Empirical Wavelet Transform Based ECG Signal Filtering Method

,


Introduction
Te electrocardiogram (ECG) is a widely utilized and noninvasive medical tool employed to assess the cardiac health of patients.Tis painless test, typically conducted in clinical settings, yields quick results.Te ECG is a key diagnostic tool for determining underlying disorders as it tracks the heart's rhythm and looks for aberrant electrical activity [1].It provides a quick and efective way to examine how the heart works and diagnoses a variety of cardiovascular conditions, including coronary heart disease, arrhythmias, and myocardial infarction.Tanks to continuous technological improvements, ECGs will remain essential for patient diagnosis and treatment, providing clinical teams with vital information about patients' health.Electrocardiography (ECG) has become indispensable for identifying cardiovascular diseases (CVDs), responsible for approximately 31% of fatalities worldwide [1,2], due to its high accuracy and cost-efectiveness.
Advancements in technology have revolutionized the monitoring of cardiac signals, providing a deeper understanding of our heart's well-being.However, during the recording of electrocardiogram (ECG) signals, interference and unwanted artifacts can contaminate the data.Tis occurs when external sources or patient movement introduces electrical noise, which is then captured by the electrodes used for data acquisition, resulting in distorted ECG tracings.Te presence of such noise can lead to misinterpretation and hinder accurate diagnosis, as it distorts waveform patterns and compromises the reliability of ECG readings.Common types of noise in ECG signals include power-line interference, motion artifacts, electrode interface issues, poor electrode-skin contact, and white Gaussian noise, all of which introduce high-frequency components that interfere with signal interpretation [3][4][5][6][7][8][9].
To address this challenge, employing proper fltering techniques becomes essential for reducing noise and obtaining reliable ECG signals.Filtering the ECG signals can unveil vital information about the heart, enabling early diagnosis, improved risk assessment, and more efective treatments for cardiovascular diseases.Filtering plays a crucial role in the interpretation process by eliminating noise and artifacts that can afect the accurate analysis of important features such as the PQRST waves.It also facilitates the isolation of signals at diferent frequencies, enabling comprehensive study and analysis.When performed correctly, fltering ECG signals empowers clinicians to make more precise diagnoses of cardiovascular conditions and diseases, thereby enhancing patient care and outcomes.Recognizing the signifcance of fltering ECG signals allows us to take proactive measures in safeguarding ourselves against this life-threatening condition.
Other notable approaches include Weiner and Kalman denoising techniques [23,24], which utilize statistical estimation to remove noise from the ECG signal.Te Liouville-Weyl flter [25] addresses noise removal through a combination of time and frequency domain processing, thereby incorporating the strengths of both domains.Te fractional Stockwell transform (FST) [26] represents a novel method that combines wavelet decomposition and short-time Fourier transform to achieve enhanced denoising capabilities.
While these methods exhibit promise in mitigating noise in ECG signals, it is important to note that each approach has its own set of advantages and disadvantages.Factors such as computational complexity, adaptability to diferent noise types and levels, preservation of signal features, and robustness against artifacts and disturbances all contribute to the distinct characteristics of these techniques.To comprehend the trade-ofs and determine the best strategy for certain applications, more research and comparison studies are required, paving the way for continued improvements in ECG signal denoising approaches.
We used the empirical mode decomposition (EMD) method, which breaks the signal down into a collection of intrinsic mode functions (IMFs), to solve the problem of noise in the signal.Tese IMFs are functions characterized by an equal number of extrema and zero-crossings, and their envelopes are defned by the local maxima and minima, exhibiting symmetry around zero.However, it is observed that the noise components tend to be distributed across several low-order IMFs, making their removal necessary for noise reduction.
In our denoising process, we selectively removed these IMFs containing noise constituents.While this step successfully reduced the noise, it came at the cost of losing some important information, leading to a potential drawback in signal reconstruction.Mode mixing, another challenge encountered in this process, further complicated the reconstruction of the original signal [10][11][12].Te phenomenon of mode mixing occurs when the diferent IMFs interact with each other, resulting in a distorted representation of the underlying signal.
It is important to acknowledge that the removal of specifc IMFs for noise reduction is a trade-of between preserving signal fdelity and reducing unwanted noise.Striking the right balance is crucial to ensure accurate signal reconstruction while efectively suppressing noise interference.Further research and exploration of alternative methods are necessary to address the issues related to information loss and mode mixing, allowing for more robust and reliable denoising techniques in the analysis of ECG signals.
Te DWT method, employing soft and hard thresholding techniques, is commonly used for noise removal in nonstationary signals.However, this method struggles to preserve edges efectively, leading to the potential loss of important signal details [16,17].On the other hand, the EMD-based method has shown better results compared to wavelet-based thresholding, but it may not entirely eliminate interferences.
To enhance the denoising capabilities, researchers have combined EMD and other methods with the Wavelet approach.However, this approach's major drawback lies in the challenging task of selecting the appropriate wavelet and threshold type.It requires careful consideration to achieve optimal results [26].
Adaptive flters ofer an alternative approach by utilizing a reference signal closely correlated to the original signal.While this method can efectively remove noise, it is not suitable for real-time applications due to the need for a reference signal [27,28].
In some studies, windowing techniques have been applied to preserve the QRS complex, the most crucial component of the ECG signal.However, this approach still results in residual noise in the QRS region [11].Other research eforts have employed a combination of DWT and the adaptive double-threshold flter (ADTF) for noise removal.Te DWT-ADTF hybrid approach combines the advantages of both the DWT and the ADTF methods to improve ECG signal fltering.Tis approach aims to handle electromyogram (EMG) noises, power-line frequency interference (50 Hz), and high-frequency noises that may interfere with the ECG signal [14].Unfortunately, this method discards the original subband obtained from the wavelet decomposition, leading to the loss of information in higher-frequency components.One major limitation in previous works is the lack of consideration for the dynamic behavior of the ECG signal.
