Journal of Analytical Methods in Chemistry
Volume 2019 (2019), Article ID 4608135, 8 pages
https://doi.org/10.1155/2019/4608135

Research Article

Development of Potentiometric Phenol Sensors by Nata de Coco Membrane on Screen-Printed Carbon Electrode

Ani Mulyasuryani and Afifah Muhimmatul Mustaghfiroh

Analytical Chemistry Laboratory, Department of Chemistry, Faculty of Mathematics and Natural Sciences, University of Brawijaya, Malang, Indonesia

Correspondence should be addressed to Ani Mulyasuryani; mulyasuryani@ub.ac.id

Received 22 March 2019; Revised 26 June 2019; Accepted 31 July 2019; Published 19 August 2019

Academic Editor: Hana Sklenarova

Copyright © 2019 Ani Mulyasuryani and Afifah Muhimmatul Mustaghfiroh. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

Abstract.

Nata de coco, a bacterial cellulose as a result of coconut water fermentation, is a conductive polymer with a electrical conductivity of 553 μS/cm and has high mechanical stability. In this study, nata de coco was used as a supporting membrane for the development of phenol sensors in potentiometry. Nata de coco membrane containing phenol is coated on the surface of the printed carbon electrode (screen-printed carbon electrode). The cross-sectional area of the carbon electrode coated with the membrane is 1.5 × 3 mm², while the reference electrode is Ag/AgCl. The thickness of the electrode membrane affects the Nernstian factor. The optimum Nernstian factor is produced by 100 μm membrane thickness containing 117.5 μg of phenol. Measurement of phenol solution was carried out at pH 11, in the concentration range of 10⁻⁸ to 10⁻² mol/L, resulting in a Nernstian factor of 41.8 ± 1.3 mV/decade. The Nernstian factor increased to 55.7 ± 0.4 mV/decade if the membrane of the electrode contained 0.1% Fe₃O₄ nanoparticles. This sensor has been applied in the real sample of river water, resulting in good accuracy and precision.

1. Introduction

Phenol is one of the organic substances produced from industrial plants such as petroleum. The high phenol concentration in the water system can cause death in certain groups of organisms [1]. The maximum limit of phenol concentration in the waters that is permitted based on the Regulation of the Minister of Environment of the Republic of Indonesia number 03/2010 is 1 mg/L [2]. The standard method that has been used to determine phenol levels in water according to SNI 06-6989.21-2004 is by the spectrophotometric method, in which phenol is reacted with 4-aminoantipyrine to form a brownish red complex [3]. In this method, sample preparation is needed from random measurements carried out in the laboratory. To facilitate the control of phenol levels, it is necessary to develop phenol sensors that are easily applied in the field.

Selective membranes are one of the main components in the development of chemical sensors, especially in potentiometry. Selective membranes are developed by incorporating ionophores and active ingredients in a polymer, such as polyvinyl chloride (PVC). Membranes must be a conducting material. PVC is a conducting material due to the presence of Cl⁻ which is polar and has good mechanical stability [4, 5]. Nata de coco is an example of a polar polymer because it has OH groups on their surface; hence, it can conduct electricity. Meanwhile, cellulose derived from bacteria is insoluble in water due to hydrogen bonds between adjacent hydroxyl groups [6]. In addition, bacterial cellulose has the advantage of being a membrane that has high purity, small pore size of 0.08–0.09 μm, high density, low toxicity, and high mechanical stability [7, 8]. Therefore, nata de coco can be applied as a membrane in sensor manufactures. The application of nata de coco as a membrane sensor has been developed for colorimetric sensors [9, 10] and fluorosity sensors [11]; hence, it is also possible to be applied to potentiometric sensors.

Chan et al. have developed phenol sensors using PVC membranes [12]. Phenol sensors have also been developed using chitosan and cetyl trimethylammonium phenol (CTAPh) membranes as ionophores [13]. In this study, nata de coco was used as a membrane. To detect the analyte ions specifically, the same ion as the analyte must be present in the nata de coco membrane [14]. Phenoxide ion, which was deposited into the nata de coco membrane, is used as the active ion. The nata de coco selective membrane is then coated on the surface of the carbon electrode. The electrode used is a type of screen-printed electrode (SPE) consisting of a Ag/AgCl reference and an indicator electrode (Figure 1).

