Journal of Toxicology
Volume 2020 (2020), Article ID 1467456, 6 pages
https://doi.org/10.1155/2020/1467456

Research Article

In Vitro Determination of Genotoxicity Induced by Brackets Alloys in Cultures of Human Gingival Fibroblasts

Juan Pablo Loyola-Rodríguez,¹ Ildelfonso Lastra-Corso,² José Obed García-Cortés,² Alejandra Loyola-Leyva,³ Rúben Abraham Domínguez-Pérez,⁴ David Avila-Arizmendi,⁵ Guillermo Contreras-Palma,⁵ and Cecilia González-Calixto⁵

¹Universidad Popular Autónoma del Estado de Puebla (UPAEP), Puebla, Mexico
²Universidad Autónoma de San Luis Potosí, San Luis Potosí, Mexico
³CIACYT, Universidad Autónoma de San Luis Potosí, San Luis Potosí, Mexico
⁴Universidad Autónoma de Quéretaro, Quéretaro, Mexico
⁵Universidad Autónoma de Guerrero, Acapulco, Guerrero, Mexico

Correspondence should be addressed to Juan Pablo Loyola-Rodríguez; juanpablo.loyola8@gmail.com

Received 9 December 2019; Accepted 22 July 2020; Published 27 August 2020

Academic Editor: You-Cheng Hseu

Copyright © 2020 Juan Pablo Loyola-Rodríguez et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

Abstract.

Orthodontic brackets release ions that can be reabsorbed in the oral mucosa, potentially causing complications, including cytotoxic effects and mutagenic alterations. The aim was to evaluate the genotoxicity induced by orthodontic appliance alloys in cultures of human gingival fibroblasts by comet assay. Eluates were obtained from the following brackets alloys: EconoLine (SS: stainless steel), MiniMirage (Ni-Ti: nickel-titanium), Nu-Edge (Co-Cr: cobalt-chromium), In-Vu (PC-polycrystals (PC) aluminum oxide), and Monocrystal IZE (monocrystalline (MC) aluminum oxide). Each bracket was sterilized and exposed to a corrosive process for 35 days. The obtained eluates were tested for genotoxicity of human gingival fibroblasts (HGFA) by the alkaline comet assay. All study groups showed genotoxic effects; there was a significant difference () among groups. The eluates obtained from Ni-Ti showed a 16-times greater genotoxic effect. There were differences in genotoxicity after comparing the Ni-Ti with SS () and Co-Cr brackets (). The ceramic was more genotoxic than metallic brackets (SS and Co-Cr), but less than the Ni-Ti. This in vitro model will be useful for further study of early DNA damage caused by brackets and other biomaterials used in the oral cavity before their introduction into the clinical setting.

1. Introduction

The appliances used for fixed orthodontic therapy include brackets, bands, and archwires. These are manufactured to have a high corrosion resistance using stainless steel, nickel-titanium, chromium, or nickel-cobalt alloys [1]. Brackets are devices build with adequate designs for each orthodontic technique, and different materials, such as metallic, ceramic, polymeric (polycarbonate and polyurethane), or a combination are used for their fabrication. The alloys of stainless steel (SS) commonly employed contained between 15 and 54% of nickel (Ni), 20–30% of chromium (Cr), and 40–60% of cobalt (Co). Ceramic brackets present optical and mechanical characteristics superior to polymeric brackets, and they can be classified according to their fabrication into polycrystalline aluminum oxide (Al₂O₃P) and monocrystalline aluminum oxide (Al₂O₃M) [2, 3].

The use of orthodontic appliances results in the release of metallic ions, such as nickel, chromium, and cobalt, into the oral cavity, which is of significant clinical concern. Electrochemical corrosion could occur when metals are in contact with mediums of electrolytic conductivity, especially in saliva and oral tissues. In this situation, metals are exposed to a degradation process promoted by thermal, microbiological, and enzymatic properties due to chemical reactions caused by food and beverage consumption [2]. Due to corrosion, metal alloys employed in orthodontics could cause an indirect effect on the DNA of cells by generating free radicals with mutagenic or carcinogenic effects. These results are due to the entry, accumulation, and the absorption of free radicals into oral and organism tissues (gingival, oral, intestinal, cutaneous, and respiratory system epithelium) [2, 4].

