Industrial Qualification Process for Optical Fibers Distributed Strain and Temperature Sensing in NuclearWaste Repositories

Temperature and strain monitoring will be implemented in the envisioned French geological repository for highand intermediate-level long-lived nuclear wastes. Raman and Brillouin scatterings in optical fibers are efficient industrial methods to provide distributed temperature and strain measurements. Gamma radiation and hydrogen release from nuclear wastes can however affect the measurements. An industrial qualification process is successfully proposed and implemented. Induced measurement uncertainties and their physical origins are quantified. The optical fiber composition influence is assessed. Based on radiation-hard fibers and carbon-primary coatings, we showed that the proposed system can provide accurate temperature and strain measurements up to 0.5 MGy and 100% hydrogen concentration in the atmosphere, over 200 m distance range. The selected system was successfully implemented in the Andra underground laboratory, in one-to-one scale mockup of future cells, into concrete liners. We demonstrated the efficiency of simultaneous Raman and Brillouin scattering measurements to provide both strain and temperature distributed measurements. We showed that 1.3 μm working wavelength is in favor of hazardous environment monitoring.


Introduction
Distributed optical fiber sensors (OFSs) [1][2][3] are a key technology for the monitoring of the planned French deep geological repository for long-lived high-level and intermediatelevel wastes, called Cigéo.Temperature and strain distributed sensing based on Raman, Rayleigh, and Brillouin scatterings offer exceptional advantages over traditional electronic sensors, especially as they provide distributed data over the entire structure and thus overcome limitations of traditional sensors, whose information is restricted to local effects.
This paper focuses on temperature and strain distributed sensing based on Raman and Brillouin scatterings in optical fibers for structural health monitoring, more precisely for nuclear industry.Although commercial off-the-shelf sensors and interrogation units are numerous, the global measuring chain may provide disappointing monitoring results to the end-users, unless a number of considerations specific to nuclear environments are taken into account.These are further developed within this paper, with an emphasis on environmental conditions influence, especially (i) temperature, (ii) gamma rays, and (iii) hydrogen influences.
Andra's (French National Radioactive Waste Management Agency) potential applications include surface and deep geological radioactive waste disposal structure monitoring, for instance within the future geological repository that would contain highly instrumented disposal cells.Intermediate-level long-lived waste cells are presently designed as 400 m long tunnels, with a 1 m thick concrete liner, placed 500 m deep in a clay rock called Callovian-Oxfordian formation.
Monitoring aims at preserving retrievability of nuclear wastes, assessing long-term safety, enabling optimization of structures all along the exploitation which is expected to last a century.A major specification of the geological repository monitoring system is durability, required to last up to a century, despite hazardous conditions: gamma rays and hydrogen release.Because of high gamma radiation doses, disposal cells are not accessible as soon as exploitation starts, and the first nuclear waste package is placed inside the disposal cell.This implies the monitoring system to be robust for decades without any maintenance.OFS are a robust technology known to handle radiations quite well.Durability is also ensured thanks to their ability to perform remote sensing, which enables maintenance of optoelectronic instruments.Finally, OFSs are very attractive for their small size and varied external coatings, which limits the preferential flow paths and reduces invasiveness, a highly important aspect to avoid affecting the long-term safety of the geological repository for nuclear wastes.
For these attractive advantages, Andra drives many research studies on distributed temperature and strain sensing with optical fiber sensors.To ensure measurement quality and lifetime of the global monitoring system, Andra has implemented a qualification procedure, which will be presented in the first part of the paper.Previous reported results gathered within this qualification process will be recalled.The second part focuses on recent results: field test where temperature compensation of strain measurements, based on combined Brillouin and Raman scatterings, was successfully implemented.The third part presents research results on hazardous condition influences on both Brillouin and Raman scatterings as a function of optical fiber types.We conclude with a recommendation on the optimized sensing system for our application.

