Neural Plasticity
Volume 2020 (2020), Article ID 3108490, 8 pages
https://doi.org/10.1155/2020/3108490

Research Article

The Influence of Cochlear Implant-Based Electric Stimulation on the Electrophysiological Characteristics of Cultured Spiral Ganglion Neurons

Na Shen,1,2 Lei Zhou,¹ Bin Lai,³ and Shufeng Li2,4

¹Department of Otolaryngology, Zhongshan Hospital, Fudan University, Shanghai, China
²ENT Institute and Department of Otolaryngology, Eye & ENT Hospital, Fudan University, Shanghai, China
³State Key Laboratory of Medical Neurobiology, Institutes of Brain Science, Fudan University, Shanghai 200032, China
⁴NHC Key Laboratory of Hearing Medicine (Fudan University), Shanghai, China

Correspondence should be addressed to Bin Lai; laibin@fudan.edu.cn and Shufeng Li; lisf@fudan.edu.cn

Received 24 March 2020; Revised 22 July 2020; Accepted 17 August 2020; Published 6 September 2020

Academic Editor: Jian Wang

Copyright © 2020 Na Shen et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

Abstract.

Background. Cochlear implant-based electrical stimulation may be an important reason to induce the residual hearing loss after cochlear implantation. In our previous study, we found that charge-balanced biphasic electrical stimulation inhibited the neurite growth of spiral ganglion neurons (SGNs) and decreased Schwann cell density in vitro. In this study, we want to know whether cochlear implant-based electrical stimulation can induce the change of electrical activity in cultured SGNs. Methods. Spiral ganglion neuron electrical stimulation in vitro model is established using the devices delivering cochlear implant-based electrical stimulation. After 48 h treatment by 50 μA or 100 μA electrical stimulation, the action potential (AP) and voltage depended calcium current (ICa) of SGNs are recorded using whole-cell electrophysiological method. Results. The results show that the ICa of SGNs is decreased significantly in 50 μA and 100 μA electrical stimulation groups. The reversal potential of ICa is nearly +80 mV in control SGN, but the reversal potential decreases to +50 mV in 50 μA and 100 μA electrical stimulation groups. Interestingly, the AP amplitude, the AP latency, and the AP duration of SGNs have no statistically significant differences in all three groups. Conclusion. Our study suggests cochlear implant-based electrical stimulation only significantly inhibit the ICa of cultured SGNs but has no effect on the firing of AP, and the relation of ICa inhibition and SGN damage induced by electrical stimulation and its mechanism needs to be further studied.

1. Introduction

Acoustic signal is transferred by cochlear hair cells from mechanical vibration, and then the signal is transmitted to the auditory cortex via SGNs. Cochlear hair cells in the cochlea are critical for hearing ability [1, 2], and most of the hearing loss is due to irreversible hair cell loss. Cochlear implants can partially replace the function of cochlear hair cells and is the most efficient clinical treatment currently for hearing loss patients. Hybrid cochlear implants, known as the electroacoustic stimulation developed to be used for the patients with high-frequency sensorineural hearing loss and low-frequency residual hearing, has a better clinical effect than the traditional full insertion cochlear implant [3–7]. The use of a shorter, thinner cochlear implant electrode array makes it possible to reduce implantation trauma in the low-frequency region of the cochlea, since the array is only inserted into the basal to the middle part of the cochlea, leaving the apical cochlea intact. When “soft” surgery techniques are used, low-frequency residual hearing can be preserved [8]. It can afford better speech and musical melody recognition than full insertion cochlear implant [6, 9–11].

However, the key point of such benefits in the hybrid cochlear implant depends on the preservation of residual hearing within the implanted inner ear. Unfortunately, more and more clinical researches showed that residual hearing would appear tardive and progressive loss after the hybrid cochlear implant [12–14]. But the mechanism of it was still not clear. Surgical trauma [15], inflammatory, or immune response can be the reasons leading to hair cell death [16]. The formation of fibrosis or new bone growth after implantation can also theoretically cause hair loss by attenuating the traveling wave [17]. But the hearing loss caused by the trauma can be reduced through the optimum-designed implanted electrodes and the enhancement of operate skills [18, 19]. Therefore, the simple trauma cannot explain the mechanism of the tardive and progressive hearing loss.