Some components of metallic alloys, especially Ni, Cr, and Co, can cause disturbances, such as hypersensibility, immunological sensibility, dermatitis, hyperplasia, gingivitis, and even asthma. Cobalt is an element capable of inducing abnormal mitosis, and Ni can inhibit cellular proliferation, lactate production, and consumption of glucose [2, 5]. Both metals are classified as carcinogenic because, in vitro, they are capable of causing harmful effects in human gingival mast cells and fibroblasts affecting their proliferation, morphology, and production of collagen [6, 7].

Cytotoxicity is a type of damage to the cell membrane exposing its cytoplasm causing cell death. Furthermore, genotoxicity is direct or indirect damage to cellular genetic material, which represents an essential part of assessing oral cancer risk and carcinogenic potential. In the oral cavity, these types of damage are caused by an increase in free radical production, mainly nickel, steel, and chromium ions. In the manufacture of brackets, different metallic materials and ceramic, polymeric, or combinations of materials are used. Due to this variety, the present study assessed various alloys of devices, such as metallic and ceramic, avoiding the use of recycled brackets [7]. A wide range of methods is used to detect early biological effects of DNA-damaging agents in environmental and occupational settings. There are well-established cytogenetic biomarkers, such as structural chromosomal aberrations, micronuclei, sister chromatid exchanges, and high-frequency cells [8]. However, during the last decade, the comet assay (CA or single cell gel electrophoresis) was based on the principle that damaged DNA moves faster than undamaged DNA in an agarose gel. A cell with DNA damage appears in the form of a comet, whereas an intact DNA appears as a halo [9]. An advantage over the cytogenetic biomonitoring techniques (micronuclei test: MNs) is that MNs are limited to circulating and proliferating cell populations, whereas the CA can be applied to both proliferating and nonproliferating cells.

There are reports that have addressed the genotoxicity induced by orthodontic materials, mainly analyzed by MNs [10, 11], but few studies have addressed this using CA [12]. An important point to note is that it is necessary to establish a rapid in vitro assay to test the biocompatibility of the different biomaterials used in clinical orthodontics. Therefore, this study aimed to analyze and compare the genotoxicity induced by eluates obtained from various brackets alloys using human gingival fibroblasts by the comet assay test.

2. Materials and Methods

2.1. Primary Culture of Human Gingival Fibroblasts Assay (HGFA)

Gingival biopsies were harvested from healthy donors who underwent a crown lengthening surgery for prosthetic reasons. Informed consent and research protocol were institutionally approved. The tissue sample was placed in 1 mL of DMEM (Dulbecco’s Modified Eagle Medium, Corning Life Sciences, NY, USA) supplemented with 10% fetal bovine serum (FBS) and antibiotics (penicillin 100 IU/ml, streptomycin 100 μg/ml, and amphotericin B 100 μg/ml). Collected tissues were washed twice with phosphate-buffered saline (PBS; 150 mM NaCl, 20 mM sodium phosphate, pH 7.2) supplemented with antibiotics (penicillin 100 IU/ml and streptomycin 100 μg/ml) and, then, cut into small pieces with a sterile surgical blade in a Petri dish. Tissue fragments were further digested by collagenase/dispase (Hoffmann-La Roche Corporation, Tucson, AZ, USA) for 1 h at 37°C in 100% of humidity, 95% oxygen, and 5% CO₂. Minced pieces of gingival tissue were washed with DMEM and FBS (10%) and explanted to 25 cm² tissue culture flasks (Falcon, Corning Life Sciences, NY, USA) containing 5 ml of supplemented DMEM and incubated at 37°C in 100% of humidity, 95% oxygen, and 5% CO₂. HGF was obtained by trypsinization of the primary cell outgrown and routinely passaged using 0.025% trypsin in PBS containing 0.02% EDTA at 90% of cell confluence. HGF between the 6th and 8th passages were used in the experiments. The cellular count was made using a Neubauer camera, and 1 × 10⁴ cells per well were cultured in plates of 48 wells (Falcon, Corning Life Sciences, NY, USA) [13].