Qualification Procedure and Previous Results
2.1.Qualification Procedure.The outstanding properties of optical fiber sensors drove major interest in structural health monitoring applications.Nevertheless, it is important to notice that optical fiber and optical installation practices used in telecommunications and other industries are significantly different from nuclear constraints and applications, which may be far less tolerant.Moreover, optical fiber sensing systems presently suffer from a lack of standardization of claimed performances.Dedicated qualification processes are not defined yet.Andra has implemented a multistage qualification procedure for each selected measurement chain.For OFS, it was chosen to study both sensing cables and optoelectronic instruments separately before pairing such elements and focusing on data processing.
The described overall process is inspired from [4].Global test sequence includes four stages.
Stage one consists in acquiring in-depth knowledge of the sensing technology, engineering solutions, and practical implementation constraints.It aims at selecting the technologies best suited to the specific requirements of monitoring the geological repositories for long-lived nuclear wastes.When off-the-shelf sensing chain performances do not fulfil requirements, Andra initiates research programs.It has been the case for distributed optical fiber sensing system whose results are presented in this paper.
Stage two consists in carrying out laboratory tests, under fully supervised and/or controlled environmental conditions, to qualify the sensitive component and assess the complete measurement chain performances.Sensors are tested alone, then embedded in the host material of interest.
Stage three consists in outdoor tests, to evaluate field implementation influence.At this stage, the sensing chain is preserved from hazardous conditions, extreme temperature, or gamma rays.Unexpected influence of various parameters might thus be revealed.
The fourth stage involves hardening in view of the application environmental conditions.In the envisioned French geological repository, temperature would range from 20 • C to 90 • C. Gamma radiation rates reach 1 Gy/h, total dose 10 7 Gy.Hydrogen release is also expected; its maximum levels could approach 100% hydrogen content in the atmosphere.

Previous
Results.This qualification methodology has been implemented for distributed temperature and strain sensing.
Andra selected Raman scattering to perform distributed temperature measurements, as it is the most advanced technology with superior temperature sensitivity, better than 0.1 • C [5].Andra selected Brillouin scattering for distributed strain sensing, since interferometric measurements are in favor of durability.The first three stages of the qualification procedure, laboratory tests, and preliminary outdoor tests were reported in [6].Two remaining questions were to be addressed to end the qualification procedure.
First, how to compensate for Brillouin scattering temperature sensitivity?The Brillouin frequency shifts are known to be proportional to temperature (ΔT) and strain (ε) variations as in (1) [7]: C T and C ε are characteristics of the optical fiber type.At the operating wavelength (1550 nm), for standard G652 singlemode fiber, C T and C ε are in the order of 1 MHz/ • C and 0.05 MHz/με [8].Instruments based on Brillouin scattering would perform either temperature or strain measurements; strain is 20 times less influent than temperature.A solution to decorrelate strain from temperature is detailed in Section 3.
Second, what are hydrogen and gamma rays influences on Raman and Brillouin scatterings?Quantitative evaluation of such influence as a function of optical fiber types is presented in Section 4. Based on these various evaluations, a last paragraph predicts lifetime of the optical fiber geological repository monitoring system, anticipating strain and temperature measurements uncertainties and distance range.

Tunnel Liner Instrumentation: Strain Measurement Temperature Compensation
Andra has created an Underground Research Laboratory to evaluate the constructability, safety, and reversibility of the potential radioactive waste disposal in the Callovian-Oxfordian clay stones.A multidisciplinary program is implemented in this 500 m underground structure.The "retaining and covering observation experiment" is taking place into a dedicated gallery with a concrete liner (Figure 1), similar with intermediate-level long-lived waste disposal cells.Its purpose is to evaluate several monitoring solutions.This experiment integrates onfield constrains (drilling environment, dust, and operational phases with limited intervention time) and a representative experiment in view of future geological repository (one-to-one scale, specific rock in its natural location).

