Determination of 8 Endogenous Alkaloid Components in Boletus Using Ultrahigh-Performance Liquid Chromatography Combined with Quadrupole-Time of Flight Mass Spectrometry

A ultrahigh performance liquid chromatography coupled with quadrupole-time of flight mass spectrometry (UPLC-Q-TOF/MS) method was developed for simultaneous determination of 8 endogenous alkaloid compounds in Boletus. Boletus samples were extracted by 50% (V/V) methanol-water solution, then separated by CORTECS UPLC HILIC column using a binary solvent system by gradient elution. *e analytes were determined by Q-TOF/MS in TOF MS and information dependent acquisition (IDA)-MS/MS mode. *e results showed that mass accuracy error of the 8 endogenous alkaloids were lower than 5.0×10, good linear relationship was got in range of 0.2–500 μg/L, and correlation coefficient was higher than 0.9990. *e limit of detection was in the range of 0.002–0.100mg/kg and the limit of quantification was in the range of 0.004–0.200mg/kg. Recovery of the method was in range of 80.1%–101.5% with spike levels of 0.004–2.00mg/kg, relative standard deviations were lower than 10%. *e method was simple, specific, and reliable. It could be used for the rapid screening and quantitative analysis of 8 endogenous alkaloids in Boletus.


Introduction
e Boletus is an edible fungus belonging to the Fungi, Basidiomycotina, Hymenomycetes, Agaricates [1]. It is mainly distributed in Yunnan Province of China. Due to a wide variety of secondary metabolites from fungus and their biological activities, extensive attention has been paid to the researches of fungus both in China and abroad [2]. Boletus is popular because of its delicious taste, appreciated nutrition value [3], and a variety of medicinal effects [4]. e chemical composition of the Boletus was mainly terpenes, flavonoid [5], phenols [6], and alkaloids, which possessed comprehensive biological activities, such as antioxidant, antifatigue, antitumor, and anti-inflammatory [7,8]. Generally, the species for which a particular alkaloid structure is characteristic are closely related and can be comprised within the next higher taxon, genus, or family.
Alkaloids have been reported as a group of basic organic substances of Boletus, containing at least one nitrogen atom in a ring structure in the molecule [9]. A number of alkaloids were discovered and reported with biological activities in Boletus. Nicotine is an important and most well-known alkaloid, and it is the most important alkaloid in tobacco. In addition, nicotine is also widely present in Boletus and other plants. Some countries restricted the nicotine content in some foods. For example, the European Commission stipulated that the maximum limit of nicotine in edible fungus fresh and dry products is 0.036 mg/kg and 1.17 mg/kg, while the maximum limit in Boletus dried product is 2.3 mg/kg. Besides nicotine, Boletus contains some important and pharmacologically secondary alkaloids. Choline is an indispensable substance for normal metabolism, which has the functions of promoting brain development and improving memory. Boletus is rich in amino acids. Arginine and ornithine produce putrescine under the decarboxylation of bacterial amino acid decarboxylase. Tryptamine is a monoamine alkaloid, based around the indole ring structure. It is regarded as the backbone for tryptamines, a group of compounds that include many pharmacologically active compounds, including serotonin (neurotransmitters), melatonin (hormone), and psilocybin (psychedelic drugs) [10]. Du et al. [11] isolated nicotinamide from the Boletinus pictus. Zhang et al. [12] found that anabasine, cotinine, and muscarine exist in Boletus. According to references, nicotine, anabasine, choline, tryptamine, putrescine, nicotinamide, muscarine, and cotinine can be detected in Boletus. However, due to a lack of research on the detection of alkaloids in Boletus, the inconsistency of the species, and the maturity of the tested samples, there are certain differences in the data.
At present, there are few studies on the detection of endogenous alkaloids in Boletus. e analytical methods of these alkaloids mainly include gas chromatography [13][14][15], high-performance liquid chromatography [16][17][18][19][20][21], and liquid chromatography-tandem mass spectrometry [22][23][24][25][26][27][28]. However, gas chromatography requires derivation and determination, and the process is relatively cumbersome; highperformance liquid chromatography has poor anti-interference ability [29]. Liquid chromatography-tandem mass spectrometry has a limited ability to identify isomers, and the ion dwell time is limited due to the scan rate. UPLC-Q-TOF/MS has the characteristics of high resolution, high sensitivity, high accuracy, and wide scanning range [30]. It has been widely used in quantitative analysis of pesticide residues, veterinary drug residues, and other research fields [31,32]. In this paper, UPLC-Q-TOF/MS was first used for fast and accurate determination of endogenous alkaloids in the Boletus. It can obtain accurate and stable test results and provide reliable technical support for enterprises and regulatory authorities. is study lays the foundation for the development and establishment of quality standards in Boletus resources, provide a reference for comprehensive quality control and evaluation of Boletus and its products, and promote sustainable development of the industry.