2.2. Sample Preparation

The brackets were immersed in a solution containing acetic acid (0.1 M) and sodium chloride (0.1 M) to produce a corrosion process, and each sample was carried out by triplicate. Samples were maintained for 35 days at room temperature and, then, sterilized with ultraviolet light. HGF cultures were exposed to 200 μl of eluates for 1 h at 37°C. After exposure, HGF was washed twice with phosphate buffered solution (PBS) [14]. Cells were treated with the same solution used for corrosion (without brackets) for the positive control group, and saline was used as the negative control. Table 1 shows the properties of the brackets analyzed in this study.

Table 1: Type and composition of brackets used in the study.

Brackets
Composition
Made by

EconoLine 0.022
Stainless steel (SS)
Ah-Kim-Pech, Miami, FL, USA

MiniMirage 0.022
Nickel titanium (Ni-Ti)
Borgatta specialties, Estado de México, México

Nu-Edge 0.022
Cobalt-chromium (Co-Cr)
TP orthodontics, Inc., La Porte, IN, USA

In-Vu 0.022
Polycrystalline aluminium oxide (PC-AO)
TP orthodontics, Inc., La Porte, IN, USA

Crystall IZE 0.022
Monocrystalline aluminium oxide (MC-AO)
Ah-Kim-Pech, Miami, FL, USA

2.3. Genotoxicity Assay

To evaluate the magnitude of DNA damage, the alkaline version of a single cell gel (comet) assay was used. To break the cell membrane, HGF was fixed in a low melting point agarose and exposed to a cell lysis solution for 24 h. Next, the cells were submitted to an unwinding solution for 20 min to denature the DNA and free the broken fragments that suffered damage. Finally, exposed cells were subjected to electrophoresis for 20 min at 25 V and 300 mA, in which the particles migrate towards the anode, making a tail, giving the cells an appearance of a comet. To determine DNA damage, samples were stained with ethidium bromide and visualized in a fluorescence microscope. A total of 100 cells per sample were analyzed (n = 10 per group). The damage was analyzed using the software Comet Assay IV (Perceptive Instruments Ltd., Haverhill, UK); this software allows the capture and analysis of images from comet assay that permits the quantification of DNA damage and repair in a single cell preparation. Tail moment was calculated by the image analysis system as the product of the tail length (DNA migration) and the fraction of DNA in the comet tail (% DNA in the tail).

2.4. Statistical Analysis

Statistical software GraphPad InSat 3.1 (GraphPad Software, Inc.) was used to express data as means and standard deviations. The Kolmogorov–Smirnov test was used to determine data distribution. The Kruskal–Wallis test followed by Dunn post hoc analysis was used to assess the difference between groups. A p value lower than 0.05 was considered statistically significant.

3. Results

3.1. Genotoxicity Assay

Table 2 shows the DNA damage in HGFA in the study groups compared with the control; all groups showed biological effects evaluated by alkaline comet assay. However, the Co-Cr group showed less fibroblast DNA damage (0.724 ± 0.015). Besides, the eluates obtained from the analyzed brackets induced up to 3 times more damage. The greatest damage was detected in the samples exposed to the Ni-Ti solution (3.588 ± 0.137), which was approximately 17 times more genotoxic than the control (Table 2 and Figure 1).

Table 2: Genotoxicity (comet assay: tail moment) in cultures of human gingival fibroblasts induced by eluates obtained from brackets.

Bracket alloys
Comet assay
Increment of genotoxicity
value

SS
0.805 ± 0.002
3.79
<0.0001

Ni-Ti
3.588 ± 0.137
16.92

Co-Cr
0.724 ± 0.015
3.41

PC-AO
0.828 ± 0.019
3.90

MC-AO
0.832 ± 0.012
3.92

Positive control
0.212 ± 0.009

Kruskal–Wallis. SS: stainless steel; Ni-Ti: nickel-titanium; Co-Cr; cobalt chromium; PC-AO: polycrystalline Al₂O₃; MC-AO: monocrystalline Al₂O₃; comet assay: DNA damage percentage is expressed in mean and standard deviation. n = 10 per each group.

(a)

(b)

(c)

(d)

(e)

(f)

Figure 1: Representative pictures of the damage (genotoxicity effect on HGFA) caused by the exposure to different eluates from brackets as follows: SS (a), Ni-Ti showed the highest DNA damage (b), Co-Cr showed the lowest DNA damage (c), PC-AO (d), MC-AO (e), and control (f).