Instrumented Structure.
The gallery is 5 m in diameter with a 50 cm thick concrete liner.Many parameters were monitored in both clay stones and covering concrete: temperature, strain, water content, and interstitial fluid pressure.
One section of the concrete liner was instrumented with collocated fiber optic sensing cables whose picture is illustrated in Figure 2.
Andra had previously tested Raman temperature sensing into single-mode fibers in a surface building slab.This test highlighted the great sensitivity of Raman scattering in single-mode fiber to curvature, which can poorly be avoided in civil engineering structures [9].This is why two different sensing lines were implemented: multimode fibers were used for Raman temperature monitoring and single-mode fibers were installed for Brillouin strain sensing.More precisely, we used three different sensing cables, two for Brillouin measurements, composed of G652 and G657 fiber types to evaluate curvature sensitivity, and one for Raman sensing.The three cables were placed redundantly so that, finally, six sensing arches were embedded inside the concrete liner.
In order to ensure accurate positioning and maintaining during concrete pouring, sensing cables were attached to a thin piece of wire mesh, spitted on the retaining concrete.
Measurements were remotely performed every 15 minutes by commercially available instruments, a Brillouin OTDA (optical time-domain analyser) and a Raman distributed temperature sensing (DTS).Both devices were set at 0.5 m spatial resolution.
Instruments were located in another gallery of the underground laboratory.The full sensing line is approximately 500 m, with only 20 m embedded inside the concrete liner.
Electronics sensors, such as vibrating wire extensometers and platinium probes, were colocated with optical fiber sensors.

Results of Early Age
Monitoring.The gallery liner was constructed in 2 steps.First, the bottom part (inverted arch) concrete was poured.One month after, the second arch completed the liner.One from the six sensing lines got damaged during construction.We did not observe any advantage of G652 compared with G657 on the strain measurement quality.
During concrete hardening, an exothermal chemical reaction takes place.Consequently, temperature increases and Brillouin frequency shifts.Thermal expansion of concrete induces strain on the optical fiber cable, which also increases Brillouin frequency.Temperature measurement acquired by Raman sensing lines (left y axis) and raw Brillouin frequency (right y axis) are illustrated in Figure 3.
To provide useful information, concrete thermal expansion must be differentiated from the parameters of interest: strain induced by stress, creep, and shrinkage, noted ε comp (comp.stands for "compensated").If α concrete is the concrete thermal expansion coefficient, temperature-compensated strain can be obtained following equation ( 2): Embedded and instrumented concrete samples [10] enabled in situ measurement of α concrete = 10 με/ • C. Assuming C T = 1 MHz/ • C and C ε = 0.05 MHz/με, we obtained the compensated strain measurements plotted in Figure 4.
The 150 μm/m compressive strain measured is the consequence of the early-age shrinkage of concrete.This value is fully consistent with the one measured by the vibrating wire extensometers placed nearby the optical fibers.
Uncertainties were evaluated analyzing 24 repeated measurements acquired during 6 h (inset Figure 4).Such analysis has been repeated.We obtain uncertainty in the order of 30 μm/m, which corresponds to the Brillouin instrument performance.Temperature uncertainty was better than 0.2 • C in the Raman sensing line (Figure 3).The Raman-Brillouin compensation method does not increase the initial strain and temperature measuring system uncertainties.
Our results were obtained with 500 m distance range and a 0.5 m spatial resolution.Similar experiments of Raman temperature compensated Brillouin strain measurements reported 3.6 • C and 80 μm/m repeatability with 5 m resolution [11].
This method also enables long-term monitoring.Concrete liner strain evolution has been acquired all along construction steps; measurements will go on in the next years to acquire as much information as possible before the repository construction.
An advantage of distributed measurements provided by optical fibers is illustrated in Figure 5.In this gallery, covering concrete is not reinforced (no steel bar).Tensile strain zones were identified and precisely located by distributed measurements at locations where there was no electronic sensor.
As conclusion, a sensing scheme combining Raman and Brillouin instruments with multiple sensing cables is a promising solution for performing simultaneous temperature and strain monitoring.This real-scale experiment demonstrates that a Raman-Brillouin temperature and strain sensing is very well suited for underground tunnel monitoring.It also appeared that chosen optical fiber cables were robust enough to put up with construction conditions.