Figure 1: Illustrated SPE and specification used in this work. The black part on the left picture is the electrode coated with nata de coco.

Signal (cell potential, Ecell) in potentiometry is produced due to differences in concentration on the surface of the inner membrane with the outside membrane, which interacts with the analyte. The difference in concentration will cause a potential difference which is then read compared to the reference electrode [15]. Phenol is a weak acid with a pKa value of 9.94 [16]; hence, phenol can be detected in the form of phenoxide ions, and their presence are influenced by pH. As an anion, the cell potential generated in the phenol sensor refers to the ESI anion, as shown in equation (1). The 0.0592 log [C₆H₅O⁻]membrane, Easy, and Ereff in equation (1) are constant; thus, the equation is simplified in equation (2):

where K = 0.0592 log [C₆H₅O⁻]membrane–Ereff + Easy in which Easy is a potential that is unpredictable due to membrane uniformity.

The difference in phenol concentration in the analyte solution with phenol concentration in the membrane is indicated from the signal or cell potential (Ecell). The mathematical function of the relationship between concentration and signal is expressed in equation (2), which is a logarithmic equation. For monovalent ions such as phenol, sensor sensitivity is expressed by the Nernstian factor, 59.2 mV/decade. The measurement principle of the phenolic ion using the nata de coco membrane is illustrated in Figure 2 (note: Eb is a function of analyte concentration because the phenol concentration in the membrane is fixed. Eb is measured and compared to the reference electrode potential, namely, Ag/AgCl, which has a fixed amount (if using 1 M KCl solution = 222 mV).

Figure 2: Illustrated measurement principle of phenol using the nata de coco membrane ( = phenol in membrane,  = phenol in the analyte, E₁ = Eanalyte, E₂ = Emembrane, Esel= Ecell, and Eb = the potential difference between membrane boundaries and analytes).

Addition of metal oxide nanoparticles into membrane is known and been reported previously. The presence of metal oxide nanoparticles alters the properties of the membrane. For example, the hydrophobic properties and pore size of the polysulfone membrane can be modified by addition of a TiO₂ and polyvinylpyrrolidone (PVP) mixture [17]. In our previous work, we also had successfully prepared the nata de coco membrane added with Fe₃O₄ on the SPCE (screen-printed carbon electrode) surface for the diazinon sensor [18]. Preparation of the Fe₃O₄-modified electrode for measurement of other ions or molecules in solution was also reported, such as for ascorbic acid [19], chlorite ions [20], and nitrite ions [21].

Specifically, several sensors [22] and biosensors [23, 24] have been incorporating Fe₃O₄ nanoparticles in their membranes. In ion selective electrodes, this nanomaterial is extensively explored as the contact material or modified materials of working electrodes due to Fe₃O₄ because of its electrical and hydrophobic properties [25]. Fe₃O₄ is believed to be able to promote rapid electron transfer between the electrode and the active site of the reaction, based on its ability to adsorb electromagnetic waves [26] and its conductivity (800 μs/cm) [27].

Fe₃O₄ nanoparticles can also act as pseudocapacitors [28]. The capacitor is a component that can store a large electrical charge for a while [29]. Pseudocapacitors are made to increase the capacitive process in the presence of chemical reactions. This type of capacitor requires material that can chemically store the charge with a fast redox reaction. Pure Fe₃O₄ nanoparticles have a specific capacitance of 33 F/g, while magnetite nanoparticles coated on carbon can increase specific capacitance up to 510 F/g [30, 31]. Specific capacitance is directly proportional to the surface area, and the addition of Fe₃O₄ nanoparticles can increase the surface area. The existence of an increase in specific capacitance can accelerate load transduction so that it can increase the sensitivity of the sensor, which in this study was stated as the Nernstian factor.

