Biochemistry Research International
Volume 2020 (2020), Article ID 9659412, 12 pages
https://doi.org/10.1155/2020/9659412

Review Article

Animal Models of Osteochondral Defect for Testing Biomaterials

Xiangbo Meng,1,2 Reihane Ziadlou,³ Sibylle Grad,³ Mauro Alini,³ Chunyi Wen,⁴ Yuxiao Lai,² Ling Qin,2,5 Yanyan Zhao,¹ and Xinluan Wang2,5

¹College of Pharmaceutical Sciences, Hebei University, Baoding, China
²Translational Medicine R&D Center, Institute of Biomedical and Health Engineering, Shenzhen Institutes of Advanced Technology, Chinese Academy of Sciences, Shenzhen, China
³AO Research Institute Davos, Clavadelerstrasse 8, 7270 Davos Platz, Switzerland
⁴Department of Biomedical Engineering, Faculty of Engineering, The Hong Kong Polytechnic University, Hung Hom, Kowloon, Hong Kong SAR, China
⁵Musculoskeletal Research Laboratory, Department of Orthopaedics & Traumatology, The Chinese University of Hong Kong, Hong Kong SAR, China

Correspondence should be addressed to Yanyan Zhao; zhaoyany606@163.com and Xinluan Wang; xl.wang@siat.ac.cn

Received 21 October 2019; Accepted 7 January 2020; Published 28 January 2020

Academic Editor: Paul W. Doetsch

Copyright © 2020 Xiangbo Meng et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

Abstract.

The treatment of osteochondral defects (OCD) remains a great challenge in orthopaedics. Tissue engineering holds a good promise for regeneration of OCD. In the light of tissue engineering, it is critical to establish an appropriate animal model to evaluate the degradability, biocompatibility, and interaction of implanted biomaterials with host bone/cartilage tissues for OCD repair in vivo. Currently, model animals that are commonly deployed to create osteochondral lesions range from rats, rabbits, dogs, pigs, goats, and sheep horses to nonhuman primates. It is essential to understand the advantages and disadvantages of each animal model in terms of the accuracy and effectiveness of the experiment. Therefore, this review aims to introduce the common animal models of OCD for testing biomaterials and to discuss their applications in translational research. In addition, we have reviewed surgical protocols for establishing OCD models and biomaterials that promote osteochondral regeneration. For small animals, the non-load-bearing region such as the groove of femoral condyle is commonly chosen for testing degradation, biocompatibility, and interaction of implanted biomaterials with host tissues. For large animals, closer to clinical application, the load-bearing region (medial femoral condyle) is chosen for testing the durability and healing outcome of biomaterials. This review provides an important reference for selecting a suitable animal model for the development of new strategies for osteochondral regeneration.

1. Introduction

Osteochondral defects (OCD) are a common condition caused by severe trauma, sports injuries, or physical diseases, leading to joint pain, deformity, and dysfunction [1]. Joint injuries caused by trauma and sports accidents often progress into osteoarthritis (OA). So, OCD are also a significant cause of OA [2]. OA has been reported to be the third most common musculoskeletal disease in the world [3]. The global prevalence of OA for persons older than 60 years is estimated at 33.6% for women and 24.3% for men [4]. As cartilage has no vasculature and lymphatic vessels and mature chondrocytes have limited proliferation and migration capabilities, cartilage regeneration remains a major challenge. OCD including lesions or degeneration of cartilage, subchondral bone, and bone-cartilage interfaces are notorious for being unable to heal. In order to repair OCD, the tissue complex of bone, cartilage, and bone-cartilage interfaces must be taken into account for repair and regeneration [5, 6]. Yet, those OCD are difficult to treat because the cartilage and the subchondral bone are tissues with different intrinsic healing capacities.

The current clinical treatments for repair of OCD are only palliative rather than curative [7]. The common goal of successful treatments is to relieve pain, repair damaged tissue, and improve joint function [8]. Current methods for treatment of cartilage lesions mainly include medical treatments (nonsteroid anti-inflammatory drugs (NSAIDs), pain killers, and hormones, etc.) and surgical treatment (arthroscopic lavage and debridement, cell-based therapy, and tissue-based therapy) [9]. Unfortunately, the medical treatments only relieve pain, rather than restoring the structural integrity of the articular cartilage [10], and the surgical treatments cannot restore neo-tissue close to normal cartilage [9]. Therefore, the treatment effect is not ideal, and the development of new treatment strategies is an urgent need. However, any new treatment strategy must be tested in animals to ensure its safety, feasibility, and effectiveness before clinical testing. It is very important to simulate human symptoms using appropriate animal models before clinical trials. At the same time, animal models are effective for developing OCD repair methods. Therefore, it is crucial to establish a suitable animal model for evaluating the effectiveness and safety of new treatment strategies.

In this review, we summarize the benefits and limitations of each species for reproducing specific defects, analyze and compare the similarities between animal models and human clinical conditions, and emphasize the factors that need to be considered when selecting animals.

2. Selection Criteria and Critical Size

2.1. General Selection Criteria

The ideal animal model should be as close to the clinical setting as possible, have biological similarity, and be a suitable model for cartilage physiology [11, 12]. A range of factors must be considered to select an applicable animal model for OCD regeneration. Before selecting an ideal animal model, it is crucial to decide whether a small or large animal model would be suitable for a particular OCD regeneration. The small animal models for OCD regeneration include rats and rabbits [13], while large animal models for OCD repair include dogs, pigs, sheep, goats, and horses [14]. Every animal has its advantages and limitations. When assessing the clinical potential of new strategies, the animal model that most closely represents human anatomy and physiology should be selected [15]. In addition, when investigating articular osteochondral repair in vivo, the factors to be considered include joint size, cartilage thickness, defect depth and diameter, skeletal maturity age, joint load distribution, and affordability and convenience of animal handling (Table 1) [16–18].

Table 1: Comparison of age, cartilage, and defect size in different species.

Species
Age of skeletal maturity
Cartilage thickness
Cartilage volume
Critical-sized defect
Common defect depth

Rat
7 months
0.1 mm
2.17 mm³
1.4 mm
1.0–2.0 mm

Rabbit
9 months
0.3 mm
53 mm³
3.0 mm
3.0–5.0 mm

Dog
12‐24 months
0.95 mm
82.39 mm³
4.0 mm
10–12 mm

Pig
18 months
1.5 mm
107.47 mm³
6.3 mm
8–10 mm

Sheep
2‐3 years
0.45 mm
359.54 mm³
7.0 mm
6–13 mm

Goat
2‐3 years
1.1 mm
251.65 mm³
6.0 mm
6–12 mm

Horse
2‐4 years
1.75 mm
334.73 mm³
4.0 mm/9.0 mm
10 mm

Monkey
10 years [16]
0.5–0.7 mm [17]
—
—
2–4 mm

Human
18–22 years
2.35 mm
552.25 mm³
—
—

2.2. Critical Size of OCD

The critical size defect is defined as the smallest defect size (in diameter) the animal cannot self-repair without intervention [19]. In animal experiments, the understanding of critical-sized defects is crucial for reducing costs and animal suffering, at the same time still providing reliable data on the research results. So, the critical size of the defect should be considered to select the appropriate animal model for OCD repair. Katagiri et al. found that, in the rat knee, OCD with a diameter of 1.4 mm and a depth of 1.0 mm could not spontaneously recover the osteochondral unit, thus defining the critical size of rat knee osteochondral injury [20], whereby the mean animal weight is about 0.3 kg. The critical-sized defect of the rabbit knee has been defined as 3 mm, which can prevent spontaneous healing [21]. This dimension has, however, been questioned due to reported spontaneous healing [11]. Larger defects with diameters of 4 mm to 5 mm may be more appropriate [22, 23]. For the canine model with a mean weight of about 30 kg, the critical size of the OCD has been considered to be 4 mm [19, 24]. Gotterbarm et al. considered that OCD of 6.3 mm should be defined as the critical-sized defect in the porcine model with a mean weight of about 38 kg [25]. The critical-sized defect in sheep models has been considered to be 7 mm, while its average weight is about 70 kg [11]. In the goat model, 6 mm OCD proved to be unable to heal spontaneously and has been defined as a critical dimension defect, while the average weight is about 48 kg [26, 27]. The critical-sized defects in the equine femoral trochlear and condyle models are considered to be around 9 mm [28, 29]. In addition, Salonius et al. [30] reported 4 mm in diameter as critical osteochondral lesion size in the equine carpal joint model. The horse is the largest animal model for articular cartilage regeneration with an average weight of 400 kg.