Compatibility with Future Hazardous Environments
Nuclear waste repository is a challenging environment due to the presence of gamma radiations which is known to degrade the optical properties of fibers through three different phenomena [12]: radiation-induced attenuation (RIA) decreases fiber transmission efficiency, radiationinduced emission (RIE) decreases signal-to-noise ratio and compaction changes refractive index (especially for very high particle fluences or doses).
The amplitudes and kinetics of these changes depend on many parameters, among which are dose rates, total doses, optical fiber type, and operating wavelength.As illustrated in Figure 6, in Andra geological repository, doses rates are moderate; however, after 100 years of monitoring in the vicinity of high-level wastes structure cells, total doses will reach 10 7 Gy.

Gamma Influence on Brillouin
Scattering.It has been demonstrated on standard G652 fibers that there is a limited influence of gamma rays on the Brillouin frequency at least up to doses of 100 kGy [14].To evaluate possible influence at total doses corresponding to our application, we performed in situ [15] and postmortem gamma-ray irradiation tests in different classes of fibers.
We selected our samples among the most commonly used fibers, like the SMF28 from Corning, since almost all commercially available Brillouin instruments are designed to operate with this type of fibers.Thus, the first selected fiber was the step-index fiber SMF28 from Corning.In view of the geological repository monitoring requiring high gamma dose tolerance, we selected a fluorine-doped cladding, pure silica-core fiber.The fiber has a 5 μm core radius and a 40 μm cladding radius with a fluorine concentration of 1.25% mol.Finally, to evaluate the influence of high concentrations of dopants, we tested a highly GeO 2 -doped core fiber at 28% mol concentration with a core radius of 1.4 μm and a pure silica cladding with radius of 62.5 μm (HGe in the following).
As expected and detailed in Table 1, measured RIA levels are important at the high doses (up to 10 7 Gy), with a factor up to 4 between the RIA measured for the two extreme cases.The most tolerant fiber, the F-doped fiber, quickly saturates, at a moderate dose level of 5 MGy, at RIA level around 50 dB/km.Impact on distance range is discussed later.
For the dose range corresponding to geological repository application, measured central Brillouin frequencies also shifted (BFS) with dose, whatever the considered optical fiber types.Main parameters of the Brillouin spectrum are listed in Table 1 as well as their changes with dose.Unlike SMF28 and HGe fiber, a clear saturation in the Brillouin frequency shift (Δν B ) occurs at ∼2 MGy for the F-doped fiber.This saturation effect is of major interest for the target application where expected doses largely exceed this saturation level dose.Moreover, for this fiber, Brillouin frequency shift remains small, in the order of 2 MHz, which corresponds to 2 • C measurement error or 40 μm/m.On the opposite, 18 • C temperature measurement error would occur if HGe fiber was selected.
Strain and temperature sensing based on Brillouin scattering involves both the central Brillouin frequency measurement and hypothesis on calibration coefficients (C ε , C T ) (see (1)).These calibration coefficients might also be impacted by radiation.To our knowledge, it is the first time radiation influence on these parameters is evaluated.
No significant effect of radiations on C T could be observed for the SMF28 and HGe fibers (Figure 7).For the Fdoped fiber, a decrease of about 6% was noted in the C T coefficient after a deposited dose of 10 7 Gy.C ε strain coefficients do not seem to depend on received gamma dose, whatever the fiber type.As a conclusion, gamma impact on calibration factors would increase measurement uncertainties by few percent only.
As a conclusion, radiation effects on the performances of strain and temperature Brillouin scattering based optical fiber sensors have been deeply investigated for different classes of optical fibers.There is a strong influence of the core composition.Thus, optical fiber type should be carefully chosen to ensure durability of the monitoring system.
The presented tests are relevant since it is well known that higher dose rates increase RIA; thus, results overestimate degradations that will endure the geological repository monitoring system.
Fluorine-doped pure-silica core fibers are able to handle dose rates and total doses during one century of monitoring within the geological repository.Strain and temperature measurement uncertainties would slightly degrade.The most impacting radiation effect is RIA.Since optoelectronic instrument optical budget is in the order of 10 dB, distance range would be reduced down to 200 m for the F-doped fiber after a century of monitoring.It suits requirements as (i) high-level long-lived waste (HLW) disposal cell will be 40 to 100 m long and (ii) in 400 m intermediate-level long-lived waste storage cell, doses are greatly reduced.
On the opposite, with SMF28 fiber, distributed sensing would be compromised after only a decade of monitoring along 100 m of HLW disposal cell, instead of the required century.
We focused our study on the response at the 1550 nm wavelength since it is the commercial instrument working wavelength.However, a global optimization of the entire system must also consider this parameter as a possible variable.Initially, optical losses are slightly more important in pristine optical fibers at the 1310 nm wavelength compared to the ones reported at 1550 nm.Yet, after a 10 MGy dose, the fibers exhibit lower propagation losses at 1310 nm; for instance, 23 dB/km versus 56 dB/km at 1550 nm for the fluorine fiber.