Previously, we had develop a modified SPCE Ppy-SiO₂ for phenol determination at a concentration of 10³–10⁻⁵ mM with a sensitivity of 7.93 μA/mM [32]. In this paper, we are reporting the utilisation of nata de coco as membrane on the screen-printed electrode (SPE) for phenol degradation. The effect of the membrane thickness, pH of phenol solution, and addition of Fe₃O₄ nanoparticles into the membrane to the sensor performance was discussed. All preparations and measurements (except the SEM analysis) were conducted at room temperature in Chemistry Laboratory of Chemistry Department, University of Brawijaya, Indonesia.

2. Materials and Methods

2.1. Materials and Instrumentations

Chemicals were used without further purification and were all obtained from Merck, namely, phenol, NaOH, (NH₄)₂SO₄, and Fe₃O₄ nanoparticles of 50–100 nm. Materials used in this study were nata de coco and distilled water. Laboratory tools and instrumentations used in this study were local chopper blender, magnetic stirrer, BI1703 Quasense screen-printed carbon electrode (SPCE), Sanwa CD800a potentiometer, Quasense electrode connector, Senz TI-13MO597 pH meter, Accumax Pro micropipette, Fourier-transform infrared (FT-IR) Shimadzu 8400S (4500–400 cm⁻¹), and scanning electron microscopy (SEM) FEI Inspect S50 (performed at State University of Malang, Indonesia).

2.2. Methods

2.2.1. Preparation of Phenol Sensors

Nata de coco was made by anaerobic fermentation of coconut water in the presence of Acetobacter xylinum for 7 days, with a composition of 1 L of coconut water, 6.7 g of sugar, 5.0 g (NH₄)₂SO₄, and 100 mL of Acetobacter xylinum starter. A 100 g of nata de coco was cut into small pieces. Then, it was added with 50 mL of distilled water with pH 7 and blended using a blender for 5 minutes. Next, the nata de coco was filtered off using filter cloth, weighed 3 g, and added with 90 mL of distilled water with pH 7. The mixture was then ground again using a blender for 15 minutes resulting in a suspension. The suspension was then added with a 0.094 g phenol in a 10 mL of distilled water. The mixture once was again stirred at room temperature using a magnetic stirrer for 24 hours. The nata de coco membrane that has been made was finally coated onto the carbon electrode as much as 2.5, 5.0, 7.5, 10.0, 12.5, and 15.0 μL (depending on the membrane thickness) and dried for 30 minutes at 50°C.

2.2.2. Preparation of Phenol Sensor + Fe₃O₄

A 0.094 g of phenol was added with 0.1 g Fe₃O₄ nanoparticles and then mixed with nata de coco suspension up to 10 mL. The mixture is shaken with a shaker for 24 hours. A total of 12.5 μL of the mixture was coated on the surface of the carbon electrode and dried for 30 minutes at 50°C.

2.2.3. Measurement of Phenol Cell Potential

The indicator electrode was connected to the negative pole on the potentiometer, while the Ag/AgCl reference electrode was connected to the positive pole. Measurement of cell potential of the phenol solution was carried out by dripping a phenol solution on 50 μL on the surface of both electrodes (indicator electrode and reference electrode). The measurement of cell potential was carried out consecutively from phenol concentrations of 10⁻⁸–10⁻¹ M at pH 11. The cell potential was measured at room temperature for 3 minutes.

2.2.4. Validation of Phenol Sensors

Four identical phenol sensors were prepared, with 100 μm nata de coco membrane thickness containing phenol and Fe₃O₄. Each sensor was used to measure 13 phenol standard solutions at pH 11 with a concentration range of 10⁻⁸–10⁻² M, and one real sample was taken from the river water. Repetition of measurements was carried out five times every five days. For validation of the river water sample, the standard addition method was used, and the calculation of sample concentration was based on the standard curve of each repetition.

2.2.5. Data Analysis

A standard of deviation (SD) was used to validate the data. Analysis of variance was used in testing between treatments on the influence of membrane thickness with 6 treatments (h = 6), five replications each, total data 30 ().

3. Results and Discussion

3.1. Preparation of Phenol Sensor

One of the advantages of using nata de coco as membrane is that nata de coco has many hydroxyl groups on their surface. These functional groups, along with other functional groups, could alter the properties of the membrane, including conductivity. The hydroxyl group in nata de coco was confirmed by the broad peak around 3200–3400 cm⁻¹. Nata de coco used in this work also has the alkyl (C-H) group and carbonyl (C=O) group, which was indicated by the sharp peak around 2900 cm⁻¹ and 1650 cm⁻¹, respectively. In detail, the infrared spectra of nata de coco are presented in Figure 3.

