is the area of the axial image, is the size-dependent conversion factor, and SSDE is the size-specific dose estimate.

To analyze the radiation dose, we listed three SSDE measurements as follows: SSDE, SSDEweight, and SSDEBMI. SSDEweight and SSDEBMI, defined as the new radiation dose indicators, required three steps. Firstly, linear regression analysis was used to generate linear equations () for body weight and BMI with across all modeled patients. Secondly, new body size metrics of , defined as and , were predicted from body weight or BMI, after which the corresponding was obtained using formula (2). Finally, SSDEweight and SSDEBMI were computed via multiplied by CTDIvol(s). All three steps were performed with a Microsoft Excel 2016 spreadsheet embedded with all linear equations of and two biometric indicators of body weight and BMI. Only plain axial CT images were analyzed in this study.

2.4. Statistical Analysis

Statistical analyses were performed using the statistical software SPSS 22.0 (IBM Corp., Armonk, NY, USA). The normality of continuous variables was determined using the Shapiro-Wilk test. Continuous variables with normal distributions were described as the , and dichotomous variables were reported as counts and percent. The Spearman Rho test was used to assess the correlation of variables that do not follow normal distribution and Pearson’s test for those which follow normal distribution. Associations between age and , body weight, and BMI with were assessed with Spearman’s rank correlation analysis and Pearson’s correlation analysis, respectively. Steiger’s test was used to compare the correlation coefficients of body weight and BMI with as the common variable in chest and abdomen-pelvis. A paired -test was used for SSDEweight and SSDEBMI in chest and abdomen-pelvic CT examinations. The Bland-Altman test was performed to determine the accuracy of the linear equations compared with the measured values (Figure 1). With SSDE from the axial images as reference values, the mean root-mean-square errors of SSDEweight and SSDEBMI were calculated for all verified patients. A value of less than 0.05 was considered to indicate a statistically significant difference.

(a)

(b)

(c)

(d)

Figure 1: Bland-Altman plots for the measured and the predicted . (a, b) Plots for chest CT; (c, d) plots for abdomen-pelvis CTs.

3. Results

3.1. Patient Characteristics, Body Size, and Radiation Dose

The number of CT examinations for chest and abdomen-pelvis was 264 and 239 on the 40-slice scanner and 352 and 323 on the 128-slice scanner, respectively. Patient demographics are listed in Table 1. CTDIvol and values of , , and SSDE, calculated using the in-house program, are shown in Table 2. Measurements of for both chest and abdomen-pelvis were considerably less than the 32 cm diameter of the standard AAPM phantom used to determine CTDIvol.

Table 1: Demographic data of 1178 patients.

Anatomic region
CT scanner
Number ()
Age (years)
Height (cm)
Body weight (kg)
BMI (kg/m²)

Chest
40- and 128- slice
616

40-slice
264

128-slice
352

Abdomen-pelvis
40- and 128-slice
562

40-slice
239

128-slice
323

Abbreviation: BMI: body mass index.

Table 2: Body size and radiation dose of 1178 patients.

Anatomic region
CT scanner
Number ()
(cm)

CTDIvol (mGy)
SSDE (mGy)

Chest
40- and 128-slice
616

40-slice
264

128-slice
352

Abdomen-pelvis
40- and 128-slice
562

40-slice
239

128-slice
323

Abbreviation: : water-equivalent diameter; : size-dependent conversion factor; CTDIvol: volume CT dose index; SSDE: size-specific dose estimate.

3.2. Correlation of with Two Biometric Characteristics

Spearman’s correlation analysis showed that no statistically significant correlations occurred between age and for chest and abdomen-pelvis CT (, -0.010, all ). Pearson’s correlation analysis indicated that both body weight and BMI were strongly correlated with for the chest (, 0.87, all ) and abdomen-pelvis (, 0.86, all ). Steiger’s test showed a statistically significant difference associated with the correlation coefficients of body weight and BMI with as a common variable for the chest (, ). However, no significant difference was observed for two correlation coefficients of body weight and BMI and as a common variable for abdomen-pelvis using Steiger’s test (, ). Figure 2 shows the correlations and linear regressions relating to body weight, BMI, and automatically derived from axial images of modeled patients. The Bland-Altman test indicated that there was good agreement between measured and predicted values based on the linear equations for chest and abdomen-pelvis CTs (Figure 1).