Hydrogen Influence on Brillouin Scattering.
As introduced previously, for nuclear waste repository instrumentation, hydrogen influence must also be quantified.
Hydrogen originates from (i) nuclear waste release and (ii) anoxic corrosion of metallic materials.Although hydrogen releases are expected small (in the order of 430 mmol/hour release for each intermediate-level nuclear waste), when ventilation stops with cell closure, concentrations would slowly yet regularly increase.Its maximum levels could approach 100% hydrogen content in the atmosphere in few months.
The chosen samples were the same F-doped and HGe fibers used for radiation studies.A G652 from iXFiber company was chosen as a reference.It is composed of a GeO 2 -doped core fiber at 3.4 mol% concentration with a 4.6 μm radius and a pure silica cladding with radius 62.5 μm.
The optical fibers were placed inside autoclave chambers, where molecular hydrogen pressure was maintained at 150 bars and temperature was regulated at 25 • C. With such conditions, hydrogen concentrations into the optical fiber core reach more than 95% of the saturation level in the fiber core after 330 h (around 13 days).
Selected pressure condition accelerates hydrogen diffusion.We also performed measurement during natural hydrogen release at the end of the experiment: 16 days (versus 13 days under pressure) were required for exposed optical fibers to retrieve their original characteristics.
During hydrogen loading, samples were removed from autoclave regularly to perform measurements: distributed Brillouin scattering and absorption losses on a large wavelength span, from 1.1 μm to 1.56 μm.
Measured spectral attenuations confirmed three absorption bands appear in the attenuation spectra of exposed optical fibers, the most important at 1245 nm and two smaller at 1165 nm and 1130 nm, consistently with [17].Losses reached 70 dB/km (resp.50 dB/km) at saturation at 1550 nm (resp.1310 nm).Standard Brillouin instruments have optical budget in the order of 10 dB.As a result, in instrumented disposal cells for long-lived nuclear wastes, maximal distance range would significantly diminish with the increase of hydrogen content in the atmosphere, from the kilometer range down to one hundred meters.Improvement could be expected if working wavelength was tuned down to 1.3 μm.
Brillouin spectra before and after hydrogen exposure are illustrated in Figure 8 for the G652 fiber.
Brillouin scattering is modified by hydrogen content in optical fibers.On top of reduced amplitude, Brillouin scattering shifts towards high frequencies.This shift is somewhat linear and reaches 21 MHz at saturation for the G652 and the F-doped fibers, 18 MHz for the HGe fiber.Assuming standard coefficients (C T = 1 MHz/ • C, C ε = 0.05 MHz/με), such a shift would induce an error in temperature (resp.strain) measurement in the order of 21 • C (resp.420 μm/m).
Fluorine fiber is revealed to be more sensitive to small and moderate hydrogen contents than other fibers.Virgin fiber Hydrogen loaded fiber Amplitude (a.u.) Brillouin frequency (GHz) Figure 8: Measured Brillouin spectra in the G652 fiber before and after 13 days exposure to hydrogenated atmosphere (at saturation level).
At saturation, F-doped fiber and G652 are similar; HGe fiber is slightly less impacted.Further work is required to conclude if dopant type, internal stress, or defect concentrations are the major cause, or if differences are induced by different diffusion rates.This result is not in favor of the application requirements since the F-doped fiber was the only sample which could endure radiations.In view of our application, we tested carbon coating which is known to prevent hydrogen migration into silica [18].Samples endured the same hydrogen exposure.As detailed in [19], carbon coating of G652 fibers proved to be fully efficient since neither variation of propagation losses nor Brillouin frequency shift could be detected along these samples.
As a conclusion, an F-doped fiber with carbon coating is mandatory for the monitoring of the future geological repository where there will be both gamma radiation and hydrogen release.