Figure 3: Infrared spectra of nata de coco used in this work.

Phenol is an OH-substituted benzene compound, and the presence of hydroxyl (−OH) and carbonyl (C=O) groups in nata de coco allows the formation of hydrogen bonds between phenol and nata de coco. The stirring process for 24 hours is expected to result in a stronger interaction between phenol and nata de coco.

3.2. Effect of Nata de Coco Membrane Thickness

The nata de coco membrane thickness is studied because it will affect membrane uniformity and will affect the amount of Easy. In this study, the thickness of the membrane studied was 60, 70, 80, 90, 100, and 110 μm. The measurements of cell potential for each membrane thickness are shown in Figure 1. As shown Figure 1, the cell potential is inversely proportional to log[C₆H₅O⁻], indicating that the measured sample is anion, namely, phenolic ion, and this corresponds to equation (2). All electrodes with various membrane thicknesses produce the identical pattern data, in which the signal was increasing significantly in the concentration range of 10⁻²–10⁻⁸ M. The Nernstian factor is calculated using data in Figure 4 with a concentration range of 10⁻²–10⁻⁸ M. The Nernstian factor values at various electrodes of different thicknesses are shown in Figure 5.

Figure 4: Relationship between the −log[C₆H₅O⁻] with the cell potential (Ecell) at various membrane thickness.

Figure 5: Relationship curve between membrane thickness and Nernst factor.

Sensor performance can be studied through the Nernstian factor value. For monovalent ions, the theoretical Nernstian factor is 59.2 mV/decade. Phenol sensors have a good performance if they produce a Nernstian factor close to the theoretical calculation. Figure 2 shows that the Nernstian factor increased directly and proportional to the membrane electrode thickness, except at 110 μm thickness. The Nernstian factor was insignificantly increased from thickness 100 to 110 μm. Thus, it was decided that the best electrodes were produced at 100 μm thickness, with a Nernstian factor of 41.8 mV/decade. The increase in the Nernst factor is caused by an increase in the phenol concentration in the membrane.

As presented in Table 1, the increase in the K level (equation (2)) is straight line with the membrane thickness. This is understandable because the setting of membrane thickness was done by increasing the volume of the membrane material coated on the surface of the carbon electrode within the same area. This also shows that the membrane regularity (in this case the spread of active ions) is much better if the thickness of the membrane increases until certain limits since its thick membrane may cause irregularities. According to Fouskaki and Chaniotakis [33], a thick membrane causes the membrane surface distance with the electrode to be farther away so that the sensor response becomes slower, and this was likely to occur at the electrode with a membrane thickness of 110 μm; hence, the data were less consistent, as indicated by the high SD value.

Table 1: The value of K and membrane conductivity for each membrane thickness.

No
Membrane thickness (μm)
K (mV)
Electrical conductivity (μS/cm)

1
60
−210
543

2
70
−205
545

3
80
−187
547

4
90
−164
551

5
100
−144
569

6
110
−135
561

There is a slight increase in membrane electrical conductivity for each increase in membrane thickness (Table 1). This is because, in thicker membrane, the amount of water is higher, and this also results in a higher membrane electrical conductivity and capacitance [34]. In addition, the increase in membrane thickness is directly proportional to the amount of phenol in the membrane. Because phenol is a polar compound, the electrical conductivity increases with increasing phenol concentration. However, the increase in electrical conductivity has little effect on the Nernstian factor, which is a sensor performance indicator or sensor sensitivity. Thus, other effort needs to be done to increase the Nernstian factor significantly and to get closer Nernstian factor to the theoretical value.