Weiner and Kalman flters are known for their efciency in preserving signal edges.However, they are prone to blurring phenomena due to the integral-based blurring models utilized [23,24].As a result, there is a signifcant loss of high-frequency information in the original signal stream [29].
Tese limitations and challenges highlight the need for further research and development of denoising methods that can efectively address the preservation of signal details, adapt to dynamic ECG characteristics, and minimize information loss in higher-frequency components.
In a study conducted by the authors of [25], the limitations of LW (Liouville-Weyl) flters were identifed.Due to their computational complexity, these flters might be diffcult to use with big datasets or high-resolution signals.Furthermore, the linearity and stationarity assumptions made by the LW technique might not always hold true in real-world situations.As a result, these methods may not exhibit optimal performance when applied to denoise nonlinear and nonstationary signals.
Another research efort [26] focused on the FST (fractional Stockwell transform) denoising method.However, this method proved to be sensitive to the selection of parameters.Choosing the fractional order and threshold value for denoising requires careful consideration, as selecting incorrect parameters can lead to subpar denoising performance.It is crucial to strike the right balance and make appropriate parameter choices to achieve efective noise reduction using the FST method.
Tese fndings highlight the importance of considering the computational complexity, linearity, stationarity, and parameter selection when utilizing LW and FST methods for denoising tasks.Future research should explore alternative approaches that can overcome these limitations and provide robust denoising performance for various types of signals, including nonlinear and nonstationary ones.By addressing these challenges, researchers can advance the development of more efcient and accurate denoising techniques in the feld of signal processing.
To mitigate the impact of noise from various sources, the utilization of appropriate signal processing techniques becomes crucial.In this regard, the empirical wavelet transform (EWT) approaches [30,31] have emerged as efective solutions.Tis method combines the ideas of wavelet transform (WT), empirical mode decomposition (EMD), and Fourier transform to decompose signals into their constituents at diferent scales.Te EWT method is rooted in the empirical analysis of signals, enabling it to capture the nonstationary and nonlinear characteristics of the signal.Tis approach has been successfully employed in diverse felds, including the detection of bearing faults [32] in various applications such as high-speed trains [33], hydraulic pumps [34], renewable energy systems [35], and biomedical signals [36,37].Tese applications have demonstrated the favorable outcomes achieved through the implementation of the EWT method.
However, it is worth noting that the potential and performance of the EWT approach have not been directly compared with the aforementioned methods.To ascertain the efectiveness of the EWT method, a comprehensive evaluation was conducted using arrhythmia data sourced from the MIT-BIH database [38].Tis evaluation encompassed various types of real-world data as well as simulated noise scenarios to thoroughly assess the suggested method's capabilities.By subjecting the EWT approach to rigorous testing and comparison, researchers aimed to establish its efcacy in denoising ECG signals and highlight its potential benefts over existing techniques.EWT is inherently adaptive and fexible.It allows the decomposition of signals into modes with varying frequency content, providing adaptability to the diverse and dynamic nature of ECG signals.It improves the segmentation of the spectra, which is crucial for identifying various disturbances in ECG signals.EWT constructs a customized wavelet basis for each signal, allowing for a tailored approach to signal analysis.Tis customization is particularly benefcial when dealing with ECG signals that may have unique characteristics and structures.EWT excels in frequency localization.By decomposing the signal into modes with well-localized frequencies, EWT facilitates a detailed analysis of diferent components of the ECG signal, helping in the separation of desired components from the unwanted noise.
Te fndings from this investigation will contribute to expanding the knowledge and understanding of the EWT method's capabilities, enabling researchers and practitioners to make informed decisions regarding its implementation in ECG signal processing.Tis approach is compared with the discrete wavelet transform with adaptive dual-threshold fltering (DWT-ADTF) technique, and the results show that our method outperforms the DWT-ADTF technique in terms of signal-to-noise ratio (SNR) and root mean square error (RMSE), and it allows for an excellent recovery of the original ECG morphology and features, making it easier to detect subtle abnormalities in the heart's electrical activity.Te comparative analysis will shed light on the strengths and advantages of the EWT approach, ultimately facilitating the development of improved denoising techniques in the feld of signal processing.
Te structure of the paper is as follows.Te Methods section provides a comprehensive overview of the theoretical principles underlying the empirical wavelet transform (EWT) and DWT-ADTF technique.In addition, this section outlines the materials utilized and presents the proposed algorithm.In order to assess the performance of the algorithm, a series of simulations were conducted, and the results are detailed in the subsequent section.
Te Results section not only quantifes but also qualitatively evaluates the efectiveness of the proposed algorithm.Trough these simulations, the performance of the algorithm is thoroughly analyzed, taking into account various metrics and criteria.Te obtained results are then discussed and interpreted in the following section.Te goal of this discussion is to give readers a greater understanding of the simulation results, as well as the algorithm's advantages, disadvantages, and room for development.
Te "Conclusion" section outlines the most important conclusions and learnings from this research.Te conclusions draw upon the presented simulations and discussions, ofering a comprehensive overview of the algorithm's performance and its implications.Te conclusion section serves as a concise summary, encapsulating the main takeaways of the research and highlighting the contributions and signifcance of the proposed approach in the context of ECG signal processing.