3. Small Animal Models

Small animal models are crucial in “proof-of-concept” studies, especially for testing biosafety. In these studies, concepts are validated and in vitro results are first translated in vivo. Small animals are inexpensive, easy to handle and feed, and often used to investigate the pathophysiology and pathogenesis of the disease [31]. However, the limitations of small animal models for OCD regeneration consist in the small size of the knee joint and the thin cartilage thickness [32, 33]. It is therefore difficult to design surgical OCD models suitable for comparison with human conditions.

3.1. Rats

The rat models used for OCD regeneration have several advantages, as rats are inexpensive, easy to handle and house, and clinically more relevant than mice. The skeletal maturity of rats is approximately 7 months [34]. Rats aged between 9 and 12 weeks have been used to evaluate the degradation rate and safety profile of biomaterials, whereby the experimental period of implants generally lasts 8–12 weeks (Table 2). The critical size of rat OCD was defined as 1.4 mm [20]. The cartilage thickness of the medial femoral condyle in rat is around 0.1 mm [11]. Most commonly, OCD of 2.0 mm diameter and 2.0 mm depth on the trochlear groove of the femur have been used for the assessment of biomaterial strategies. However, their small joint size and thin cartilage remain the main limitations for testing of biomaterials in the rat OCD model [20]. Therefore, the rat model seems to be applicable for preliminary in vivo evaluation but not for preclinical studies.

Table 2: Examples of studies using rat osteochondral defect models.

Authors
Age
Defect size (diameter × depth)
Location
Endpoint
Material tested

Lee and Im [35]
12 weeks
2 mm × 2 mm
The trochlear groove of the femur
8 weeks
SOX trio-co-transduced ASCs

Muttigi et al. [36]
12 weeks
2 mm × 2 mm
The center of the groove
12 weeks
Matrilin-3/mesenchymal stem cell

Mahmoud et al. [37]
10 weeks
2 mm × 2 mm
The patellar groove of the femur
4, 12 weeks
Muse cells

Dahlin et al. [38]
10–12 weeks
2 mm × 2 mm
The center of the trochlear groove
4, 8 weeks
PCL scaffold/MSC

Li et al. [39]
12 weeks
1.5 mm × 2 mm
The trochlear groove
6, 12 weeks
PLGA/HA-MSC

3.1.1. Experimental Protocol of Animal Surgeries

In typical procedures, animals were anaesthetized and shaved and the knee was disinfected. A medial temporal medial longitudinal incision was made to expose the synovium of the knee joint, and then the trochlear groove was further exposed after the lateral patellar luxation. The defect (1.5–2 mm diameter and 2 mm depth) was drilled in the center of the trochlear groove. The biomaterials were implanted, after irrigating the joint with sterile isotonic saline. Lastly, the patella was relocated and the wound sutured in layers [40].

3.1.2. Applications of Rat OCD Model for Testing of Osteochondral Repair Materials

Using a 12-week-old rat model, Lee and Im [35] found that SOX trio-co-transduced adipose tissue derived stem cells (ASCs) in fibrin gel promoted the OCD (2 mm diameter and 2 mm depth) regeneration and attenuated the progression of OA caused by surgery. Muttigi et al. [36] created an OCD of 2 mm diameter and 2 mm depth in the patellar groove of the femur. The model was created to assess the effect of matrilin-3 codelivery with ASCs. They found that matrilin-3 codelivery with ASCs enhanced the formation of cartilage tissue and concluded that matrilin-3 may be an attractive biochemical factor that promotes stem cell repair of articular cartilage. Mahmoud et al. [37] used 10-week-old rat to create an OCD model in the femur patellar groove (2 mm diameter and 2 mm depth) to test the efficacy of multilineage-differentiating stress-enduring (Muse) cell transplantation for OCD repair. They found that injection of Muse cells was a promising method to repair an OCD, especially when subchondral bone is covered by fibrous tissue. Dahlin et al. [38] cultured bovine articular chondrocytes with rat mesenchymal stem cells (MSCs) onto electrospun poly(3-caprolactone) (PCL) scaffolds and implanted them into OCD (2 mm diameter and 2 mm depth) in the rat trochlear groove. The results showed cocultures of articular chondrocytes and MSCs have the potential to repair cartilage defects in vivo. Li et al. [39] combined poly(lactide-coglycolide)/hydroxyapatite (PLGA/HA) composite scaffolds with MSCs to successfully repair cartilage defects, while these implants may also be valuable for other clinical applications.

3.2. Rabbit

The rabbit model provides a suitable small animal model for assessing the repair of OCD, as rabbits have larger joints for surgical procedures [41]. The age of skeletal maturity in rabbits is 9 months. Rabbits aged between 3 and 8 months have been used to evaluate the degradation rate and safety of biomaterials, and the experimental period of implants generally lasted 8–24 weeks (Table 3). The cartilage of rabbit is relatively thin, showing an average cartilage thickness of 0.44 ± 0.08 mm for the trochlear groove and 0.3 ± 0.07 mm for the medial femoral condyle [47]. In addition, the subchondral bone of the rabbit trochlea (386 ± 160 μm) is similar to the human medial femoral condyle (213 ± 116 μm), and both have a relatively thin bone plate and a more porous and lower density subchondral bone [48]. The relative length of the trochlear groove is greater compared with the human knee joint, which is probably related to the mainly squatting posture of the animal. Besides, the rabbit has faster skeletal change and bone turnover in comparison with other species [49]. Defects have been created in the femoral trochlea [50, 51], the medial femoral condyle [52, 53], and the lateral femoral condyle [54]. OCD of 3.0–5.0 mm diameter and 2.0–5.0 mm depth are often used to evaluate biomaterials in rabbit models.

Table 3: Examples of studies using rabbit osteochondral defect models.

Authors
Age/weight
Defect size (diameter × depth)
Location
Endpoint
Material tested

Liao et al. [42]
2–2.5 kg
4 mm × 3 mm
The trochlear groove
6, 12, and 18 weeks
CSMA/PECA/GO hybrid scaffold

Bauer et al. [43]
8 months
4 mm × 5 mm
The medial trochlear groove
4 and 12 weeks
Hyaluronic acid thioester

Ruan et al. [44]
6 months
4 mm × 3 mm
The medial trochlear groove
4, 8, and 12 weeks
SF/CS/nHA phase scaffold

Meng et al. [45]
4–6 months
4 mm × 2 mm
The trochlear groove
6, 12, and 24 weeks
AMP-E7/BM-MSC

Zhang et al. [46]
2.5–3 kg
4 mm × 4 mm
The patellar groove
6 and 12 weeks
COL-nanofiber and COL scaffolds

3.2.1. Experimental Protocol of Animal Surgeries

In most studies, the creation of an OCD was based on the following protocol. The rabbits were anaesthetized; then, a medial peripatellar incision was made to expose the knee joint. The patella was dislocated laterally, and the articular surface of the distal femur was exposed. A cylindrical OCD was made using an electrical trephine in the trochlear groove (Figure 1). After irrigating the joint with sterile isotonic saline, the biomaterials were implanted. Lastly, the patella was relocated and the wound sutured in layers [50, 51].

Figure 1: The process of the OCD regeneration in rabbits. A: the OCD were generated by electric drill in the femoral patellar groove; B: a 3.2 mm in diameter and 3.0 mm deep OCD was obtained; C: the biomaterial was implanted into the OCD.