3.3. Sensor Performance

One of the factors that can affect the performance of the sensor is pH, considering that phenolic ions can only form perfectly (α = 1) at pH ≥ 10 (). For this reason, the pH of the phenol solution was adjusted to pH 11 by using the 0.01 M carbonate buffer and using NaOH solution separately. Figure 6 shows the Ecell difference obtained using different buffers at pH 11. As shown in Figure 6, the linear concentration was obtained in the range 10⁻⁵–10⁻⁸ M for the carbonate buffer. This result leaves uncertainty, does the presence of CO₃⁻² and HCO₃⁻ ions block the interaction of phenolic ions with the membrane? The purpose of using the 0.01 M carbonate buffer is to increase the activity of the solution so that it is expected to improve the sensor performance, but the data were not inconclusive. It is possibly due to high concentration of the carbonate buffer, which is 100 times higher than the NaOH concentration in phenol solution. In addition, CO₃⁻² and HCO₃⁻ ions geometrically have a larger size than Na⁺ and OH⁻ ions; thus, it is possible that the CO₃⁻² and HCO₃⁻ ions inhibit the interaction of phenolic ions with the membrane.

Figure 6: Potential cell of phenol solution at pH = 11 in carbonate and in NaOH buffers.

In Figure 6, it can also be seen that the signal generated from the NaOH solution is greater than the signal generated from the 0.01 M carbonate buffer. The interaction of the electrode interface with the electrolyte solution can produce a double layer of electricity on the surface of the electrode, producing a capacitor [35–37]. In the nonstirring process, the velocity of the capacitor formation depends on the mobility of the ions naturally. Mobility of OH⁻ (20.64 × 10⁻⁸ m²s⁻¹v⁻¹) is higher than the mobility of CO₃²⁻ (7.18 × 10–8 m²s⁻¹v⁻¹); hence, OH⁻ ions have a greater ability to get to the surface of the electrode [36]. This can cause the capacitance on the electrode surface to be higher in NaOH solution, than in a carbonate buffer solution, if observed at the same time. This concluded the signal produced in the NaOH solution is greater.

3.4. Addition of Fe₃O₄ Nanoparticles

Another factor that can affect sensor performance is the nature of the membrane, including selective membranes and conductive membranes. In an effort to increase membrane conductivity, Fe₃O₄ nanoparticles were added. The SEM images of the sensor surface with and without addition of Fe₃O₄ nanoparticles are presented in Figure 7. Both images show significant difference, in which nata de coco has homogeneous fibery surface, whereas due to addition of Fe₃O₄ nanoparticles, some of the surface area were covered with the granule shape. This also indicates that the Fe₃O₄ has been incorporated into the membrane. Based on SEM images, Fe₃O₄ makes the electrode surface to become rougher and uneven which eventually affects the sensor performance. The electrode surface profile can affect the amount of Easy [15], thus affecting the value of K.

(a)

(b)

Figure 7: SEM images (mag: 10000x) of the sensor surface without (a) and with (b) addition of Fe₃O₄ nanoparticles.

Furthermore, it is needed to investigate whether the nata de coco membrane is a selective membrane or not without the addition of phenolic ions; hence, it is necessary to also make a sensor with a membrane without phenol. Figure 8 shows a comparison of the potential cell (signal) produced by three electrodes made of different membranes, namely, electrodes with membranes made of nata de coco only (A), made of nata de coco and phenol (B), and made of nata de coco, phenol, and Fe₃O₄ (C). The electrode membrane thickness of A, B, and C was set equal to 100 μm.

Figure 8: Signal data profile on changes in phenol concentration in each membrane electrodes. Electrodes with membranes made of nata de coco only (A), nata de coco + phenol (B), and nata de coco + phenol + Fe₃O₄ (C).

3.5. Validation of Phenol Sensors

Phenol sensor validation was carried out for 20 days using 4 identical sensors. Based on the five repetitions, the average signal of four sensors and phenol concentration are showing the same profile, as seen in Figure 9. Sensitivity, which was indicated by the value of the Nernstian factor, tends to decrease in less than 5%, except for the 5th repetition in which the decrease was up to 7%. Based on Table 2, it can be seen that phenol sensors have a good sensitivity for up to 4 times measurement, day 15 (each sensor was used to measure 64 standard phenol solutions and samples). Although the sensitivity of the sensor tends to decrease in each repetition, the accuracy of the sensor (or reproducibility of the measurement results) of the 4 sensors is reasonably very good; it can be seen from the RSD which is not more than 1%, even if the measurement results are repeated for 5 times.