Denoising Methods: Empirical Wavelet Transform
Te EWT is a technique used to decompose signals into their constituents at diferent scales, and this method was frst introduced by the authors of [30] in 2013.It combines the ideas of WT, EMD, and the Fourier transform, allowing the signal to be decomposed into a mode series.Based on the EWT analysis, the decomposition result is dependent on the spectrum segmentation.Terefore, the improvement in the segmentation of the spectra is crucial for the identifcation of various disturbances.It applies an adaptive flter that can be tuned according to the signal of interest and produces more accurate representations than traditional transform techniques, such as Fourier or wavelets [30][31][32][33][34][35].As its name implies, this method relies on empirical data instead of predefned functions to produce its results.Tis allows for improved fexibility in analyzing a wide variety of signals with minimal computational efort and complexity.Te EWT method assumes that the Fourier spectrum of the AM-FM component has a good stability [30][31][32][33][34][35].Te bandpass flter set construction can be employed to segment and fltrate the spectra, and the signal can be decomposed adaptively via several modes.
Te EWT technique is described in the following steps: (1) We frst determine the spectrum X(ω) of x(t) by FFT.Te frequency spectrum is standardized in the interval of [0, π].Ten, we locate the local maxima in the X(ω) with their associated frequencies, and we denote them as local peak frequencies.Assuming that X(ω) is composed of N local maxima with corresponding local peak frequencies Ω j , j � 1, 2, . . . . . . . . ., N, we order the detected maxima in descending order of frequency according to the amplitude of the local maxima [31,32].(2) On the base of the frequency dominant points of (3) Having identifed the segment, the empirical scaling function Υ 1 (ω) and the empirical wavelet function Η n (ω) can be built by the Littlewood-Paley and Meyer wavelets.Furthermore, their frequency domain forms are Υ 1 (ω) and respectful.
Η n (ω) � where τ n � cω n valid argument should be larger than zero and sufciently small, while β(x) represents a transition function as defned in reference [27], and this function is given by the following equation (4) Te number of levels of decomposition is dependent on the number of dominating frequency points, and the approximation coefcients and detail coefcients are available by empirical wavelets with low awareness. 4 Journal of Electrical and Computer Engineering Te detail coefcients are as follows: where <, > denotes the internal product.Te exponent -denotes the conjugate of the variables.( 5) Lastly, the mode components of u(t) may be generated as in equation ( 6), which shall be organized as a function of the frequency.
where * is the convolution calculation.
Te purpose of the analytic beneft of the EWT is to be able to recognize the diferent distribution of signal components from spectral information and to build an adaptive flter bank that is suitable for the extraction of modes at diferent frequencies.Te dominating point of frequency is the values of the frequencies of the various components of the signal target, which is the signifcant information implied in the frst phase of the EWT [36].
Te procedure for eliminating noise from ECG signals includes the steps outlined in this section, as depicted in Figure 2.