Gamma Influence on Raman Sensing. Unlike Brillouin
sensing which is wavelength encoded, for Raman sensing phenomena, RIA not only degrades distance range, but also induces measurement error.Indeed, temperature values are obtained by taking the ration of the two Raman components (Antistokes and Stokes) as detailed in (3) [20,21]: where λ AS and λ S are Raman Antistokes and Stokes wavelengths, Δυ = 13.2THz for silica, h is the Planck's constant, c is the speed of light in vacuum, k is Boltzmann's constant, T is the temperature of the optical fiber, and α refers to propagation losses.Gamma radiation induces differential losses at Stokes and Antistokes wavelengths, which then induce temperature measurement errors, consistently with (3), as detailed in [20,21].Promoted solutions were to use double-ended configuration (closed loop).
However, single-ended arrangement (open loop) suits geological repository specification much better since (i) waste disposal cells are not always accessible at both ends and (ii) ability to perform measurements up to breaking points is important for durability in the order of a century.This is why we turned towards single-ended configuration and chose to focus on radiation-hard optical fibers.
Multimode silica-based fibers with 50 μm core diameters were selected: standard (OM4 type) and radiation-hard fibers.Two samples were extracted from each optical fiber coil; one has been left pristine, the second sample has been irradiated.
We used a Co source at a dose rate of 0.65 kGy/h.A 67 kGy total dose was deposited in two steps.Postmortem temperature measurements were performed on pristine and irradiated samples of the two optical fiber types, placed inside a climatic chamber.Results are illustrated in Figure 9 for the 50 • C temperature step.
Temperature measurements performed on standard pristine fiber matches imposed oven temperature (50 • C) with a 1 • C shift.Once irradiated, large measurement errors appear; error increases with distance, up to 10 • C error cumulated in 20 m.OTDR (optical time-domain reflectometer) measurements revealed 80 dB/km at 850 nm and 21 dB/km at 1300 nm.As shown in Figure 9, raw data may be processed to take into account this 0.06 dB/m differential loss value, applying (3).It limits significantly temperature error, but linear drift still remains.Since Raman instrument works at 1064 nm, propagation losses measurements should rather be performed at the real Stokes and Antistokes wavelengths, 1110 nm and 1020 nm.It will be performed in a near future.Pristine as well as irradiated radiation hard-fiber samples provide successful temperature measurements (Figure 9), however, with poor accuracy, in the order of 3 • C.
We increased irradiation dose.Second irradiation run was performed in MOL facilities, simultaneously with singlemode fibers.It consisted in gamma rays, with a dose of 1,5 kGy/h during 341 hours.Total dose of 551,5 kGy was thus deposited on top of the previous 67 kGy, which provided 618 kGy.Sample was submitted to temperature changes, 23 • C, 40 • C, 50 • C, and 60 • C. As illustrated in Figure 10, temperature sensing remains possible even after 0.6 * 10 7 Gy value that approaches a hundred years of monitoring in the geological repository.
Similarly, with the pristine radiation-hard fiber, temperature measurement quality is poor, degraded by (i) a bias to the oven temperature value and (ii) a linear drift that increases with distance.This linear error is positive, whereas it was negative for standard fiber.Indeed, measured losses at 850 nm and 1300 nm were, respectively, 51 dB/km and 74 dB/km (larger at 1300 nm); thus, differential losses are opposite to previous case.