Figure 9: Profile of Ecell toward phenol concentrations at pH 11. Average signal data on first measurement (I), second repetition (II), third repetition (III), fourth repetition (IV), and fifth repetition (V).

Table 2: The average of Nernstian factor of 4 phenol sensors.

Days to
Nernstian factor (mV/decade)
RSD (%)
Bias (%)

1
55.71 ± 0.43
0.77
—

5
54.43 ± 0.27
0.50
2.30

10
53.75 ± 0.12
0.22
3.52

15
53.42 ± 0.48
0.90
4.12

20
51.86 ± 0.12
0.23
6.91

The measurement accuracy can be observed from the recovery of standard samples added to river water samples. This was done so that the environmental conditions of the standard phenol electrolyte are the same as that of river water samples. As shown in Table 3, the average recovery is 93.50% for 4 μM standard phenol and 93.25% for 8 μM (the data are the average of the measurement obtained from 4 phenol sensors with 4 repetitions each). Based on Tables 2 and 3, it can be highlighted that phenol sensors produced from the nata de coco and Fe₃O₄ membrane have a good precision and accuracy.

Table 3: Phenol concentration in river water sample and the recovery percentage.

Solution
Phenol concentration (μM)
Recovery (%)

Sample
1.60 ± 0.04
—

Sample + 4 μM phenol standard
5.34 ± 0.04
93.50 ± 0.75

Sample + 8 μM phenol standard
9.06 ± 0.04
93.25 ± 0.44

4. Conclusions

Nata de coco can be used as a membrane material in a phenol sensor selective membrane by potentiometry. Membrane thickness of up to 100 μm increases membrane electrical conductivity, and this corresponds to an increase in the Nernstian factor. The highest Nernstian factor was produced by electrodes with a membrane of 100 μm. Setting the pH of the phenol solution to 11 using NaOH solution produces a better signal than using the carbonate buffer of 0.01 M. The addition of 0.1% Fe₃O₄ nanoparticles in the electrode membrane improves the sensor performance. In the concentration range of 10⁻⁸–10⁻² M, the Nernstian factor increases from 41.8 ± 1.3 to 55.7 ± 0.4 mV/decade. This sensor, which was prepared from nata de coco and Fe₃O₄ membrane, shows good precision and accuracy.

Data Availability

All data used to support the finding of this study are available from the corresponding author on reasonable request.

Conflicts of Interest

The authors declare that there are no conflicts of interest regarding the publication of this article.

Acknowledgments

We would like to thank Brawijaya University, Indonesia, for providing the laboratory facilities and technical support.

References

X. Dong, Q. Hong, L. He, X. Jiang, and S. Li, “Characterization of phenol-degrading bacterial strains isolated from natural soil,” International Biodeterioration & Biodegradation, vol. 62, no. 3, pp. 257–262, 2008.

Kementerian Lingkungan Hidup–Republik Indonesia, PP. Nomor 3 Tahun 2010, Baku Mutu Air Limbah Bagi Kawasan Industri, KLH, Jakarta, Indonesia, 2010.

Badan Standarisasi Nasional–Republik Indonesia, Air Dan Air Limbah-Bagian 21: Cara Uji Kadar Fenol Secara Spektrofotometri (SNI 06-6989.21-2004), BSN, Tangerang, Indonesia, 2004.

E. T. Eftimie, S. Catalina, and M. Alexandra, “Ion selective polymeric membranes for chemical sensor. I,” Proceedings of the Recent Researches in Energy Environment and Sustainable Development, vol. 1, pp. 161–165, 2012, https://pdfs.semanticscholar.org/54de/eb82d55b999099d557cd7e14df89d2fa1abc.pdf.

M. A. Simon and R. P. Kusy, “The molecular, physical and mechanical properties of highly plasticized poly(vinyl chloride) membranes,” Polymer, vol. 34, no. 24, pp. 5106–5115, 1993.

I. Sulaeva, U. Henniges, T. Rosenau, and A. Potthast, “Bacterial cellulose as a material for wound treatment: properties and modifications. A review,” Biotechnology Advances, vol. 33, no. 8, pp. 1547–1571, 2015.