Physiology Signals
In this section, diferent ECG denoising methods, including DWT-ADTF, FST, Kalman, LW, Weiner, and EWT, are performed on the two ECG databases.Te two ECG databases employed in this study are the MIT-BIH arrhythmia and noise stress test databases.Tese databases are available to the public in [39].A short overview of these databases is presented in the following: Te ECG signal is considered nonlinear and nonstationary.Tis means that the shape of an ECG waveform (its morphology) can change depending on several factors such as heart rate, level of physical activity, age, or gender; furthermore, this variability makes it difcult to make general assumptions about how any given cycle should look like.Te most common source of noise that can attack ECG signals is power-line interference (PLI).PLI is a type of electromagnetic disturbance that can afect the signals in electronic devices located near the power lines.PLI occurs when electrical energy from an adjacent cable radiates or couples onto another device and interferes with its normal operation.Motion artifact (MA) is an electrical disturbance created when a patient or subject moves during the recording of electrocardiographic (ECG) signals.Tis kind of interference can be especially detrimental to data acquisition as it masks important features in the signal, and thus, misinterpretations can result from this type of noise.Te baseline wander (BW) is an undesired drift of the baseline value in ECG recordings.Tis happens when there are sudden changes or shifts in electrical potentials that cause fuctuations and disruptions in signal waveforms over time.Muscle artifacts are electrical disturbances caused by muscle movements that interfere with the quality of ECG tracings during data acquisition.Tese types of interferences can signifcantly impact accurate diagnostics as they tend to mask or distort important features in waveforms such as PQRST complexes due to their amplitude and frequency.Tis may result in delayed potential diagnoses if they are not properly fltered out before or during analysis.
Te denoising performance of the suggested method in comparison to the techniques cited above is evaluated for certain parameters such as SNR out , SNR imp , MSE, PRD, and RMSE.
Te MSE is generally computed to verify the diference between the original and denoised signals.Tis is given in the following formula: RMSE is generally computed to verify the diference between the original and denoised signals.Tis is given in the formula as follows: PRD is computed to verify the distortion of the denoised signal with the original signal and is provided by the following formula: Journal of Electrical and Computer Engineering Here, z(n) is the ECG signal before adding noise and z(n) is the reconstructed signal after fltering.
Te SNR Out equation is obtained by using the following equation: Let z(n) represent the original ECG signal before noise addition, z(n) denote the results after denoising the ECG signals, and N indicate the total number of biomedical signals utilized.
Te SNR improvement, or signal-to-noise ratio improvement, is a measure of how much better a signal is compared to the noise that is present in the environment.Te SNR imp equation is given in the following equation: where z a is the noisy signal.
It is a measure of the signal strength relative to the noise level, and the higher the SNR improvement, the better the signal quality.