Hydrogen Influence on Raman
Scattering.Hydrogen influence on Raman distributed temperature sensing has been described in the literature [22] and is expected to induce differential losses, thus temperature reading errors similar with radiation effects.Based on hydrogen tests performed with single-mode fibers (Section 4.2), Andra will supply carbon-coated multimode radiation-hard fibers in the future.4.5.Discussion.In order to predict distributed temperature and strain measurement performances in the future geological repository for long-lived nuclear waste, we considered separately gamma radiation and hydrogen influences on Brillouin and Raman scatterings in various optical fiber types.
However, combined influence of hydrogen and radiation is expected to be in favor of measuring performances [23].Presented results are worst cases.
Carbon coating should also be in favor of radiation tolerance.
Temperature, in the order of 90 • C in the vicinity of highlevel waste disposal cell, will accelerate hydrogen diffusion in optical fibers [24]; for ease of manipulation, pressure, instead of temperature, was used in the test.Equivalence will need to be checked.
Tests on radiation influence on Raman sensing were limited to postmortem measurements.These preliminary results will be augmented with on-line measurements on hydrogenloaded carbon-coated radiation-hard fiber.Spectral transmission measurements will also be included to take full advantage of the compensation model.

Conclusion
We developed a strategy to design a durable monitoring system for nuclear structures, based on truly-distributed optical fiber sensors.
We demonstrated Brillouin strain measurement can be efficiently compensated from temperature influence installing a parallel sensing line paired with a Raman instrument.
State-of-the-art results on gamma and hydrogen influences on Rayleigh, Raman, and Brillouin scatterings were obtained.Fluorine-doped pure-silica-core fibers coated with carbon will be mandatory to handle high gamma dose (10 7 Gy) and hydrogen release in the vicinity of high-level long-lived waste disposal cells.After a century of monitoring based on Brillouin scattering, maximal range would decrease down to a hundred meter but strain measurement uncertainty should remain stable.With other fiber types or primary coatings, large strain measurement errors would ruin the monitoring system reliability.
To reduce hydrogen and radiation influences, a significant future improvement could be achieved by choosing the operating wavelength at 1.3 μm instead of 1.55 μm.
We showed that radiation-hard fibers enabled single-end Raman measurements up to 0.6 MGy; however, temperature measurement uncertainty, presently in the order of few degrees, must be improved.A postprocessing to compensate for differential loss induced by radiation has been proposed and will, in the future, take benefit of spectral measurements.
Next step of Andra qualification procedure is to incorporate these special fibers into strain sensing cables and run (i) combined hydrogen-gamma influence test and (ii) another outdoor test, to evaluate if curvatures or splicing may get more sensitive than with standard fibers.

Figure 1 :Figure 2 :
Figure 1: Scheme of gallery instrumentation by distributed sensing and picture of the gallery 6 months after liner casting.

Figure 3 :
Figure 3: Raman temperature measurement and Brillouin frequency shift during concrete hardening.

Figure 4 :Figure 5 :
Figure 4: Processed data: temperature compensated strain acquired during concrete pouring (x axis is dates, from October to November 2011).

Figure 6 :
Figure 6: Overview of typical applications as a function of total dose and dose rate [13].

Figure 7 :
Figure 7: Temperature (C T -up) and strain (C ε -down) Brillouin calibration coefficients, with received gamma dose for the three optical fiber types.

Figure 9 :
Figure 9: Temperature measurements based on Raman scattering in OM4 and radiation-hard fibers, pristine, and irradiated samples placed in an oven at 50 • C.

Figure 10 :
Figure 10: Temperature measurements based on Raman scattering in the radiation-hard fiber irradiated at 618 kGy total dose.

Table 1 :
Summary of radiation-induced attenuations and Brillouin frequency shifts.