(a)

(b)

(c)

(d)

Figure 2: Scatterplots representing correlations between and two body size metrics of body weight and BMI. (a) Correlation of body weight with for chest. (b) Correlation of BMI with for chest. (c) Correlation of body weight with for abdomen-pelvis. (d) Correlation of BMI with for abdomen-pelvis. Linear equations and coefficients of determination () from the linear regression analysis are also shown.

3.3. Accuracy of SSDEweight and SSDEBMI

SSDEweight and SSDEBMI were  mGy and  mGy in chest CT examination, respectively (, ); SSDEweight and SSDEBMI were  mGy and  mGy in abdomen-pelvic CT examination, respectively (, ). Compared with reference values of SSDE, slightly greater dispersions were observed for SSDEweight and SSDEBMI with respect to standard deviation, range, and coefficient of variation, as shown in Table 3. Maximal mean root-mean-square errors of SSDEweight and SSDEBMI were less than 11%, as shown in Table 4.

Table 3: Mean values and variation of radiation dose in chest and abdomen-pelvic CT examination across all verified patients.

Parameters
Chest
Abdomen

SSDE
SSDEweight
SSDEBMI
SSDE
SSDEweight
SSDEBMI

Mean values (mGy)

Range (mGy)
13.77
16.83
14.41
10.62
10.70
12.51

COV (%)
27.78
29.39
28.60
12.19
14.34
14.14

Abbreviation: SSDE: size-specific dose estimate; SSDEweight: size-specific dose estimate based on body weight; SSDEBMI: size-specific dose estimate based on body mass index; COV (%): the coefficient of variation calculated using standard deviation divided by mean values. Note: mean values are reported as the . Range is the difference between maximum and minimum values of radiation dose.

Table 4: Mean root-mean-square errors of SSDEweight and SSDEBMI across all verified patients (mGy, %).

Radiation dose
All patients
Males
Females
Underweight
Normal weight
Overweight
Obese

Chest

    Numbers
308
213
95
38
135
127
8

    SSDEweight
0.62 (6.10%)
0.53 (5.19%)
0.77 (7.69%)
0.41 (5.05%)
0.48 (4.99%)
0.71 (6.38%)
1.49 (10.86%)

    SSDEBMI
0.57 (5.65%)
0.62 (6.07%)
0.46 (4.60%)
0.40 (4.93%)
0.60 (6.24%)
0.57 (5.13%)
0.84 (5.25%)

Abdomen-pelvis

    Numbers
281
168
113
32
124
120
5

    SSDEweight
0.76 (5.61%)
0.62 (4.55%)
0.89 (6.64%)
0.61 (5.15%)
0.71 (5.57%)
0.83 (5.67%)
1.14 (6.63%)

    SSDEBMI
0.71 (5.22%)
0.55 (4.02%)
0.67 (5.00%)
0.73 (6.23%)
0.65 (5.10%)
0.73 (4.98%)
1.23 (7.15%)

Abbreviation: SSDEweight: size-specific dose estimate based on body weight; SSDEBMI: size-specific dose estimate based on body mass index.

4. Discussion

Prior to estimating patients’ absorbed radiation dose, accurate measurements of patient size are necessary, as the radiation dose is closely related to patient factors as well as the output of the CT scanner. Similarly to previous studies [18, 23], we found that the actual patient sizes for chest and abdomen-pelvis were considerably less than the 32 cm diameter standard AAPM phantom. Irrespective of inhomogeneous X-ray attenuation, a 32 cm diameter cannot accurately represent a realistic patient size in terms of geometric dimension. Hence, the radiation doses expressed via CTDIvol were underestimated compared to actual values [18, 23, 24].