Experiment and Analysis
Tis section looks into how the EWT performs under various contaminations.Te suggested algorithm is applied to the ECG signal taken from the MIT-BIH database, and this signal 115 is obtained from the PhysioNet dataset.Te signal lines transporting the ECG signal from a patient to the viewing equipment are afected by electromagnetic disturbances from the 50/ 60 Hz power line noise that can be simulated as sinusoids [18].Tis sinusoid model of the 50 Hz power line is given as where f is the power line frequency, A is the amplitude, and N(t) is the noise on the power lines.Te peak-to-peak amplitude is the noise level.Te power line has a 50 Hz frequency and A � 0.15 mV.Similar to how power-line interference was modeled in the previous equation, this one does as well.

Qualitative Results.
As an important step in the evaluation of the ECG denoising approach, the qualitative analysis of the results allows a signifcant study of the performances of an approach.Tis is due to the analysis of the distortion recovery quality of this approach.
Figure 2 shows a case of the ECG signal denoising using the proposed method.Te subject signal is the MIT-BIH record 119, and the additive noise is the synthetic power-line signal.Figure 3(a) presents the original signal of 119, Figure 3(b) presents the original signal with the interfered noise of the synthetic power-line signal, and Figure 3(c) presents the reconstructed signal.
As shown in this fgure, the proposed method allows an excellent recovery of the original ECG morphology and features.Tis is a result of the efective combination of discrete wavelet transform (DWT) and fast Fourier transform (FFT), which enables a robust performance in pinpointing the unwanted noise spectrum and removing it at the appropriate decomposition level.By integrating DWT and FFT techniques, we have achieved a high level of accuracy in identifying where the undesirable noise is present within the signal and removing it at the most suitable stage of signal decomposition.Tis synergy between DWT and FFT allows us to precisely isolate and eliminate the noise, enhancing the overall performance of the process.Figure 3 presents a case of the ECG denoising of 15 dB SNR level of the white Gaussian noise WGN based on the proposed method.Te analyzed signal, in this case, is the MIT-BIH record 103. Figure 4 Tis fgure allows observing the correct reconstruction of the original ECG features as the QRS complex and the other amplitude waves.Based on these results, the proposed method shows interesting performances in of qualitative results, which are illustrated in Figures 3 and 4 1 showcase the performance of the proposed method in correcting power-line interference in MIT-BIH record 115, characterized by noise N(t) with an amplitude (A) of 0.15 mV and a frequency (f ) of 50 Hz.Te comparison includes several recent methods, namely, RL, AZP, FZP, and ADTF-DWT, as documented in [18].