Electrical stimulation may be an important reason to the residual hearing loss after implantation. Electrical stimulation can promote the neural stem cell’s differentiation into neurons and can promote the maturation of newly generated neuron [20, 21], and also can directly excite the SGNs and their peripheral processes, and the residual hair cells in the low-frequency area [22]. Thus, we suppose that during the electrical-acoustic stimulation, the electrical stimulation may spread to the low-frequency area, then excite the cochlear hair cells and SGNs. This excitation can overlap with the acoustic stimulation and induce the excitatory toxicity. Like the noise-induced deafness, the main mechanisms of the noise-induced deafness are the glutamate excite-toxicity, calcium overload, and oxidative stress [23–25]. Electrical stimulation can both induce voltage-gating calcium channels (VDCCs) opening, and the calcium influx and multiple types of VDCCs are involved in the neurite growth inhibition of SGNs [26] induced by the electrical stimulation. The continuous electrical stimulation may cause excessive Ca²⁺ influx and lead to toxic effect which induces SGN death [27]. This was verified by our previous study [26] that charge-balanced biphasic electrical stimulation inhibited the neurite growth of SGNs and decreased Schwann cell density in vitro. But the effect of cochlear implant-based electrical stimulation on the electrophysiological characteristics, the base of neuronal signal transmission, of spiral ganglion neurons is unclear.

In this study, we use the electrophysiological method to study whether cochlear implant-based electrical stimulation can induce the change of the voltage-dependent calcium currents (ICa) and the AP firing characteristics of SGNs.

2. Materials and Methods

2.1. Spiral Ganglion Cultures

Postnatal day 4 (P4) to P6 rat pups of both sexes were provided and bred by the Laboratory Animal Center of the Eye, Ear, Nose and Throat Hospital (Shanghai, China) under routine conditions according to the institute’s ethical and environmental guidelines. Dissociated spiral ganglion cultures were prepared as follows. Ganglia were dissected from the rat pups after being sacrificed by decapitation, dissociated with trypsin and collagenase, plated on polyornithine/laminin-coated 4-well or 8-well culture chambers (Nalge Nunc International, Naperville, IL), and maintained in high glucose Dulbecco’s Modified Eagle’s Medium (DMEM) (Gibco, Grand Island, US) with N2 supplement (Invitrogen, Carlsbad, CA) with 10% fetal bovine serum and fresh insulin (10 μg/ml, Sigma-Aldrich, St. Louis, MO) in a humidified incubator with 37°C, 5.0% CO2. The cultures were placed in the incubator for twenty-four hours, and then maintained for another 48 hr with electrical stimulation to fix for current-clamp experiments.

2.2. Electrical Stimulation

Two platinum wires were placed into one well which we wanted to give the electrical stimulation. And then, it should be connected to the artificial cochlea device (Reseat Medical Tech. Co., Ltd, Shanghai, CN) when we gave the electrical stimulation (Figure 1(a)). The parameters for the electrical stimulation were as follows: the pulse width was 65 ms, the frequency was 200 ms (Figure 1(b)), and the current intensity was 50 μA or 100 μA.

(a)

(b)

Figure 1: The cultured spiral ganglion neurons (SGNs) are treated with charge-balanced biphasic electrical stimulation. (a) Four-well chamber slides were used in this culture system. Four holes adjacent to the floor were made on two opposite walls of each chamber to introduce two platinum-iridium wires at the two opposite borders. The wires were connected to charge-balanced biphasic pulse generators. (b) The biphasic pluses used for the electrical stimulation were 50 μA or 100 μA amplitude, 65 μs pulse width, 8 μs open-circuit interphase gap, 4862 μs short-circuit phase, and 200 Hz frequency.