Table 3 shows a comparison of the genotoxicity between the different brackets alloys studied. We detected significant differences in the Ni-Ti brackets compared with SS (), Co-Cr (), and with the control group. Likewise, we detected a significant difference between Co-Cr and MC-AO groups (). The comparison of MC-AO and PC-AO groups showed a statistically significant difference () when compared with the control group. However, there was no statistical difference between MC-AO and PC-AO groups (). Therefore, it is assumed that ceramic brackets (MC-AO, PC-AO) are most genotoxic than the metallic brackets (SS and Co-Cr), but less than the Ni-Ti.

Table 3: Comparisons of the difference in genotoxicity among the different brackets alloys used.

Brackets alloys
Difference
value

Stainless steel vs. Ni-Ti
2.783
<0.01

Stainless steel vs. Co-Cr
0.081
>0.05

Stainless steel vs. PC-Al₂O₃
0.023
>0.05

Stainless steel vs. MC-Al₂O₃
0.027
>0.05

Stainless steel vs. control
0.593
>0.05

Ni-Ti vs. Co-Cr
2.864
<0.001

Ni-Ti vs. PC-Al₂O₃
2.76
>0.05

Ni-Ti vs. MC-Al₂O₃
2.756
>0.05

Ni-Ti vs. control
3.376
<0.001

Co-Cr vs. PC-Al₂O₃
0.104
>0.05

Co-Cr vs. MC-Al₂O₃
0.108
<0.01

Co-Cr vs. control
0.512
>0.05

PC-Al₂O₃ vs. MC-Al₂O₃
0.004
>0.05

PC-Al₂O₃ vs. control
0.616
<0.001

MN-Al₂O₃ vs. control
0.620
<0.001

Dunn’s test. Ni-Ti: nickel-titanium; Co-Cr; cobalt chromium; PC-Al₂O₃: polycrystalline aluminium oxide; MC-Al₂O₃: monocrystalline aluminium oxide.

4. Discussion

Ulcers and periodontal pathologies are the main clinical concerns regarding the effects of corrosion of the orthodontic devices due to damage caused to the tissues of the oral cavity [8, 15]. Therefore, it is essential to determine the levels of biocompatibility because brackets are placed in the proximity of the gum and even within the gum (when retained dental organs are pulled). Any metal placed within the oral cavity will suffer corrosion due to the formation of organometallic compounds, which depends on composition, temperature, humidity, and pH [16]. Furthermore, there are intrinsic (material composition, surface roughness, hardness, and other orthodontic devices) and extrinsic factors (cell renewal, proteins, enzymes, ions in saliva, bacteria, friction between orthodontic devices, toothpaste, and some mouthwashes) that contribute to the corrosion process [17, 18].

In vitro [19] and in vivo [20, 21] studies have evaluated the release of metals such as iron, nickel, and chrome from permanent orthodontic devices. It has been reported that ion release occurs during the first months of orthodontic treatment producing cytotoxic, immunogenic, and mutagenic effects. These effects are caused because materials accumulate in the adjacent soft tissue and could be absorbed, especially Ni-Ti ions through the transport system , calcium, and iron channels or by phagocytosis. Furthermore, it has been reported that Ni-Ti induces apoptosis (by the inhibition of ) among gingival fibroblasts, and that this involves caspase-3 activation [22].

There are reports of cellular models about the biocompatibility of different types of metallic and ceramic brackets [23, 24]. However, these studies only assess macrostructure and cell functions (viability and morphology), but not genotoxicity. Conversely, it has also been noted that some brackets are cytotoxic and genotoxic for fibroblast cultures, except titanium brackets [25]. Considering that brackets from different manufacturers have different corrosion behavior, it has been reported that Fe and Cr ions had the most abundant corrosion products of the evaluated materials [17, 19, 23]. However, the brackets that had more significant damage to their external structure were, in ranked order, NiTi, Co-Cr, SS, MC-AO, and PC-AO (data not shown).