Quantitate Results. Te results presented in Table
Te signal-to-noise ratio (SNR) is a crucial metric indicating the quality of the corrected signal.A higher SNR out value signifes a better ability to recover the original signal from noise.Te proposed method outperforms all other methods (RL: 6.54/AZP: 12.17/FZP: 14.25/ADTF-DWT: 23.29/proposed method: 27.6265), including the competitive ADTF-DWT approach, demonstrating its efectiveness in mitigating powerline interference, and preserving signal integrity.
Te mean squared error provides a quantitative measure of the diference between the corrected signal and the true signal.A lower MSE indicates a closer match to the original signal.Te proposed method achieves the smallest MSE among all methods (RL: 0.0754/AZP: 0.0206/FZP: 0.0128/ ADTF-DWT: 0.0015/proposed method: 0.0007), emphasizing its accuracy in minimizing the deviation from the true signal.
Table 2 presents a comprehensive comparison between the proposed method and Liouville-Weyl fltering (LW) in the context of denoising ECG signal record 122.Te simulated sinusoid's amplitude is varied across three levels (A � 0.3 mV, A � 0.35 mV, and A � 0.4 mV).Te results highlight the efcacy of the suggested approach in comparison to the LW fltering method.
Te SNR out values demonstrate the superiority of the proposed method over LW fltering at each level of simulated noise amplitude (A � 0.  Journal of Electrical and Computer Engineering signify better noise reduction and improved signal quality.Te proposed method consistently outperforms LW, indicating its robustness across diferent noise levels. Te MSE values further support the efectiveness of the proposed method, consistently exhibiting lower error than LW fltering (A � 0.3 mV: proposed method � 0.0201, LW � 0.0235/A � 0.35 mV: proposed method � 0.0288, LW � 0.0344/A � 0.4 mV: proposed method � 0.0292/ LW � 0.0405).Lower MSE values indicate a closer match to the true signal, emphasizing the proposed method's accuracy in denoising ECG signals under varying noise amplitudes.
Table 3 provides a comprehensive comparison between the proposed method and two well-known denoising flters, namely, Weiner and Kalman flters, applied to the ECG signal records 103, 105, and 121.Tese signals are afected by both power-line interference and muscle artifact (MA), with a simulated sinusoidal interference of amplitude A � 0.15 mV.Te results highlight the robustness of the proposed approach, which consistently outperforms the compared flters even in the presence of more complex noises.
Te proposed method consistently demonstrates signifcantly higher SNR out values than Weiner and Kalman flters, indicating superior noise reduction and preservation of signal quality in the presence of power-line interference.
Te proposed method consistently yields lower MSE and PRD values than Weiner and Kalman flters for all three records.Te proposed method outperforms Weiner and Kalman flters in denoising MA interference, as evidenced by higher SNR out and lower MSE and PRD values for all three records.Tis suggests that the proposed method is more efective in handling complex noise scenarios associated with muscular activity during ECG signal recording.
Table 4 presents the results of the denoising process applied to signal 115 from the MIT-BIH database, focusing on EMG interferences.Te comparison includes various denoising flters, specifcally RL, AZP, FZP, ADTF-DWT, and the proposed method.Te evaluation metrics, SNR out and MSE, provide insights into the performance of each method in reducing the impact of EMG interferences on the signal.
Te SNR out values reveal that the proposed method achieves a signifcantly higher signal quality than the other denoising flters.A higher SNR out indicates a better noise reduction and improved fdelity of the signal.Te proposed method outperforms RL, AZP, FZP, and ADTF-DWT, thereby reinforcing its efectiveness in handling EMG interferences.
Te MSE values further emphasize the superior performance of the proposed method.Lower MSE values indicate a closer match between the denoised signal and the true signal.Te proposed method achieves the lowest MSE among all compared flters, highlighting its accuracy in preserving signal details.
Table 5 presents compelling fndings from an investigation focused on comparing the hybridization of adaptive dual-threshold fltering (ADTF) and discrete wavelet transform (DWT) applied to various signals extracted from the Arrhythmia dataset.Te study specifcally introduces additive white Gaussian noise (WGN) at a signal-to-noise ratio (SNR) level of 5 dB.Te evaluation metrics employed for this comparative analysis include mean squared error (MSE), root mean squared error (RMSE), percentage residual diference (PRD), and signalto-noise ratio improvement (SNR imp ).Te results demonstrate the promising performance of the proposed method, leveraging multidecomposition levels of analysis through both DWT and empirical mode decomposition (EMD) approaches.Te hybridization of ADTF and DWT, when applied to signals afected by additive WGN, exhibits favorable outcomes based on the specifed evaluation metrics.Tese fndings underscore the potential efcacy of the proposed method in enhancing the denoising process and preserving signal integrity in the presence of simulated noise conditions.
Table 6 provides a comparative analysis of several denoising flters, namely, Stockwell transform (ST), fractional Stockwell transform (FST), and the proposed method, applied to correct muscular artifact (MA) interferences in signal 222 from the MIT-BIH database.Te evaluation metrics include the mean squared error (MSE) and root mean squared error (RMSE) [26,30].
Te results for MA interference denoising demonstrate the superior performance of the proposed method compared to FST and ST.Te signifcantly lower MSE and RMSE values for the proposed method suggest that it is more effective in preserving the original signal while reducing the impact of muscular artifacts.Tis indicates that the proposed method outperforms both FST and ST in terms of accuracy and fdelity in MA interference correction.
Similarly, for electrode motion (EM) interference denoising, the proposed method exhibits competitive performance.While FST achieves an impressively low MSE and RMSE, the proposed method maintains a good balance between accuracy and robustness.ST, on the other hand, has higher MSE and RMSE values, indicating comparatively less efectiveness in handling EM interference.
Te table presents the signal-to-noise ratio improvement (SNR imp ) for the three denoising methods (EEMD, CEEMDAN, and EWT) across diferent SNR levels.In Tables 6-8, the proposed method demonstrates noteworthy outcomes when compared to various widely utilized and recently published methods [26].Te adaptability showcased by this method enables a versatile approach to treating ECG noise contamination, leading to competitive performance.Furthermore, the proposed method allows for a lower level of processing complexity in comparison to the CEEMDAN and EEMD processes, leveraging their empirical approaches in the proposed treatment.
In summary, the results suggest that the proposed method exhibits promising performance, ofering adaptability and competitive outcomes in managing ECG noise.In addition, the method introduces a favorable balance by achieving efective noise reduction comparable to CEEMDAN and EEMD while maintaining a lower processing complexity.Tis underscores its potential utility in practical applications where both performance and computational efciency are critical considerations.

