Hardware Removal in Maxillofacial Trauma: A Retrospective Study

Background Miniplates are commonly used for the fixation of maxillofacial fracture segments. Removal of the hardware is controversial. A retrospective study of medical records was performed to observe the reasons for plate removal. Materials and Methods A 10-year retrospective study of medical records was performed. Demographics, type of fracture, location, type of miniplate used, the time gap between the insertion and removal, and causes of hardware removal were assessed. Results About 1472 patients had undergone internal fixation for the facial fractures. Stainless steel miniplate was used in 489 patients, and titanium was used in 983 patients. Out of the 42 cases, 22 cases involved the removal of titanium hardware and 20 patients involved the removal of stainless steel hardware. Infection/osteomyelitis was the main cause of hardware removal. The maximum amount of hardware failure was in the mandible. 78.6% of hardware removal was performed in males. Conclusion Based on our study, routine removal of titanium miniplates can be performed in children to avoid growth disturbances, not indicated in adult patients unless symptomatic.


Introduction
Management of maxillofacial trauma has proved the best results by internal fixation. e available types of internal fixation methods are macroplates, miniplates, and microplates. e materials used for miniplates are either titanium or stainless steel. e main advantages of internal fixation are prompt stabilization and pain-free mastication. In recent years, titanium miniplates have gained popularity over other modalities. However, the use of plates is not without complications. ey cause obstacles to the future imaging procedure when left in place. Screw loosening, palpable plate, and plate fracture will require a second surgical procedure for plate removal. Infected hardware causes localized abscess, fistula formation, nonunion, and osteomyelitis.
As reported in the literature, stainless steel or cobalt chrome plates can cause corrosion, metal allergy, toxicity, or malignant transformation [1]. Titanium and its alloy have the highest biocompatibility, excellent ductility, tensile strength, nontoxic, and resistance to corrosion than other metals [2][3][4]. ese properties are because of the passive layer of self-regenerating oxide surface called titanium dioxide layer [5]. Titanium plates do not interfere with CT scan, MRI, or radiography. ese properties make titanium popular to use. e purpose of this study is to review the patients who have undergone plate removal postmaxillofacial trauma at our hospital.

Materials and Methods
For this retrospective study, records of patients who had undergone hardware removal over a period of 10 years (1999-2019) were collected after obtaining institutional ethical committee approval.
Medical records were screened to only include patients who required plate removal postopen reduction and internal fixation for maxillofacial trauma treated using stainless steel and titanium hardware.
Plate removal postorthognathic surgery or any other causes were not included in this study.
Data collected included age, sex, chief complaint, a history of presenting illness, medical history, history of the accident, duration of surgery from the date of fracture, type of anesthesia used (LA/GA), tooth in line of fracture, and tooth extracted if any. e fractures were divided as a type of fracture, location of fracture, and number of fractures. e cause of hardware removal was then classified as removal due to screw loosening/hardware failure (plate or screw fracture)/local osteomyelitis/plate exposure/any other. e correlations between the causes for hardware removal and the associating factors were then determined using appropriate statistical analysis.

Results
From the year 1999-2019, 42 patients with a history of craniomaxillofacial fractures secondary to trauma underwent hardware removal. Out of the 42 patients, 33 (78.6%) were male and nine (21.4%) were female. e maximum amount of plate removal was seen in the patients of the age group 31-40 years (35.7%) and the least was for patients < 10 years of age (4.8%). e most common cause of trauma was RTA (road traffic accident) which was seen in 37 (88%) of the patients and the rest of the five (12%) patients were injured due to a fall (Table 1). e location of fracture for 19 (45.2%) patients was the mandibular region and for six (14.3%) patients was the maxillary region. For four (9.5%) patients, the fracture was in the zygomatic region ( Table 1).
Out of the 42 patients, the pus discharge was the chief complaint given by 17 (40.4%) patients. Four (9.5%) underwent elective hardware removal which was prophylactic in a growing patient. Two (4.8%) patients complained of heaviness and metallic feeling in the mouth. One (2.4%) patient complained of loss of sensation. One (2.4%) patient complained about the mobility of the fracture segment ( Table 2).
ere was a higher incidence of hardware removal within seven months to one year which involved 12 (28.5%) patients ( Table 3).
Out of the 42 cases, 22 cases involved the removal of titanium hardware (52.4%) and 20 patients involved the removal of stainless steel hardware (47.6%) ( Table 4).
In nine cases, the tooth was in the line of the fracture and was extracted (21.4%).
e association between the indication of hardware removal specifically soft tissue infection/osteomyelitis and the type of hardware used was found to be statistically significant (p < 0.05). e correlation between the indication of hardware removals such as exposed hardware and screw loosening and the type of hardware used was not statistically significant.
is correlation was determined using the chi-square test.
Certain observations made in our study were as follows: (1) Soft tissue infection/osteomyelitis was the main cause of hardware removal in patients with stainless steel hardware (55%) (2) Only one case required hardware removal due to nerve injury (3) e maximum amount of hardware failure was in the mandible (4) 78.6% (33 cases) hardware removal was performed in males