There is controversy regarding the evaluation of genotoxicity in oral epithelial cells [2, 26]. Mouse lymphoma L5178Y or Chinese Hamster Ovary A common cells are commonly used for in vitro studies due to their ease of handling and cultivation [14]. However, human gingival fibroblasts are more sensible to evaluate side effects of dental materials compared to cell lines derived from animal tissues [27]. An important point to note is that HGFA could be useful to analyze the genotoxic effects of different metal appliances (implants, orthodontic appliances, and oral surgery appliances) that generate nanoparticles and micron-sized particles released in oral tissues [27].

A few studies have assessed the in vitro genotoxicity of SS brackets and other metal alloys and ceramics; there is information about the cytotoxicity of polycarbonate and ceramic brackets, but not about genotoxicity. In agreement with the present results, it has been reported that metal brackets produced genotoxicity in HGF when the comet assay was used [25] and others have observed that nanoparticles of cobalt-chromium induced increased DNA damage in cell cultures [35]. By contrast, it has been shown that none of these cited alloys cause DNA alterations in immortalized gingival keratinocytes [17]. The comet assay is a sensitive method to determine the genotoxicity throughout the detection of a primary lesion in the DNA chain of individual cells, which repair efficiently 1 h after their exposure to a toxic agent, mainly when it occurs in regions of active transcription. Primary lesions and double-strand breaks represent initial DNA damage events that can lead to chromosomal aberrations [12, 14, 28]. It has been described that Olive Tail Moment is one of the best indicators of DNA damage which is a practical measure to calculated DNA migration distance as a genotoxic effect [12].

Alloys with a high content of Ni are more susceptible to corrosion than steel, chromium, and magnesium alloys [15, 17, 25]. It has been reported that Ni is the most cytotoxic, and it releases ions that penetrate cells and subsequently affect the functionality and causes hypersensitivity reactions, such as dermatitis, stomatitis, asthma, and burning sensation in the esophagus and neck areas. Besides, the Ni cytotoxic effect reduces the sense of taste and acts as carcinogenic in the nasal cavity and respiratory system [25]. Manganese is a corrosion product from SS brackets that leads to mitochondrial dysfunction that could result in cellular oxidative stress. Producing intra- and extracellular accumulation of glycosaminoglycans (GAG) affects adhesion, migration, growth, and cellular differentiation [23, 29]. Therefore, other alloys, such as pure titanium (that is more inert and stable in the oral environment than SS), have been introduced. Titanium has high corrosion resistance, improved biocompatibility, and does not produce allergic responses [15].

On the other hand, ceramic brackets made of aluminum oxide can cause numerical chromosomal aberrations, and they are not inert materials, as previously thought. However, it has been reported that such brackets show higher biocompatibility in cell cultures due to the monocrystalline composition, which eliminates contamination or impurities during the manufacture. All brackets are a source of oxidative stress; conventional ceramic brackets are more biocompatible compared to polymeric and metal brackets [29].

The comet assay is a rapid, simple, and sensitive technique for measuring DNA breakage with a small number of cells and detects intercellular differences in DNA damage [30]. In vitro studies are simple and inexpensive, provide a significant amount of information, and are suited to elucidating the mechanisms of cellular toxicity presented in vivo [14]. Therefore, we believe that further in vitro studies must be performed with different brands of brackets to determine the type of production and quality of the corrosion resistance. Also, it will be necessary to detect and quantify the ions present and observe the type of genotoxic damage in oral tissues.

5. Conclusions

All types of orthodontic brackets, regardless of the constituent materials, produce DNA damage when its eluates are exposed to HGFA using the comet assay test. These findings show that ceramic brackets were more genotoxic than metallic brackets but less than the Ni-Ti. This in vitro model will be useful for further study of early DNA damage caused by brackets and other biomaterials used in the oral cavity before their introduction into the clinical setting.

Data Availability

The comet assay data used to support the findings of this study are available from the corresponding author upon request.

Conflicts of Interest

The authors declare that they have no conflicts of interest.

References

E. Toy, S. Yuksel, F. Ozturk, O. H. Karatas, and M. Yalcin, “Evaluation of the genotoxicity and cytotoxicity in the buccal epithelial cells of patients undergoing orthodontic treatment with three light-cured bonding composites by using micronucleus testing,” The Korean Journal of Orthodontics, vol. 44, no. 3, pp. 128–135, 2014.