2.3. Electrophysiology Experiments

Action potential (AP) and voltage-activated Ca²⁺ current (ICa) were performed using an Axopatch 200B amplifier (Molecular Devices). Electrodes (4-5 MΩ) were pulled from borosilicate glass with P-97 (Sutter, USA). Extracellular solution for APs contained the following (in mM): 145 NaCl, 6 KCl, 2 CaCl₂, 1 MgCl₂, 10 Glucose, and 1 HEEPS, at pH 7.3. The tips of the pipettes were filled with the internal solution containing the following (in mM): 133 K-gluconate, 8 NaCl, 0.6 EGTA, 2 Mg·ATP, 0.3 Na3·GTP, and 10 HEPES, at pH 7.3. Extracellular solution for ICa contained the following (in mM): 120 Choline chloride, 20 TEACl, 5 4-AP, 0.02 linopirdine, 2 CsCl, 1.8–5 CaCl₂, 0.5MgCl₂, 10 HEPES, and 5 D-glucose, at pH 7.4 with NaOH. The tips of the pipettes were filled with the internal solution containing the following (in mM): 70 CsCl, 70 N-methyl-D-glucamine (NMDG), 1 MgCl2, 10 HEPES, 2–5 EGTA, 1 CaCl₂, and 4Cs2ATP, at pH 7.2 with CsOH. Recordings were made from neuronal somata at room temperature. Because the cultures contain neuronal and nonneuronal cells, neurons were confirmed by the presence of a large, transient inward sodium current in whole-cell voltage-clamp mode. A holding potential of -70 mV was chosen to assess responses at a level in which there is minimal voltage-dependent ion channel activation. For AP recording, the APs were induced by injecting the current (100 pA, 2000 ms duration) under current clamp mode. ICa current traces were generated with depolarizing voltage steps from a holding potential of -70 mV to 80 mV and stepped to varying positive potentials (). Whole-cell Ca²⁺current amplitudes at varying test potentials were measured at the peak and steady-state levels using a peak and steady-state detection routine; the current magnitude was divided by the cell capacitance (pF) to generate the current density–voltage relationship. Voltage traces and currents were amplified, filtered (bandpass 2–10 kHz), and digitized at 5–500 kHz using an analog-to-digital convertor Digidata 1200 (Molecular Devices); Data were analyzed using clamp-fit 10.0 software. Data are presented as .

2.4. Statistical Analysis

Graphs were prepared and statistical analysis was done using GraphPad Prism 5.01 (GraphPad Software, Inc.; CA, USA). The significance of differences among all conditions was compared by unpaired -test and One-way ANOVA Kruskal-Wallis test ().

3. Results

3.1. The Electrical Stimulation Inhibits the Voltage-Dependent Calcium Currents (ICa) of SGNs

The ICa was recorded on the cultured SGNs after different electrical stimulation (Figures 2(a) and 2(b)), and the current-voltage curves of ICa were analyzed. The result showed that the inhibited effect of electrical stimulation on the ICa was obvious when the membrane potential was depolarized from the range between 0 mV and +80 mV. The ICa was activated at -60 mV and reached a peak at 20 mV. From -20 mV to +80 mV, the ICa was significantly decreased in 50 μA and 100 μA groups compared with the control group. The ICa has no difference between 50 μA and 100 μA groups. The reversal potential of ICa was nearly in the control group, but after electrical stimulation, the reversal potentials were and in 50 μA and 100 μA groups correspondingly (Figures 2(c) and 2(d)).

(a)

(b)

(c)

(d)

Figure 2: Electrical stimulation decreases voltage-dependent calcium currents (ICa) of SGNs. (a) One SGN was patched with the tip of the microelectrode. (b) The stimulation parameters of voltage-dependent calcium current. (c, d) The inhibitory effect of electrical stimulation was obvious when the membrane potential was depolarized with the step ranging between 0 mV and +80 mV. The ICa was activated at -60 mV and reached the peak at 20 mV, the reversal potential of ICa was nearly in control neuron, but electrical stimulation could change the reversal potential to and in 50 μA and 100 μA groups ().