M. S. Dayal and J. M. Catchmark, “Mechanical and structural property analysis of bacterial cellulose composites,” Carbohydrate Polymers, vol. 144, no. 6, pp. 447–453, 2016.

Q. Yang, H. Ma, Z. Dai et al., “Improved thermal and mechanical properties of bacterial cellulose with the introduction of collagen,” Cellulose, vol. 24, no. 9, pp. 3777–3787, 2017.

S. Pirsa and S. Chavoshizadeh, “Design of an optical sensor for ethylene based on nanofiber bacterial cellulose film and its application for determination of banana storage time,” Polymers for Advanced Technologies, vol. 29, no. 5, pp. 1385–1393, 2018.

P. Milindanuth and P. Pisitsak, “A novel colorimetric sensor based on rhodamine-B derivative and bacterial cellulose for the detection of Cu(II) ions in water,” Materials Chemistry and Physics, vol. 216, no. 8, pp. 325–331, 2018.

S. Abbasi-Moayed, H. Golmohammadi, and M. R. Hormozi-Nezhad, “A nanopaper-based artificial tongue: a ratiometric fluorescent sensor array on bacterial nanocellulose for chemical discrimination applications,” Nanoscale, vol. 10, no. 5, pp. 2492–2502, 2018.

W. H. Chan, A. W. M. Lee, and M. S. Wong, “Some observations on the determination of phenol using ion selective electrodes,” Microchemical Journal, vol. 40, no. 3, pp. 322–327, 1989.

A. A. Tyas, Z. Aini, and W. S. Wari, “Determination of phenol in water using phenol sensor,” Jurnal Penelitian Saintek, vol. 20, no. 1, pp. 53–56, 2015.

E. Pretsch, M. Badertscher, M. Welti, T. Maruizumi, W. E. Morf, and W. Simon, “Design features of ionophores for ion selective electrodes,” Pure and Applied Chemistry, vol. 60, no. 4, pp. 567–574, 1988.

E. Bakker and Y. Qin, “Electrochemical sensors,” Analytical Chemistry, vol. 78, no. 12, pp. 3965–3984, 2006.

Z. Rappoport, The Chemistry of Phenols, John Wiley & Sons, New Jersey, NJ, USA, 2003.

N. N. Said, F. Hamzah, N. A. Ramlee, and N. N. Yunus, “The effect of TiO₂ particles addition on the characteristics of polysulfone membrane,” International Journal on Advanced Science, Engineering and Information Technology, vol. 8, no. 3, pp. 825–831, 2018.

D. Pramana, A. Mulyasuryani, and R. Triandi Tjahjanto, “Membranes of nata de coco-nanoparticles Fe₃O₄ for diazinon sensors,” The Journal of Pure and Applied Chemistry Research, vol. 7, no. 3, pp. 275–281, 2018.

F. G. Nejad, H. Beitollahi, and S. Shakeri, “Magnetic core–shell Fe₃O₄@SiO₂/graphene nanocomposite modified carbon paste electrode for voltametric determinantion of ascorbic acid in the presence of L-cystein,” Analytical and Bioanalytical Electrochemistry, vol. 8, no. 3, pp. 318–328, 2016.

M. H. Mashhadizadeh and H. Jeilanpour, “Voltammeric investigation of ketotifen using carbon paste electrode modified by magnetic core-shell Fe₃O₄@TMSPT nanoparticles,” Analytical and Bioanalytical Electrochemistry, vol. 6, pp. 308–320, 2014.

E. M. Rosida, A. Mulyasuryani, and R. T. Tjahjanto, “Modification of screen printed carbon electrode (SPCE) with Fe₃O₄ for the determination of nitrite (NO2-) in squarewave voltammetry,” Molekul, vol. 12, no. 2, pp. 139–145, 2017.

Y. Yang, Y. Zhang, J.-C. Shen, H. Yang, Z.-G. Zhou, and S.-P. Yang, “A highly selective magnetic sensor with functionalized Fe/Fe₃O₄ nanoparticles for detection of Pb²⁺,” Chinese Chemical Letters, vol. 27, no. 6, pp. 891–895, 2016.