Previous studies have shown that the measurement of some patient size metrics, including age, height, body weight, BMI, body circumference, and body diameter, could be integrated into routine clinical practice for dose optimization [15, 26, 27]. As suggested by AAPM, and are suitable for quantifying patient size [10, 11]. Compared with , can be considered the gold standard for patient size, as it can be adapted to different body shapes, including complex deformities and asymmetric contours, and accounts for X-ray attenuations of heterogeneous tissues within the body. However, owing to the unavailability of commercial or in-house software in many health institutions worldwide, the measurement of is often a complex and time-consuming process for routine CT radiation dose calculations in radiological practice. The need for a simplified and efficient method to obtain is clear, and, in the meantime, accurate SSDEs are required.

The primary aim of this study was to determine the correlations between body weight, BMI, and . Because age is an easily obtainable factor, the correlation between age and was also assessed to determine whether age can be used as a surrogate to estimate SSDEs. Unfortunately, unlike the observation that increases with age for the pediatric population [28], no significant correlation was observed between and age for chest and abdomen-pelvis CT, which is attributed to the lack of growth of with the increasing ages of adult patients. However, similar to previous results reported by Menke [20, 25], we also found that both body weight and BMI strongly correlated with in chest and abdomen-pelvis, despite different populations in Asian-Pacific and European-American regions. Thus, age cannot be used to generate for calculating SSDEs in adult patients. In contrast, both body weight and BMI can be used as surrogates for accurate patient size, while accounting for geometric dimensions and X-ray attenuation characteristics. However, it is important to note that an automated software program is required to calculate , which must be verified first. Only then is the implementation of surrogates recommended in clinical practice.

In terms of correlation coefficients, a significant difference was found between body weight and BMI with a common variable of in chest. However, was in good agreement with the measured , comparable to for chest. Our results also reveal that accurate measurements of patient size, i.e., , can be predicted using both body weight and BMI in abdomen-pelvis. With regard to the accuracy of SSDEs based on body weight and BMI, overall mean root-square-mean errors were less than 6.10% in chest and abdomen-pelvis CTs across all verified patients, although SSDEweight and SSDEBMI had slightly greater dispersion than the reference SSDEs. Additionally, a higher correlation coefficient between BMI and generated a smaller overall mean root-square-mean error of SSDEBMI compared with SSDEweight in chest and abdomen-pelvis across all verified patients, even when the two values of SSDEweight and SSDEBMI were similar. This result may be attributed to the fact that BMI, which takes height into account, is a better indicator of body size. For males, SSDEweight had a smaller overall mean root-square-mean error than SSDEBMI in the chest CT, while the opposite was true for the abdomen-pelvis region. For females, SSDEBMI had a smaller overall mean root-square-mean error than SSDEweight in either chest or abdomen-pelvis regions. These different results of males and female may be related to the distribution of fat. In general, fat is more distributed in the abdomen-pelvic region, and females tend to have relatively more fat content compared with males [29]. Furthermore, the phenomena can be characterized by BMI to some extent [29]. Hence, our results demonstrate that body weight surrogates were preferable for males in the chest region, and BMI for males in the abdomen-pelvic region and for females in chest and abdomen-pelvic regions, although differences between SSDEweight and SSDEBMI were relatively small. In this study, we further investigated the accuracy of SSDEweight and SSDEBMI for eight different patient BMI populations. Our results indicate that the maximal mean root-square-mean errors were achieved for obese patients, which may be related to the smaller sample size of obese patients in this study.

This study had some limitations. First, it was a retrospective study, and patient choice was a possible bias. However, the patient population was retrieved consecutively. It is likely that the values of and SSDE obtained were approximated, compared with a prospective study. Second, although the patient population had a wide BMI spectrum, a relatively small sample size of underweight and obese patients was available and no pediatric patients were enrolled. Therefore, further study is needed to determine whether our findings are applicable for overly small and overly large patients, and our results apply only for adult patients. Third, patient data in this study were acquired only from one health institution. As different CT scanners and subtly different CT protocol details, including pitch factor, dose modulation mode, scan range, and respiratory phase, are available in individual health institutions, our results need to be confirmed using data from multiple health institutions. Finally, the BMI of our patients was lower than that of Westerners, although some overweight and obese patients were enrolled in the study. Our results presented correspond to Asian-Pacific body types and may not be generalized for other body types.