3.2. The Influence of Electrical Stimulation on the AP Firing Characteristics of SGNs

A total of 34 SGNs were recorded (control group: ; 50 μA group: and 100 μA group: ). Three type action potentials release modes could be found under a depolarized current (100 pA; 2000 ms during) in all neurons; Type I neuron had only one AP after depolarized current stimulation, type II had 2~6 AP and type III could continue firing during the test pulse (Figure 3(a)). The firing characteristics of the first AP were analyzed including AP amplitude, AP decay time, AP half width, and AP rise time. The AP amplitude had no statistically significant difference in all three groups (control group: ; 50 μA group: and 100 μA group: ). The AP latency (control group: ; 50 μA group: and 100 μA group: ) and AP duration (control group: ; 50 μA group: and 100 μA group: ) also had no statistically significant difference in all three groups (Figures 3(b)–3(e)).

(a)

(b)

(c)

(d)

(e)

Figure 3: Influence of electrical stimulation on the firing characteristics of SGNs. (a) Type I neuron had only one action potential after electrical stimulation, type II had 2~6 action potentials, and type III could continue firing APs during the test pulse. (b–e) AP amplitude, latency, and duration had no statistically significant differences in all three groups ().

4. Discussion

In mammal’s inner ear, cochlear hair cells and SGNs are two key cell types for hearing ability, Cochlear hair cells convert the mechanical sound vibrations into electronic neural signals, and SGNs transmit these electronic signals to the auditory cortex. In mammal’s inner ear, cochlear hair cells and SGNs are sensitive for multiple stress and injuries, including noise, gene mutation, ototoxic drugs, inflammation, and aging [28–31]. On the other hand, the mammal’s cochlea only has very limited hair cell and SGN regeneration ability, most of the hair cell loss and SGN loss are permanent and cannot be reversed [32, 33]. Thus, most of the hearing loss is irreversible, and there is no clinical treatment to perfectly cure hearing loss in the clinic by far. Cochlear implant (CI) is an artificial instrument, which is the most widely used neural prosthetic by delivering electrical signals converted from sound information to spiral ganglion neurons and can partially replace the function of cochlear hair cells, and thus is the most efficient clinical treatment currently for hearing loss patients. In the last decade, electric-acoustic stimulation (EAS) technology was newly developed for patients with severe or profound high-frequency hearing loss and residual low-frequency hearing. EAS technology significantly improves music appreciation and speech recognition in background noise through the preservation of residual low-frequency [3, 34]. Unfortunately, clinical trials showed that 30–75% of EAS recipients experienced delayed progressive loss of residual low-frequency hearing over time after the activation of EAS [12–14]. However, the mechanism of this delayed hearing impairment is still not clear by far. In this study, we explored the influence of cochlear implant-based electric stimulation on the electrophysiological characteristics of cultured spiral ganglion neurons.

Calcium overload and oxidative stress are the main causes for SGNs death [27] and the important mechanisms of delayed neuronal death [35]. Our previous study found that charge-balanced biphasic electrical stimulation inhibited the neurite growth of SGNs and decreased Schwann cell density in vitro, and calcium influx through multiple types of VDCCs was involved in the electrical stimulation-induced neurites growth inhibition in SGNs [26]. In this study, we found that ICa was significantly reduced after 48 h electrical stimulation in both 50 μA and 100 μA group. There may be two reasons for the inhibition of ICa caused by electrical stimulation. Firstly, electrical stimulation can decrease voltage-dependent calcium channel expression on the membrane of SGNs, which may be a self-protection mechanism of neurons to reduce the increase of intracellular calcium. Secondly, ICa is directly proportional to the difference of calcium ion concentration inside and outside of the neuron. Electrical stimulation can lead to the release of calcium ions from the intracellular calcium storage; the increase of intracellular calcium concentration can decrease the difference of calcium ion concentration inside and outside of the neuron and leads to the decrease of ICa. Although calcium overload and calcium deficiency are the opposite, they can all be caused by the same stimulus, but only in different stage [36, 37]. There are several evidences for calcium deficiency inducing the apoptosis of neuron. Nakamura et al. found that there was no calcium overload of the neurons in the late stage of apoptosis [38] and even the resting calcium level is lower than normal, and the voltage-dependent calcium influx is significantly reduced [39]. Decreasing extracellular calcium concentration or blocking voltage-gated calcium channels can both trigger neuronal apoptosis [40]. Therefore, neuronal survival may depend on appropriate intracellular calcium set points [41]. Excessively high or low calcium is not conducive to the survival of neurons [42, 43]. In conclusion, the change in voltage-dependent calcium current induced by electrical stimulation reflects an imbalance in intracellular calcium homeostasis, which may be a major cause of neuronal death and apoptosis in the later period, although there is no significant abnormality in AP firing in SGNs.