3.2.2. Applications of Rabbit OCD Models for Testing of Osteochondral Repair Materials

Liao et al. [42] prepared a novel hybrid scaffold composed of methacrylated chondroitin sulfate (CSMA), poly(ethylene glycol) methyl ether-ε-caprolactone-acryloyl chloride (MPEG-PCL-AC, PECA was used as abbreviation for MPEG-PCL-AC), and graphene oxide (GO) and evaluated its application for cartilage regeneration using the rabbit OCD model. Micro-CT and histological observations showed that the CSMA/PECA/GO scaffold group had better chondrocyte morphology, integration, and continuous subchondral bone and thicker newly formed cartilage. Bauer et al. [43] used a 4 mm diameter and 5 mm depth rabbit OCD model to test hyaluronic acid thioester to promote articular cartilage regeneration. Ruan et al. [44] synthesized a novel biphasic scaffold, which contained a silk-fibroin/chitosan (SF/CS) and an osteoblastic phase (SF/CS/nHA). Bone marrow derived mesenchymal stem cells (BMSCs) showed high cell viability on this scaffold. This scaffold may be an attractive implant that has potential applications in the treatment of OCD. Meng et al. [45] established a functional scaffold named APM-E7 by conjugating a BMSCs affinity peptide (E7) onto the acellular peritoneum matrix (APM). Then, they established a full-thickness OCD model, 4 mm in diameter and 2 mm in height, in 6-month-old rabbits to test the APM-E7 scaffold. The results showed APM-E7 scaffold could support cell attachment. Zhang et al. [46] fabricated a bilayer microporous scaffold with collagen and electrospun poly-L-lactic acid nanofibers (COL-nanofiber) and applied it in a rabbit OCD model. The results showed that implantation of COL-nanofiber scaffold with cells induced cartilage and subchondral bone formation.

4. Large Animal Models

The large animals, such as goats, sheep, pigs, dogs, and horses, have the advantages of joint size and cartilage thickness and also have the most similar clinical lesions to humans [55]. Although large animals may be closer to human clinical conditions, they require greater logistic, financial, and ethical considerations. When planning in vivo studies, a multivariate analysis should be performed for each animal model. Ultimately, the scientific goals are crucial for determining the appropriate animal model [31]. According to available reports, the mean volume of human cartilage defects is around 552.25 mm³, and the diameter of human cartilage defects requiring treatment is usually 10 mm or more [56, 57]. However, in common animal models, the cartilage volume and cartilage thickness are smaller than in humans (Table 2) [11, 58].

4.1. Dog

The dog is considered to be a very friendly and loving partner over the world. The social and ethical issues associated with the use of dogs as preclinical and translational animal models are main reasons for their limited use [14]. Dogs are susceptible to cartilage diseases such as exfoliative osteochondritis and osteoarthritis, and dogs lack the ability to repair cartilage defects intrinsically [31]. Therefore, using this model to study osteoarthritis may be closer to humans. Dogs are also suitable for studies that require specific sports and rehabilitation protocols. Dog’s skeleton mature age is about 12 to 24 months. The thickness of the cartilage on the medial condyle of the dog has been reported to be 0.95 mm [11]. Defects have been located in the femoral trochlea [59], the medial femoral condyle [60], and both condyles concurrently and medial tibial plateau [61]. Defect diameters have ranged from 2 to 10 mm, and 4 mm is the most common one (Table 4).

Table 4: Examples of studies using dog osteochondral defect models.

Authors
Age
Defect size (diameter × depth)
Location
Endpoint
Material tested

Lv and Yu [59]
12 months
6 mm × 12 mm
The right knee joint
12 and 24 weeks
Nano-β-TCP/Col I/Col II/BMSCs

McCarty et al. [60]
—
4.5 mm × 10 mm
The medial femoral condyle
12 months
Osteochondral allograft

Salkeld et al. [61]
1.6 years
6 mm × 11 mm
The medial femoral condyle and medial tibial plateau surfaces
12, 24, and 52 weeks
Pyrolytic carbon scaffold and Co-Cr alloy scaffold

Yamazoe et al. [62]
1–3 years
5 mm × 4.5 mm
The femoral condyles
2, 4, and 10 weeks
Atelocollagen gel/MSCs

4.1.1. Experimental Protocol of Animal Surgeries

Dogs were anaesthetized intravenously. The dog was fixed on the operating table in a supine position and the hair was shaved over the knee joint. The operating field was disinfected, and an incision was created in the skin of the knee joint. The knee flexion was approximately 70°; a defect was created in the femoral trochlea, the medial femoral condyle, or condyles concurrently and medial tibial plateau. Scaffolds were implanted, and the wound layer was sutured [59].

4.1.2. Applications of Dog OCD Models for Testing of Osteochondral Repair Materials

Lv and Yu [59] investigated the articular OCD (6 mm diameter and 12 mm depth) repair using a composite lamellar scaffold of nano-β-tricalcium phosphate (β-TCP)/collagen (col) I and II with BMSCs in the canine knee joint. The composite lamellar scaffold was gradually degraded and absorbed, while new cartilage tissue was formed. Salkeld et al. [61] used a 6 mm diameter and 11 mm deep OCD in the medial femoral condyle of the canine knee to test a pyrolytic carbon implant. They found that the pyrolytic carbon as a hemiarthroplasty implant material was superior to cobalt-chromium (Co-Cr) alloy. In addition, pyrolytic carbon implants reduced wear, degradation, and cellular changes at the surface of the tibial cartilage. Yamazoe et al. [62] proposed that autologous transplantation of an atelocollagen gel containing canine-derived mesenchymal stem cells could not promote the repair of canine knee joint but rather the subchondral bone regeneration.

4.2. Pig

Pigs are considered to be a suitable animal model for mimicking human diseases and have widely been used in biomedical research [63, 64]. The pig joint size, weight requirements, and cartilage thickness are closer to humans than dogs and smaller animal models. In addition, the bone apposition rate and trabecular thickness of the mini-pig are similar to human bones. However, purchase and maintenance of pigs are very expensive. Pigs generally reach skeletal maturity in around 18 months [14]. Fisher et al. [65] reported a cartilage thickness of 1.5 mm at the medial femoral condyle level in mini-pig. Gotterbarm et al. [25] showed that 6.3 mm diameter OCD did not spontaneously heal in mini-pig, confirming the applicability of this pig breed to articular cartilage research. The large majority of the cartilage regeneration studies in the mini-pig are performed on the joint knee, involving the medial [66] or femoral condyles [67, 68], or femoral trochlea. Generally, 6 mm to 8 mm diameter or larger dimensions OCD are created, and the postoperative follow-up period is usually between 3 and 24 months (Table 5).

Table 5: Examples of studies using pig osteochondral defect models.

Authors
Age
Defect size (diameter × depth)
Location
Endpoint
Material tested

Christensen et al. [68]
19.8 months
6 mm × 8 mm
The medial trochlear and the lateral trochlear
6, 24 months
Autologous dual-tissue transplantation/autologous cartilage chips

Betsch et al. [66]
18–30 months
6 mm × 10 mm
The medial femoral condyle
26 weeks
EPO/BMAC/scaffold

Jagodzinski et al. [69]
14 months
7 mm × 10 mm
The medial or lateral femoral condyles
3 months
Bone marrow derived cell concentrates

4.2.1. Experimental Protocol of Animal Surgeries

After animals were anaesthetized, a 5 cm incision was created in the skin to expose the medial condyle. A cylindrical OCD was created in the knee joint. The implant was placed into the defect and taken care of to ensure that the scaffold was flushed with the surface of the surrounding articular cartilage. Lastly, the wound was sutured in layers [66].