5. Conclusion

Body weight and BMI were found to closely correlate with and can be used as surrogates of anatomic regions of the chest and abdomen-pelvis. On the basis of the linear equations of body weight, BMI, and , SSDEs can be calculated efficiently and accurately in chest and abdomen-pelvis CTs, and the predicted values of SSDEweight and SSDEBMI are comparable to the reference SSDEs derived from axial images with an overall mean root-mean-square error of less than 0.76 mGy or 6.10% for chest and abdomen-pelvis CT. Therefore, we conclude that body weight and BMI can be used to simply and accurately estimate SSDE in the absence of software that automatically calculates SSDEs.

Data Availability

The data used within the present study are available from the first author.

Conflicts of Interest

The authors declare that there is no conflict of interest regarding this paper.

Acknowledgments

The authors would like to acknowledge the technical team in Shenshi Technology, Co., Ltd for their technical assistance in the in-house program for this paper.

References

K. Shimada and M. Kai, “Calculating disability-adjusted life years (DALY) as a measure of excess cancer risk following radiation exposure,” Journal of Radiological Protection, vol. 35, no. 4, pp. 763–775, 2015.

V. K. Ivanov, V. V. Kashcheev, S. Y. Chekin et al., “Estimating the lifetime risk of cancer associated with multiple CT scans,” Journal of Radiological Protection, vol. 34, no. 4, pp. 825–841, 2014.

R. A. Nievelstein, “Radiation exposure and mortality risk from CT and PET imaging of patients with malignant lymphoma,” European Radiology, vol. 22, no. 9, pp. 1946–1954, 2012.

D. A. Gibson, R. E. Moorin, J. Semmens, and D.'. A. J. Holman, “The disproportionate risk burden of CT scanning on females and younger adults in Australia: a retrospective cohort study,” Australian and New Zealand Journal of Public Health, vol. 38, no. 5, pp. 441–448, 2014.

D. D. Zhou, P. Sun, Z. Jia et al., “Multisection computed tomography: results from a Chinese survey on radiation dose metrics,” Journal of the Chinese Medical Association, vol. 82, no. 2, pp. 155–160, 2019.

W. A. Kalender, “Dose in X-ray computed tomography,” Physics in Medicine and Biology, vol. 59, no. 3, pp. R129–R150, 2014.

N. E. Bolus, “NCRP report 160 and what it means for medical imaging and nuclear medicine,” Journal of Nuclear Medicine Technology, vol. 41, no. 4, pp. 255–260, 2013.

C. H. McCollough, S. Leng, L. Yu, D. D. Cody, J. M. Boone, and M. F. McNitt-Gray, “CT dose index and patient dose: they are not the same thing,” Radiology, vol. 259, no. 2, pp. 311–316, 2011.

J. A. Seibert, J. M. Boone, S. L. Wootton-Gorges, and R. Lamba, “Dose is not always what it seems: where very misleading values can result from volume CT dose index and dose length product,” Journal of the American College of Radiology, vol. 11, no. 3, pp. 233–237, 2014.

American Association of Physicists in Medicine, Report of AAPM TG 204: Size-specific dose (SSDE) in pediatric and adult body CT examinations, AAPM, USA, 2011.

American Association of Physicists in Medicine, Report of AAPM TG 220: use of water equivalent diameter for calculating patient size and size-specific dose estimates (SSDE) in CT, AAPM report, USA, 2014.

J. Boos, R. S. Lanzman, P. Heusch et al., “Does body mass index outperform body weight as a surrogate parameter in the calculation of size-specific dose estimates in adult body CT?” The British Journal of Radiology, vol. 89, no. 1059, 2016.

R. D. A. Khawaja, S. Singh, B. Vettiyil et al., “Simplifying size-specific radiation dose estimates in pediatric CT,” AJR American Journal of Roentgenology, vol. 204, no. 1, pp. 167–176, 2015.