Conclusion
In this study, we introduce the empirical wavelet transform (EWT) method as a promising approach for denoising ECG signals.Te denoising results obtained using the EWT method are compared and extensively discussed in relation to several conventional flters.Te evaluation metrics, including the smallest PRD, MSE, and RMSE values, as well as high SMP imp and SNR out , demonstrate the superior signal reconstruction capabilities of the proposed method.
Te efectiveness of the EWT approach is further highlighted by its successful performance in the presence of various disturbances commonly encountered in ECG signals, such as additive white Gaussian noise, muscular artifacts (MA), EMG, and PLI.Te EWT exhibits strong noise reduction capabilities and demonstrates outstanding robustness.Overall, our fndings suggest that the proposed approach excels in both visual assessment and quantitative measurements, delivering exceptional results in signal visualization and analysis.

Journal of Electrical and Computer Engineering
In conclusion, our study presents the empirical wavelet transform (EWT) method as an efective solution for denoising ECG signals.Te comparative analysis demonstrates its superiority over conventional flters, yielding improved signal reconstruction with minimized distortions.Moreover, the EWT showcases remarkable performance in handling diverse noise sources, making it a valuable tool for enhancing the visualization and quantitative measurements of ECG signals [42].

( 1 )
MIT-BIH Arrhythmia Database.It includes a group of 48 samples taken from 47 persons, where two records were obtained from the same person, and each sample had a length of 30 minutes.Tese records have a sampling frequency of 360 Hz and are digitized by using an 11-bit resolution over a 10 mV range [40].(2) MIT-BIH Sound Stress Test Database.It comprises a total of 15 records, 12 of which are ECG recordings of half an hour and the other 3 are typical noises, i.e., baseline drift, muscle noise (EMG), and electrode movement artifact.Tese noises are taken from the records by choosing their range, which is afected by the noise [40].

Figure 2 :
Figure 2: Flow diagram of the EWT method.
(a) presents the original signal of 103, Figure 4(b) presents the original signal interfered with the Gaussian noise signal, and Figure 4(c) presents the reconstructed signal. .

Figure 3 :
Figure 3: Te denoising outcomes of the PLI, including the original signal (119) (a), the infected signal (b), and the reconstructed signal (c).

Figure 4 :
Figure 4: Te results of the WGN's 15 dB of denoising: the original signal (103) (a), the infected signal (b), and the reconstructed signal (c).
can be found by various techniques.Te manner of segmenting the specter is shown in Figure1, and N continued intervals Λ n � [ω n−1 , ω n ] are obtained, where 2τ n is the transitional part of each ω n border (represented in the green region of Figure2) and n ∈ [1, N] and

Table 1 :
Results of the comparison of diferent fltering techniques.

Table 2 :
Results of the comparison of diferent fltering techniques (EWT and LW).

Table 3 :
Results of the comparison of diferent fltering techniques (EWT, Weiner, and Kalman).

Table 5 :
Te results of the proposed method for denoising artifcial interferences (WGN of 5 dB) in certain signals from the MIT-BIH database.

Table 6 :
Results of the comparison of diferent fltering techniques (ST, FST, and proposed method).
Te bold signifed the good results.
Te bold signifed the good results.