4.2.2. Applications of Pig OCD Models for Testing of Osteochondral Repair Materials

Several studies on cartilage and cartilage defects have been reported using min-pig. Christensen and coauthors [68] created OCD of 6 mm diameter and 8 mm depth in the medial trochlear to investigate the role of cartilage chips. They found that the cartilage chips promoted the formation of fibrocartilage rather than fibrous tissue. Betsch et al. [66] found that the combination of erythropoietin (EPO) and bone marrow aspirate concentrate (BMAC) could promote osteochondral healing in mini-pig OCD. Jagodzinski et al. [69] found that stem cell concentrates enhanced the attachment of new bone but did not enhance the mechanical properties and histological appearance of cartilage regenerates in mini-pig OCD models.

4.3. Sheep

Sheep is one of the commonly used animal models in orthopaedic research. The anatomy of the knee is similar to humans. However, due to the thinness of the cartilage, most of the defects are located in the subchondral bone, and the skeletal maturation is later, representing certain limitations [11]. Sheep aged between 2 and 3 years have been used to evaluate the degradation rate and safety profile of biomaterials, and the experimental period of implants generally lasted for 16–52 weeks. The critical-sized defect has been reported as 7 mm. The cartilage thickness of the medial femoral condyle is approximately 0.45 mm. The location of the cartilage defects in the sheep model has involved the medial femoral condyle [67, 70, 71], both femoral condyles [72, 73], and the femoral trochlea [70]. OCD with a diameter of 6–8 mm and a depth of 5–13 mm were used for the assessment of biomaterial strategies (Table 6).

Table 6: Examples of studies using sheep osteochondral defect models.

Authors
Age
Defect size (diameter × depth)
Location
Endpoint
Material tested

Schlichting et al. [70]
2 and 3 years
7.3 mm × 10 mm
The femoral condyles
3, 6 months
Stiff scaffold

Bernstein et al. [71]
2–4 year
7 mm × 25 mm
The femoral condyles
6, 12, 26, and 52 weeks
β-TCP/chondrocytes

Mohan et al. [72]
>3.5 years
6 mm × 6 mm
MFCs and LFCs
1 year
PLGA/β-TCP

Yucekul et al. [74]
—
8 mm × 10 mm
The lateral condyles
3, 6 and 12 months
PLLA/PCL/β-TCP

Mrosek et al. [75]
—
8 mm × 13 mm
The medial femoral condyle
16 weeks
Trabecular metal with an autologous periosteum graft

4.3.1. Experimental Protocol of Animal Surgeries

The sheep were anaesthetized; then, sheep were placed in dorsal recumbency. The skin on the right knee was sterilized and was ready for sterile surgery. The lateral para-aortic joint was incised to expose the medial and lateral femoral condyles. An ideal OCD was created in the medial and lateral femoral condyles using a suitable drill bit. After irrigating the joint with sterile isotonic saline, the biomaterials were implanted. Lastly, the wound was sutured in layers [72].

4.3.2. Applications of Sheep OCD Models for Testing of Osteochondral Repair Materials

Schlichting et al. [70] created an 8 mm in diameter and 15 mm deep OCD in the femoral condyles of 24 sheep to prove that stiff scaffolds could improve bone and cartilage regeneration. Bernstein et al. [71] indicated that microporous β-TCP scaffolds with chondrocytes were favorable for the treatment of OCD using the sheep model. Mohan et al. [72] compared microfracture and osteochondral methods using microsphere-based gradient plugs in sheep models. They found that gradient scaffolds had better cartilage repair capacity for OCD. Yucekul et al. [74] investigated a biodegradable, trilayered poly(glycolic acid) mesh/poly(l-lactic acid)-colorant tidemark layer/collagen type I and ceramic microparticle coated poly(l-lactic acid)-poly(ε-caprolactone) monolith) osteochondral plug indicated for the repair of cartilage defects (8 mm × 10 mm) in sheep. The scaffold proved to have a significant positive effect on the healing of osteochondral lesions. Mrosek et al. [75] demonstrated that trabecular metal (TM) was a very suitable material for reconstructing bone defects. TM enabled excellent bone ingrowth and rapid integration.

4.4. Goat

Goats are similar to sheep and are easy to raise and manage. The skeletal maturity of goats is similar to that of sheep, namely, about 2 to 3 years [11]. Goats aged between 2 and 4 years have been used to evaluate the degradation rate and safety profile of biomaterials, and the experimental period of implants generally lasted for 6–12 months (Table 7). The thickness of cartilage in goat is greater than that in sheep, and the subchondral bone is softer than that in sheep, which renders goats prone to osteochondral bone defects. Goat joints are usually larger than canine joints, and the most common defect size is 6 mm in diameter; this size has been proven to be unable to heal spontaneously. Defects have been created in the femoral trochlea, the medial femoral condyle, the lateral femoral condyle, and the talus [76, 77, 81]. If the limitations of large animal models can be overcome, including higher costs and adequate facility requirements, the goat model is a viable large animal model for cartilage and osteochondral lesions. However, the size of the lesions is still smaller than the human-related clinical diagnosis (Table 7).

Table 7: Examples of studies using goat osteochondral defect models.

Authors
Age/weight
Defect size (diameter × depth)
Location
Endpoint
Material tested

Zhang et al. [76]
12 months
6 mm × 8 mm
Knee joint
12, 24 weeks
BMSC-integrated osteochondral scaffolds

van Bergen et al. [77]
4-year-old
6 mm × 6 mm
Knee joint
24 weeks
Demineralized bone matrix

Kon et al. [78]
2-year-old
6 mm × 10 mm
The load-bearing medial femoral condyle
24 weeks
Aragonite-hyaluronate

Sun et al. [79]
22.5 kg
9 mm × 3 mm
The weight bearing area of the medial femoral condyle
24 weeks
Gene enhanced tissue engineering followed mosaicplasty

Pei et al. [80]
—
6 mm × 12 mm
The femoral medial condyle weight-bearing area
12 and 24 weeks
Tissue-engineered osteochondral graft

4.4.1. Experimental Protocol of Animal Surgeries

Surgery was performed under general anesthesia via joint surgery. Using retractors with the limb placed at maximal flexion, the implantation site was exposed. Defect was created and an implant was inserted via a surgical tool. The implant reached its final position in a press-fit manner, slightly below the articular surface. The knee capsule and skin were then sutured.

4.4.2. Applications of Goat OCD Models for Testing of Osteochondral Repair Materials

Goat has been successfully used as a model for OCD to evaluate new implants. Zhang et al. [76] fabricated BMSC-integrated osteochondral scaffolds that could promote the repair of OCD in goats. van Bergen et al. [77] used a 6 mm OCD in the talus goat model to evaluate the effectiveness of demineralized bone matrix (DBM) with and without platelet-rich plasma (PRP). They found that PRP would further enhance the regenerative capacity of DBM. Kon et al. [78] created critical-sized defects of 6 mm diameter and 10 mm depth in the medial femoral condyle of the knee joint. The defect model was created to test the in vivo effect of aragonite-hyaluronate (Ar-HA) scaffolds. They found that the Ar-HA scaffold might induce cartilage and subchondral bone regeneration. Sun et al. [79] evaluated the efficacy of gene enhanced tissue engineering following mosaicplasty in a goat model. They found that gene enhancement could effectively restore a 9 mm diameter OCD in a goat model. Pei et al. [80] used the goat OCD model and implanted a tissue-engineered osteochondral (TEO) graft to investigate its reparative efficacy. Their results showed that this TEO was a promising substitute biomaterial for osteochondral regeneration.

4.5. Horse

As horses are robust and long-lived animals, they are suitable models for assessing the repair of superficial cartilage and subchondral bone in chronic injuries in weight-bearing conditions. Similar to humans, the horses suffer from cartilage diseases and have very weak cartilage self-repairing ability [82]. It is reported that the thickness of articular cartilage is 1.75 mm, which is closest to human cartilage thickness (2.35 mm). Cartilage and OCD of 15 to 20 mm can be assessed in horses. In addition, the upright knee joint with large joint size, thick joint cartilage, and fully straightened gait process is closer to the human knee anatomy than the other animal models. The age of skeletal maturity in the horse is 2–4 years. The age of horses used ranges from 2 to 6 years. Defects have been created in the femoral trochlea [83], the medial femoral condyle [84], the lateral trochlear ridge [85], and the medial surface of lateral trochlea of the talus [86]. A 10 mm in diameter and 5 mm–10 mm deep defect has often been created to simulate osteochondral defects. The major disadvantages of equine models include high cost, inconvenient management, and long-term care during and after surgery. High joint load conditions, high prices, and the need for highly specialized facilities limit the use of horse models for researchers (Table 8).