F. Faccioni, P. Franceschetti, M. Cerpelloni, and M. E. Fracasso, “In vivo study on metal release from fixed orthodontic appliances and DNA damage in oral mucosa cells,” American Journal of Orthodontics and Dentofacial Orthopedics, vol. 124, no. 6, pp. 687–693, 2003.

S. Zinelis, O. Annousaki, T. Eliades, and M. Makou, “Elemental composition of brazing alloys in metallic orthodontic brackets,” The Angle Orthodontist, vol. 74, no. 3, pp. 394–399, 2004.

P. Wolowiec, K. Chojnacka, B. W. Loster, and M. Mikulewics, “Do dietary habits influence trace elements release from fixed orthodontic appliances?” Biological Trace Element Research, vol. 180, no. 2, pp. 214–222, 2017.

E. Fernández-Miñano, C. Ortiz, A. Vicente, J. L. Calvo, and A. J. Ortiz, “Metallic ion content and damage to the DNA in oral mucosa cells of children with fixed orthodontic appliances,” Biometals, vol. 24, no. 5, pp. 935–941, 2011.

H. S. Hafez, E. M. N. Selim, F. H. Kamel Eid, W. A. Tawfik, E. A. Al-Ashkar, and Y. A. Mostafa, “Cytotoxicity, genotoxicity, and metal release in patients with fixed orthodontic appliances: a longitudinal in-vivo study,” American Journal of Orthodontics and Dentofacial Orthopedics, vol. 140, no. 3, pp. 298–308, 2011.

P. Downarowicz and M. Mikulewicz, “Trace metal ions release from fixed orthodontic appliances and DNA damage in oral mucosa cells by in vivo studies: a literature review,” Advances in Clinical and Experimental Medicine, vol. 26, no. 7, pp. 1155–1162, 2017.

R. A. Mateuca, I. Decordier, and M. Kirsch-Volders, “Cytogenetic methods in human biomonitoring: principles and uses,” Methods in Molecular Biology, vol. 817, pp. 305–334, 2012.

M. Baricevic, I. Ratkaj, M. Mladinic et al., “In vivo assessment of DNA damage induced in oral mucosa cells by fixed and removable metal prosthodontic appliances,” Clinical Oral Investigations, vol. 16, no. 1, pp. 325–331, 2012.

M. G. Flores-Bracho, C. S. Takahashi, W. O. Castillo et al., “Genotoxic effects in oral mucosal cells caused by the use of orthodontic fixed appliances in patients after short and long periods of treatment,” Clinical Oral Investigations, vol. 23, no. 7, pp. 2913–2919, 2019.

P. G. Francis, M. Thomas, V. Antony, M. Shaloob, K. J. Hasam, and G. Roshan, “Cytomorphometric analysis on the effects of components of orthodontic appliances on the epithelial cells of the buccal mucosa,” Journal of International Society of Preventive and Community Dentistry, vol. 7, no. 3, pp. 142–146, 2017.

T. S. Gonçalves, L. M. de Menezes, C. Trindade et al., “Cytotoxicity and genotoxicity of orthodontic bands with or without silver soldered joints,” Mutation Research/Genetic Toxicology and Environmental Mutagenesis, vol. 762, pp. 1–8, 2014.

U. S. Barreras, F. T. Méndez, R. E. M. Martínez, C. S. Valencia, P. R. M. Rodríguez, and J. P. L. Rodríguez, “Chitosan nanoparticles enhance the antibacterial activity of chlorhexidine in collagen membranes used for periapical guided tissue regeneration,” Materials Science and Engineering: C, vol. 58, no. 1, pp. 1182–1187, 2016.

F. Angelieri, J. P. C. Marcondes, D. C. de Almeida, D. M. F. Salvadori, and D. A. Ribeiro, “Genotoxicity of corrosion eluates obtained from orthodontic brackets in vitro,” American Journal of Orthodontics and Dentofacial Orthopedics, vol. 139, no. 4, pp. 504–509, 2011.

C.-T. Kao, S.-J. Ding, Y. Min, T. C. Hsu, M.-Y. Chou, and T.-H. Huang, “The cytotoxicity of orthodontic metal bracket immersion media,” The European Journal of Orthodontics, vol. 29, no. 2, pp. 198–203, 2007.