Discussion
Plate removal in patients is controversial; in some cases, surgical removal is performed only if there are clinical symptoms, and in some cases, it is performed as a routine procedure after the patient has been explained about the risks and informed consent is taken [6][7][8][9][10][11][12]. Some authors believe that the plate can act as a foreign object. is might cause a restriction in growing patients which is a factor to be considered [12,13]. It is important to note that removal of the plate does not imply failure of treatment [6]. Symptoms may develop after the bony union has occurred in when the  [14]. is 10-year retrospective study was performed to determine the reasons for hardware removal in patients who had undergone open reduction and plate fixation postcraniomaxillofacial fractures secondary to trauma. A total of 1,472 patients had undergone internal fixation for facial fractures. Stainless steel miniplate was used in 489(33%) patients, and titanium was used in 983(66%) patients. Of the 42 patients who required plate removal, the most common cause of trauma in these patients was road traffic accident and 33 (78.6%) were males. is could be attributed to the fact that young males are more prone to be involved in RTA's [15].
During the study, we noticed that from the year 1999 to 2011, stainless steel plates were most commonly used, and from 2011 to 2019, titanium miniplates were popularly used.
is may be due to the unavailability of titanium in the early period. ough stainless steel plates are less expensive, the disadvantages include less durability, bulky, cause radiographic scatter at MRI, galvanic potential, and more reactive than titanium and show increased rate of infection [14,16]. Titanium plates have improved biocompatibility when compared to stainless steel plates and excellent osseointegration along with high corrosion resistance [14,16]. A study showed that corrosion in titanium plates is caused mainly due to manufacturing or handling of the materials by the surgeon, and if these two factors are kept in mind during the   e Scientific World Journal placement, then long-term stability of titanium is guaranteed, and routine removal is not required in asymptomatic patients [12]. We do agree the handling errors do occur while drilling the holes and contouring the plate, especially with the unexperienced hands in educational institutions. In our study, a higher number of plates were removed from the mandibular region. Out of 42 patients, 24 patients underwent plate removal from the mandibular region. Mandible fractures are due to severe traumatic forces which lead to further complications which can explain the increased number of hardware removal in this region [17]. Contributing factors for hardware failure in the mandibular region can be the mobility of the mandible which interferes with the stability of the prosthesis, inability to maintain proper oral hygiene, collection of saliva or food in the vestibule [17], proximity to tooth roots [18], constant irritation to the surrounding tissues [18], and trauma due to continuous masticatory forces [10]. ese can cause infections and eventually failure and removal of hardware in this region. Nerve injury might also lead to hardware removal in some cases if the plates were placed in close proximity to the nerves in the mandible [18][19][20]. Other factors can also be the inability to maintain routine follow-up which might lead to improper follow-up measures and local infection [18,21,22].
Another cause of hardware failure in our study was hardware exposure and palpable hardware which was commonly seen in the zygomatic and infraorbital region.
is could be attributed to the zygomaticofrontal region, and the infraorbital region has thin skin, and this leads to the screws becoming more visible over time [17]. Exposure might be commonly seen in cases with thin, soft tissue covering over the plates. Continued exposure leads to palpable hardware and secondary infection [7].
Hardware loosening is also an indication for plate removal during the healing period which can be treated by replacement of the hardware or by rigid external fixation [7].
Increased incidence of plate removal was seen in patients who underwent open reduction and internal fixation with an intraoral surgical approach [15,18,23]. Our study showed that this was five times more than those who underwent open reduction and internal fixation using the extraoral approach. is may be because of the fact that the thickness of soft tissue is much lesser to cover the foreign body. e intraoral approach leads to reduced exposure of the field along with reduced visibility which can lead to further complications such as damage to nerves, tooth roots, and improper placement of the plates which will eventually lead to an increased chance of infection, hardware failure, and removal [15,18]. us, the extraoral approach can be used to ensure a more sterile environment along with increased visibility and fewer complications [15,18,23]. However, there is always a risk of scar formation.
Proper oral hygiene, constant follow-up, and radiographic examination of the teeth in the line of fracture followed by endodontic treatment if necessary should be performed in all cases to reduce the chance of complications [15,18]. Stability by using adequate rigid fixation of the fracture leads to decreased mobility of the foreign body and contributes to a lower risk of infection.
is is very important for mandibular fractures as the mandible is loadbearing and mobile [18].
According to the statistical analysis, our study showed that majority of the plate-related problems usually developed within the first seven months to one year. is is in line with the data observed from other studies [8,11,13,14].
us, informed consent of the patient should be taken at the time of the initial surgery, and plate removal can be mentioned as a possible postoperative complication that may require the patient to undergo a second surgery [8,11,21].
A comparative study by Deepak et al. showed that more complications were seen in patients having stainless steel prosthesis when compared to titanium. is was because the time taken to adapt stainless steel plates was far greater, and this is attributed to the pliability of the material. is could be the factor leading to higher infection rates and wound dehiscence [16].
Our study also showed that the number of removal of titanium plates, when compared to stainless steel, was less. Of the 983 patients, titanium plates amounted to 22 (0.022%) cases only, and of the 489 patients, stainless steel plate removal was performed in 20 (0.04%) cases. is is similar to the data seen in other studies [8,16]. Studies have also shown that titanium plates do not require routine removal unless clinically indicated [6,[9][10][11][12]

Conclusion
Based on the results seen in our study, most of the cases showed complications within the first year of placement irrespective of the material used. us, routine removal is to be considered when the bony union has occurred to prevent any disturbance in the growth of the facial bones in children. Hence, introducing the protocol for routine removal can be a preventive measure, which requires a second surgical procedure. Our current protocol is the removal of titanium plates only in symptomatic adult patients as titanium is most biocompatible and has osseointegration properties.

e Scientific World Journal
Data Availability e data used to support the findings of this study are available from the corresponding author upon request.

Conflicts of Interest
e authors declare that they have no conflicts of interest.