Table 8: Examples of studies using horse osteochondral defect models.

Authors
Age
Defect size (diameter × depth)
Location
Endpoint
Material tested

Seo et al. [83]
3.6 ± 2.3 years
10 mm × 5 mm
The medial condyle
6 months
GT/MSCs/BMP-2/PRP implantation

Bolanos et al. [84]
6 years
11 mm × 10 mm
The middle aspect of medial femoral trochlear ridge
6 months
CDM/CaP

McCarrel et al. [85]
2–5 years
10 mm × 10 mm
The lateral trochlear ridge
4, 12, and 24 months
Biphasic cartilage repair device

Maninchedda et al. [86]
3 years
10 mm × 5 mm
The medial surface of lateral trochlea of talus
6 months
Type II collagen

4.5.1. Experimental Protocol of Animal Surgeries

Horse was positioned in dorsal recumbence. General anesthesia was maintained and a 5 cm incision made between the middle and medial patellar ligaments. OCD were created using a power-driven drill. Defect site and joints were flushed with saline solution before implantation. Scaffolds were press-fit implanted into each defect. Wounds were sutured in four layers (joint capsule, deep fascia, superficial fascia, and skin) and a stent bandage was applied over the incision [84].

4.5.2. Applications of Horse OCD Models for Testing of Osteochondral Repair Materials

Bolanos et al. [84] used a horse model to investigate the effect of decellularized cartilage-derived matrix (CDM) scaffolds with a calcium phosphate (CaP) base for the repair of OCD. Seo et al. [83] evaluated the efficacy of a synovial flap and gelatin/β-tricalcium phosphate (GT) sponge loaded with mesenchymal stem cells (MSCs), bone morphogenetic protein-2 (BMP-2), and platelet-rich plasma (PRP) for repairing of OCD in horses. The results showed that the GT/MSCs/BMP-2/PRP implantation promoted osteochondral regeneration in the equine model. McCarrel et al. [85] used a 10 mm in diameter and 10 mm deep equine model to test a biphasic cartilage repair device (CRD) for feasibility of arthroscopic implantation and long-term repair of OCD. Maninchedda et al. [86] established a 10 mm in diameter and 5 mm deep OCD model in 3-year-old horses, and the defect was filled with chitosan-GP. After 180 days, they found that the implanted chitosan-GP did not cause any important inflammatory reaction and allowed cell growth.

5. Nonhuman Primate Model

Most animal models differ in biomechanical functions and/or physiological responses from human, limiting the ability to extrapolate data to clinical practice. The nonhuman primate (NHP) models overcomes many of these limitations, as they have similar genetic, physiological, and behavioral characteristics to humans and can highly mimic human health issues [87, 88]. Some reports have used NHP to study cartilage regeneration. Kagimoto et al. used a monkey model to assess the safety and efficacy of the xenotransplantation of human cartilage progenitor cells. They found that autologous transplantation of cartilage progenitor cells may be effective in repairing elastic cartilage [89]. Buckwalter et al. used skeletally mature cynomolgus monkeys to create 3.2 mm in diameter and 4.0 mm deep osteochondral defects of the articular surfaces of the patella (PA) and the medial femoral condyle (FC) in both knees and then treated them with intermittent passive motion (IPM) or cast-immobilization (CI). However, they found that repair of acute osteochondral damage in primates failed to restore normal articular surfaces within eight weeks [90]. Ma et al. suggested that the chondrogenic clonal MSC-loaded monkey acellular dermal matrix (MSC-ADM) scaffold can improve cartilage damage in cynomolgus monkey models and can be used to repair similar human cartilage defects [91]. Jiang et al. made 3 mm in diameter and 2 mm deep cartilage defects on the distal femurs surface of cynomolgus monkeys and treated them with autologous selected chondrogenic clonal MSCs (sC-MSCs). They found that sC-MSCs can effectively improve the healing of cartilage damage in monkey OA induced by collagenase [92] (Table 9). Despite having big similarity to humans, NHP have been seldom utilized in cartilage regeneration research, due to scarcity, high costs, ethical consideration, and high profile in animal welfare and also because these are often unable to provide additional information beyond the aforementioned large animal models.

Table 9: Examples of studies using monkey cartilage or osteochondral defect models.

Authors
Age
Defect size (diameter × depth)
Location
Endpoint
Treatment

Buckwalter et al. [90]
—
3.2 mm × 4 mm
The patella and the medial femoral condyle
8 weeks
Intermittent passive motion (IPM) or cast-immobilization (CI)

Ma et al. [91]
3–5 years old
3.2 mm × 2 mm
Knee joints
24 weeks
MSC-loaded ADM scaffold

Jiang et al. [92]
3–5 years old
3 mm × 2 mm
The surface of distal femurs
24 weeks
Autologous selected chondrogenic clonal MSCs

6. Selecting an Appropriate Animal Model Based on Multiple Factors

The selection of animal models is critical to promote translational research to the clinical application of biomaterials. Generally, small animal models including rats and rabbits are beneficial for early-phase testing, such as testing degradation, biocompatibility, and interaction of implanted biomaterials with host tissues. Because they are economical and easy to handle and have short time for healing (usually 12 weeks for rabbits) [19], large animals are more suitable for late-phase translational research because their articular cartilage structure is much similar to the mechanical load on humans [93, 94]. However, large animal study is often limited by high costs, long duration (at least 24 weeks), or even ethics. For example, it is difficult to obtain ethical permission to use dogs in some countries or districts pertaining to their companion animal status. Multiple factors should be considered for selecting the appropriate animal models to achieve specific study objectives, such as the size and location of the defect, age, study duration, and surgical considerations. Besides scientific evaluation, the choice is also influenced by practical aspects such as ethics, costs, and housing.

7. Conclusion

In this review, we summarize the benefits and limitations of each species for reproducing specific defects, analyze and compare the similarities between animal models and human clinical situations, and emphasize the factors we need to consider when choosing animals. This review provides an important reference for selecting a suitable animal model(s) for the development of new strategies for osteochondral regeneration.

Conflicts of Interest

The authors declare that they have no conflicts of interest.

Acknowledgments

This work was supported by the National Natural Science Foundation of China (81773964), Sino-Swiss collaborative project from Ministry of Science and Technology (2015DFG32200), the Swiss National Science Foundation under the SSSTC program (156362), “3D-OCT” of AO-Research Institute, and Shenzhen Double Chain Project for Innovation and Development Industry supported by Bureau of Industry and Information Technology of Shenzhen (201908141541).

References

C. Deng, J. Chang, and C. Wu, “Bioactive scaffolds for osteochondral regeneration,” Journal of Orthopaedic Translation, vol. 17, pp. 15–25, 2019.

W. Yuk-wai Lee and B. Wang, “Cartilage repair by mesenchymal stem cells: clinical trial update and perspectives,” Journal of Orthopaedic Translation, vol. 9, pp. 76–88, 2017.

T. Vos, A. D. Flaxman, M. Naghavi, and R. Lozano, “Years lived with disability (YLDs) for 1160 sequelae of 289 diseases and injuries 1990–2010: a systematic analysis for the global burden of disease study 2010,” Lancet, vol. 380, no. 9859, pp. 2163–2196, 2012.

D. Pereira, B. Peleteiro, J. Araújo, J. Branco, R. A. Santos, and E. Ramos, “The effect of osteoarthritis definition on prevalence and incidence estimates: a systematic review,” Osteoarthritis and Cartilage, vol. 19, no. 11, pp. 1270–1285, 2011.