T. Eliades, H. Pratsinis, D. Kletsas, G. Eliades, and M. Makou, “Characterization and cytotoxicity of ions released from stainless steel and nickel-titanium orthodontic alloys,” American Journal of Orthodontics and Dentofacial Orthopedics, vol. 125, no. 1, pp. 24–29, 2004.

P. Tomakidi, U. Koke, R. Kern et al., “Assessment of acute cyto- and genotoxicity of corrosion eluates obtained from orthodontic materials using monolayer cultures of immortalized human gingival keratinocytes,” Journal of Orofacial Orthopedics, vol. 61, no. 1, pp. 2–19, 2000.

G. Cortés-Sandoval, G. A. Martínez-Castañón, N. Patiño-Marín, P. R. Martínez-Rodríguez, and J. P. Loyola-Rodríguez, “Surface roughness and hardness evaluation of some base metal alloys and denture base acrylics used for oral rehabilitation,” Materials Letters, vol. 144, pp. 100–105, 2015.

A. C. S. S. Haddad, A. Tortamano, A. L. de Souza, and P. V. de Oliveira, “An in vitro comparison of nickel and chromium release from brackets,” Brazilian Oral Research, vol. 23, no. 4, pp. 399–406, 2009.

F. Amini, A. Jafari, P. Amini, and S. Sepasi, “Metal ion release from fixed orthodontic appliances–an in vivo study,” The European Journal of Orthodontics, vol. 34, no. 1, pp. 126–130, 2012.

R. M. de Souza and L. M. de Menezes, “Nickel, chromium and iron levels in the saliva of patients with simulated fixed orthodontic appliances,” Angle Orthodontics, vol. 78, no. 2, pp. 345–350, 2008.

F. Amini, E. Asadi, D. Hakimpour, and A. Rakhshan, “Salivary nickel and chromium levels in orthodontic patients with and without periodontitis: a preliminary historical cohort study,” Biological Trace Element Research, vol. 191, no. 1, pp. 10–15, 2018.

M. T. Costa, M. A. Lenza, C. S. Gosch, I. Costa, and F. Ribeiro-Dias, “In vitro evaluation of corrosion and cytotoxicity of orthodontic brackets,” Journal of Dental Research, vol. 86, no. 5, pp. 441–445, 2007.

L. B. Retamoso, T. B. Luz, D. R. Marinowic et al., “Cytotoxicity of esthetic, metallic, and nickel-free orthodontic brackets: cellular behavior and viability,” American Journal of Orthodontics and Dentofacial Orthopedics, vol. 142, no. 1, pp. 70–74, 2012.

A. J. Ortiz, E. Fernandez, A. Vicente, J. L. Calvo, and C. Ortiz, “Metallic ions released from stainless steel, nickel-free, and titanium orthodontic alloys: toxicity and DNA damage,” American Journal of Orthodontics and Dentofacial Orthopedics, vol. 140, no. 3, pp. 115–122, 2011.

G. H. Westphalen, L. M. Menezes, D. Prá et al., “In vivo determination of genotoxicity induced by metals from orthodontic appliances using micronucleus and comet assays,” Genetics and Molecular Research, vol. 7, no. 4, pp. 1259–1266, 2008.

I. Papageorgiou, C. Brown, R. Schins et al., “The effect of nano- and micron-sized particles of cobalt-chromium alloy on human fibroblasts in vitro,” Biomaterials, vol. 28, no. 19, pp. 2946–2958, 2007.

R. R. Tice, E. Agurell, D. Anderson et al., “Single cell gel/comet assay: guidelines for in vitro and in vivo genetic toxicology testing,” Environmental and Molecular Mutagenesis, vol. 35, no. 3, pp. 206–221, 2000.

Z. I. Buljan, S. P. Ribaric, M. Abram, A. Ivankovic, and S. Spalj, “In vitro oxidative stress induced by conventional and self-ligating brackets,” The Angle Orthodontist, vol. 82, no. 2, pp. 340–345, 2012.

A. Katarkar, S. Mukherjee, M. H. Khan, J. G. Ray, and K. Chaudhuri, “Comparative evaluation of genotoxicity by micronucleus assay in the buccal mucosa over comet assay in peripheral blood in oral precancer and cancer patients,” Mutagenesis, vol. 29, no. 5, pp. 325–334, 2014.