N. Sanaeifar, M. Rabiee, M. Abdolrahim, M. Tahriri, D. Vashaee, and L. Tayebi, “A novel electrochemical biosensor based on Fe₃O₄ nanoparticles-polyvinyl alcohol composite for sensitive detection of glucose,” Analytical Biochemistry, vol. 519, pp. 19–26, 2017.

W. Zhang, X. Li, R. Zou et al., “Multifunctional glucose biosensors from Fe₃O₄ nanoparticles modified chitosan/graphene nanocomposites,” Scientific Reports, vol. 5, Article ID 11129, pp. 1–9, 2015.

T. Yin and W. Qin, “Applications of nanomaterials in potentiometric sensors,” TrAC Trends in Analytical Chemistry, vol. 51, pp. 79–86, 2013.

A. Kaushik, R. Khan, P. R. Solanki et al., “Iron oxide nanoparticles–chitosan composite based glucose biosensor,” Biosensors and Bioelectronics, vol. 24, no. 4, pp. 676–683, 2008.

O. Yavuz, M. K. Ram, and M. Aldissi, “Chapter 9. Electromagnetic applications of conducting and nanocomposite materials,” in The New Frontiers of Organic and Composite Nanotechnology, V. Erokhin, M. K. Ram, and O. Yavus, Eds., pp. 435–475, Elsevier, Oxford, UK, 2008.

P. Tipsawat, U. Wongpratat, S. Phumying, N. Chanlek, K. Chokprasombat, and S. Maensiri, “Magnetite (Fe₃O₄) nanoparticles: synthesis, characterization and electrochemical properties,” Applied Surface Science, vol. 446, pp. 287–292, 2018.

M. Chanif and S. Sarwito, “Analisa pengaruh penambahan kapasitor terhadap proses pengisian baterai wahana bawah laut,” Jurnal Teknik ITS, vol. 3, no. 1, pp. 1–6, 2014.

N.-L. Wu, S.-Y. Wang, C.-Y. Han, D.-S. Wu, and L.-R. Shiue, “Electrochemical capacitor of magnetite in aqueous electrolytes,” Journal of Power Sources, vol. 113, no. 1, pp. 173–178, 2003.

S.-Y. Wang, K.-C. Ho, S.-L. Kuo, and N.-L. Wu, “Investigation on capacitance mechanisms of Fe[sub 3]O[sub 4] electrochemical capacitors,” Journal of The Electrochemical Society, vol. 153, no. 1, p. A75, 2006.

E. Pratiwi, A. Mulyasuryani, and A. Sabarudin, “Modification of screen printed carbon electrode (SPCE) with polypyrrole (Ppy)-SiO₂ for phenol determination,” The Journal of Pure and Applied Chemistry Research, vol. 7, no. 1, pp. 12–18, 2018.

M. Fouskaki and N. A. Chaniotakis, “Thick membrane, solid contact ion selective electrode for the detection of lead at picomolar levels,” Analytical Chemistry, vol. 77, no. 6, pp. 1780–1784, 2005.

A. R. Rebelo, A. J. Archer, X. Chen, C. Liu, G. Yang, and Y. Liu, “Dehydration of bacterial cellulose and the water content effects on its viscoelastic and electrochemical properties,” Science and Technology of Advanced Materials, vol. 19, no. 1, pp. 203–211, 2018.

W. Huang, A. K. Diallo, J. L. Dailey, K. Besar, and H. E. Katz, “Electrochemical processes and mechanistic aspects of field-effect sensors for biomolecules (a review),” Journal of Materials Chemistry C, vol. 3, no. 25, pp. 6445–6470, 2015.

F. Faridbod, A. Shafaat, and M. R. Ganjali, “Conducting polymer all solid state potentiometric sensor for the tramadol assay,” International Journal of Electrochemical Science, vol. 11, no. 12, pp. 10123–10134, 2016.

T. Ghosh, H.-J. Chung, and J. Rieger, “All-solid-state sodium-selective electrode with a solid contact of chitosan/prussian blue nanocomposite,” Sensors, vol. 17, no. 11, pp. 1–17, 2017.