C. Deng, H. Zhu, J. Li et al., “Bioactive scaffolds for regeneration of cartilage and subchondral bone interface,” Theranostics, vol. 8, no. 7, pp. 1940–1955, 2018.

D. J. Huey, J. C. Hu, and K. A. Athanasiou, “Unlike bone, cartilage regeneration remains elusive,” Science, vol. 338, no. 6109, pp. 917–921, 2012.

S. P. Nukavarapu and D. L. Dorcemus, “Osteochondral tissue engineering: current strategies and challenges,” Biotechnology Advances, vol. 31, no. 5, pp. 706–721, 2013.

R. A. Magnussen, W. R. Dunn, J. L. Carey, and K. P. Spindler, “Treatment of focal articular cartilage defects in the knee: a systematic review,” Clinical Orthopaedics and Related Research, vol. 466, no. 4, pp. 952–962, 2008.

S.-S. Seo, C.-W. Kim, and D.-W. Jung, “Management of focal chondral lesion in the knee joint,” Knee Surgery & Related Research, vol. 23, no. 4, pp. 185–196, 2011.

J. H. Guettler, C. K. Demetropoulos, K. H. Yang, and K. A. Jurist, “Osteochondral deflects in the human knee—influence of defect size on cartilage rim stress and load redistribution to surrounding cartilage,” The American Journal of Sports Medicine, vol. 32, no. 6, pp. 1451–1458, 2004.

B. J. Ahern, J. Parvizi, R. Boston, and T. P. Schaer, “Preclinical animal models in single site cartilage defect testing: a systematic review,” Osteoarthritis and Cartilage, vol. 17, no. 6, pp. 705–713, 2009.

Y. Li, S.-K. Chen, L. Li, L. Qin, X.-L. Wang, and Y.-X. Lai, “Bone defect animal models for testing efficacy of bone substitute biomaterials,” Journal of Orthopaedic Translation, vol. 3, no. 3, pp. 95–104, 2015.

A. da Silva Morais, J. M. Oliveira, and R. L. Reis, “Small animal models,” Osteochondral Tissue Engineering, vol. 1059, pp. 423–439, 2018.

I. R. Dias, C. A. Viegas, and P. P. Carvalho, “Large animal models for osteochondral regeneration,” Osteochondral Tissue Engineering, vol. 1059, pp. 441–501, 2018.

M. Rudert, “Histological evaluation of osteochondral defects: consideration of animal models with emphasis on the rabbit, experimental setup, follow-up and applied methods,” Cells Tissues Organs, vol. 171, no. 4, pp. 229–240, 2002.

R. J. Colman, M. A. Lane, N. Binkley, F. H. Wegner, and J. W. Kemnitz, “Skeletal effects of aging in male rhesus monkeys,” Bone, vol. 24, no. 1, pp. 17–23, 1999.

J. Malda, J. C. de Grauw, K. E. M. Benders et al., “Of mice, men and elephants: the relation between articular cartilage thickness and body mass,” PLoS One, vol. 8, no. 2, 2013.

C. J. Moran, A. Ramesh, P. A. Brama, J. M. O’Byrne, F. J. O’Brien, and T. J. Levingstone, “The benefits and limitations of animal models for translational research in cartilage repair,” Journal of Experimental Orthopaedics, vol. 3, no. 1, p. 1, 2016.

J. L. Cook, C. T. Hung, K. Kuroki et al., “Animal models of cartilage repair,” Bone & Joint Research, vol. 3, no. 4, pp. 89–94, 2014.

H. Katagiri, L. F. Mendes, and F. P. Luyten, “Definition of a critical size osteochondral knee defect and its negative effect on the surrounding articular cartilage in the rat,” Osteoarthritis and Cartilage, vol. 25, no. 9, pp. 1531–1540, 2017.

Y.-B. Park, C.-W. Ha, J.-A. Kim et al., “Effect of transplanting various concentrations of a composite of human umbilical cord blood-derived mesenchymal stem cells and hyaluronic acid hydrogel on articular cartilage repair in a rabbit model,” PLoS One, vol. 11, no. 11, Article ID e0165446, 2016.

Y. Du, H. Liu, Q. Yang et al., “Selective laser sintering scaffold with hierarchical architecture and gradient composition for osteochondral repair in rabbits,” Biomaterials, vol. 137, pp. 37–48, 2017.

T. Chen, J. Bai, J. Tian, P. Huang, H. Zheng, and J. Wang, “A single integrated osteochondral in situ composite scaffold with a multi-layered functional structure,” Colloids and Surfaces B: Biointerfaces, vol. 167, pp. 354–363, 2018.

D. Kazemi, K. Shams Asenjan, N. Dehdilani, and H. Parsa, “Canine articular cartilage regeneration using mesenchymal stem cells seeded on platelet rich fibrin,” Bone & Joint Research, vol. 6, no. 2, pp. 98–107, 2017.

T. Gotterbarm, S. J. Breusch, U. Schneider, and M. Jung, “The minipig model for experimental chondral and osteochondral defect repair in tissue engineering: retrospective analysis of 180 defects,” Laboratory Animals, vol. 42, no. 1, pp. 71–82, 2008.

T. J. Levingstone, A. Ramesh, R. T. Brady et al., “Cell-free multi-layered collagen-based scaffolds demonstrate layer specific regeneration of functional osteochondral tissue in caprine joints,” Biomaterials, vol. 87, pp. 69–81, 2016.

D. W. Jackson, P. A. Lalor, H. M. Aberman, and T. M. Simon, “Spontaneous repair of full-thickness defects of articular cartilage in a goat model—a preliminary study,” The Journal of Bone and Joint Surgery-American Volume, vol. 83, no. 1, pp. 53–64, 2001.

A. J. Nixon, E. Rickey, T. J. Butler, M. S. Scimeca, N. Moran, and G. L. Matthews, “A chondrocyte infiltrated collagen type I/III membrane (MACI® implant) improves cartilage healing in the equine patellofemoral joint model,” Osteoarthritis and Cartilage, vol. 23, no. 4, pp. 648–660, 2015.

C. W. McIlwraith, L. A. Fortier, D. D. Frisbie, and A. J. Nixon, “Equine models of articular cartilage repair,” Cartilage, vol. 2, no. 4, pp. 317–326, 2011.

E. Salonius, L. Rieppo, M. J. Nissi et al., “Critical-sized cartilage defects in the equine carpus,” Connective Tissue Research, vol. 60, no. 2, pp. 95–106, 2019.

C. R. Chu, M. Szczodry, and S. Bruno, “Animal models for cartilage regeneration and repair,” Tissue Engineering Part B Reviews, vol. 16, no. 1, pp. 105–115, 2010.

C. A. Vilela, C. Correia, J. M. Oliveira, R. A. Sousa, J. Espregueira-Mendes, and R. L. Reis, “Cartilage repair using hydrogels: a critical review of in vivo experimental designs,” ACS Biomaterials Science & Engineering, vol. 1, no. 9, pp. 726–739, 2015.

A. M. McCoy, “Animal models of osteoarthritis: comparisons and key considerations,” Veterinary Pathology, vol. 52, no. 5, pp. 803–818, 2015.

R. X. Fan, H. Gong, R. Zhang, J. Z. Gao, Z. B. Jia, and Y. J. Hu, “Quantification of age-related tissue-level failure strains of rat femoral cortical bones using an approach combining macrocompressive test and microfinite element analysis,” Journal of Biomechanical Engineering-Transactions of the Asme, vol. 138, no. 4, 2016.

J.-M. Lee and G.-I. Im, “SOX trio-co-transduced adipose stem cells in fibrin gel to enhance cartilage repair and delay the progression of osteoarthritis in the rat,” Biomaterials, vol. 33, no. 7, pp. 2016–2024, 2012.

M. S. Muttigi, B. J. Kim, B. Choi et al., “Matrilin-3 codelivery with adipose-derived mesenchymal stem cells promotes articular cartilage regeneration in a rat osteochondral defect model,” Journal of Tissue Engineering and Regenerative Medicine, vol. 12, no. 3, pp. 667–675, 2018.