Electrical stimulation generated by CI itself may also be one of the important factors to affect the survival of residual SGNs. Previous studies have reported that the electric conductive biomaterials and the electrical stimulation have very obvious effects on regulating the proliferation and differentiation of neural stem cells [44, 45], as well as on regulating the survival and maturation of neurons, including SGNs [46]. In this study, we found that electrical stimulation could decrease the ICa but had no effect on the action potential firing in cultured SGNs. This result suggests that, although CI can simulate sound stimulation and active the SGNs, but long-term stimulation of CI will break the calcium balance of SGNs and affect the long-term survival of SGNs negatively.

In addition, Schwann cells, as the main glial cells of the peripheral nervous system, have been shown to secrete a variety of nerve growth factors and axon protection factors, such as glial cell line-derived neurotrophic factor (GDNF) and brain Brain-derived neurotrophic factor (BDNF), can promote the growth of nerve cells [47]. Therefore, electrical stimulation may also affect the surrounding Schwann cells, causing them to degenerate and further aggravate the inhibitory effect of electrical stimulation on the growth and function of spiral neurons.

The SGNs are the first-class neurons of the auditory system [48]. Two types of SGNs were found to compose the first neural elements in the auditory pathway [49]. Type I SGNs exhibit input from only one inner hair cell, whereas type II SGNs extend long projections and receive input from dozens of outer hair cell [50]. Reid et al. proved that there were two types of SGNs with different electrophysiological firing pattern. Type I SGNs could be consistently classified as rapidly accommodating at stimulation and firing only one action potential, while type II SGNs fired significantly more action potentials in response to stimulation [49]. In this study, we also found SGNs with one action potential and 2~6 action potential after electrical stimulation. According to the previous reports, these two types of SGNs can be defined. But one type of SGNs that could continue firing was found in this study. Can it be the third type of SGNs? We will confirm this in the further study.

The firing characteristics of the first AP were analyzed including AP amplitude, AP decay time, AP half width, and AP rise time. There were no statistical significance in all the three groups. This may prompt that electrical stimulation can affect the axon length of SGNs [26] by changing the status of calcium ion channel, but it could not change the AP firing of SGNs by affecting the voltage-dependent sodium channel and potassium channel. The axon retraction may damage the intact between SGNs and cochlear hair cells and cause hearing loss.

5. Conclusion

Our study suggests cochlear implant-based electrical stimulation only significantly inhibit the ICa of cultured SGNs but has no effect on the release of AP, and the relation of ICa inhibition and SGN damage induced by electrical stimulation and its mechanism needs to be further studied.

Data Availability

The data used to support the findings of this study are available from the corresponding author upon request.

Conflicts of Interest

The authors declare that they have no conflicts of interest.

Authors’ Contributions

NS and LZ contributed to the experiments of this paper as well as related statistical analysis. BL and SL contributed to the design of all the experiments, the writing of the manuscript, and the preparation of figures. Na Shen and Lei Zhou contributed equally to this paper.

Acknowledgments

We thank Changjian Xu and Zengjun Sun from Shanghai Cochlear Engineering Technology Research Center, for their technique support of adjust of multichannel charge-balanced biphasic pulse generator. This study was supported by the National Natural Science Foundation of China (NSFC, 31970956, 81171482 and 81670927), the Research Special Fund for Public Welfare Industry of Health (201202001), the Fund of the Science and Technology Commission of Shanghai Municipality (12DZ2251700), and the Shanghai Pujiang Program (18PJD004).

References

Y. Liu, J. Qi, X. Chen et al., “Critical role of spectrin in hearing development and deafness,” Science advances, vol. 5, no. 4, 2019.

J. Qi, Y. Liu, C. Chu et al., “A cytoskeleton structure revealed by super-resolution fluorescence imaging in inner ear hair cells,” Cell discovery, vol. 5, no. 1, 2019.