Instruments and Equipment.
e high-speed refrigerated centrifuge (CR22N, HITACHI, Germany), the vortex mixer (Vortex Genius 3, IKA, Germany), and the ultrasonic cleaner (Elmasonic P300H, Elma, Germany) were used in the procedure of extraction. e separation of compounds was carried out on a LC-30AD UPLC system equipped with a binary solvent manager, sample manager, and column manager (Shimadzu, Japan). Quantitative analysis of target compounds was conducted on a TripleTOFTM 5600 + quadrupole/time of flight mass spectrometry (AB Sciex, USA).

Sample Preparation.
e sample was pulverized uniformly on a small pulverizer, and one gram of pulverized sample was weighed and transferred to a 50 mL centrifuge tube. Following spiking 20 mL of 50% methanol-water solution, vortexing for 1 min, and sonicating for 20 min, the sample was centrifuged at 8000 r/min for 5 min. After centrifugation, the supernatant was filtered through a 0.22 μm nylon membrane before UPLC-Q-TOF/MS detection.

Mass Spectrometry Conditions.
e MS analysis was performed using an electrospray ion source (ESI) in positive ionization mode. e optimized parameters of ion source were as follows: the ionization voltage was 5.5 kV, the source temperature was 400°C, the pressure of curtain gas was 35 psi, the pressure of nebulizer gas was 50 psi, and the pressure of auxiliary gas was 55 psi; TOF MS conditions were as follows: scan range was 50∼500 m/z, duration time was 15 min, and accumulation time was 0.15 s. IDA-MS/MS conditions were as follows: accumulation time was 0.05 s, high sensitivity mode was set, exclude isotopes were within 4 Da. Declustering potential was 80 V, collision energy was 35 V, and the dynamic energy of collision was 15 eV. Before each experiment, the instrument performed mass accuracy calibration through the CDS system. During the experiment, every five samples were run, and the mass accuracy calibration was automatically performed. e mass spectrum information of 8 endogenous alkaloids was shown in Table 1.

Optimization of Sample Extraction.
To find out an optimum extraction method for samples, this experiment investigated the extraction effect when acetonitrile, methanol, ethanol, water, 50% acetonitrile-water solution, 50% methanol-water solution, and 50% ethanol-water solution were used as the extraction solvent, and the extraction time of 10, 15, 20, and 30 min was compared.

Optimization of Liquid Chromatography and Mass Spectrometry Conditions.
ree analytical columns were tested prior to the selection of any additional chromatographic parameters, the Acquity BEH C18 column (100 mm × 2.1 mm, 2.5 μm), the Acquity UPLC HSS T3 column (100 mm × 2.1 mm, 1.8 μm), and the Cortecs UPLC HILIC column (100 mm × 2.1 mm, 1.6 μm) were used. e mobile phase was evaluated. First, methanol and acetonitrile were tested as organic solvents in the mobile phase. In relation to the aqueous phase, an aqueous solution with various concentrations of NH 4 OAc (0, 5 mmol/L, 10 mmol/L, 15 mmol/L, 20 mmol/L) were tested.
Syringe injection was used to inject 8 endogenous alkaloids directly into the mass spectrometer. e experiment investigated the response of target compounds under positive and negative ionization modes. For this method, TOF MS and IDA MS/MS modes were adopted for the detection of a target analyte. Under TOF MS mode, the experiment investigated the response of target compounds under different declustering potentials (50-300 V) and found the declustering potentials with the highest response.