E. E. Mahmoud, N. Kamei, R. Shimizu et al., “Therapeutic potential of multilineage-differentiating stress-enduring cells for osteochondral repair in a rat model,” Stem Cells International, vol. 2017, Article ID 8154569, 8 pages, 2017.

R. L. Dahlin, L. A. Kinard, J. Lam et al., “Articular chondrocytes and mesenchymal stem cells seeded on biodegradable scaffolds for the repair of cartilage in a rat osteochondral defect model,” Biomaterials, vol. 35, no. 26, pp. 7460–7469, 2014.

H. Li, Q. Zheng, Y. Xiao, J. Feng, Z. Shi, and Z. Pan, “Rat cartilage repair using nanophase PLGA/HA composite and mesenchymal stem cells,” Journal of Bioactive and Compatible Polymers, vol. 24, no. 1, pp. 83–99, 2009.

K.-S. Park, B.-J. Kim, E. Lih et al., “Versatile effects of magnesium hydroxide nanoparticles in PLGA scaffold-mediated chondrogenesis,” Acta Biomaterialia, vol. 73, pp. 204–216, 2018.

X. Wang, J. D. Mabrey, and C. M. Agrawal, “An interspecies comparison of bone fracture properties,” Bio-Medical Materials and Engineering, vol. 8, no. 1, pp. 1–9, 1998.

J. Liao, Y. Qu, B. Chu, X. Zhang, and Z. Qian, “Biodegradable CSMA/PECA/graphene porous hybrid scaffold for cartilage tissue engineering,” Scientific Reports, vol. 5, no. 1, p. 9879, 2015.

C. Bauer, V. Jeyakumar, E. Niculescu-Morzsa, D. Kern, and S. Nehrer, “Hyaluronan thiomer gel/matrix mediated healing of articular cartilage defects in New Zealand white rabbits—a pilot study,” Journal of Experimental Orthopaedics, vol. 4, no. 1, p. 14, 2017.

S.-Q. Ruan, L. Yan, J. Deng, W.-L. Huang, and D.-M. Jiang, “Preparation of a biphase composite scaffold and its application in tissue engineering for femoral osteochondral defects in rabbits,” International Orthopaedics, vol. 41, no. 9, pp. 1899–1908, 2017.

Q. Meng, X. Hu, H. Huang et al., “Microfracture combined with functional pig peritoneum-derived acellular matrix for cartilage repair in rabbit models,” Acta Biomaterialia, vol. 53, pp. 279–292, 2017.

S. Zhang, L. Chen, Y. Jiang et al., “Bi-layer collagen/microporous electrospun nanofiber scaffold improves the osteochondral regeneration,” Acta Biomaterialia, vol. 9, no. 7, pp. 7236–7247, 2013.

T. Rasanen and K. Messner, “Regional variations of indentation stiffness and thickness of normal rabbit knee articular cartilage,” Journal of Biomedical Materials Research, vol. 31, no. 4, pp. 519–524, 1996.

A. Chevrier, A. S. M. Kouao, G. Picard, M. B. Hurtig, and M. D. Buschmann, “Interspecies comparison of subchondral bone properties important for cartilage repair,” Journal of Orthopaedic Research, vol. 33, no. 1, pp. 63–70, 2015.

S. Castañeda, R. Largo, E. Calvo et al., “Bone mineral measurements of subchondral and trabecular bone in healthy and osteoporotic rabbits,” Skeletal Radiology, vol. 35, no. 1, pp. 34–41, 2006.

J. Radhakrishnan, A. Manigandan, P. Chinnaswamy, A. Subramanian, and S. Sethuraman, “Gradient nano-engineered in situ forming composite hydrogel for osteochondral regeneration,” Biomaterials, vol. 162, pp. 82–98, 2018.

X. Liu, X. Jin, and P. X. Ma, “Nanofibrous hollow microspheres self-assembled from star-shaped polymers as injectable cell carriers for knee repair,” Nature Materials, vol. 10, no. 5, pp. 398–406, 2011.

H. Schmal, J. M. Kowal, M. Kassem et al., “Comparison of regenerative tissue quality following matrix-associated cell implantation using amplified chondrocytes compared to synovium-derived stem cells in a rabbit model for cartilage lesions,” Stem Cells International, vol. 2018, Article ID 4142031, 12 pages, 2018.

T. J. Levingstone, E. Thompson, A. Matsiko, A. Schepens, J. P. Gleeson, and F. J. O’Brien, “Multi-layered collagen-based scaffolds for osteochondral defect repair in rabbits,” Acta Biomaterialia, vol. 32, pp. 149–160, 2016.

M. Ramallal, E. Maneiro, E. Lopez et al., “Xeno-implantation of pig chondrocytes into rabbit to treat localized articular cartilage defects: an animal model,” Wound Repair and Regeneration, vol. 12, no. 3, pp. 337–345, 2004.

B. Schneider-Wald, A. K. von Thaden, and M. L. R. Schwarz, “Defect models for the regeneration of articular cartilage in large animals,” Der Orthopäde, vol. 42, no. 4, pp. 242–253, 2013.

E. B. Hunziker, “Articular cartilage repair: basic science and clinical progress. A review of the current status and prospects,” Osteoarthritis and Cartilage, vol. 10, no. 6, pp. 432–463, 2002.

P. S. J. M. Bouwmeester, R. Kuijer, G. N. Homminga, S. K. Bulstra, and R. G. T. Geesink, “A retrospective analysis of two independent prospective cartilage repair studies: autogenous perichondrial grafting versus subchondral drilling 10 years post-surgery,” Journal of Orthopaedic Research, vol. 20, no. 2, pp. 267–273, 2002.

D. D. Frisbie, M. W. Cross, and C. W. McIlwraith, “A comparative study of articular cartilage thickness in the stifle of animal species used in human pre-clinical studies compared to articular cartilage thickness in the human knee,” Veterinary and Comparative Orthopaedics and Traumatology, vol. 19, no. 3, pp. 142–146, 2006.

Y. M. Lv and Q. S. Yu, “Repair of articular osteochondral defects of the knee joint using a composite lamellar scaffold,” Bone & Joint Research, vol. 4, no. 4, pp. 56–64, 2015.

E. C. McCarty, R. R. Fader, J. J. Mitchell, R. E. Glenn, H. G. Potter, and K. P. Spindler, “Fresh osteochondral allograft versus autograft: twelve-month results in isolated canine knee defects,” The American Journal of Sports Medicine, vol. 44, no. 9, pp. 2354–2365, 2016.

S. L. Salkeld, L. P. Patron, J. C. Lien, S. D. Cook, and D. G. Jones, “Biological and functional evaluation of a novel pyrolytic carbon implant for the treatment of focal osteochondral defects in the medial femoral condyle: assessment in a canine model,” Journal of Orthopaedic Surgery and Research, vol. 11, no. 1, p. 155, 2016.

K. Yamazoe, H. Mishima, K. Torigoe et al., “Effects of atelocollagen gel containing bone marrow-derived stromal cells on repair of osteochondral defect in a dog,” Journal of Veterinary Medical Science, vol. 69, no. 8, pp. 835–839, 2007.

L. V. Sondergaard, F. Dagnaes-Hansen, and M. S. Herskin, “Welfare assessment in porcine biomedical research—suggestion for an operational tool,” Research in Veterinary Science, vol. 91, no. 3, pp. e1–e9, 2011.

F. J. van der Staay, B. Pouzet, M. Mahieu, R. E. Nordquist, and T. Schuurman, “The d-amphetamine-treated Gottingen miniature pig: an animal model for assessing behavioral effects of antipsychotics,” Psychopharmacology, vol. 206, no. 4, pp. 715–729, 2009.

M. B. Fisher, N. S. Belkin, A. H. Milby et al., “Cartilage repair and subchondral bone remodeling in response to focal lesions in a mini-pig model: implications for tissue engineering,” Tissue Engineering Part A, vol. 21, no. 3-4, pp. 850–860, 2015.