S. O’Neill, R. G. Kavanagh, B. W. Carey, N. Moore, M. Maher, and O. J. O’Connor, “Using body mass index to estimate individualised patient radiation dose in abdominal computed tomography,” European Radiology Experimental, vol. 2, no. 1, 2018.

S. Kritsaneepaiboon, S. Eng-chuan, and S. Yoykaew, “Can patient's body weight represent body diameter for pediatric size-specific dose estimate in thoracic and abdominal computed tomography?” Journal of Clinical Imaging Science, vol. 9, p. 24, 2019.

B. Karmazyn, H. Ai, P. Klahr, F. Ouyang, and S. G. Jennings, “How accurate is size-specific dose estimate in pediatric body CT examinations?” Pediatric Radiology, vol. 46, no. 9, pp. 1234–1240, 2016.

C. S. Burton and T. P. Szczykutowicz, “Evaluation of AAPM Reports 204 and 220: estimation of effective diameter, water-equivalent diameter, and ellipticity ratios for chest, abdomen, pelvis, and head CT scans,” Journal of Applied Clinical Medical Physics, vol. 19, no. 1, pp. 228–238, 2017.

J. Boos, P. Kröpil, O. T. Bethge et al., “Accuracy of size-specific dose estimate calculation from center slice in computed tomography,” Radiation Protection Dosimetry, vol. 178, no. 1, pp. 8–19, 2018.

I. Ikuta, G. I. Warden, K. P. Andriole, R. Khorasani, and A. Sodickson, “Estimating patient dose from x-ray tube output metrics: automated measurement of patient size from CT images enables large-scale size-specific dose estimates,” Radiology, vol. 270, no. 2, pp. 472–480, 2014.

S. Sarmento, B. Mendes, and M. Gouvêa, “Automatic calculation of patient size metrics in computed tomography: what level of computational accuracy do we need?” Journal of Applied Clinical Medical Physics, vol. 19, no. 1, pp. 218–227, 2017.

A. Iriuchijima, Y. Fukushima, T. Nakajima, Y. Tsushima, and A. Ogura, “Simple method of size-specific dose estimates calculation from patient weight on computed tomography,” Radiation Protection Dosimetry, vol. 178, no. 2, pp. 208–212, 2018.

WHO, “Appropriate body-mass index for Asian populations and its implications for policy and intervention strategies,” The Lancet, vol. 363, no. 9403, pp. 157–163, 2004.

S. Leng, M. Shiung, X. Duan, L. Yu, Y. Zhang, and C. H. McCollough, “Size-specific dose estimates for chest, abdominal, and pelvic CT effect of intrapatient variability in water-equivalent diameter,” Radiology, vol. 276, no. 1, pp. 184–190, 2015.

J. Boos, C. Thomas, E. Appel et al., “Institutional computed tomography diagnostic reference levels based on water-equivalent diameter and size-specific dose estimates,” Journal of Radiological Protection, vol. 38, no. 2, pp. 536–548, 2018.

J. Menke, “Comparison of different body size parameters for individual dose adaptation in body CT of adults,” Radiology, vol. 236, no. 2, pp. 565–571, 2005.

B. Alikhani, T. Getzin, T. F. Kaireit et al., “Correlation of size-dependent conversion factor and body-mass-index using size-specific dose estimates formalism in CT examinations,” European Journal of Radiology, vol. 100, pp. 130–134, 2018.

P. L. Kleinman, K. J. Strauss, D. Zurakowski, K. S. Buckley, and G. A. Taylor, “Patient size measured on CT images as a function of age at a tertiary care children's hospital,” AJR American Journal of Roentgenology, vol. 194, no. 6, pp. 1611–1619, 2010.

W. Huda and S. V. Tipnis, “Doses metrics and patient age in CT,” Radiation Protection Dosimetry, vol. 168, no. 3, pp. 374–380, 2016.

W. S. Lee, “Body fatness charts based on BMI and waist circumference,” Obesity, vol. 24, no. 1, pp. 245–249, 2016.