B. J. Gantz and C. W. Turner, “Combining acoustic and electrical hearing,” Laryngoscope, vol. 113, no. 10, pp. 1726–1730, 2003.

J. Mueller, R. Behr, C. Knaus, C. Milewski, F. Schoen, and J. Helms, “Electrical stimulation of the auditory pathway in deaf patients following acoustic neurinoma surgery and initial results with a new auditory brainstem implant system,” Advances in Oto-Rhino-Laryngology, vol. 57, pp. 229–235, 2000.

J. Kiefer, M. Pok, O. Adunka et al., “Combined electric and acoustic stimulation of the auditory system: results of a clinical study,” Audiology and Neurotology, vol. 10, no. 3, pp. 134–144, 2005.

C. W. Turner, L. A. J. Reiss, and B. J. Gantz, “Combined acoustic and electric hearing: preserving residual acoustic hearing,” Hearing Research, vol. 242, no. 1-2, pp. 164–171, 2008.

S. Irving, L. Gillespie, R. Richardson, D. Rowe, J. B. Fallon, and A. K. Wise, “Electroacoustic stimulation: now and into the future,” BioMed Research International, vol. 2014, Article ID 350504, 17 pages, 2014.

C. Tanaka, A. Nguyen-Huynh, K. Loera, G. Stark, and L. Reiss, “Factors associated with hearing loss in a normal-hearing guinea pig model of hybrid cochlear implants,” Hearing Research, vol. 316, pp. 82–93, 2014.

B. S. Wilson, D. T. Lawson, J. M. Müller, R. S. Tyler, and J. Kiefer, “Cochlear implants: some likely next steps,” Annual Review of Biomedical Engineering, vol. 5, no. 1, pp. 207–249, 2003.

C. W. Turner, B. J. Gantz, C. Vidal, A. Behrens, and B. A. Henry, “Speech recognition in noise for cochlear implant listeners: benefits of residual acoustic hearing,” The Journal of the Acoustical Society of America, vol. 115, no. 4, pp. 1729–1735, 2004.

K. E. Gfeller, C. Olszewski, C. Turner, B. Gantz, and J. Oleson, “Music perception with cochlear implants and residual hearing,” Audiology and Neurotology, vol. 11, no. 1, pp. 12–15, 2006.

B. J. Gantz, M. R. Hansen, C. W. Turner, J. J. Oleson, L. A. Reiss, and A. J. Parkinson, “Hybrid 10 clinical trial,” Audiology & Neuro-Otology, vol. 14, no. 1, pp. 32–38, 2009.

W. Gstoettner, S. Helbig, C. Settevendemie, U. Baumann, J. Wagenblast, and C. Arnoldner, “A new electrode for residual hearing preservation in cochlear implantation: first clinical results,” Acta Oto-Laryngologica, vol. 129, no. 4, pp. 372–379, 2009.

P. L. Santa Maria, C. Domville-Lewis, C. M. Sucher, R. Chester-Browne, and M. D. Atlas, “Hearing preservation surgery for cochlear implantation--hearing and quality of life after 2 years,” Otology & Neurotology, vol. 34, no. 3, pp. 526–531, 2013.

P. S. Roland and C. G. Wright, “Surgical aspects of cochlear implantation: mechanisms of insertional trauma,” in Cochlear and Brainstem Implants, vol. 64, pp. 11–30, 2006.

A. A. Eshraghi, C. Gupta, T. R. van de Water et al., “Molecular mechanisms involved in cochlear implantation trauma and the protection of hearing and auditory sensory cells by inhibition of c-Jun-N-terminal kinase signaling,” Laryngoscope, vol. 123, Supplement 1, pp. S1–14, 2013.

C. H. Choi and J. S. Oghalai, “Predicting the effect of post-implant cochlear fibrosis on residual hearing,” Hearing Research, vol. 205, no. 1-2, pp. 193–200, 2005.

O. Adunka, J. Kiefer, M. H. Unkelbach, T. Lehnert, and W. Gstoettner, “Development and evaluation of an improved cochlear implant electrode design for electric acoustic stimulation,” Laryngoscope, vol. 114, no. 7, pp. 1237–1241, 2004.