M. Betsch, S. Thelen, L. Santak et al., “The role of erythropoietin and bone marrow concentrate in the treatment of osteochondral defects in mini-pigs,” PLoS One, vol. 9, no. 3, Article ID e92766, 2014.

S. Pilichi, S. Rocca, R. R. Pool et al., “Treatment with embryonic stem-like cells into osteochondral defects in sheep femoral condyles,” BMC Veterinary Research, vol. 10, no. 1, p. 301, 2014.

B. B. Christensen, C. B. Foldager, M. L. Olesen, K. C. Hede, and M. Lind, “Implantation of autologous cartilage chips improves cartilage repair tissue quality in osteochondral defects: a study in gottingen minipigs,” The American Journal of Sports Medicine, vol. 44, no. 6, pp. 1597–1604, 2016.

M. Jagodzinski, C. Liu, D. Guenther et al., “Bone marrow-derived cell concentrates have limited effects on osteochondral reconstructions in the mini pig,” Tissue Engineering Part C: Methods, vol. 20, no. 3, pp. 215–226, 2014.

K. Schlichting, H. Schell, R. U. Kleemann et al., “Influence of scaffold stiffness on subchondral bone and subsequent cartilage regeneration in an ovine model of osteochondral defect healing,” The American Journal of Sports Medicine, vol. 36, no. 12, pp. 2379–2391, 2008.

A. Bernstein, P. Niemeyer, G. Salzmann et al., “Microporous calcium phosphate ceramics as tissue engineering scaffolds for the repair of osteochondral defects: histological results,” Acta Biomaterialia, vol. 9, no. 7, pp. 7490–7505, 2013.

N. Mohan, V. Gupta, B. P. Sridharan et al., “Microsphere-based gradient implants for osteochondral regeneration: a long-term study in sheep,” Regenerative Medicine, vol. 10, no. 6, pp. 709–728, 2015.

M. Caminal, D. Peris, C. Fonseca et al., “Cartilage resurfacing potential of PLGA scaffolds loaded with autologous cells from cartilage, fat, and bone marrow in an ovine model of osteochondral focal defect,” Cytotechnology, vol. 68, no. 4, pp. 907–919, 2016.

A. Yucekul, D. Ozdil, N. H. Kutlu, E. Erdemli, H. M. Aydin, and M. N. Doral, “Tri-layered composite plug for the repair of osteochondral defects: in vivo study in sheep,” Journal of Tissue Engineering, vol. 8, 2017.

E. H. Mrosek, H.-W. Chung, J. S. Fitzsimmons, S. W. O’Driscoll, G. G. Reinholz, and J. C. Schagemann, “Porous tantalum biocomposites for osteochondral defect repair: a follow-up study in a sheep model,” Bone & Joint Research, vol. 5, no. 9, pp. 403–411, 2016.

T. Zhang, H. Zhang, L. Zhang et al., “Biomimetic design and fabrication of multilayered osteochondral scaffolds by low-temperature deposition manufacturing and thermal-induced phase-separation techniques,” Biofabrication, vol. 9, no. 2, Article ID 025021, 2017.

C. J. A. van Bergen, G. M. M. J. Kerkhoffs, M. Özdemir et al., “Demineralized bone matrix and platelet-rich plasma do not improve healing of osteochondral defects of the talus: an experimental goat study,” Osteoarthritis and Cartilage, vol. 21, no. 11, pp. 1746–1754, 2013.

E. Kon, G. Filardo, J. Shani et al., “Osteochondral regeneration with a novel aragonite-hyaluronate biphasic scaffold: up to 12-month follow-up study in a goat model,” Journal of Orthopaedic Surgery and Research, vol. 10, no. 1, p. 81, 2015.

J. Sun, X.-K. Hou, and Y.-X. Zheng, “Restore a 9 mm diameter osteochondral defect with gene enhanced tissue engineering followed mosaicplasty in a goat model,” Acta Orthopaedica et Traumatologica Turcica, vol. 50, no. 4, pp. 464–469, 2016.

Y. Pei, J.-J. Fan, X.-Q. Zhang, Z.-Y. Zhang, and M. Yu, “Repairing the osteochondral defect in goat with the tissue-engineered osteochondral graft preconstructed in a double-chamber stirring bioreactor,” BioMed Research International, vol. 2014, Article ID 219203, 11 pages, 2014.

A. He, L. Liu, X. Luo et al., “Repair of osteochondral defects with in vitro engineered cartilage based on autologous bone marrow stromal cells in a swine model,” Scientific Reports, vol. 7, no. 1, p. 40489, 2017.

T. G. Koch and D. H. Betts, “Stem cell therapy for joint problems using the horse as a clinically relevant animal model,” Expert Opinion on Biological Therapy, vol. 7, no. 11, pp. 1621–1626, 2007.

J.-P. Seo, Y. Kambayashi, M. Itho et al., “Effects of a synovial flap and gelatin/beta-tricalcium phosphate sponges loaded with mesenchymal stem cells, bone morphogenetic protein-2, and platelet rich plasma on equine osteochondral defects,” Research in Veterinary Science, vol. 101, pp. 140–143, 2015.

R. A. V. Bolanos, S. M. Cokelaere, J. M. E. McDermott et al., “The use of a cartilage decellularized matrix scaffold for the repair of osteochondral defects: the importance of long-term studies in a large animal model,” Osteoarthritis and Cartilage, vol. 25, no. 3, pp. 413–420, 2017.

T. M. McCarrel, S. L. Pownder, S. Gilbert et al., “Two-year evaluation of osteochondral repair with a novel biphasic graft saturated in bone marrow in an equine model,” Cartilage, vol. 8, no. 4, pp. 406–416, 2017.

U. Maninchedda, O. M. Lepage, M. Gangl et al., “Development of an equine groove model to induce metacarpophalangeal osteoarthritis: a pilot study on 6 horses,” PLoS One, vol. 10, no. 2, Article ID e0115089, 2015.

I. Messaoudi and D. K. Ingram, “Overview of aging research using nonhuman primate models,” Age, vol. 34, no. 5, pp. 1047–1049, 2012.

R. M. Anderson and R. J. Colman, “Prospects and perspectives in primate aging research,” Antioxidants & Redox Signaling, vol. 14, no. 2, pp. 203–205, 2011.

S. Kagimoto, T. Takebe, S. Kobayashi et al., “Autotransplantation of monkey ear perichondrium-derived progenitor cells for cartilage reconstruction,” Cell Transplantation, vol. 25, no. 5, pp. 951–962, 2016.

J. A. Buckwalter, J. A. Martin, M. Olmstead, K. A. Athanasiou, M. P. Rosenwasser, and V. C. Mow, “Osteochondral repair of primate knee femoral and patellar articular surfaces: implications for preventing post-traumatic osteoarthritis,” Iowa Orthopedic Journal, vol. 23, pp. 66–74, 2003.

A. Ma, L. Jiang, L. Song et al., “Reconstruction of cartilage with clonal mesenchymal stem cell-acellular dermal matrix in cartilage defect model in nonhuman primates,” International Immunopharmacology, vol. 16, no. 3, pp. 399–408, 2013.

L. Jiang, A. L. Ma, L. J. Song et al., “Cartilage regeneration by selected chondrogenic clonal mesenchymal stem cells in the collagenase-induced monkey osteoarthritis model,” Journal of Tissue Engineering and Regenerative Medicine, vol. 8, no. 11, pp. 896–905, 2014.

L. Cong, F. A. Ran, D. Cox et al., “Multiplex genome engineering using CRISPR/Cas systems,” Science, vol. 339, no. 6121, pp. 819–823, 2013.

I. Tessaro, V. T. Nguyen, A. Di Giancamillo et al., “Animal models for cartilage repair,” Journal of Biological Regulators and Homeostatic Agents, vol. 32, no. 6 Suppl. 1, pp. 105–116, 2018.