Method Validation.
Validation of the whole analytical method was performed with linearity, the limit of detection (LOD), the limit of quantification (LOQ), precision, accuracy, and recovery. e linearity of the method was evaluated by constructing calibration curves with different concentrations of 8 alkaloids. e LOD and LOQ under the present chromatographic conditions were calculated on the basis of the response and slope of each regression equation at signalto-noise ratios (S/N) of 3 : 1 and 10 : 1, respectively. e recoveries of analytes were evaluated by adding the standard solutions with three different concentration levels (1x LOQ, 3x LOQ, 10x LOQ) to the known amounts of fungus samples, and each level was repeated six times. e precision of the method was expressed by a relative standard deviation (RSD).         Figure 1 shows, the extraction efficiency of extraction solvents for various endogenous alkaloids was different. 50% methanolwater solution yielded the best reproducibility and recovery of the 8 alkaloids. erefore, a 50% methanol-water solution was selected as the extraction solvent.

Optimization of Extraction
Times. As Figure 2 shows, when the ultrasonic extraction time was increased from 10 min to 20 min, the extraction efficiency of alkaloids was greatly improved. After 20 min, the growth trend decelerates, and the extraction efficiency of alkaloids changes little. Considering the extraction efficiency of the experiment and time cost, 20 min was chosen as extraction time.

Optimization of Liquid Chromatography Conditions.
e results showed that the nicotine, anabasine, choline, tryptamine, putrescine, and muscarine were not retained on the Acquity BEH C 18 column and the Acquity UPLC HSS T3 column. ey can be retained on the CORTECS UPLC HILIC column, and the separation effect was good. Because they are strong polar hydrophilic compounds, which were weakly retained on the C 18 and T3 columns, so it is necessary to replace hydrophilic columns for separation. erefore, the CORTECS UPLC HILIC column (100 mm × 2.1 mm, 1.6 μm) was selected as the analytical columns. e mobile phase was evaluated. First, methanol and acetonitrile were tested as organic solvents in the mobile phase, observing that methanol and acetonitrile as mobile phases have little effect on the signal intensity, but when methanol was used as the mobile phase for gradient elution, column pressure varies widely, and equilibration takes     longer. In relation to the aqueous phase, the results showed 20 mmol/L NH 4 OAc as the aqueous solution showed better separation and elution capabilities. Considering that under the positive mode, the addition of formic acid could increase the ionization of the compounds, which improved the separation efficiency and the intensity of the mass spectrometry signal. erefore, 20 mmol/L NH 4 OAc solution containing 0.1% (v/v) formic acid and acetonitrile containing 0.1% (v/v) formic acid were selected as the mobile phase for subsequent experiments.

Optimization of Mass Spectrometry Parameters.
e TripleTOF ™ 5600 + instrument has a CDS automatic calibration infusion system. e reference spray of the DuoSpray ™ ion source was used to input the calibration solution for automatic system calibration. e DuoSpray ™ ion source has two types: ESI and atmospheric pressure chemical ionization source (APCI). In this experiment, the ESI was selected as the detection ion source, and the APCI was used as the calibration ion source. rough the automatic calibration system, automatic batch calibration was performed to ensure the accurate mass of the system was stable for a long time.
e results showed that compounds have a higher response in positive ionization mode, so the positive ionization mode was used for detection. Under TOF MS mode, the experiment investigated the response of target compounds under different declustering potentials (50-300 V), with the findings that at the fragmentor voltage of 80 V, compound response was the highest; relatively low fragmentor voltage was unfavorable for ion transmission, and overly high fragmentor voltage would cause the compound to fragment within the source. e accurate mass, retention time, and isotope ratio was obtained. Figure 3 shows the extracted ion chromatograms of 8 alkaloids (500 ng/mL). e accurate mass deviations of target compounds were less than 5.0 × 10 −6 ( Table 1). e MS/MS spectra of the target were used for the final confirmation of the initial screening results of the accurate mass ( Figure 4).

Method Validation
3.3.1. Linearity and Sensitivity. As Table 2 shows, the linear range was studied by preparing a calibration curve with a concentration range of 0.2-500 μg/L for each compound,    Table 2.