S. Berrettini, F. Forli, and S. Passetti, “Preservation of residual hearing following cochlear implantation: comparison between three surgical techniques,” The Journal of Laryngology & Otology, vol. 122, no. 3, pp. 246–252, 2008.

R. Guo, S. Zhang, M. Xiao et al., “Accelerating bioelectric functional development of neural stem cells by graphene coupling: implications for neural interfacing with conductive materials,” Biomaterials, vol. 106, pp. 193–204, 2016.

Z. He, S. Zhang, Q. Song et al., “The structural development of primary cultured hippocampal neurons on a graphene substrate,” Biointerfaces, vol. 146, pp. 442–451, 2016.

S. Y. Kang, D. J. Colesa, D. L. Swiderski, G. L. Su, Y. Raphael, and B. E. Pfingst, “Effects of hearing preservation on psychophysical responses to cochlear implant stimulation,” Journal of the Association for Research in Otolaryngology, vol. 11, no. 2, pp. 245–265, 2010.

H. W. Lin, A. C. Furman, S. G. Kujawa, and M. C. Liberman, “Primary neural degeneration in the Guinea pig cochlea after reversible noise-induced threshold shift,” Journal of the Association for Research in Otolaryngology, vol. 12, no. 5, pp. 605–616, 2011.

Q. Wang and S. H. Green, “Functional role of neurotrophin-3 in synapse regeneration by spiral ganglion neurons on inner hair cells after excitotoxic trauma in vitro,” Journal of Neuroscience, vol. 31, no. 21, pp. 7938–7949, 2011.

J. L. Puel, J. Ruel, C. G. d’Aldin, and R. Pujol, “Excitotoxicity and repair of cochlear synapses after noise-trauma induced hearing loss,” Neuroreport, vol. 9, no. 9, pp. 2109–2114, 1998.

N. Shen, Q. Liang, Y. Liu et al., “Charge-balanced biphasic electrical stimulation inhibits neurite extension of spiral ganglion neurons,” Neuroscience Letters, vol. 624, pp. 92–99, 2016.

J. L. Hegarty, A. R. Kay, and S. H. Green, “Trophic support of cultured spiral ganglion neurons by depolarization exceeds and is additive with that by neurotrophins or cAMP and requires elevation of [Ca2+]i within a set range,” The Journal of Neuroscience, vol. 17, no. 6, pp. 1959–1970, 1997.

L. Liu, Y. Chen, J. Qi et al., “Wnt activation protects against neomycin-induced hair cell damage in the mouse cochlea,” Cell Death & Disease, vol. 7, no. 3, 2016.

C. Zhu, C. Cheng, Y. Wang et al., “Loss of ARHGEF6 causes hair cell stereocilia deficits and hearing loss in mice,” Frontiers in Molecular Neuroscience, vol. 11, p. 362, 2018.

W. Liu, X. Xu, Z. Fan et al., “Wnt signaling activates TP53-induced glycolysis and apoptosis regulator and protects against cisplatin-induced spiral ganglion neuron damage in the mouse cochlea,” Antioxidants & Redox Signaling, vol. 30, no. 11, pp. 1389–1410, 2019.

Z. H. He, S. Y. Zou, M. Li et al., “The nuclear transcription factor FoxG1 affects the sensitivity of mimetic aging hair cells to inflammation by regulating autophagy pathways,” Redox Biology, vol. 28, p. 101364, 2020.

C. Cheng, Y. Wang, L. Guo et al., “Age-related transcriptome changes in Sox2+ supporting cells in the mouse cochlea,” Stem Cell Research & Therapy, vol. 10, no. 1, p. 365, 2019.

F. Tan, C. Chu, J. Qi et al., “AAV-ie enables safe and efficient gene transfer to inner ear cells,” Nature Communications, vol. 10, no. 1, p. 3733, 2019.

C. Vonilberg, J. Kiefer, J. Tillein et al., “Electric-acoustic stimulation of the auditory system,” ORL, vol. 61, no. 6, pp. 334–340, 1999.

D. W. Choi and S. M. Rothman, “The role of glutamate neurotoxicity in hypoxic-ischemic neuronal death,” Annual Review of Neuroscience, vol. 13, no. 1, pp. 171–182, 1990.

J. M. Lee, G. J. Zipfel, and D. W. Choi, “The changing landscape of ischaemic brain injury mechanisms,” Nature, vol. 399, no. 6738, pp. A7–14, 1999.

S. P. Yu, L. M. Canzoniero, and D. W. Choi, “Ion homeostasis and apoptosis,” Current Opinion in Cell Biology, vol. 13, no. 4, pp. 405–411, 2001.

T. Nakamura, H. Minamisawa, Y. Katayama et al., “Increased intracellular Ca2+ concentration in the hippocampal CA1 area during global ischemia and reperfusion in the rat: a possible cause of delayed neuronal death,” Neuroscience, vol. 88, no. 1, pp. 57–67, 1999.

J. A. Connor, S. Razani-Boroujerdi, A. C. Greenwood, R. J. Cormier, J. J. Petrozzino, and R. C. S. Lin, “Reduced voltage-dependent Ca2+ signaling in CA1 neurons after brief ischemia in gerbils,” Journal of Neurophysiology, vol. 81, no. 1, pp. 299–306, 1999.

J. Y. Koh and C. W. Cotman, “Programmed cell death: its possible contribution to neurotoxicity mediated by calcium channel antagonists,” Brain Research, vol. 587, no. 2, pp. 233–240, 1992.

W. He, J. F. Sun, R. Y. Jiang, Z. J. Zhang, Q. Chen, and Y. H. Huang, “Effect of calcium-sensing receptor on the apoptosis of rat spinal cord neurons in anoxia/reoxygenation injury and its significance,” Zhongguo yi xue ke xue Yuan xue bao. Acta Academiae Medicinae Sinicae, vol. 39, no. 5, pp. 623–628, 2017.

T. Koike and S. Tanaka, “Evidence that nerve growth factor dependence of sympathetic neurons for survival in vitro may be determined by levels of cytoplasmic free Ca2+,” Proceedings of the National Academy of Sciences of the United States of America, vol. 88, no. 9, pp. 3892–3896, 1991.

L. Jiang, J. Zhong, X. Dou, C. Cheng, Z. Huang, and X. Sun, “Effects of ApoE on intracellular calcium levels and apoptosis of neurons after mechanical injury,” Neuroscience, vol. 301, pp. 375–383, 2015.

Q. Fang, Y. Zhang, X. Chen et al., “Three-dimensional graphene enhances neural stem cell proliferation through metabolic regulation,” Frontiers in Bioengineering and Biotechnology, vol. 7, 2020.

J. Zhao, M. Tang, J. Cao et al., “Structurally tunable reduced graphene oxide substrate maintains mouse embryonic stem cell pluripotency,” Advanced Science, vol. 6, no. 12, p. 1802136, 2019.

R. Guo, X. Ma, M. Liao et al., “Development and application of cochlear implant-based electric-acoustic stimulation of spiral ganglion neurons,” ACS Biomaterials Science & Engineering, vol. 5, no. 12, pp. 6735–6741, 2019.

P. C. Roehm and M. R. Hansen, “Strategies to preserve or regenerate spiral ganglion neurons,” Current Opinion in Otolaryngology & Head and Neck Surgery, vol. 13, no. 5, pp. 294–300, 2005.

D. J. Zha, Y. Lin, D. J. Zha et al., “Infrared video patch-clamp technique for spiral ganglion neurons in rat cochlear slices,” Acta Otolaryngol, vol. 129, no. 5, pp. 527–532, 2009.

M. A. Reid, J. Flores-Otero, and R. L. Davis, “Firing patterns of type II spiral ganglion neurons in vitro,” Journal of Neuroscience, vol. 24, no. 3, pp. 733–742, 2004.

S. Sun, T. Babola, G. Pregernig et al., “Hair cell mechanotransduction regulates spontaneous activity and spiral ganglion subtype specification in the auditory system,” Cell, vol. 174, no. 5, pp. 1247–1263.e15, 2018.